Source: http://homedocbox.com/Environmental_Safety/66532016-English-version-central-secretariat-rue-de-stassart-35-b-brussels.html
Timestamp: 2018-11-17 17:36:11
Document Index: 184536558

Matched Legal Cases: ['art 35', 'art 35', 'art 2', 'art 35', 'art 2', 'art 2', 'art 4', 'art 1', 'art 051', 'art 1', 'art 503', 'art 1']

English version. Central Secretariat: rue de Stassart 35, B Brussels - PDF
Download "English version. Central Secretariat: rue de Stassart 35, B Brussels"
1 TECHNICAL REPORT CLC/TR RAPPORT TECHNIQUE TECHNISCHER BERICHT December 2008 ICS English version Passive components to be used in optical fibre communication systems - Product specifications - Part 2-1: SC(SC2)-PC connector-type fixed optical attenuators using IEC Category B1.1 singlemode fibre Composants passifs destinés à être utilisés dans les systèmes de communication par fibres optiques - Spécifications de produit - Partie 2-1: Affaiblisseurs optiques fixes de type connecteur SC(SC2)-PC utilisant une fibre unimodale de la catégorie B1.1 de la CEI Passive Bauteile zur Verwendung in Lichtwellenleiter- Kommunikationssystemen - Produktnormen - Teil 2-1: Bauart SC(SC2)-PC Steckverbinder mit optischem Dämpfungsglied zum Anschluss an Einmodenfasern der Kategorien B1.1 nach IEC This Technical Report was approved by CENELEC on CENELEC members are the national electrotechnical committees of Austria, Belgium, Bulgaria, Cyprus, the Czech Republic, Denmark, Estonia, Finland, France, Germany, Greece, Hungary, Iceland, Ireland, Italy, Latvia, Lithuania, Luxembourg, Malta, the Netherlands, Norway, Poland, Portugal, Romania, Slovakia, Slovenia, Spain, Sweden, Switzerland and the United Kingdom. CENELEC European Committee for Electrotechnical Standardization Comité Européen de Normalisation Electrotechnique Europäisches Komitee für Elektrotechnische Normung Central Secretariat: rue de Stassart 35, B Brussels 2008 CENELEC - All rights of exploitation in any form and by any means reserved worldwide for CENELEC members. Ref. No. CLC/TR :2008 E
2 CLC/TR : Foreword This Technical Report was prepared by the Technical Committee CENELEC TC 86BXA, Fibre optic interconnect, passive and connectorised components. The text of the draft was circulated for voting in accordance with the Internal Regulations, Part 2, Subclause (simple majority) and was approved by CENELEC as CLC/TR on
3 - 3 - CLC/TR :2008 Passive components to be used in optical fibre communication systems Product specifications Part 2-1: SC(SC2)-PC connector-type fixed optical attenuators using IEC Category B1.1 singlemode fibre Description Performance Type: fixed Application: EN :2007, Category U Style: plug style ES (see 1.3) Operating wavelength: Coupling mechanism: ( ) nm and ( ) nm push pull Attenuation: 1 db, 5 db and 15 db Configuration: plug-attenuatoradaptor-plug Fibre category: special fibre Return loss grade: W: 45 db References: EN ) Connector sets and interconnect components to be used in optical fibre communication systems - Product specifications - Part 4-4: Type SC-PC simplex terminated on IEC category B1.1, B1.3 and B6a singlemode fibre, with full zirconia ferrule Category U EN series Fibre optic interconnection devices and passive components - Basic test and measurement procedures (IEC series) EN Fibre optic interconnecting devices and passive components performance standard - Part 1: General and guidance for performance standards (IEC ) EN Fibre optic interconnecting devices and passive components performance standard - Part 051-3: Single-mode fibre, plug-style fixed attenuators for Category U - Uncontrolled environment (IEC ) EN Fibre optic connector optical interfaces - Part 1: Optical interfaces for single mode non-dispersion shifted fibres - General and guidance (IEC ) ETSI ES Transmission and Multiplexing (TM); Passive optical components; Connectortype optical fixed attenuators for single-mode optical fibre communications systems; Common requirements and conformance testing ETSI EN Equipment Engineering (EE); Environmental conditions and environmental tests for telecommunications equipment Outline: 1) At draft stage.
