Document ID: chunk:federal_register_of_legislation:F2024C00311:clause:1_5
Version: federal_register_of_legislation:F2024C00311
Segment Type: clause
Provision Reference: sch 1 cl 5
Character Range: 196572–197336

5     10 000  25 000       50

Division 4—Vehicle tanks

Tanks used only for sullage
 1 The maximum permissible error for verification or in‑service inspection for each scale mark on a sight tube is ±0.5 scale interval.

Vehicle tanks except for sullage
 2 The maximum permissible error applicable to a vehicle tank for verification or in‑service inspection is:
 (a) for a tank with a capacity mark—±0.2% of the indicated volume; and
 (b) for a tank with a dipstick—±0.5 scale interval for each scale mark on the dipstick.

Division 5—Liquid‑measuring systems

Accuracy classes
 1 Liquid‑measuring systems are classified into 5 accuracy classes as set out in table 4.
Table 4 Liquid‑measuring systems—Accuracy classes

Item  Accuracy class  Field of application