Source: https://patents.google.com/patent/WO2010129095A2/en
Timestamp: 2019-08-20 01:51:41
Document Index: 252141074

Matched Legal Cases: ['art 212', 'art 212', 'art 212', 'art.\n22', 'art.\n83', 'art.\n85', 'art.\n87', 'art.\n97', 'art.\n106', 'art.\n108', 'art.\n117', 'art.\n122']

WO2010129095A2 - Impedance based anatomy generation - Google Patents
WO2010129095A2
WO2010129095A2 PCT/US2010/027436 US2010027436W WO2010129095A2 WO 2010129095 A2 WO2010129095 A2 WO 2010129095A2 US 2010027436 W US2010027436 W US 2010027436W WO 2010129095 A2 WO2010129095 A2 WO 2010129095A2
PCT/US2010/027436
WO2010129095A3 (en
2009-05-08 Priority to US12/437,812 priority Critical
2009-05-08 Priority to US12/437,794 priority
2009-05-08 Priority to US12/437,794 priority patent/US8103338B2/en
2009-05-08 Priority to US12/437,812 priority patent/US8571647B2/en
2010-03-16 Application filed by Rhythmia Medical, Inc. filed Critical Rhythmia Medical, Inc.
2010-11-11 Publication of WO2010129095A2 publication Critical patent/WO2010129095A2/en
2011-01-13 Publication of WO2010129095A3 publication Critical patent/WO2010129095A3/en
Attorney Docket No.: 20240-0013WOl Client Ref. No.: RMD-INVD-8A
Conventional 3D mapping techniques include contact mapping and non-contact mapping. In contact mapping techniques one or more catheters are advanced into the heart. Physiological signals resulting from the electrical activity of the heart are acquired with one or more electrodes located at the catheter distal tip after determining that the tip is in stable and steady contact with the endocardium surface of a particular heart chamber. Location and electrical activity is usually measured sequentially on a point-by- point basis at about 50 to 200 points on the internal surface of the heart to construct an electro-anatomical depiction of the heart. The generated map may then serve as the basis for deciding on a therapeutic course of action, for example, tissue ablation, to alter the propagation of the heart's electrical activity and to restore normal heart rhythm. On the other hand, in non-contact-based mapping systems a multiple electrodes catheter is percutaneously placed in the heart chamber of interest. Once in the chamber, the catheter is deployed to assume a 3D shape. Using the signals detected by the non-contact electrodes and information on chamber anatomy and relative electrode location, the system provides physiological information regarding the endocardium of the heart chamber. Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
In some aspects, a method includes inserting a catheter into a heart, the catheter comprising three or more electrodes. The method also includes moving the catheter to each of multiple, different positions in the heart. The method also includes for each of the different catheter positions, causing current to flow between at least some of the electrodes and in response to current flow, measuring an electrical signal at each of one or more of the electrodes. The method also includes determining anatomical information about the heart based on positions of the catheter electrodes and the measured signals at the different catheter positions. Embodiments can include one or more of the following.
Determining the anatomical information can include detecting a boundary of the heart. Determining the anatomical information can include, for each of the different catheter positions, determining a surface based on the measured signals, the surface representing a surface at which the conductivity changes. The surface can be a closed and parameterized surface around at least a portion of the catheter. The surface can be an ellipsoid. The surface can be a curvilinear surface. The surface can provide a boundary between a region represented by a first conductivity inside the surface and a region represented by a second conductivity outside the surface, the first conductivity being different from the second conductivity. Attorney Docket No. 20240-00 BWOl Client Reference RMD-INVD-8A
Selecting the one or more regions can include selecting the one or more regions based at least in part on a distance between a portion of the surface and the catheter. Selecting the one or more regions can include selecting the one or more regions based at least in part on a magnitude of a distortion field, the distortion field being based at least in part on a difference between a field calculated based on the measurements and a field in a homogonous medium. Selecting the one or more regions can include selecting the one or more regions based at least in part on an error calculation in an optimization used to generate the surface. Determining the anatomical information can include joining the regions of the determined surfaces corresponding to an expected chamber boundary to generate the anatomical information. Joining the regions can include using a meshing algorithm. The method can also include using the multiple electrodes on the catheter to measure cardiac signals at the catheter electrodes in response to electrical activity in the heart.
In some additional aspects, a method can include inserting a catheter into a heart, the catheter comprising multiple, spatially distributed electrodes including multiple sets of electrodes each set comprising at least two electrodes. The method can also include Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
for each of the multiple different sets of electrodes, causing current to flow between at least some of the electrodes and in response to current flow, measuring an electrical signal at each of one or more measuring electrodes. The method can also include determining anatomical information based on the measured signals. Embodiments can include one or more of the following.
The determination can account for a first conductivity inside the cardiac chamber boundary and a second conductivity outside the cardiac chamber boundary. Determining the anatomical information can include determining the anatomical information based at least in part on impedance information generated based on the measured signals. The impedance information can be based on a conductivity contrast between blood and surrounding tissue. The anatomical information can include a representation of at least a portion of a boundary of the heart. Determining the anatomical information can include detecting a boundary of the heart. Determining the anatomical information can include determining a surface based on the measured signals, the surface representing a surface at which the conductivity value changes. The surface can be a closed and parameterized surface around at least a portion of the catheter. The surface can be an ellipsoid. The surface can be a curvilinear surface. The surface can provide a boundary between a region represented by a first conductivity inside the surface and a region represented by a second conductivity outside the surface, the first conductivity being different from the second conductivity.
Determining the anatomical information can include for each of the determined surfaces selecting one or more regions of the surface corresponding to an expected boundary of the heart. Selecting the one or more regions can include selecting the one or more regions based at least in part on a distance between a portion of the surface and the catheter. Selecting the one or more regions can include selecting the one or more regions based at least in part on a magnitude of a distortion field, the distortion field being based at least in part on a difference between a field calculated based on the measurements and a field in a homogonous medium. Selecting the one or more regions can include Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
selecting the one or more regions based at least in part on an error calculation in an optimization used to generate the surface.
In some aspects, a method includes inserting a catheter into a heart, the catheter comprising three or more electrodes. The method also includes causing current to flow between at least some of the electrodes and in response to current flow, measuring an electrical signal at each of one or more of the electrodes. The method also includes determining a boundary of at least a portion of the heart based on the measured electrical signals. The method also includes displaying a portion of less than the entire boundary of the heart. Embodiments can include one or more of the following.
