Source: http://www.google.com/patents/US20110092804?dq=patent:+7360079
Timestamp: 2016-09-27 22:39:25
Document Index: 341429030

Matched Legal Cases: ['Application No. 60', 'Application No. 60', 'Application No. 60', 'Application No. 60', 'Application No. 60', 'Application No. 60', 'art 502']

Patent US20110092804 - Patient-Specific Pre-Operative Planning - Google PatentsSearch Images Maps Play YouTube News Gmail Drive More »Sign inPatentsA method for pre-operative orthopedic planning includes obtaining only a high-resolution knee-joint scan of a patient, determining hip rotation center and ankle rotation center from anthropometric data based on personal data of the patient, and determining a mechanical axis of the knee joint based on...http://www.google.com/patents/US20110092804?utm_source=gb-gplus-sharePatent US20110092804 - Patient-Specific Pre-Operative PlanningAdvanced Patent SearchTry the new Google Patents, with machine-classified Google Scholar results, and Japanese and South Korean patents.Publication numberUS20110092804 A1Publication typeApplicationApplication numberUS 12/973,214Publication dateApr 21, 2011Filing dateDec 20, 2010Priority dateFeb 27, 2006Also published asUS9345548, US20160213491Publication number12973214, 973214, US 2011/0092804 A1, US 2011/092804 A1, US 20110092804 A1, US 20110092804A1, US 2011092804 A1, US 2011092804A1, US-A1-20110092804, US-A1-2011092804, US2011/0092804A1, US2011/092804A1, US20110092804 A1, US20110092804A1, US2011092804 A1, US2011092804A1InventorsRyan J. Schoenefeld, Brian A. Uthgenannt, Bryan Morrison, Robert MetzgerOriginal AssigneeBiomet Manufacturing Corp.Export CitationBiBTeX, EndNote, RefManPatent Citations (137), Non-Patent Citations (1), Referenced by (138), Classifications (22), Legal Events (1) External Links: USPTO, USPTO Assignment, EspacenetPatient-Specific Pre-Operative Planning
[0001] This application is a continuation-in-part of U.S. application Ser. No. 12/955,361 filed Nov. 29, 2010, which is a continuation-in-part of U.S. application Ser. Nos. 12/938,905 and 12/938,913, both filed Nov. 3, 2010, and which are continuation-in-part of U.S. application Ser. No. 12/893,306 filed Sep. 29, 2010, which is a continuation-in-part of U.S. application Ser. No. 12/888,005, filed Sep. 22, 2010, which is a continuation-in-part of U.S. application Ser. No. 12/714,023, filed Feb. 26, 2010, which is a continuation-in-part of U.S. application Ser. No. 12/571,969, filed Oct. 1, 2009, which is a continuation-in-part of U.S. application Ser. No. 12/486,992, filed Jun. 18, 2009, and is a continuation-in-part of U.S. application Ser. No. 12/389,901, filed Feb. 20, 2009, which is a continuation-in-part of U.S. application Ser. No. 12/211,407, filed Sep. 16, 2008, which is a continuation-in-part of U.S. application Ser. No. 12/039,849, filed Feb. 29, 2008, which: (1) claims the benefit of U.S. Provisional Application No. 60/953,620, filed on Aug. 2, 2007, U.S. Provisional Application No. 60/947,813, filed on Jul. 3, 2007, U.S. Provisional Application No. 60/911,297, filed on Apr. 12, 2007, and U.S. Provisional Application No. 60/892,349, filed on Mar. 1, 2007; (2) is a continuation-in-part U.S. application Ser. No. 11/756,057, filed on May 31, 2007, which claims the benefit of U.S. Provisional Application No. 60/812,694, filed on Jun. 9, 2006; (3) is a continuation-in-part of U.S. application Ser. No. 11/971,390, filed on Jan. 9, 2008, which is a continuation-in-part of U.S. application Ser. No. 11/363,548, filed on Feb. 27, 2006; and (4) is a continuation-in-part of U.S. application Ser. No. 12/025,414, filed on Feb. 4, 2008, which claims the benefit of U.S. Provisional Application No. 60/953,637, filed on Aug. 2, 2007.
[0005] This application is also a continuation-in-part of U.S. application Ser. No. 12/255,945, filed on Oct. 22, 2008.
[0007] The present teachings provide various methods of pre-operative planning for orthopedic procedures customized for particular patients.
[0008] The present teachings provide a method for pre-operative orthopedic planning includes obtaining only a high-resolution knee-joint scan of a patient, determining hip rotation center and ankle rotation center from anthropometric data based on personal data of the patient, and determining a mechanical axis of the knee joint based on the anthropometric data. The method also includes preparing at least a two-dimensional image model of the knee joint using the knee-joint scan and the determined mechanical axis, and preparing a pre-operative surgical plan based on the image model of the knee joint.
[0009] The present teachings provide a method for pre-operative orthopedic planning that includes obtaining only a high-resolution knee-joint scan of a patient, taking a digital photographic image of the patient, and determining a hip rotation center and an ankle rotation center from the digital photographic image. The method also includes determining a mechanical axis of the knee joint from the knee-joint scan and the hip and ankle rotation centers, preparing a at least a two-dimensional image model of the knee joint using the knee-joint scan and the determined mechanical axis, and preparing a pre-operative surgical plan based on the image model of the knee joint.
[0010] The present teachings provide a method for pre-operative orthopedic planning that includes creating a diseased bone database, obtaining personal data of a patient, and selecting a best-fit bone model from the diseased bone database based on the personal data of the patient. The method also includes transforming the best fit model to a customized model matching the personal data of the patient, and preparing a pre-operative surgical plan based on the customized model without obtaining image data of the patient.
