Source: https://www.law.cornell.edu/cfr/text/21/870.5210
Timestamp: 2019-06-19 20:05:06
Document Index: 635657214

Matched Legal Cases: ['§ 870', '§ 870', '§ 870', 'art 807', '§ 870', 'art 807', '§ 870']

21 CFR § 870.5210 - Cardiopulmonary resuscitation (CPR) aid. | CFR | US Law | LII / Legal Information Institute
Subpart F. Cardiovascular Therapeutic Devices
Section 870.5210. Cardiopulmonary resuscitation (CPR) aid.
21 CFR § 870.5210 - Cardiopulmonary resuscitation (CPR) aid.
§ 870.5210 Cardiopulmonary resuscitation (CPR) aid.
(a)CPR aid without feedback -
(1)Identification. A CPR aid without feedback is a device that performs a simple function such as proper hand placement and/or simple prompting for rate and/or timing of compressions/breathing for the professionally trained rescuer, but offers no feedback related to the quality of the CPR being provided. These devices are intended for use by persons professionally trained in CPR to assure proper use and the delivery of optimal CPR to the victim.
(2)Classification. Class I (general controls). The device is exempt from the premarket notification procedures in subpart E of part 807 of this chapter subject to the limitations in § 870.9.
(b)CPR aid with feedback -
(1)Identification. A CPR Aid device with feedback is a device that provides real-time feedback to the rescuer regarding the quality of CPR being delivered to the victim, and provides either audio and/or visual information to encourage the rescuer to continue the consistent application of effective manual CPR in accordance with current accepted CPR guidelines (to include, but not be limited to, parameters such as compression rate, compression depth, ventilation, recoil, instruction for one or multiple rescuers, etc.). These devices may also perform a coaching function to aid rescuers in the sequence of steps necessary to perform effective CPR on a victim.
(2)Classification. Class II (special controls). The special controls for this device are:
(3)Premarket notification. The CPR Aid with feedback device is exempt from the premarket notification procedures in subpart E of part 807 of this chapter if it does not contain software (e.g., is mechanical or electro-mechanical) and is in compliance with the special controls under paragraph (b)(2) of this section, subject to the limitations of exemptions in § 870.9.