The method can also include displaying the physiological information at multiple locations of the boundary of the heart. Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
Determining the anatomical information can include joining the regions of the determined surfaces corresponding to an expected chamber boundary to generate the anatomical information. Attorney Docket No. 20240-0013WOl Client Reference RMD-INVD-8A
Displaying the portion of less than the entire boundary of the heart can include using a visual indicia to indicate a valid area of the boundary. The method can also include determining a valid area of the boundary.
Displaying the portion of less than the entire boundary of the heart comprises using a visual indicia to indicate the determined valid area of the boundary.
Embodiments can include one or more of the following. The processing unit can be configured to account for a change in conductivity at a cardiac chamber boundary in the determination of the anatomical information.
The processing unit can be configured to account for a first conductivity inside the cardiac chamber boundary and a second conductivity outside the cardiac chamber boundary. The processing unit can be configured to determine the anatomical information based at least in part on impedance information. The impedance information can be based on a conductivity contrast between blood and surrounding tissue.
The anatomical information can include a representation of at least a portion of a boundary of the heart. The processing unit can be configured to determine the anatomical information by determining a surface based on the measured signals, the surface representing a surface at which the conductivity changes.
The processing unit can be configured to, for each of the determined surfaces, select one or more regions of the surface corresponding to a boundary of a portion of the heart. Attorney Docket No. 20240-0013WO1 Client Reference RMD-INVD-8A
The processing unit can be configured to determine the anatomical information based at least in part on impedance information generated based on the measured signals at the different catheter positions. Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
The processing unit can be configured to select the one or more regions based at least in part on an error calculation in an optimization used to generate the surface. The processing unit can be configured to join the regions of the determined surfaces corresponding to an expected chamber boundary to generate the anatomical information.
The electrodes on the catheter can be further configured to measure cardiac signals in response to electrical activity in the heart. The processing unit can be configured to determine physiological information at multiple locations of the boundary of the heart based on the determined positions of the catheter electrodes and the measured cardiac signals.
In some aspects, a system includes a catheter that includes two or more electrodes configured to inject a current and to measure electrical signals. The system also includes a processing unit configured to determine a boundary of at least a portion of the heart Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
based on the measured electrical signals and a display device configured to display a portion of less than the entire boundary of the heart.
The display device can be further configured to display the physiological information at multiple locations of the boundary of the heart. The system can be configured to display the physiological information at multiple locations of the boundary of the heart for only the determined valid area.
The processing unit can be configured to account for a change in conductivity at the cardiac chamber boundary in the determination of the boundary. .
The processing unit can be configured to determine the boundary based at least in part on impedance information generated based on the measured signals at the different catheter positions. The impedance information can be based on a conductivity contrast between blood and surrounding tissue.
The processing unit can be configured to determine the boundary by determining a surface based on the measured signals, the surface representing a surface at which the conductivity changes. The processing unit can be configured to for each of the determined surfaces select one or more regions of the surface corresponding to a boundary of a portion of the heart.
The processing unit can be configured to select the one or more regions based at least in part on a distance between a portion of the surface and the catheter. The processing unit can be configured to select the one or more regions based at least in part on a magnitude of a distortion field, the distortion field being based at least in Attorney Docket No. 20240-0013WOl Client Reference RMD-INVD-8A
part on a difference between a field calculated based on the measurements and a field in a homogonous medium.
The two or more electrodes can be further configured to measure cardiac signals in response to electrical activity in the heart. The processing unit can be configured to determine physiological information at multiple locations of the boundary of the heart based on the determined positions of the catheter electrodes and the measured cardiac signals.
Embodiments of the system may also include devices, software, components, and/or systems to perform any features described above. Embodiments of the methods and systems generally disclosed herein can be applied to determining the position of any object within an organ in a patient's body such as the patient's heart, lungs, brain, or liver.
As used herein, the "position" of an object means information about one or more of the 6 degrees of freedom that completely define the location and orientation of a three- dimensional object in a three-dimensional coordinate system. For example, the position of the object can include: three independent values indicative of the coordinates of a point of the object in a Cartesian coordinate system and three independent values indicative of the angles for the orientation of the object about each of the Cartesian axes; or any subset of such values. As used herein, "heart cavity" means the heart and surrounding tissue.
As used herein, the "anatomical information" means information about an anatomy of an organ, information about a boundary of an organ, and/or information about an anatomy for ablation.
Unless otherwise defined, all technical and scientific terms used herein have the same meaning as commonly understood by one of ordinary skill in the art to which this Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
invention belongs. In case of conflict with documents incorporated herein by reference, the present document controls.
FIG 1 is an exemplary schematic diagram of an arrangement for positioning electrodes in a patient's heart. FIGS. 2a-2c show perspective, end, and side views, respectively, of a deployed catheter with multiple current injection electrodes (CIE) and multiple potential measuring electrodes (PME).
FIGS. 3A and 3B are schematic diagrams of two current injection electrodes (CIE) pair constellations. FIGS. 4A-4E are schematic diagrams of a process for generating anatomical information including a reconstructed anatomy.
FIG 5 is a schematic diagram of a model for determining a parameterized surface.
FIG 6 is a schematic diagram of a process for determining a parameterized surface. FIG 7 is an exemplary graph of a confidence model.
FIG 8 is a schematic diagram of a dipole at a distance from an infinite plane.
FIG 9 is a schematic diagram of a partial boundary determination.
FIG 10 is a schematic diagram of an ellipsoidal local parameterized surface.
FIG 11 is a schematic diagram of a curvilinear local parameterized surface. FIG 12 is a schematic diagram of valid patches relative to a catheter location.
Like reference symbols in the various drawings indicate like elements. Attorney Docket No. 20240-0013WO 1 Client Reference RMD-INVD-8A
Embodiments disclosed herein include a method and system for generating patient specific anatomical information such as cardiac anatomy. The patient specific cardiac anatomy can be used, for example, during the generation of an electro anatomical map (EAM) of heart tissue which can be used to identify the site of origin of an arrhythmia followed by a targeted ablation of the site. While patient specific anatomy is a necessary component in the generation of an EAM for the catheter ablation treatment of arrhythmia, accurate representation of cardiac anatomy is useful for other medical applications such as congestive heart failure, injection of biologies into the heart and scar tissue, anatomical guidance of biopsies, minimally invasive valve repair and replacement, and the like.