[0014] FIG. 1A is a flowchart of an implant manufacturing method according to the present teachings;
[0015] FIG. 2 is a diagram illustrating a computer interface for an implant manufacturing method according to the present teachings;
[0016] FIG. 3 is perspective view of a generic casting of an implant according to the present teachings;
[0017] FIG. 4 is a posterior view of a generic casting according to the present teachings;
[0018] FIG. 4A is a posterior view of a generic casting including a plurality of additional modifiable features according to the present teachings;
[0019] FIG. 5 is a plan view of a generic casting according to the present teachings;
[0020] FIG. 5A is a plan view of a generic casting including a plurality of additional modifiable features according to the present teachings;
[0021] FIG. 5B is a distal view of a three-dimensional image of the patient's distal femur showing planned resection planes according to the present teachings;
[0022] FIG. 5C is a anterior view of a three-dimensional image of the patient's distal femur showing planned resection planes according to the present teachings;
[0023] FIG. 6 is a flow chart for an osteophyte/protrusion removal control method according to the present teachings;
[0024] FIG. 7 is a representative image of a patent's anatomy showing osteophyte/protrusion control tools for modifying the image;
[0025] FIGS. 8 and 9 are representative images of a patent's anatomy showing exemplary osteophyte/protrusion locations;
[0026] FIG. 10 is a representative image of a patent's anatomy showing representative depth control selections for surgeon manipulation;
[0027] FIGS. 11 and 12 are representative images of a patent's anatomy after osteophyte/protrusion removal with exemplary implants attached thereon;
[0028] FIGS. 13A and 13B illustrate a flowchart of a method of implant and guide design;
[0029] FIG. 14 is a schematic view of hardware and a user;
[0030] FIG. 15 is a plan view of a display of a device;
[0031] FIG. 16 is an exemplary illustration of a patient in preparation for a knee joint arthroplasty according to a method of the present teachings;
[0032] FIG. 17 is an anterior skeletal view of an exemplary leg of a patient illustrating anatomic and mechanical axes;
[0033] FIG. 18 is a flowchart of a method associated with FIG. 16 according to the present teachings;
[0034] FIG. 19 is an exemplary illustration of a patient in preparation for a knee joint arthroplasty according to a method of the present teachings;
[0035] FIG. 20 is a flowchart of a method associated with FIG. 19 according to the present teachings;
[0036] FIG. 21 is a flowchart of a method according to the present teachings.
[0037] FIG. 22A is an exemplary illustration of a best-fit diseased bone model associated with block 1174 of FIG. 21; and
[0038] FIG. 22B is an exemplary illustration of a transforming the best fit model of FIG. 21A to a patient-specific bone model associated with block 1176 of FIG. 21.
[0039] The following description is merely exemplary in nature and is in no way intended to limit the present teachings, applications, or uses. For example, although some of the present teachings are illustrated for a knee implant, the present teachings can be used for any orthopedic implant.
[0040] The present teachings provide a manufacturing method that integrates patient's anatomic and medical information with interactive participation by a surgeon to select and manufacture an implant and, optionally, related surgical instruments, for a particular patient from generally three options: a custom made implant specific to the patient, an implant that is only partially custom-made or a semi-custom implant, and a standard off-the shelf implant. Similarly, off-the-shelf, custom-made, or semi-custom-made instrumentation (e.g. alignment guides, drill guides, cutting guides or other instruments) can be selected and manufactured, as recommended by the surgeon, for the surgical procedure. All the implant components, alignment guides, and other disposable instruments can be included in a package provided to a surgeon for a specific patient.
[0041] Referring to FIG. 1, an exemplary flowchart of an interactive implant manufacturing method according to the present teachings is illustrated. The portion of the patient's anatomy related to the orthopedic procedure and the implant is characterized and detailed at 100. The characterization can be performed with various imaging methods capable of obtaining a representation of the affected anatomy, including, for example, soft and hard tissues. The tissues can include bone, bone joints with or without cartilage, ligaments, or other soft tissue. The imaging methods can include, for example, MRI, CT, ultrasound, radiography or X-ray, cameras and other devices. Newer methods can also be used, including, for example, T-ray computed tomography and T-ray diffraction tomography. T-ray is a pulsed terahertz (THz) radiation that can be used to image three-dimensional (3D) structures in the far-infrared region. The THz-wave computed tomography system provides sectional images of objects similar to conventional CT techniques such as x-ray, but without the harmful effects of ionizing radiation. See, for example, Ferguson et al, T-ray Computed Tomography, Opt Lett. 2002 Aug. 1; 27(15):1312-4.
[0042] The image information for the patient can be obtained at a medical facility or a doctor's office and can be sent to the manufacturer in an electronic and/or digital form contained. The image information can be stored on a physical medium, such as a CD, DVD, flash memory device (e.g. memory stick, compact flash, secure digital card), or other storage device. The information may alternatively, or in addition, be transmitted electronically with the Internet or worldwide web using appropriate transfer protocols. Also, electronic transmissions can include e-mail or other digital transmission to any appropriate type of computer device, smart phone, PDA or other devices in which electronic information can be transmitted.
[0043] Appropriate handheld devices (used as illustrated in FIG. 13), can include handheld mobile device or portable communication devices, such as the iPhone� handheld mobile device sold by Apple Inc., a corporation of California, USA; the LG Shine� handheld mobile device sold by LG Corp. a corporation of REPUBLIC OF KOREA; or the Blackberry Bold� handheld mobile device sold by Research In Motion Limited a corporation of CANADA. The handheld device can be those that are held in the palm of a hand of a user, such as a surgeon (see FIG. 14). The surgeon can then enter data with a stylus, keyboard, touch screen, etc. The handheld device can use local area networks, cell phone networks, or other data transmission systems to communicate with a main memory and processor of a service provider (see FIG. 14).
[0044] Appropriate handheld devices can provide access to electronic communication or file transfer protocols, such as internet or electronic mail, to transfer or access information files. The handheld devices can have installed programs that can be used to manipulate the information files, as discussed herein. Alternatively, or in addition thereto, the handheld devices can access servers that process data files while receiving input through the handheld devices and displaying images to the surgeon or user via the handheld device. In certain instances, the handheld device may only be a client that does not process and edit a data file of the pre-op plan.
[0045] With continued reference to FIG. 1, at 110, the information collected at 100 can be used to create a three-dimensional model or image of the bone or joint with or without associated soft tissue or related anatomy using commercially available computer modeling software from various vendors or developers, such as, for example, from Materialise USA, Ann Arbor, Mich. The three-dimensional model of the patient's anatomy can be viewed on a computer display or other electronic screen and can also reproduced as a hard copy on film or other medium and viewed by direct or indirect or backlight illumination. The model can be sized for viewing on any appropriate screen size and may be cropped, rotated, etc. as selected by the individual (e.g. the surgeon) viewing the screen.
