Document ID: chunk:federal_register_of_legislation:F2024C01214:reg:1:p1
Version: federal_register_of_legislation:F2024C01214
Segment Type: reg
Provision Reference: reg 1 (pt 1/3)
Character Range: 55257–59251

1      Lion Alcometer SD400       Storing:
                                  The device is to be stored in clean and dry conditions and not exposed to extreme weather conditions.
                                  Testing:
                                  The device is to be tested in a manner not inconsistent with the manufacturer's instructions provided that compliance with any action recommended by the manufacturer in the instructions is mandatory.
                                  Maintaining:
                                  The device is to be maintained in a manner not inconsistent with the manufacturer's instructions for use of the device, provided that compliance with any action recommended in those instructions is to be regarded as mandatory.
                                  The device is to be cleaned only with a slightly damp cloth and not exposed to cleaning products that contain solvents.
                                  Operating:
                                  The device is to be operated in a manner not inconsistent with the manufacturer's instructions for use of the device, provided that:
                                     * calibration and configuration of the device is to be undertaken only by the manufacturer or an agent approved by it
                                     * the device is to be configured so that:
                                     –     Test Mode is set to Active Sampling
                                     –     Digital Display Mode is set to Reveal
                                     –     Display Format is set to Positive or Negative
                                     –     Measurement ratio is set to 2100:1
                                     –     Display Mask is set to declare a result at or below 0.009 as Negative
                                     –     Breath Sampling Parameters are set to be:
                                              for discard volume parameter — between 1 and 5 litres
                                              for drop or flow rate parameter – between 20 and 50 litres/minute
                                     –     Display Reset Time is set to 10 seconds
                                     –     Calibration Lock is to render the device inactive if recalibration has not been undertaken within the last 6 months
                                     –     Device Firmware is dongle protected and accessible only by the manufacturer or an agent approved by it
                                     * the following optional functions are not to be activated:
                                     –     Passive Test Mode
                                     –     Hide Digital Display Mode
                                     –     Actual Alcohol Units Display
                                     –     Timed Sample Capture
                                     –     Precursory Mode
                                     –     Manual Sampling
                                     * the device is only to be used in Analyse Mode
                                     * the device is to be used only with one‑use‑only mouthpieces and not with the reusable sampling cup
                                     * a clean unused mouthpiece is to be used for every test
                                     * only mouthpieces manufactured or supplied by Lion Laboratories Limited are to be used
                                     * the device is not to be used with "online‑400" PC software
                                     * printing of test results is not required.
 2     Drager Alcotest 9510 Aus   Storing:
                                  The device is to be stored in its travel case in clean and dry conditions and not exposed to extreme weather conditions.
                                  Testing:
                                  The device is to be verified to National Measurement Institute standard R126 so that after verification only the time and date