Embodiments disclosed herein include methods and systems for generating anatomical information including patient specific anatomy by deriving impedance based information from a tracked multi-electrode array (MEA) catheter. FIG. 1 shows a schematic diagram of an exemplary embodiment of a anatomy generation system 200. The system 200 includes a moveable catheter 210 having multiple spatially distributed electrodes (described below). During the anatomy generation process, a physician or medical professional 202 inserts the catheter 210 into a chamber of interest (e.g., the heart 212) of a patient 214. The catheter 210 is displaced to one or multiple locations within the chamber of interest (e.g., the heart 212). In some embodiments, the distal end of the catheter 210 is fitted with multiple electrodes spread somewhat uniformly over the catheter (e.g., as described in more detail below). The catheter 210 includes both current injection electrodes (CIE) and potential measuring electrodes (PME). In the example of anatomy generation for the heart chamber, due to the conductivity contrast between blood and the myocardium and surrounding tissue, the measurements collected by the catheter 210 can be analyzed to detect chamber boundary in the vicinity of the catheter 210. In order to reconstruct the chamber anatomy, system 200 includes a processing unit 206 which performs operations pertaining to anatomy determination based on a location of the catheter 210 (e.g., a location provided by a tracking system) and the measurements from the electrodes on the catheter 210. As the tracked catheter 210 is moved inside the chamber, a partial or complete representation of the chamber anatomy is constructed. For example, a representation of the boundary of the chamber can be constructed. The Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
chamber anatomy 208 can be displayed on a display device 204 which is connected to the processing unit 206. In addition, embodiments disclosed herein include methods and systems for generating an EAM using the generated anatomy.
As described herein, due to the conductivity contrast between blood and the myocardium and surrounding tissue, measurements collected by catheter 210 located inside the heart 212 of a patient 214 can be analyzed to detect a chamber boundary in the vicinity of the catheter 212. These measurements can be used to generate anatomical information representing the anatomy of the heart. Other exemplary methods for generating an anatomy of the heart (or other organ) include point collection, ultrasound, rotational angiography, computed tomography (CT) and magnetic resonance (MR). In general, point collection is a method in which a tracked catheter is moved inside the heart and numerous catheter locations are collected. Over time the outline of the heart is traced based on the collection of points. Exemplary drawbacks of point collection for generation of an anatomy can include that the process is time consuming and can suffer from limited accuracy and uncertainty. Ultrasound is a method in which a tracked intra cardiac ultrasound/echo catheter is moved inside the heart, the 2D images collected with the catheter are segmented to identify chamber boundary and combined using the 3D tracking data. Exemplary drawbacks of ultrasound for generation of an anatomy can include that the process can be time consuming, segmentation is manual, and a proprietary ultrasound catheter must be introduced into the organ solely for determining anatomy. Rotational Angiography is a method in which a contrast agent is injected into chamber of interest and a fluoroscopy c-arm is rotated to obtain volumetric representation. Segmentation is used to detect endocardial boundary. Exemplary Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
drawbacks of rotational angiography can include that its use can lengthen procedure time, may suffer from limited accuracy, it can be difficult to image all chambers, and/or excessive injection of contrast may be toxic. When CT & MR are used, a volumetric chamber representation is acquired ahead of procedure, volume segmentation provides cardiac chamber of interest, and registration to tracking system is performed during procedure. Exemplary drawbacks of CT and MR can include that the anatomy is generated prior to the procedure which may lead to inaccuracy in anatomy at the time of the procedure, the process often requires use of contrast agent, and/or not all chambers can easily be imaged.
It is believed that generating the anatomy based on measurements collected by a catheter located inside the heart of a patient and calculations based on the conductivity contrast between blood and the myocardium and surrounding tissue (e.g., as described herein) can provide one or more of the following advantages. The generation of the anatomy can be relatively fast, since a large area of the anatomy can be acquired in each heartbeat. In another example, the anatomy information is acquired (and may be re- acquired) during the procedure as opposed to being acquired prior to the procedure, hi another example, these methods can provide the advantage enabling the same catheter to be used for anatomy generation, electrical reconstruction, and/or tracking.
In some embodiments, a patient specific anatomy is generated using impedance based information from a tracked multi-electrode array (MEA) catheter. The system includes an MEA catheter that provides mechanical support for an array of current injecting electrodes (CIE) and potential measuring electrodes (PME). For example, the catheter may be configured with multiple electrodes and used for cardiac mapping, such as described in commonly owned patent application serial no. 11/451,898, entitled "NON-CONTACT CARDIAC MAPPING, INCLUDING MOVING CATHETER AND MULTI-BEAT INTEGRATION" and filed June 13, 2006, application serial no. 11/451,908, entitled "NON-CONTACT CARDIAC MAPPING, INCLUDING PREPROCESSING" and filed June 13, 2006, application serial no. 11/451,871 entitled "NON-CONTACT CARDIAC MAPPING, INCLUDING RESOLUTION MAP" and filed June 13, 2006, and application serial no. 11/672,562 entitled "IMPEDANCE REGISTRATION AND CATHETER TRACKING" and filed February 8, 2007, the Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
contents of each of which are incorporated herein by reference. The system also includes electronics for driving current and measuring potential, a tracking system that provides 3D location of catheter electrodes, and a computer with appropriate software algorithm to collect data and process the data in order to construct a 3D representation of the chamber anatomy.
To generate the excitation pattern necessary for anatomy construction, a sequence of linearly independent current injection patterns are generated by utilizing current injecting electrodes ("CIE") on the catheter. Simultaneously potential measurement is performed on potential measuring electrodes ("PME"), also mounted on the catheter. It should be noted that PME and CIE may be one and the same.
A number (>6) of current injecting electrodes (CIE) are mounted on catheter 110. For example, 3 orthogonal CIE pairs may be mounted on the catheter. Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
The CIE are designated 119, while electrodes 118 are used as potential measuring electrodes (PME). The purpose of the CIEs is to inject current into the heart cavity. For example, each CIE pair can define a source and sink electrode, respectively, for injecting current into the heart cavity. It should be noted that any low impedance electrode can be used for current injection and in a case where many or all electrodes are capable of injecting current the designation of such electrodes as CIE on the catheter only indicates that these electrodes are actually being used for current injection. It should be further appreciated that other configuration sets of CIE are possible as long as these configurations are known and can be accounted for. Examples of such configurations could be quadruples involving 4 CIE, or even a non-symmetrical configuration involving 3 CIE in known positions on the catheter. For simplicity the method of using electrode pairs will be explained, but the same method can be applied using other configurations.