[0046] At 120, soft tissue associated with the affected anatomy can be modified, or removed or repaired, to restore alignment of the joint, for example, or to remove torn or diseased tissue, or to cut or repair ligaments, or to provide natural or artificial ligament grafts. Soft tissue information can be optionally used as an additional design parameter or input for the implant design, at 125. For example, a custom or patient-specific bearing articulation of a knee joint can be designed based on the kinematic profile and the soft tissue/ligament information available for a particular patient. Further, kinematic information for the patient can be obtained by an actual gait analysis of the patient, and can also be obtained by computer modeling software that uses the MRI images of the patient's joints and associated ligaments, muscle or other soft tissue to derive kinematic analysis of the patient and corresponding recommendations for soft tissue modification, such as releasing a ligament, for example. Such software is commercially available from the Biomechanics Research Group, Inc., of San Clemente, Calif.
[0047] At 130, a preliminary pre-operative plan of the surgical procedure can be prepared for surgeon or other medical user or technician review, including the planning of various bone resections, sizes and types of implants, and various geometric requirements including relevant dimensions, such as height, width, orientation of particular features, etc. The preliminary pre-operative surgical plan can include a recommendation of particular implants and associated instruments to be used in the surgical procedure, as discussed below. The preliminary pre-operative surgical plan can be in the form of digital images that can be viewed interactively using a computer modeling software, such as the software referenced above. The preliminary pre-operative plan and any further changes or a finalized pre-operative plan can be a plan devised to obtain a healthy or as close to healthy anatomical orientation after an operative procedure. The healthy anatomy can be based on natural or pre-injury anatomy or mechanically correct or efficient anatomical orientation.
[0048] At 140, the preliminary pre-operative surgical plan can be submitted to the surgeon (or other user) for review, either electronically or by land mail, and either in digital or hard copy form, as discussed above in connection with transmitting imaging information. In particular, the surgeon can review the resection planes shown in image of the patient's anatomy, make changes in the location, size and orientation of the resection planes and, generally, work interactively until the pre-operative plan from 130 is surgeon-approved. Specifically, the surgeon may approve the image of the patient's anatomy showing corresponding resection planes. As shown in FIGS. 7 and 8, the patient's anatomy 510, as represented in the image, can be, for example, a distal femur with approved resection planes including medial and lateral anterior chamfer planes 513, medial and lateral anterior cut planes 511, medial and lateral posterior chamfer planes 512 and medial and lateral posterior cut planes 514. Following the surgeon's approval of the anatomy and the resection planes at 140, the surgeon is provided with the opportunity to remove one or more osteophytes/protrusions from the image of the patient's anatomy 510 at surgeon-selected locations and depths at 500 (See FIG. 6). Removal of such protrusions and smoothening of the joint surface that receives the implant can parallel the intra-operative joint preparation by the surgeon and improve the actual fit of a surgeon-selected implant, whether patient-specific, semi custom, or off the shelf.
[0049] An automated osteophyte/protrusion removal control module 500 can be incorporated in the planning stage of the manufacturing method illustrated in FIG. 1. The automated osteophyte/protrusion removal control module 500 can be provided as a separate pre-operative planning module, as shown in FIG. 6, or it can be incorporated and/or fully integrated with the manufacturing method illustrated in FIG. 1.
[0050] Certain parts of the bone, including various bone bumps, protrusions, growths and osteophytes can be generally removed from the three-dimensional reconstruction of a patient's anatomy before designing a patient-specific implant or semi-custom implant, or before selecting an off the shelf implant. The automated osteophyte/protrusion removal control module can replace a time-consuming and potentially less accurate manual modification of the three-dimensional image to remove such bone growths or osteophytes by an experienced image or CAD technician. The automated osteophyte/protrusion removal control module 500 can provide more accurate and faster removal of such bone irregularities, which can vary in shape, location and size from patient to patient. It will be appreciated that the osteophyte/protrusion removal control module 500 can be used for smoothing out a bone surface by removing any type of bone protrusion, including bumps, irregularities and osteophytes. According to the present teachings, osteophytes are illustrated as exemplary, but not exclusive, candidates for complete or partial removal.
[0051] Referring to FIG. 6, the osteophyte/protrusion removal control module 500 can start 502 with an input of the three-dimensional image of the patient's anatomy 510 including resection planes, as shown in FIGS. 7-9, after review and approval of the resection planes by the surgeon (or other user, including other professionals or technicians) at 140 of FIG. 1. In the exemplary illustration of FIG. 7, the image of the patient's anatomy 510 can be analyzed to identify osteophyte/protrusion locations 530 (at 504 of FIG. 6) by determining tissue or bone overhang protruding past outer edges 532 of the various resection planes, such as the resection planes illustrated at 511, 513, 512 and 514 in FIGS. 7-9. If such osteophyte/protrusions 530 extend beyond the edges of the resection planes in the direction of the planned or anticipated implant location, the osteophyte/protrusions 530 can interfere with implant fitting.
[0052] Referring to FIGS. 6, 7 and 10, in addition to identifying the location of osteophytes/protrusions 530, the osteophyte/protrusion removal control module 500 can provide visual control for the surgeon to select the aggressiveness of osteophyte/protrusion removal, or the degree of smoothening and/or flattening of the corresponding joint anatomy. Specifically, by fine-tuning the osteophyte/protrusion locations, at 506 of FIG. 6, the surgeon can control the depth of the osteophyte/protrusion removal in a continuous or discrete manner. In one aspect, a landmark location 540 for each osteophyte/protrusion 530 can be identified and pegged for measuring from and initiating a continuous series of constant or variable depth contours 542 to aid the surgeon in selecting the depth of osteophyte/protrusion removal. The depth contours can be automatically generated by the computer software that generates a three-dimensional model or image of the anatomy, such as the software commercially available, for example, from Materialise USA, Ann Arbor, Mich. The landmark location 540 can be a location of lowest possible depth in the vicinity of the identified osteophyte/protrusion, a minimum, or a valley location, as shown in FIG. 10. Although the depth contours 542 are shown as discrete in FIG. 10, it will be appreciated that a continuous removal control can be provided, such that the surgeon can exercise unlimited choices of depth contours for removal. The depth contours 542 can represent curved smoothed-out surfaces under the original osteophyte/protrusion 530 and can be exposed after an overlying area is shaved or peeled in the image of the patient's anatomy 510 by the operation of graphical or visual removal tools provided on the image of the patient's anatomy 510. The surgeon or other user can manipulate the graphical removal tools with a user interface, such as a mouse, touch screen, joystick, slide pad, or other user interface.
