Source: http://www.readbag.com/accessdata-fda-cdrh-docs-pdf5-k052867
Timestamp: 2019-07-17 10:57:40
Document Index: 366980908

Matched Legal Cases: ['arts 800', 'art 807', 'arts 801', 'art 820', 'art 807', 'art 21', 'art 0']

Read K052867.pdf
Read K052867.pdf text version
Ortho-Clinical Diagnostics DEL 2 3 2005
Rochester. New York 14626-5101
coimny
This summary of 510(k) safety and effectiveness information is being submitted in accordance with the requirements of SMDA 1990 and 21 CFR 807.92. The assigned 5 10(k) number is:
1. Submitter name, address, contact
Ortho-Clinical Diagnostics, Inc. 100 Indigo Creek Drive Rochester, New York 14626-5 101 (585) 453-4253 Contact Person: Darlene Phillips
2. Preparation Date 3. Device name
October 10, 2005 Trade or Proprietary Names: VITROS Chemistry Products dTIBC Reagent VITROS Chemistry Products Calibrator Kit 29 VITROS Chemistry Products Performance Verifiers I &amp; 11 Common Names: Total iron binding capacity assay Classification Names: Iron-binding capacity test system (862.1415): Class: I Calibrator (862.1150): Calibrator Kk 29: Class II. Quality Control material (assayed and unassayed) (862.1660): Class I (general controls). Since these devices (VITROS Performance Verifiers I &amp; 1I) are assayed controls, they meet the reserved criteria under Section 510(1) of the Food, Drug, and Cosmetic Act.
Continued on next page Ortho-Clinical Diagnostics, Inc VITROS Chemistry Products dTIBC assay and controls Page l ot'5
510(k) Summary (continued) 4. Predicate Devices The VITROS Chemistry Products dTIBC assay is substantially equivalent to the Dade Behring Total Iron Binding Capacity (IBCT) Flex® assay on the Dimension® clinical chemistry systems. The VITROS Chemistry Products Performance Verifiers I and 1I are substantially equivalent to the VITROS Chemistry Products Performance Verifiers I and 11. The VITROS Chemistry Products dTIBC Reagent is used in conjunction with the VITROS Chemistry Products Calibrator Kit 29 on the VITROS 5,1 FS Chemistry Systems for the quantitative measurement of TIBC in human serum. The VITROS Chemistry Products dTIBC Reagent is a dual chambered package containing ready-to-use liquid reagents. Reagent 1, an acidic buffer containing ferric ions bound to chromazurol B (iron-binding dye) is added to the sample. The acidic pH releases iron from transferrin and the released iron binds to the excess chromazurol B. Reagent 2, a neutral buffer is added, shifting the pH, which results in increased affinity'of transferrin for iron. Serum transferrin rapidly extracts iron from the dye- iron complex. The decrease in absorbance of the colored dye-iron complex is directly proportional to the total iron-binding capacity of the sample and is measured spectrophotometrically at 660 nm. Once a calibration has been performed, the TIBC concentration in each unknown sample can be determined using the stored calibration curve and the measured absorbance obtained in the assayof the sample. The VITROS Chemistry Products Calibrator Kit 29 is a two level standard used to calibrate VITROS 5,1 FS Chemistry Systems for the quantitative measurement of total iron binding capacity (TIBC). VITROS Calibrator Kit 29 level I is an aqueoim solution containing processed bovine serum albumin, and preservative. VITROS Calibrator Kit 29 level 2 is a lyophilate containing processed human serum, proteins, enzymes, organic compounds, electrolytes, immunoglobulins, inorganic compounds, hormones, and metals. The VITROS Chemistry Products FS Reconstitution Diluent is processed water used to reconstitute the VITROS Calibrator Kit 29 level 2. The VITROS Chemistry Products Performance Verifiers I and LI are lyophilized materials prepared from processed human serum to which enzymes, electrolytes, stabilizers, preservatives, and other organic analytes have been added. The lypohilate is reconstituted using diluent manufactured from processed water to which inorganic salts have been added. These are assayed quality control materials are used to monitor the performance of the VITROS dTIBC assay on the VITROS 5,1 FS System.
Ortho-Clinical Diagnostics. Inc.