It should be appreciated that configurations other than orthogonal pairs may be used for either method, and that more than 2 CIE may participate in current injection at a given time. FIGS. 3A and 3B show two different CIE pair constellations. FIG. 3B shows the 3 pair constellation described above while FIG. 3 A shows 7 pairs. The 7 pairs are the same 3, plus 4 additional diagonal pairs.
The MEA catheter is connected to electronics hardware capable of driving the necessary current and detecting both signals originating from the heart as well as those used for anatomy construction. There is a need to distinguish between the two signals in order to separate the signal being used for the anatomy determination from the cardiac signal. The CIE are therefore coupled to electronics injecting the current at a frequency higher than cardiac activation (cardiac activation <2kHz, CIE > 4kHz, e.g. 5kHz) such that the two types of signals can be easily distinguished using frequency analysis. It should be noted that other methods for distinguishing between the CIE signal and the cardiac activation signal can be used, such as injecting a spread-spectrum signal having a low energy level in the frequency range of the cardiac activation signal, and detecting this spread-spectrum signal in the signal collected by the all PME. Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
A multitude of separate known configurations of CIE need to inject current in order to interrogate the medium. There is a need to determine the source of the injected signal and to trace it to a specific CIE configuration. For example, 3 pairs of CIE can inject the current sequentially, one pair at a time, so that it is possible to trace the source of the measured PME signals to a specific pair. This is called time division multiplexing. In the case of time division multiplexing, CIE are activated in sequence such that at one point in time one pair is activated and at the next point in time another pair is activated. The switching between pairs may occur every cycle (e.g., l/5kHz=200μs) or every few cycles (e.g., 20 cycles, 20x200μs=4mS). It should be noted that frequency or code division (spread spectrum) multiplexing, rather than time division may be used to separate the signals emanating from different CIE pairs. In the case of frequency multiplexing all CIE pairs may inject the current at the same time, but each pair uses a different carrier frequency. The signal collected at the PME is filtered according to the frequency, and the signal measured in each frequency is then associated with the appropriate originating pair.
The amount of current injected needs to be large enough to provide a reasonable signal to noise ratio, while also small enough not to stimulate cardiac tissue. A current level of 50μA at 5kHz is believed to be appropriate.
In order to combine information collected from multiple catheter locations, the MEA electrode locations are tracked relative to the cardiac chamber in three dimensional space. This task is accomplished by a tracking subsystem. Such subsystems have been shown previously and may include any of a magnetic, impedance, x-ray and ultrasound subsystems. For example, the tracking system can be a tracking system such as described in commonly owned patent application serial no. 12/061,297, entitled "Intracardiac Tracking System" and filed April 2, 2008, the contents of which is incorporated herein by reference. Attorney Docket No. 20240-0013 W01 Client Reference RMD-INVD-8A
FIG. 4 A shows a schematic description of true chamber boundary 250, a single catheter 254 and a local parameterized surface 252 reconstructed from the single catheter position. The local parameterized surface 252 is generated based on measurements collected by the PME on catheter 254. As described above, impedance characterization does not gather information from great distances and therefore the local parameterized Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
surface 254 provides only a portion of the chamber anatomy. More particularly, since a single catheter location cannot detect the entire chamber anatomy, the local parameterized surface 242 is a surface that provides a best fit to PME measurements at a particular catheter location (e.g., as described in more detail below). As shown in FIG. 4B, after the local surface 252 is generated for a particular catheter location, the system detects and marks regions of the local parameterized surface 252 that are valid. Valid regions of the local surface 252 are region(s) that are expected to lie on the true chamber boundary 250. These regions are detected and marked as valid patches (e.g., patches 256a and 256b). As shown in FIG. 4C, the catheter 254 is then moved to another location (e.g., moved from location 258a to location 258b) and the process of construction of local anatomy and detection of valid patches is repeated. As shown in FIG. 4C, the system detects another valid patch 256c based on the information collected by catheter 254 at location 258b. As shown in FIG. 4D, to reconstruct the entire chamber anatomy or portion of interest (e.g., a portion of the chamber anatomy, but not the entire chamber anatomy), the catheter is moved to multiple locations, using the tracking system for position information in each location. At each location, the system detects additional valid patches corresponding to additional region(s) that are expected to lie on the true chamber boundary 250. As shown in FIG. 4E, once the catheter has been moved around the entire chamber or the portion of interest, the chamber boundary 260 is reconstructed by connecting the valid patches, hi some embodiments, the valid patches can be connected using a surface meshing algorithm such as the algorithm described in US patent no. 6,226,542, the contents of which are hereby incorporated by reference.
Cardiac contraction changes the anatomy of the chamber. For that reason, in some embodiments, the anatomy generation is gated according to the cardiac cycle. Gating information can be obtained from electrical measurements of the cardiac cycle (e.g., by the use of surface ECG or intracardiac signal from a stable location) and triggering on a constant marker in the cardiac phase (e.g., using an R-wave detection algorithm, a threshold criterion, a maximum criterion or correlation to a template). Another option is to use a measurement that is affected directly by the mechanical movement of the heart, such as the measurement of the impedance between CIE, and triggering on a constant marker in the cycle. Once a trigger is determined, the cardiac Attorney Docket No 20240-0013 WO 1 Client Reference RMD-INVD-8A
cycle is divided into m slices (e.g., m=10), and the mentioned process is repeated for each slice separately. Alternatively, a particular slice of interest may be chosen (e.g., end diastole) and other data is not collected and/or is discarded. It is also important to note that all CIE sets may be scanned quickly enough (e.g., 4mS) such that it can be assumed that the heart did not move substantially in that period.
Though sinusoidal signals are used to modulate the signal such that it does not interfere with cardiac signal, the frequency is relatively low, and so the validity of the quasi-static approximation via the Laplace's equation remains intact: V - (σ -i-]ωε)Vφ = Q (Eq. 1) Where Ψ describes the electric potential field, *> is the conductivity, ε is the permittivity and &> is the angular frequency.