[0053] Referring to FIG. 7, various visual removal tools can be provided for on-screen manipulation and control by the surgeon, at 508 of FIG. 6. For example, a removal tool corresponding to each edge of a resection plane can be provided and used to visually/graphically remove a portion of an osteophyte/protrusion associated with a particular edge 532. In FIG. 7, four such exemplary removal tools 520 a, 520 b, 520 c, 520 d (collectively referenced as 520) are shown, each removal tool associated with an edge of a resection plane, such as lateral and medial chamfer plane and lateral and medial cut plane. Although the removal tools 520 are illustrated as straight sliders in FIG. 7, the amount removed follows a depth contour 542, as illustrated in FIG. 10. The removal tools 520 can include a visual indicator 525 that can provide information to the surgeon in the form of a number on a scale indicative of the depth of aggressiveness of osteophyte/protrusion removal. In another aspect, the indicator 525 can provide visual information in terms of variable color in shades gradually changing from minimum depth removal (green, for example) to maximum depth removal (red, for example).
[0054] After the surgeon completes the osteophyte/protrusion removal, the surgeon can manipulate and superimpose implant images in relation to the modified patient's anatomy 510. In FIGS. 11 and 12, exemplary images of a resected femur 510 and tibia 515 referenced relatively to a mechanical axis 522 are illustrated. The femur image illustrates the patient's anatomy 510 after the osteophytes/protrusions 530 shown in FIGS. 8 and 9 have been removed and a femoral component 560 is placed on the resulting smoothed out surface that follows one of the depth contours 542 shown in FIG. 9.
[0055] Based on the preliminary pre-operative surgical plan and the patient information, the surgeon can make a recommendation regarding the design of the implant at 150, and any desired associated alignment guides at 160. At 150, the surgeon can recommend a method of designing an implant. Specifically, the surgeon can select one of the following three options: a first option of a custom or patient-specific implant at 170 or a second option of a semi-custom made implant at 180, or a third option of a standard or off-the-shelf implant at 190. It will be appreciated that, based on the surgeon's recommendation at 140, the preliminary pre-operative surgical plan can be modified at 130 and then resubmitted to the surgeon for approval.
[0056] A custom-made implant is a patient-specific, one of a kind implant specifically made for a particular patient, and consequently there is no inventory associated with such implant. Standard or off-the-shelf-implants are available and stocked in a number of sizes, typically six or more, and a number of configurations or types, including bilateral or unilateral implants, constrained, semi-constrained, mobile, etc. Because of the variety of sizes and configurations that are kept in stock to be accommodate different patients, a large inventory of standard implants is created, and several molds for each type and size of implant may be used. As described below in detail, semi-custom implants provide an intermediate solution between custom-made and off-the-shelf implants. Semi-custom implants reduce the size of inventory and molds required for production, while allowing some degree of patient-specific customization.
[0057] Custom or patient-specific implants, when approved by surgeon at 170 for a specific patient, can be manufactured for the patient by rapid prototyping methods, such as stereolithography or other similar methods, or by CNC milling, or other automated or computer-controlled machining, or by robotic methods, at 250. Manufacturing can take place at a manufacturing center or facility in situ or at remote or off-site location. It will be understood that in situ manufacturing is used as a short hand for a manufacturing site of the original equipment manufacturer (OEM), but can be physically located at a different facility of the OEM. Off-site or remote manufacturing will be understood to refer to facilities operated by other manufacturers who are contracted by the OEM for manufacturing all or some of the components or parts for the surgical procedure.
[0058] Off-the-shelf implants, when approved by the surgeon a 190, can be manufactured by standard casting methods from bar stock or other stock material at 200, then shaped to a final shape and size by grinding or milling at 210, polished at 220, and then cleaned/passivated at 230. Such off-the-shelf implants can be part of an existing inventory, or mass-produced, or produced by just-in-time agile manufacturing methods.
[0059] Semi-custom implants, when approved by the surgeon at 180, can be made from a generic casting at 240, as described below, or by modifying existing standard implant designs to match various features or parameters based on the anatomy of the patient, as described in co-pending patent application entitled Patient-Modified Implant and Associated Method, Ser. No. 12/103,834, filed on Apr. 16, 2008, the disclosure of which is incorporated by reference herein. After the generic casting is modified for certain parameters of a patient, it can be processed at aspects 210-230 to a passivated form. Patient-specific parameters can include parameters relating to the size of the implant, including height, width, various articulation parameters or angles, etc., as discussed in specific example below in reference to FIGS. 3-5.
[0060] The surgeon's review of the surgical plan at 140 may further include, at 160, a request for one or more patient-specific alignment guides to be used with the implant. Patient-specific alignment guides are described in co-pending patent application Ser. No. 11/756,057, filed on May 31, 2007, Ser. No. 11/971,390, filed on Jan. 9, 2008, Ser. No. 12/025,414, filed on Feb. 4, 2008, and Ser. No. 12/039,849 filed on Feb. 29, 2008. The alignment guides can be manufactured at 260 with by rapid prototyping methods, such as stereolithography or other similar methods or by CNC milling, or other automated or computer-controlled machining or robotic methods, and cleaned at 270. The alignment guides, the implants and optionally other disposable instruments can be packaged and sterilized at 280, and forwarded to the surgeon or the surgeon's medical facility for implantation at 290.
[0061] Referring to FIG. 2, a computer interface 400 to a computer program for the management of the manufacturing method is illustrated diagrammatically. An orthopedic system manager 402 can be in the form of software or other computer program associated with the original equipment manufacturer. The orthopedic system manager 402 can be accessible locally via dedicated computer machines or computer terminal directly communicated with software either by hard wire or wirelessly. The orthopedic system manager 402 can also be accessible remote remotely via the Internet or other remote communication portals using any electronic or other devices that can connect to the Internet or other web-based network, or other similar communication networks, including cable, satellite and telephone-based networks.
