Document ID: chunk:federal_register_of_legislation:F2024C00311:clause:1_3
Version: federal_register_of_legislation:F2024C00311
Segment Type: clause
Provision Reference: sch 1 cl 3
Character Range: 208229–209128

3     2.0    ±1.0%                                          ±2.0%

Catch weighers
 5 The maximum permissible error for any load equal to or greater than the minimum capacity and equal to or less than the maximum capacity in automatic operation is:
 (a) if the national instrument test procedures that apply to catch weighers eliminate the need for digital rounding—the maximum permissible error set out in table 13 minus a verification scale interval of 0.5 e; or
 (b) in any other case—set out in table 13.
Note: The national instrument test procedures are defined in the Act and are available at www.nmi.gov.au.
Table 13 Automatic catch weighing instruments

Item  Load (m) expressed in verification scale intervals (e)  Maximum permissible error for class Y (a) or Y (b) instruments—verification or in‑service inspection
      Class Y (a)                                             Class Y (b)