The anatomical information provides a three dimensional distribution of the electrical conductivity of the surrounding media. The goal of the algorithm is to find the location and orientation of the internal boundary of the heart chamber, that is, to find the boundary where the conductivity value changes from the value associated with blood. For example, on average the conductivity contrast between blood σb and surrounding tissue σs is 'σs ~ . Theoretically, an impedance tomography technique, which is able to provide the distribution of the conductivity surrounding the medium, could be used to detect the anatomical features. However, due to the smoothing effect of current's distribution in material as governed by Laplace's equation, impedance tomography techniques provide blurry representation of the medium and therefore are not ideal for the anatomy reconstruction process. For example the impedance tomography techniques often employ a Tikhonov regularization operator as part of the inverse solution for conductivity. This type of regularization promotes limited change in electrical conductivity over a neighborhood and therefore contributes to the blurriness of the final representation. In general, regularization techniques in inverse problems usually introduce extra equations or terms, which express a priory expectation about the solution and at the same time mathematically convert the underlying equation system from underdetermined to over determined, thereby making it solvable. In other words, Attorney Docket No. 20240-0013WOl Client Reference RMD-1NVD-8A
regularization techniques are needed in inverse problems to handle the overly high number of unknown degrees of freedom, which is used to represent the conductivity distribution.
Instead of using an overly high number of degrees of freedom, in some embodiments the abrupt conductivity distribution is represented in a way that uses only a few parameters. For example, as shown in FIG. 5, this can be achieved by an explicit representation of a closed and parameterized local surface 276 around the catheter 278. This representation divides the 3D space into two regions 274 and 272. The region 274 in which the catheter 278 is located is associated with the conductivity of blood (Ωbiood) while the outside region 272 is associated with an unknown conductivity (Ωextemai) since it varies from patient to patient and between different regions in the cardiac chamber. This local parameterized surface 276 and the conductivity values (Ωbiood and Ωextemai) constitute a local forward model used in the inverse solver. This model aims to reconstruct the shape and value of the conductivity distribution. This inverse problem is nonlinear and requires the use of an iterative solver. The inverse solver is an optimizer that determines both the surface's parameters as well as the unknown external conductivity (Ω external)- The choice of the local parameterized surface 276 is only limited by its number of parameters or degrees of freedom. For example, a fully parameterized 3D ellipsoid introduces nine degrees of freedom: three axial parameters and six parameters for rigid body translation and rotation. It is also possible to use polynomial representation, Bezier, NURBS or curvilinear finite elements to represent the parameterized surface 276.
Attorney Docket No. 20240-0013WO 1 Client Reference RMD-rNVD-8A
Where H is the vector of measured electrode potentials 286, Z 1 . is the vector of potential samples predicted by the forward model 296 and dependent on 2, which is the vector of various parameters of the forward model 288 itself. During the measurement phase it is assumed that neither the location nor the orientation of the catheter is changing significantly, therefore the local forward model 288 assumes steady position and orientation of the catheter 280. Both the measured potentials 286 and predicted potential values 296 can be arranged in structured vectors and so the measure of difference can be defined by the sum of squares of corresponding differences. Minimization problems of this type can be efficiently solved by the Levenberg-Marquardt (LM) method (e.g., using LM optimizer 294). The LM algorithm is formulated in terms of the residual vector and the first derivative of this vector, which is called the Jacobian. During the search for the optimal parameter values, the forward solver 292 repeatedly solves for the ever changing local forward model 288. The solution of the forward model 288 is governed by the Laplace's equation and provides the potential distribution for the domain defined by the local forward model 288, and so the potential values at the electrodes of the model- Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
catheter as well. Different approximation methods such as Spherical Harmonics, Finite Elements, Boundary Elements or Multiple MultiPoles can be used to discretize Laplace's equation, resulting in an algebraic linear equation system in Equation 2.
Kx — f (Equation 2) The solution of the linear equation system provides the values for the so called degree of freedoms (χ ), which in turn along with the corresponding field approximation functions constitute the overall approximation of the electrical field. K is the so called stiffness matrix collecting the contribution from the discretized differential operator, / represents excitation due to sources or boundary conditions. The LM algorithm requires the derivatives of the residual vector in terms of the parameters of the local forward model. There are two possible approaches to obtaining the derivatives. They can be estimated using finite differentiation techniques, or using a direct differentiation method in the forward solver, hi testing the direct approach was shown to be superior in terms of speed and accuracy due to the larger approximation error of the finite differentiation technique.
After successful optimization of the local parameterized surface 276, the local parameterized surface 276 is expected to be similar or partially similar to a portion of the true chamber boundary (e.g., as shown above in FIGS. 4A-4E). The region of the local parameterized surface that approximates the chamber boundary with sufficient accuracy is described herein as a valid patch. The rest of the local parameterized surface bears no significant influence on the electric field-pattern predicted by the local forward model.
Such surface regions may be present in a relatively distant location from the catheter because, as explained earlier, the injected current pattern will not diffuse sufficiently into those far regions and so the field pattern will not be influenced by features located in those regions.
The parts of the local parameterized surface that are less likely to approximate the true chamber boundary are filtered out so only the valid patches remain that are most responsible for the field pattern measured by the electrodes and reconstructed by the Attorney Docket No. 20240-0013WO 1 Client Reference RMD-INVD-8A
optimized local forward model. The final boundary of the chamber is reconstructed by forming a surface (e.g., a triangular-mesh based surface) so that it fits the set of previously obtained valid patches while also maintaining reasonable surface-smoothness properties.
The selection of the valid patches can be a mostly a heuristic procedure. In some embodiments, the local parameterized surface is subdivided and only certain regions are kept for the final reconstruction step as valid patches. The main criterion for selecting the valid patches is based on the strength of a distortion field. The distortion field is the difference between the field in presence of inhomogeneity and the field which corresponds to the homogenous medium. The higher the strength of the distortion field on a certain location of the local parameterized surface, the more the field pattern resulting on the PME is sensitive to the location and orientation of that surface location. Therefore, the distortion field strength is believed to be a good indicator of the surface location's validity (strength of influence). The main criterion is then simply to select patches that exhibit higher distortion field than a certain threshold (dftkmskoid ). This criterion is then combined with additional side-criteria in order to increase the certainty of the decision. In some embodiments, two side-criteria are used: one criteria is based on the distance of the surface location from the center of the catheter, and the other criteria is based on the final residual error of the LM optimization. It should be noted that additional quantities may also contribute to the accuracy of the filtering process. The side criteria are formulated and combined via the fuzzy logic approach.