[0062] The system manager 402 can provide access to patient file information, including lists of all current patients at 403, and surgery dates, surgeons, and approval status of the surgical plan for each patient, at 404. Each patient file can include personal and medical information of the patient, such as, for example, weight, height, gender, age, lifestyle, pertinent medical records and medical history, as well as information on patient assessment that includes physical and kinematic evaluation pertaining to the orthopedic procedure at 406, and soft and hard tissue analysis at 408, including information provided at aspects 120 and 125 of FIG. 1, as discussed above. Imaging center information for patient scans, as discussed in relation to aspects 100 and 110 of FIG. 1, can added or modified at 410, and an imaging center for each specific patient can be specified at 412. Surgeon profiles, including surgeon preferences regarding anatomic axes alignment or implant and instrument preferences that can be taken into account when preparing the preliminary pre-operative plan discussed at aspect 130 of FIG. 1, can be created and edited at 414. Information and selection of manufacturing centers can be accessed at 416 for manufacturing the implants and or alignment guides as discussed in relation to aspects 260, 250, 240, and 210-230 of FIG. 1. The preliminary pre-operative surgical plan for each patient can be provided at 418, as discussed above at 140 in reference to FIG. 1, and e-mailed or otherwise communicated to the patient's surgeon at 420.
[0063] As discussed above at aspects 150 to 190 of FIG. 1, one implant option includes manufacturing semi-custom implants by generic casting. Illustrative examples of generic casting of a semi-custom femoral component are shown in FIGS. 3-5. A generic casting 300 of the implant is a casting that is more specialized than ordinary bar stock, from which any size of component can be made, but less specialized than the off-the-shelf components that are available in a particular number of sizes, typically six-to ten sizes and are finished from specific castings of those sizes. The generic casting can be made in a size and shape that can accommodate a range of variable features for the component, and at the same time can be machined to multiple sizes, such as three or four smaller sizes. In contrast, off-the-shelf implants require a mold or casting for each offered size, and a larger inventory of available sizes for each implant component. The generic casting can generally include geometric features which are size/shape and/or patient-independent or universal, and also features that are size/shape or patient-specific, as discussed in the examples below. More particularly, the generic casting can include at least one geometric feature that will remain unchanged for any patient or universal feature, and at least one geometric feature that can be specifically customized for and is specific to a particular patient.
[0064] Referring to FIGS. 4 and 5, an exemplary generic casting 300 of a femoral component is illustrated. In this example, the generic casting 300 can have an anterior flange 302 of medial-lateral width W, and/or a height H and/or other geometric dimensions to accommodate multiple sizes of femoral components. For example, multiple sizes of left-sided implants 304 a, 304 b, and various sizes of right-sided implants 306 a, 306 b can be formed by a single generic casting. Appropriate markings or indentations or score lines for cutting to size can be provided, such as height markings 330, for example. The implant for a particular patient can be formed from the generic casting 300 by selecting particular features, such as the width W or height H, or other geometric features for a particular patient and machining the generic casting 300 to provide the size, dimension or shape, or combinations thereof for that particular geometric feature.
[0065] Referring to FIG. 5, the generic casting 300 does not include a patella track feature, but provides an area in which a custom patella track 308 can be machined at a custom angle for each specific patient. The generic casting 300 can also include additional material in the intercondylar notch area 310 to allow for custom machining of the intercondylar notch area 310 to accommodate various types of articulation or constraint in relation to a tibial component, such cams or intercondylar boxes, and other contact areas for articulation with the tibial component in accordance with a kinematic plan for the joint of the specific patient. Separate molds for posterior stabilized and cruciate retaining articulations can be made, each mold capable of accommodating multiple sizes of the corresponding implant type. For example, the intercondylar notch area 310 can be machined for line or area contact with the articular surfaces of a tibial component of various degrees of flexion. Exemplary articulations are disclosed in commonly assigned U.S. Pat. Nos. 6,589,283, 6,413,279, and 6,165,223, and in co-pending U.S. patent application Ser. No. 10/840,765 filed on May 6, 2004, all of which are incorporated herein by reference. Various markings 332 corresponding to different sizes can be provided.
[0066] Referring to FIG. 3, the generic casting 300 can include at least one patient-independent or universal feature, such as, for example, universal cement wells 312 or other universal features. Such universal features can be used with any internal geometry 314, which can be machined into the generic casting 300 to accommodate the appropriate shape and/or size for a specific patient.
[0067] Referring to FIGS. 3, 4A, 5A, 5B and 5C, a semi-custom implant (304 a, 304 b, 306 a, 306 b) can also be generated from a generic casting 300 by customizing a plurality of features based on the patient's anatomy during the pre-operative planning stage in addition to or instead of the parameters discussed above. The generic casting 300 can include a standard, non-custom articulation surfaces for the medial and lateral condyles and an internal bone engagement surface 300 a for engaging the resected femur and including five internal planes 511 a, 513 a, 517 a, 512 a and 514 a (FIG. 3) corresponding to anterior cut plane 511′, anterior chamfer plane 513′, distal cut plane 517′, posterior chamfer plane 512′ and posterior cut plane 514′ (FIGS. 5B and 5C). The internal bone engagement surface 300 a and corresponding internal planes of the generic casting 300 can correspond to standard, i.e., non-custom, sizes for the femoral component. By selecting a relatively large number of parameters corresponding to the patient's anatomy, as captured in the three-dimensional image of the corresponding joint portion of the patient, the semi-custom implant can be made to correspond to the patient's anatomy as closely as desired while retaining the internal geometry corresponding to the resection planes in the sagittal view. More specifically, in an exemplary embodiment, at least nine parameters of the implant can be patient-specific, including three lengths and six angles, as discussed below in reference to the preoperative plan for the specific patient and the various planned resection planes.