VITROS Chenistry Products dTIBC assay and controls
510(k) Summary (continued)
The VITROS dTIBC assay utilizes VITROS Chemistry Products FS Diluent Pack 2 (BSA/Saline), a common reagent that is used by multiple assays on the VITROS 5,1 FS System. This is a dual chambered package containing two ready-to-use liquid diluents. Diluent I (Saline) is prepared from processed water to which inorganic salt has been added. Diluent 2 (BSA) is prepared from processed water to which bovine serum albumin, inorganic salts and preservatives have been added. The VITROS 5,1 FS Chemistry System is a clinical chemistry instrument that provides automated use of the VITROS Chemistry Products MicroTip® and MicroSlides® range of products. The VITROS 5,1 FS Chemistry System was cleared for market by a separate 510(k) premarket notification (K031924). VITROS Chemistry Products dTIBC Reagent: For in vitro diagnostic use only. VITROS Chemistry Products dTIBC Reagent is used to quantitatively measure total iron-binding capacity (TIBC) in human serum. VITROS Chemistry Products Calibrator Kit 29: For in vitro diagnostic use only. VITROS Chemistry Products Calibrator Kit 29 is used to calibrate VITROS 5,1 FS Chemistry Systems for the quantitative measurement of total iron-binding capacity (TIBC) using VITROS Chemistry Products dTIBC Reagent. VITROS Chemistry Products Performance Verifiers I &amp; lI: For in vitro diagnostic use only. VITROS Chemistry Products Performance Verifiers are assayed controls used to monitor the performance of VITROS Chemistry Systems. The VITROS Chemistry Products dTIBC Reagent and VITROS Chemistry Products Calibrator Kit 29 are substantially equivalent to the Dade Behring Total Iron Binding Capacity (IBCT) Flex reagent cartridge (K994115) on the Dimension® clinical chemistry systems. (K994115) and Dade Behring IBCT Calibrator (K994114) (predicate devices) which were cleared by the FDA for IVD use. The relationship between the VITROS dTIBC assay and the predicate device, determined by least squares linear regression, is: VITROS dTIBC =0.94 X + 12.99 (ltg/dL), with a correlation coefficient of 0.981, where X is the Dade Behring Total Iron Binding Capacity (IBCT) Flex® assay on the Dimension® clinical chemistry system.
6. Device intended uses
7. Comparison to predicate devices
VITROS Chemistry Products dTIBC assay and controls
510(k) Summary (continued) 7. Comparison to predicate devices
The VITROS Chemistry Products Performance Verifiers I &amp; 1I are substantially equivalent to the VITROS Chemistry Products Performance Verifiers I &amp; II (predicate device) which was cleared by the FDA (K042006) for IVD use. In addition to correlation studies, bench testing was performed to determine assay precision, linearity, specificity, expected values, limit of detection, dilution and specimen matrix of the VITROS dTIBC assay. Tables 1 and 2 summarize the similarities and differences between the new devices and the predicates.
Similarities and differences between the VITROS dTIBC assay (new device) and the DADE IBCT assay (predicate device). Similarities For in vitro diagnostic use only. Used to quantitatively measure total iron-binding capacity (TIBC) in human samples. The iron binding capacity is useful in the differential diagnosis of anemia, iron deficiency anemia, thalassemia, sideroblastic anemia, and iron poisoning. None required Clinical chemistry analyzer Traceable to NIST SRM 937
Sample Pretreatment Instrumentation Calibration
Device Characteristic Sample Type Reportable Range Calibrators
VITROS dTIBC assay (New device) Human Serum 60- 650 gg/dL Two levels
DADE IBCT assay (Predicate device) Human Serum and plasma 0 - 1000 [tg/dL Three levels
510(k) Summary (continued) Table 2 Similarities and differences between the VITROS Performance Verifiers (new device) and the VITROS Performance Verifiers (predicate device). Matrix Similarities The lyophilized performance verifiers are prepared from processed human serum to which enzymes, electrolytes, stabilizers, preservatives, and other organic analytes have been added. Manufactured from processed water to which inorganic salts have been added. Two Differences VITROS Performance Verifiers (New Device) For in vitro diagnostic use only. Assayed controls used to monitor the performance on VITROS Chemistry Systems. New additional intended use: to monitor performance of VITROS dTIBC assay on VITROS 5,1 FS Chemistry Systems. VITROS dTIBC MicroTip assay in addition to all of those supported by the predicate device.
Diluent Number of levels
Device Characteristic Indications for Use
VITROS Performance Verifiers (Predicate Device) For in vitro diagnostic use only. Assayed controls used to monitor the performance on VITROS Chemistry Systems.
VITROS assays supported
Several assays including VITROS ALB, TP, TRIG, URIC, LAC, and dHDL, and TIBC MicroSlide® assays and VITROS dHDL and dLDL MicroTip® assays.