FIG. 7 shows an exemplary graph 300 of "Membership functions" of "confidence" in which the x-axis 304 represents the distance or residual while the y-axis 302 represents the confidence that can be used which can be represented by:
C(x. t. a} = -r— T 1 + ea<x~ύ
Where t and oc are parameters of the membership function; f is the turn-point 306 and # determines the slope of the membership function at the turnpoint 306. Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
After proper normalization of the quantities involved in the side criteria, the evaluation of the membership functions result in confidence factors for each criterion (Creszdvai and C d-.srance). These factors are then used as multipliers of the distortion field strength (<*/ ) in the final acceptance criterion for valid patch selection: df x Crβsιduaι x Cd,sta nce> dftitrgskoιd (Equation 3)
The LM residual is expected to be smaller when the optimization is more successful and therefore the local parameterized surface is expected to approximate the chamber boundary better. For this reason, smaller values for the LM residual should produce higher confidence levels: one or close to one. Increasing residual, on the other hand, should eventually switch the corresponding confidence level down to zero. In order to establish what "small" is, the LM residual is normalized. This normalization is such that when the local forward model is the homogenous case, the corresponding normalized residual would be exactly one. In other words, the residual measures how much the distortion field is reconstructed by the optimized local forward model at the PME compared to the homogenous case, which is regarded as complete lack of reconstruction. The normalization described here is only the first step to make sure the membership function produces meaningful confidence levels. The missing additive is the parameter of the membership function itself, called turn-point. In some examples, for the LM residual a turn-point 306 of 0.05 is believed to be adequate. This is the normalized LM residual value, which produces confidence level of exactly one half. The appropriate a for this criteria is believed to be 80.
As shown in FIG. 8, similar deductions can be given for the confidence level generated by the distance of the surface location relative to the catheter. For this criterion confidence should decrease as distance from the center of the catheter is increased. The surface location distance is normalized by expressing it relative to the radius of the Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
catheter. For example, a reasonable turn-point is believed to be two times the catheter radius. For example, an appropriate a for this criteria is believed to be 0.5.
For the subdivision of the local parameterized surface a surface triangulation is used as a mesh. The mesh is as uniform as possible, and the average size of the triangles should match the desired resolution for the filtering step. In some embodiments, an average edge length of 2mm was found to be sufficient in balancing run-time with accuracy. Other subdivision of the local parameterized surface could also be used.
Electro-Anatomical Map ("EAM") Construction
The chamber boundary reconstructed using impedance measurements disclosed herein can be used as the surface onto which electrical information is projected. This electrical information may be collected using a contact scheme or non-contact scheme (e.g., as described in US patent 7,505,810, the contents of which are hereby incorporated Attorney Docket No. 20240-0013WO1 Client Reference RMD-INVD-8A
by reference). In the case of a non-contact scheme, both electrical and anatomical data may be collected simultaneously by the MEA catheter thus expediting the EAM generation process. Electrical information displayed on the EAM can include any of a number of isopotentials, bipolar maps, local activation time, voltage map, dominant frequency map, and the like.
In some applications, it is necessary to construct only a partial EAM of the chamber (e.g., construct an EAM of less than the entire chamber). That is, to save procedure time, only a portion of the chamber known to participate in the arrhythmogenic mechanism needs to be provided on the EAM. For example, in the case of scar related ventricular tachycardia, only the scarred area and its immediate surrounding tissue may be required for clinical treatment. Such portion can represent under 25% of total chamber area and be collected with a limited number of catheter locations. For example, less than 10 catheter locations may be used to generate the partial EAM (e.g., 8 catheter locations or less, 6 catheter locations or less, 5 catheter locations or less, 4 catheter locations). In such case, as shown in FIG. 9, it is possible to construct a partial anatomy 336 by meshing a closed surface around the valid patches (e.g., patches 332a, 332b, and 332c). Since the surface 336 is not complete in this case, such closed mesh also contains areas 334 where no valid patches exist nearby. However, those are known from the valid patch selection process described above and marked invalid. Invalid areas 334 in the mesh can be either transparent or rendered differently (e.g. show only mesh edges and render mesh faces transparent, display gray, make semi-transparent).
Once the partial chamber anatomy is constructed, electrical information can then be displayed only on the valid areas (e.g., area 336) of the anatomy using either a non- contact or contact scheme. It is important to note that the added information of valid and invalid zones is crucial to the construction of a partial EAM map. If invalid areas of the anatomy are not marked as such, the EAM may lead to wrong clinical interpretation. The information regarding the validity of the map provided with the impedance scheme is unavailable with other point collection schemes making partial maps more difficult to interpret. Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
FIG. 12 shows valid patches detected relative to a catheter location. FIGS. 13A-D, provides multiple projections of an experiment reconstructing left ventricular ("LV") anatomy. FIG. 13A provides a three dimensional representation of the reconstructed anatomy of the LV, FIG. 13B provides a two dimensional representation of the reconstructed anatomy of FIG. 13A in the x-y plane, FIG. 13C provides a two dimensional representation of the reconstructed anatomy of FIG. 13A in the x-z plane, and FIG. 13D provides a two dimensional representation of the reconstructed anatomy of FIG. 13 A in the y-z plane. The construction of the anatomy was accomplished using a MEA catheter with 6 CIE and 64 PME in a constellation similar to the one shown in FIG. 3. The catheter was moved to 51 locations within the left ventrical (LV). The synthetic measurements were generated using a Finite Element Method simulator. As shown in the figures, even areas that were sparsely sampled by the catheter are reasonably reconstructed.
In some aspects, the catheter used to generate the anatomical information and/or generate the EAM information can additionally include an electrode for delivering ablation energy for ablating tissue. As such, a single catheter can generate an EAM map (including generating the anatomical information used for the EAM map) and perform ablation of identified regions of the organ. It is believed that this can provide the advantage of limiting the number of catheters inserted into the organ of the patient. For example, an ablation procedure can involve mapping of electrical activity in the heart (e.g., based on cardiac signals), such as at various locations on the endocardium surface ("cardiac mapping"), to identify the site of origin of the arrhythmia followed by a targeted ablation of the site. A single catheter inserted into the patient's heart chamber Attorney Docket No. 20240-0013WO 1 Client Reference RMD-INVD-8A
can be used both to perform such cardiac mapping (including generating the anatomy of the heart) and to perform the ablation.
The methods and systems described herein are not limited to a particular hardware or software configuration, and may find applicability in many computing or processing environments. The methods and systems can be implemented in hardware, or a combination of hardware and software, and/or can be implemented from commercially available modules applications and devices. Where the implementation of the systems and methods described herein is at least partly based on use of microprocessors, the methods and systems can be implemented in one or more computer programs, where a computer program can be understood to include one or more processor executable instructions. The computer program(s) can execute on one or more programmable processors, and can be stored on one or more storage medium readable by the processor (including volatile and non- volatile memory and/or storage elements), one or more input devices, and/or one or more output devices. The processor thus can access one or more input devices to obtain input data, and can access one or more output devices to communicate output data. The input and/or output devices can include one or more of Attorney Docket No. 20240-0013WOl Client Reference RMD-INVD-8A
the following: Random Access Memory (RAM), Redundant Array of Independent Disks (RAID), floppy drive, CD, DVD, magnetic disk, internal hard drive, external hard drive, memory stick, or other storage device capable of being accessed by a processor as provided herein, where such aforementioned examples are not exhaustive, and are for illustration and not limitation.