[0068] In an exemplary embodiment of the preoperative plan, an image of the patient's anatomy, i.e. the distal femur 510′ of the patient indicating the planned cut planes is illustrated in FIGS. 5B and 5C. FIG. 5B represents a distal view of the three-dimensional image of the distal femur 510′, as reconstructed by MRI, CT or other scans. The planned resections indicated on the image include medial and lateral sides of a posterior chamfer plane 512′, of a distal cut plane 517′, and of an anterior chamfer plane 513′. The most anterior point of the intercondylar notch is indicated by a tangent line at 907. FIG. 5C represents an anterior view of the three-dimensional image of the distal femur 510′, and illustrates the distal cut plane 517′, anterior chamfer plane 513′, and an anterior cut plane 511′. The straight edges of the various cut planes are indicated as follows in reference to FIG. 5C: posterior distal edge 950 is the posterior edge of the distal cut plane 517′; anterior distal edge 952 is the anterior edge of distal cut plane 517′; distal anterior edge 954 is the distal edge of the anterior cut plane 511′; and proximal anterior edge 956 is the proximal edge of the anterior cut plane 511′. FIG. 5B also illustrates the posterior distal edge 950, the anterior distal edge 952, and the distal anterior edge 954.
[0069] A representative, but not exhaustive list of patient-specific parameters that can be selected at the preoperative stage for incorporation into the implant design after a particular size implant has been selected is indicated in FIGS. 4A and 5A-C.
[0070] Specifically, a central plane 800 perpendicular to the epicondylar axis 801 of the distal femur 510′ can be identified through the center of and orthogonal to the medial-lateral width W of the generic casting 300. The central plane 800 can be used as a reference plane for the bone cuts and the various patient-specific parameters to be used in modeling the semi-custom implant. The specified parameters can include, for example, three distances/lengths of the implant and six angles of the implant, each of which can be modified pre-operatively to be patient-specific such that they correspond to conform to the particular patient's anatomy. It will be appreciated that a lesser or greater number of parameters or different parameters can be use to customize and optimize the implant for the patient. The following exemplary parameters are illustrated in FIGS. 4A and 5A-C: medial condyle width 802, lateral condyle width 804, notch height 806, first, second and third medial angles 808, 810, 812; and first, second and third lateral angles 814, 816, 818.
[0071] In particular, and referring to FIGS. 5B and 5C, the medial condyle width 802 is defined as the distance between the central plane 800 and the farthest medial point/tangent line 902 (but not part of an osteophyte) on the posterior distal edge 950. The lateral condyle width 804 is defined as distance between the central plane 800 and the farthest lateral point/tangent 904 on the posterior distal edge 950 (but not part of an osteophyte). The notch height 806 defined as the distance along the central plane 800 from the posterior cut plane 514′ at edge 906 to the most anterior point/tangent line 907 on the notch (but not part of an osteophyte). The first medial angle 808 is defined as the angle between the central plane 800 and the line 908 connecting the medial-most points on the posterior distal edge 950 and the anterior distal edge 952. The second medial angle 810 is defined as the angle between the central plane 800 and the line 910 connecting the medial-most points on the anterior distal edge 952 and the distal anterior edge 954. The third medial angle 812 is defined as the angle between the central plane 800 and the line 912 connecting the medial-most points on the distal anterior edge 954 and the proximal anterior edge 956. The first lateral angle 814 is defined as the angle between the central plane 800 and the line 914 connecting the lateral-most points on the posterior distal edge 950 and the anterior distal edge 952. The second lateral angle 816 is defined as the angle between the central plane 800 and the line 916 connecting the lateral-most points on the anterior distal edge 952 and the distal anterior edge 954. The third lateral angle 818 defined as the angle between the central plane 800 and the line 918 connecting the lateral-most points on the distal anterior edge 954 and the proximal anterior edge 956.
[0072] Referring to FIG. 1A, when the above parameters 800, 802, 804, 806, 808, 810, 812, 814, 816, 818 are identified in the pre-operative planning stage, an image of a semi custom implant customized with these parameters can be displayed at 135 for review and comparison with the image of the patient's anatomy. The values of the parameters 800, 802, 804, 806, 808, 810, 812, 814, 816, 818 can be adjusted, as desired, and the implant with the adjusted or finalized parameters can selected at 137 and forwarded to the surgeon for review at 140 together with the preoperative plan, as described above in connection with FIG. 1. The preoperative planning procedure can proceed as discussed above.
[0073] As discussed above, each semi-custom implant has an internal bone engagement geometry including five planes 511 a, 513 a, 517 a, 512 a and 514 a corresponding to the resection planes and at least nine patient-specific parameters, including the three lengths and six angles described above.
[0074] It will be appreciated from the above discussion that generic casting can greatly reduce inventory, machining costs and investment in mold tooling, while at the same time accommodating sizes and geometric features specific to a patient. Specifically, each implant type can be formed from a generic casting that can accommodate multiple AP sizes corresponding to medial-lateral widths, such as four sizes, for example. For implants that are available in eight sizes, generic casting can reduce inventory by a half, using two molds total for eight sizes. Further, additional reductions in inventory can be obtained by combining right and left side implants into a single generic casting, as discussed above in relation to FIG. 4.
[0075] The process can then be followed as illustrated in FIGS. 13A and 13B as discussed above in relation to FIG. 1. The blocks in FIGS. 13A and 13B that are illustrated with the same reference numerals as in FIG. 1, but augmented with a prime are not discussed in further detail, but are discussed above in FIG. 1 and include substantially similar processes. In addition to the various applications discussed above, input from a surgeon or other appropriate user can be provided with a handheld device, as discussed above. As illustrated in FIGS. 13A and B, a handheld device can be used by the surgeon to review the pre-operative plan at 140′. It will be understood, however, that the process for providing a selected implant and tools can be similar to that discussed above, for example, with reference to FIG. 1 [0076] With reference to FIG. 14, a schematic diagram illustrating main or exemplary hardware components for the process illustrated in FIGS. 13A and 13B is illustrated. The pre-operative plan or preliminary pre-operative plan from block 130 and 130′ can be developed or produced by a service provider 700. The service provider 700 can own, operate, manage, or the like a main processor 702 and a main memory 704. The main processor and main memory 702, 704 can be at the service provider 700, in communication with the service provider 700, or otherwise controlled, maintained, or used by the service provider 700. Further, the main processor and main memory 702, 704 can be incorporated into a single server system. Regardless, the main processor 702 can process or execute a program, such as a program to develop the pre-operative plan, accept inputs from the surgeon, and augment or generate the final pre-operative plan. In addition, the main processor 702 can also be used to design and output the implant and alignment guide in blocks 150, 150′, and 160, 160′.