The data presented in the premarket notification provide a reasonable assurance that the VITROS dTIBC assay and the VITROS Chemistry Products Performance Verifiers are safe and effective for the stated intended uses and are sub stantially equivalent to the cleared predicate devices. Equivalence to predicates was demonstrated using commercially available reagents along with patient samples.
Ms. Darlene Phillips Regulatory Associate Ortho-Clinical Diagnostic, Inc. 100 Indigo Creek Drive Rochester, New York 14626-5101
k052867 Trade/Device Name: VITROS Chemistry Products dTIBC Regent VITROS Chemistry Products Calibrator Kit 29 VITROS Chemistry Products Chemistry Performance Verifiers I &amp; II Regulation Number: 21 CFR 862.1415 Regulation Name: Iron-binding capacity test system Regulatory Class: Class I Product Code: JMO, JIS, JJY Dated: October 7, 2005 Received: October 11, 2005
Dear Ms. Phillips: We have reviewed your Section 510(k) premarket notification of intent to market the device referenced above and have determined the device is substantially equivalent (for the indications for use stated in the enclosure) to legally marketed predicate devices marketed in interstate commerce prior to May 28, 1976, the enactment date of the Medical Device Amendments, or to devices that have been reclassified in accordance with the provisions of the Federal Food, Drug, and Cosmetic Act (Act) that do not require approval of a premarket approval application (PMA). You may, therefore, market the device, subject to the general controls provisions of the Act. The general controls provisions of the Act include requirements for annual registration, listing of devices, good manufacturing practice, labeling, and prohibitions against misbranding and adulteration. If your device is classified (see above) into either class IL(Special Controls) or class III (PMA), it may be subject to such additional controls. Existing major regulations affecting your device can be found in Title 2 1, Code of Federal Regulations (CFR), Parts 800 to 895. In addition, FDA may publish further announcements concerning your device in the Federal Register. Please be advised that FDA's issuance of a substantial equivalence determination does not mean that FDA has made a determination that your device complies with other requirements of the Act or any Federal statutes and regulations administered by other Federal agencies. You must comply with all the Act's requirements, including, but not limited to: registration and listing (21 CFR Part 807); labeling (21 CFR Parts 801 and 809); and good manufacturing practice requirements as set forth in the quality systems (QS) regulation (21 CFR Part 820).
Page 2 This letter will allow you to begin marketing your device as described in your Section 510(k) premarket notification. The FDA finding of substantial equivalence of your device to a legally marketed predicate device results in a classification for your device and thus, permits your device to proceed to the market. If you desire specific information about the application of labeling requirements to your device, or questions on the promotion and advertising of your device, please contact the Office of In Vitro Diagnostic Device Evaluation and Safety at (240) 276-0484. Also, please note the regulation entitled, &quot;Misbranding by reference to premarket notification&quot; (21CFR Part 807.97). You may obtain other general information on your responsibilities under the Act from the Division of Small Manufacturers, International and Consumer Assistance at its toll-free number (800) 638-2041 or (301) 443-6597 or at its Internet address http://www.fda. gov/cdrh/industrv/supportlindex .html.
Director Division of Chemistry and Toxicology Office of In Vitro Diagnostic Device Evaluation and Safety Center for Devices and Radiological Health Enclosure
510(k) Number (if known): Device Name(s):
K052867 VITROS Chemistry Products dTIBC Reagent VITROS Chemistry Products Calibrator Kit 29 VITROS Chemistry Products Performance Verifiers I &amp; II For in vitro diagnostic use only. VITROS Chemistry Products dTIBC Reagent is used to quantitatively measure total iron-binding capacity (TIBC) in human serum. The iron binding capacity is useful in the differential diagnosis of anemia, iron deficiency anemia, thalassemia, sideroblastic anemia, and iron poisoning. For in vitro diagnostic use only. VITROS Chemistry Products Calibrator Kit 29 is used to calibrate VITROS 5,1 FS Chemistry Systems for the quantitative measurement of total iron-binding capacity (TIBC) using VITROS Chemistry Products dTIBC Reagent. For in vitro diagnostic use only. VITROS Chemistry Products Performance Verifiers are assayed controls intended for use in monitoring performance on VITROS Chemistry Systems.
-X_Prescription Use _ (Part 21 CFR 801 Subpart 0)
PAGE IF (PLEASE DO NOT WRITE BELOW THIS LINE-CONTINUE ON ANOTHER NEEDED)
Division Sign-10ff Office of In Vitro Diagnostic Device Evaluation and Safety 5HO0k)._ 4&lt; 05! --