References to "a microprocessor" and "a processor", or "the microprocessor" and "the processor," can be understood to include one or more microprocessors that can communicate in a stand-alone and/or a distributed environment(s), and can thus be configured to communicate via wired or wireless communications with other processors, where such one or more processor can be configured to operate on one or more processor-controlled devices that can be similar or different devices. Furthermore, references to memory, unless otherwise specified, can include one or more processor- readable and accessible memory elements and/or components that can be internal to the processor-controlled device, external to the processor-controlled device, and can be accessed via a wired or wireless network using a variety of communications protocols, and unless otherwise specified, can be arranged to include a combination of external and internal memory devices, where such memory can be contiguous and/or partitioned based on the application. Accordingly, references to a database can be understood to include one or more memory associations, where such references can include commercially available database products (e.g., SQL, Informix, Oracle) and also proprietary databases, Attorney Docket No. 20240-0013WO1 Client Reference RMD-INVD-8A
and may also include other structures for associating memory such as links, queues, graphs, trees, with such structures provided for illustration and not limitation.
Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8AWHAT IS CLAIMED IS:
1. A method comprising: inserting a catheter into a heart, the catheter comprising three or more electrodes; moving the catheter to each of multiple, different positions in the heart; for each of the different catheter positions, causing current to flow between at least some of the electrodes and in response to current flow, measuring an electrical signal at each of one or more of the electrodes; and determining anatomical information about the heart based on positions of the catheter electrodes and the measured signals at the different catheter positions.
6. The method of claim 4, wherein the impedance information is based on a permittivity contrast between blood and surrounding tissue. Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
15. The method of claim 10, wherein determining the anatomical information further comprises, for each of the determined surfaces selecting one or more regions of the surface corresponding to a boundary of a portion of the heart. Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
21. The method of claim 1 , further comprising using the multiple electrodes on the catheter to measure cardiac signals at the catheter electrodes in response to electrical activity in the heart.
22. The method of claim 21 , further comprising determining physiological information at multiple locations of the boundary of the heart based on the determined positions of the catheter electrodes and the measured cardiac signals at the different catheter positions.
23. The method of claim 1 , further comprising using one or more electrodes on the catheter for delivering ablation energy for ablating tissue. Attorney Docket No. 20240-0013WO 1 Client Reference RMD-INVD-8A
27. A method comprising: inserting a catheter into a heart, the catheter comprising multiple, spatially distributed electrodes including multiple sets of electrodes each set comprising at least two electrodes; for each of the multiple different sets of electrodes, causing current to flow between at least some of the electrodes and in response to current flow, measuring an electrical signal at each of one or more measuring electrodes; and determining anatomical information based on the measured signals.
30. The method of claim 29, wherein determining the anatomical information comprises determining the anatomical information based at least in part on impedance information generated based on the measured signals. Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
40. The method of claim 36, wherein the surface provides a boundary between a region represented by a first conductivity inside the surface and a region represented by a Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
second conductivity outside the surface, the first conductivity being different from the second conductivity.
47. The method of claim 27, further comprising using the multiple electrodes on the catheter to measure cardiac signals at the catheter electrodes in response to electrical activity in the heart. Attorney Docket No. 20240-0013WOl Client Reference RMD-INVD-8A
52. A system comprising: a catheter comprising one or more electrodes configured to inject a current and to measure electrical signals in response to the injected current; a system configured to determine a position of the catheter electrodes; and a processing unit configured to determine anatomical information about the heart based on positions of the catheter electrodes and measured electrical signals at different catheter positions.
54. The system of claim 53, wherein the processing unit is configured to account for a first conductivity inside the cardiac chamber boundary and a second conductivity outside the cardiac chamber boundary. Attorney Docket No 20240-0013WO1 Client Reference RMD-INVD-8A
63. The system of claim 62, wherein the processing unit is configured to select the one or more regions based at least in part on a magnitude of a distortion field, the Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
distortion field being based at least in part on a difference between a field calculated based on the measurements and a field in a homogonous medium.
68. A system comprising: a catheter comprising multiple, spatially distributed electrodes including multiple sets of electrodes each set comprising at least two electrodes, the electrodes being configured to inject a current and to measure electrical signals in response to the injected current; and a processing unit configured to determine anatomical information about the heart based on the measured signals.
69. The system of claim 68, wherein the processing unit is configured to account for a change in conductivity at the cardiac chamber boundary in the determination of the anatomical information. Attorney Docket No. 20240-0013WOl Client Reference RMD-INVD-8A
77. The system of claim 76, wherein the processing unit is configured to select the one or more regions based at least in part on a magnitude of a distortion field, the distortion field being based at least in part on a difference between a field calculated based on the measurements and a field in a homogonous medium. Attorney Docket No. 20240-0013WOl Client Reference RMD-INVD-8A
82. A method comprising: inserting a catheter into a heart, the catheter comprising three or more electrodes; causing current to flow between at least some of the electrodes and in response to current flow, measuring an electrical signal at each of one or more of the electrodes; determining a boundary of at least a portion of the heart based on the measured electrical signals; and displaying a portion of less than the entire boundary of the heart.
83. The method of claim 82, further comprising: measuring cardiac signals at the catheter electrodes in response to electrical activity in the heart; and determining physiological information at multiple locations of the boundary of the heart based the measured cardiac signals. Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
84. The method of claim 83, further comprising displaying the physiological information at multiple locations of the boundary of the heart.
85. The method of claim 83, further comprising displaying the physiological information at multiple locations of the boundary of the heart for only a determined valid area.
86. The method of claim 82, wherein displaying the boundary comprises displaying the boundary for a portion of less than all of the heart.
87. The method of claim 82, wherein the determination of the boundary accounts for a change in conductivity at the cardiac chamber boundary.
88. The method of claim 87, wherein the determination accounts for a first conductivity inside the cardiac chamber boundary and a second conductivity outside the cardiac chamber boundary.
89. The method of claim 82, wherein determining the boundary comprises determining the anatomical information based at least in part on impedance information generated based on the measured signals.
90. The method of claim 89, wherein the impedance information is based on a conductivity contrast between blood and surrounding tissue.