[0077] As discussed above, and further herein, the pre-operative plan can be delivered or accessed by the surgeon via notification or surgeon access in block 600′, 602′. The access or delivery of the pre-op plan can be via an internet or worldwide web connection 706 that uses a first communication method 708 from the service provider 700 and a second communication method 710 to a handheld device 712. It will be understood that the first and second communication method 708, 710 can be wired or wireless and can both be the same. Alternatively, or in addition to an internet connection a cell or mobile phone connection system 714, such as a tower, cell phone, antenna, can be provided. A first communication line 716 can transmit a communication through the mobile phone connection system 714. It will be understood that the service provider can communicate directly with the mobile phone connection system 714 via connection 716 or though an indirect connection 716′, such as an internet connection. A second communication line 718 can be used by a surgeon 720 with the handheld device 712. Again, it will be understood, that the first and second communication system 716, 716′, and 718 can be wired or wireless and can be the same or different. In addition, an intermediate system, such as a laptop or desktop computer 722 can be in communication with a system, such as the internet 706 via a first communication system 724 and the handheld device 712 can be interconnected with the computer 722 via a second communication system 726. Again, the first and second communication system 724, 726 can be the same or different and be wired or wireless.
[0078] With further reference to FIGS. 13A, 13B, and 14, the pre-operative plan from block 130′ can be delivered to a surgeon or accessed by a surgeon in any appropriate manner, such as via the internet 706 or cell communication 714. The pre-operative plan can be the preliminary pre-operative plan as discussed above. The pre-operative plan can include or be saved as a data file, in the main memory 704 associated with the main processor 702 of the service provider 700, of an appropriate type including image data, patient data, resection area data, etc. The pre-operative plan can be generated and stored by the service provider 700. The service provider 700 can be any appropriate service, such as an implant and/or guide manufacturer or specification producer. A specification producer can be a service that provides specifications for an implant or guide to a manufacturer for production.
[0079] The service provider 700 can notify the surgeon 720 or user that the preliminary pre-operative plan is ready for review in block 600′. The notification that the pre-operative plan is prepared can be performed in any appropriate manner. For example, an electronic mail notification can be sent to the surgeon 720, a text message can be sent to the surgeon 720, a telephone call can be made to the surgeon 720 via landline or a wireless connection, as illustrated in FIG. 14. Regardless, the surgeon can be notified that the pre-operative plan is ready for review in block 600 through the use of the mobile device 712.
[0080] Once the surgeon 720 is notified that the pre-operative plan is ready for review, the surgeon 720 can access the pre-operative plan in block 602′. The surgeon can access the pre-operative plan in one or a plurality of ways in block 140′. For example, the surgeon 720 can download the pre-operative plan to the handheld device in block 604′. Alternatively, or in addition thereto, the surgeon 720 can access the main processor/memory 702, 704 to review the pre-operative plan in the main memory 704 in block 606′ with the handheld device 712. It will be further understood that the surgeon 720 may also access the plan with the computer or terminal 722 by downloading the pre-operative plan data file to the computer 722 on which appropriate software is installed to access the pre-operative plan. The surgeon 720 may also view a printout of the pre-operative plan for manipulating or commenting on the pre-operative plan, or any other appropriate manner.
[0081] If the surgeon 720 downloads the file to the handheld device 712, the file can be downloaded to the handheld device 712 using any appropriate transfer protocol or communication system, as illustrated in FIG. 14. For example, the handheld device 712 can be connected to the computer 722 through an appropriate communications cable or protocol 726, such as Bluetooth�, a wireless communication protocol or a Universal Serial Bus (USB) cable. Once the file is downloaded to the handheld device 712, a program on the handheld device 712 can execute or read the file and display images for the surgeon 720. The surgeon 720 can then review the plan in block 140′. For example, as illustrated in FIG. 7, a view of a bone to be resected can be displayed along with the slider bars 520 for allowing editing or augmentation of the pre-operative plan by the surgeon.
[0082] The view of the images, including the slider bars 520, can be augmented for the handheld device 712. For example, a cropped and zoomed image may only include a portion of the bone or region to be cut or resected for viewing by the surgeon. For example, as illustrated in FIG. 15, the handheld device 712 can include a view screen 740 that displays an image of the bone to be resected, but only includes a portion thereof, such as a lateral portion of a distal femur. The zoomed and cropped image can include fewer than all of the slide bars 520, such as only the slide bars 520′c and 520′d. The slide bars 520′c and 520′d can be illustrated and accessed by the surgeon to change data in the file for creation of appropriate instruments and implants for the specific patient or for augmentation of semi-custom or selection of an off-the-shelf instrument and implant.
[0083] The surgeon 720 can access or change the pre-operative plan file using the slide bars 520′c and 520′d through any appropriate access or manipulation process. For example, the screen 740 of the handheld device 712 can be a touch screen. Accordingly, the surgeon 720 can touch the screen with a finger or stylus to move the slider or marker portion 525 on the slider bars 520′c and 520′d. Alternatively, or in addition thereto, a trackball or other pointer device 742 can be provided to access and move the slider bars 520′c and 520′d. It will be understood that the handheld device 712 can have any appropriate input devices, such as an external or connected input devices, that can be mapped for appropriate command inputs into the system for augmenting the pre-operative plan file. The changes, if made by the surgeon, can then be saved to the pre-operative plan file to generate an edited pre-operative plan file.
[0084] If the surgeon 720, after review of the pre-operative plan in block 140′, finds the plan to be unacceptable in block 620′, the NO path 622′ can be followed for the surgeon 720 to edit the plan in block 624′. The surgeon 720 can edit the plan in block 624 in any appropriate manner, such as by moving the slider bars 520′c and 520′d, or any other appropriate slider bars. Alternatively, the surgeon 720 can edit the pre-operative plan in any appropriate manner, including those discussed above. Moreover, the surgeon 720 can input changes into the pre-operative plan using any appropriate input portion, such as touching the screen 740. It will be understood, that the pre-operative plan file that is downloaded to the handheld device 712 can then be saved with the edits as the edited pre-operative plan and transmitted to the service provider 700 for appropriate edits to be re-reviewed or finalized, as discussed above. It will be understood, however, that the handheld device 712, which can include the appropriate program, can save the file in the appropriate format and transmit it back to the service provider 700.