4 CLC/TR : Contents 1 Scope Test samples Test program Spectral attenuation loss (according to EN ) Polarisation dependent loss (PDL) (according to EN , option 1) Test laboratories involved in this RRT Measurement results Individual test laboratory results with reference connectors Laboratory A results Laboratory B results Laboratory C results Laboratory D results Laboratory E results Laboratory F results Laboratory G results Overview of all attenuation test results Attenuation measurements summary Attenuation: Random mating versus reference connectors Overview of PDL results Mechanical interface issues with SC plug style attenuators Conclusions... 27
5 - 5 - CLC/TR : Scope This document reports the measurement results of a round robin test program carried out on SC/PC plug style fixed attenuators. The work was initiated at CENELEC TC 86BXA meeting in June 2003 in order to get a clear understanding on the accuracy and repeatability of the spectral attenuation loss measurements on fixed attenuators. Out of these results some recommendations are made for attenuation tolerance values that can be used in the performance standards. 2 Test samples In total 18 SC/PC plug style fixed attenuators were collected for this round robin test: - 6 attenuators with nominal attenuation of 15 db (labelled 1, 2, 3, 4, 5 and 6); - 6 attenuators with nominal attenuation of 5 db (labelled 7, 8, 9, 10, 11 and 12); - 6 attenuators with nominal attenuation of 1 db (labelled 13, 14, 15, 16, 17 and 18). The attenuators were obtained from various suppliers. The attenuating principle of all devices is based on the use of high attenuating fibre. The performance grade of these attenuators is defined as: - operating wavelength range: nm nm and nm nm; - attenuation tolerance: 0,5 db for attenuators 5 db, 10 % of nominal attenuation value for attenuators > 5 db. 3 Test program All participating test laboratories measured spectral attenuation and polarisation dependent loss for each attenuator. 3.1 Spectral attenuation loss (according to EN ) For the ease of the data processing the measured values were reported for the discrete wavelengths: nm, nm, nm, nm, nm, nm, nm, nm, nm, nm and nm. The spectral width was set at 2 nm. Some laboratories also reported values with a spectral width of 10 nm. Each test laboratory performed the measurements with 2 sets of reference connectors and adapters: - measurements with common reference connectors and adapter (same references for all the test laboratories); - measurements with own test laboratory reference connectors and adapter. Uncertainty of each loss measurement at the above mentioned wavelength range was better than 0,1 db. 3.2 Polarisation dependent loss (PDL) (according to EN , option 1) PDL was measured at nm and nm, with common reference connectors. The accuracy of each PDL measurement was better than 0,1 db. 4 Test laboratories involved in this RRT The following laboratories were involved in this round robin test (in alphabetic order): - Adamant Kogyo Co. Ltd (Japan); - Diamond (Switzerland); - Huber and Suhner (Switzerland); - Telekomunikacja Polska (Poland); - TILab (Telecom Italia Laboratories) (Italy); - Tyco Electronics-AMP (the Netherlands); - Tyco Electronics-Raychem (Belgium).
English version. Central Secretariat: Avenue Marnix 17, B Brussels
TECHNICAL REPORT CLC/TR 62453-503-2 RAPPORT TECHNIQUE TECHNISCHER BERICHT November 2009 ICS 25.040.40; 35.100.05; 35.110 English version Field device tool (FDT) interface specification - Part 503-2: Communication
TECHNICAL REPORT RAPPORT TECHNIQUE TECHNISCHER BERICHT CEN/TR 15371-1 May 2017 ICS 97.200.50 Supersedes CEN/TR 15371-1:2015 English Version Safety of toys - Interpretations - Part 1: Replies to requests