91. The method of claim 89, wherein the impedance information is based on a permittivity contrast between blood and surrounding tissue.
92. The method of claim 89, wherein the impedance information comprises complex impedance information based on both a permittivity contrast and a conductivity contrast between blood and surrounding tissue. Attorney Docket No. 20240-0013WOl Client Reference RMD-INVD-8A
93. The method of claim 89, wherein determining the boundary comprises, determining a surface based on the measured signals, the surface representing a surface at which the conductivity value changes.
94. The method of claim 93, wherein the surface comprises a closed and parameterized surface around at least a portion of the catheter.
95. The method of claim 93, wherein the surface provides a boundary between a region represented by a first conductivity inside the surface and a region represented by a second conductivity outside the surface, the first conductivity being different from the second conductivity.
96. The method of claim 93, wherein determining the anatomical information further comprises, for each of the determined surfaces selecting one or more regions of the surface corresponding to an expected boundary of the heart.
97. The method of claim 93, wherein selecting the one or more regions comprises selecting the one or more regions based at least in part on a distance between a portion of the surface and the catheter.
98. The method of claim 97, wherein selecting the one or more regions comprises selecting the one or more regions based at least in part on a magnitude of a distortion field, the distortion field being based at least in part on a difference between a field calculated based on the measurements and a field in a homogonous medium.
99. The method of claim 98, wherein selecting the one or more regions comprises selecting the one or more regions based at least in part on an error calculation in an optimization used to generate the surface. Attorney Docket No. 20240-0013WOl Client Reference RMD-INVD-8A
100. The method of claim 98, wherein determining the anatomical information further comprises, joining the regions of the determined surfaces corresponding to an expected chamber boundary to generate the anatomical information.
101. The method of claim 98, further comprising: determining a valid area of the boundary; and using a visual indicia to indicate the determined valid area of the boundary.
102. The method of claim 89, wherein displaying the portion of less than the entire boundary of the heart comprises using a visual indicia to indicate a valid area of the boundary.
103. The method of claim 89, further comprising determining a valid area of the boundary.
104. The method of claim 103, wherein displaying the portion of less than the entire boundary of the heart comprises using a visual indicia to indicate the determined valid area of the boundary.
105. A system comprising: a catheter comprising two or more electrodes configured to inject a current and to measure electrical signals in response to the injected current; and a processing unit configured to determine a boundary of at least a portion of the heart based on the measured electrical signals; and a display device configured to display a portion of less than the entire boundary of the heart.
106. The system of claim 105, wherein: the two or more electrodes are further configured to measure cardiac signals in response to electrical activity in the heart; and Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
the processing unit is further configured to determine physiological information at multiple locations of the boundary of the heart based the measured cardiac signals.
107. The system of claim 106, wherein the display device is further configured to display the physiological information at multiple locations of the boundary of the heart.
108. The system of claim 106, wherein the system is configured to display the physiological information at multiple locations of the boundary of the heart for only the determined valid area.
109. The system of claim 105, wherein the processing unit is configured to account for a change in conductivity at the cardiac chamber boundary in the determination of the boundary.
110. The system of claim 109, wherein the processing unit is configured to account for a first conductivity inside the cardiac chamber boundary and a second conductivity outside the cardiac chamber boundary.
111. The system of claim 105, wherein the processing unit is configured to determine the boundary based at least in part on impedance information generated based on the measured signals at the different catheter positions.
112. The system of claim 111, wherein the impedance information is based on a conductivity contrast between blood and surrounding tissue.
113. The system of claim 111, wherein the impedance information is based on a permittivity contrast between blood and surrounding tissue.
1 14. The system of claim 1 11, wherein the impedance information comprises complex impedance information based on both a permittivity contrast and a conductivity contrast between blood and surrounding tissue. Attorney Docket No. 20240-0013WOl Client Reference RMD-INVD-8A
115. The system of claim 108, wherein the processing unit is configured to determine the boundary by determining a surface based on the measured signals, the surface representing a surface at which the conductivity changes.
116. The system of claim 115, wherein the processing unit is configured to for each of the determined surfaces select one or more regions of the surface corresponding to a boundary of a portion of the heart.
117. The system of claim 115, wherein the processing unit is configured to select the one or more regions based at least in part on a distance between a portion of the surface and the catheter.
118. The system of claim 117, wherein the processing unit is configured to select the one or more regions based at least in part on a magnitude of a distortion field, the distortion field being based at least in part on a difference between a field calculated based on the measurements and a field in a homogonous medium.
119. The system of claim 118, wherein the processing unit is configured to select the one or more regions based at least in part on an error calculation in an optimization used to generate the surface.
120. The system of claim 118, wherein the processing unit is configured to join the regions of the determined surfaces corresponding to an expected chamber boundary to generate the anatomical information.
121. The system of claim 108, wherein the two or more electrodes are further configured to measure cardiac signals in response to electrical activity in the heart.
122. The system of claim 121, wherein the processing unit is configured to determine physiological information at multiple locations of the boundary of the heart Attorney Docket No. 20240-0013 WO 1 Client Reference RMD-INVD-8A
based on the determined positions of the catheter electrodes and the measured cardiac signals.
PCT/US2010/027436 2009-05-08 2010-03-16 Impedance based anatomy generation WO2010129095A2 (en)
US12/437,812 2009-05-08
US12/437,794 2009-05-08
US12/437,794 US8103338B2 (en) 2009-05-08 2009-05-08 Impedance based anatomy generation
WO2010129095A2 true WO2010129095A2 (en) 2010-11-11
WO2010129095A3 WO2010129095A3 (en) 2011-01-13
ID=43050686
EP (1) EP2427106B1 (en)
WO (1) WO2010129095A2 (en)
WO2019035023A1 (en) 2017-08-17 2019-02-21 Navix International Limited Field gradient-based remote imaging
2010-03-16 WO PCT/US2010/027436 patent/WO2010129095A2/en active Application Filing
2010-03-16 EP EP10772414.8A patent/EP2427106B1/en active Active
See references of EP2427106A4 *
CN104144641A (en) * 2011-12-23 2014-11-12 心脏起搏器股份公司 Physiological status indicator apparatus and method
EP2427106A2 (en) 2012-03-14
WO2010129095A3 (en) 2011-01-13
EP2427106A4 (en) 2013-06-05
EP2427106B1 (en) 2017-04-26
Ref document number: 10772414
2011-12-07 REEP
Ref document number: 2010772414