Document ID: chunk:federal_register_of_legislation:F2024C00311:clause:1_13
Version: federal_register_of_legislation:F2024C00311
Segment Type: clause
Provision Reference: sch 1 cl 13
Character Range: 207150–208097

13    50 kg     ‑                                                                     ±4.5 g       ±20 g           ±25 g

 3 The maximum permissible error for even‑arm scales must be half the amount specified in table 11 for loads not more than half capacity and the whole amount specified for loads more than half to maximum capacity.

Other unclassified instruments
 4 The maximum permissible errors for self‑indicating weighing instruments and graduated non‑self‑indicating weighing instruments, with an instrument adjusted to zero within ±0.25 scale interval at no load must be:
 (a) ±0.5 scale interval for the first 500 scale intervals; and
 (b) ±1 scale interval for more than 500 but not more than 2 000 scale intervals; and
 (c) ±1.5 scale intervals for more than 2 000 scale intervals.

Belt weighers
Table 12 Belt weighers

Item  Class  Maximum permissible error for or verification  Maximum permissible error for in‑service inspection