[0085] Alternatively, or in addition to downloading the pre-operative plan file, the handheld device 712 can access the pre-operative plan which is stored in the main storage 704 associated with the service provider in block 606 and separate from the handheld device 712. If the handheld device 712 accesses the pre-operative plan on the main processor, memory 702, 704, the handheld device 712 need only display an image representing a portion of the file on the display screen 740. That is, the pre-operative plan and any edits or processing made to the pre-operative plan can be done solely or substantially by the main processor 702 that executes a program to manipulate and display the file. The main processor 702 and the main memory 704 need not be physically near or connected to the handheld device 712.
[0086] The handheld device 712 can be provided to display the image, such as an image of the bone for resection, for the surgeon 720. Therefore, the handheld device 712 may not be required to process the pre-operative plan file from the service provider 700, but only be provided to display the pre-operative plan file and receive and transmit input from the surgeon 720. Accordingly, even if the pre-operative plan is accessed from the service provider in block 606, the slide bars 520′c and 520′d can be displayed on the display 742 of the handheld device 712 for input by the surgeon 720. The inputs, however, can be directly transmitted to the main processor 702 for processing augmentation or editing of the file. This is in addition or alternative to augmenting or editing the file that has been downloaded to the handheld device 712 for re-transmission of the edited pre-operative plan to the service provider 700.
[0087] By only or substantially accessing the pre-operative plan file from the main processor/memory 702, 704 data transmission can be minimized from the main memory 704 or provider 700 to the handheld device 712 of the surgeon 720. Decreased data transmission can provide increased speed and decreased data usage costs or bottlenecks in a system. In addition, the handheld device 712 can be provided or include limited memory and processing capabilities when the pre-operative plan file is only accessed with the handheld 712 and only small amounts of information are transferred, for example, regarding slide bar location and smaller portions of an image file. Accordingly, it can be provided, that a complete or pre-operative plan is transmitted to the handheld device 712, processed completely on the handheld device 712, edited on the handheld device 712, saved and re-transmitted back to the service provider 700 or the handheld device 712 can only access the pre-operative plan file saved at the main memory 702 and transmit edits to the server.
[0088] Data transmission and processing can also be reduced by limiting or cropping the pre-operative plan data file. For example, as illustrated in FIG. 15, only a lateral and distal portion of the bone is illustrated on the display device or display screen 740 of the handheld device 712. It will be understood that an image file or image information can include the entire bone or other data saved in the image file. Accordingly, the image file can be cropped at the server or at the provider and only a portion of the image file transmitted to the handheld device 712. This can be done repeatedly for different portions of the image data to allow for smaller file packet size or file size for transmission to the handheld device 712. The cropping and compression of the data file can be done in substantially real-time by the server for a substantially seamless viewing and manipulation by the surgeon.
[0089] Even if the surgeon 720 accesses the data file on the main memory 704, the surgeon 720 can review the pre-operative plan block 140′, as discussed above, and make a determination of whether the pre-operative plan is acceptable in block 620′. As discussed above, if the pre-operative plan is not acceptable, the NO path 622′ can be followed to allow for surgeon edits in block 624′.
[0090] Further, regardless of the method of review of the pre-operative plan in block 140′, the pre-operative plan can be determined to be acceptable in block 620′ and follow the YES path 626′. When following the YES path 626′, the implant can be designed and an alignment guide can be designed in blocks 150′ and 160′.
[0091] The various methods described above in connection with the preparation a pre-operative plan may employ high resolution imaging of the patient's joint, as well as various other anatomic landmarks for obtaining the mechanical and anatomic axes of a specific patient. In knee arthroplasty, for example, MRI or CT of the hip, knee and ankle as well as an X-ray of the entire leg may be performed pre-operatively and used with an imaging protocol to prepare a detailed three-dimensional image of the knee joint for the pre-operative plan. T-ray CT (Terahertz Computed Tomography) may also be used for quick, non-ionizing radiation scans using portable, battery operated sources of T-ray radiation. T-ray scanners, T-ray sources and related instruments are commercially available, for example, from Advanced Photonix, Inc, Ann Arbor, Mich.
[0092] Referring to FIGS. 16-22B, various methods of reconstructing the patient's anatomy during the pre-operative plan while reducing the use of pre-operative scanning and imaging equipment are illustrated according to the present teachings.
[0093] In one embodiment, illustrated in reference to FIGS. 16-18, a high-resolution image of the knee joint only is used together with anthropometric data to prepare a three-dimensional model of the knee joint. Anthropometric data are publicly available from many sources and can include, among other things, lengths for body segments, density, mass and inertial properties, and centers of mass and axes of rotation. See, for example, David Winter, Biomechanics and Motor Control of Human Movement, 4th Edition, Chapter 4, Anthropometry, 2009, John Wiley & Sons, Inc. FIG. 4.1 of Winter's book provides, for example, various body segment lengths expressed as a fractions of body height. The Department of Defense maintains a collection of anthropometry resources. See for example the website of the Defense Technical Information Center (DTIC) at “dtic.mil/dticasd/anthro.html#data”.
[0094] In another embodiment, illustrated in reference to FIGS. 16 and 19-20, a high-resolution image of the knee joint only is combined with a digital photograph of the patient's leg. A third embodiment, illustrated in the flowchart of FIG. 22, avoids pre-operative imaging of the patient for the purpose of the pre-operative plan.
[0095] The use of anthropometric and personal data for determining a mechanical axis of the patient is illustrated in FIG. 17, in which a center of rotation of the femoral joint or hip center 76, a center of rotation of the knee joint or knee center 75 and a center of rotation of the ankle joint or ankle center 78 are indicated. Corresponding distances between these centers, such as hip height H1 (vertical distance between hip center 76 and knee center), hip offset D1 (horizontal distance between hip center 76 and knee center), tibial height H2 (vertical distance between ankle center 78 and knee center), and ankle offset D2 (horizontal distance between ankle center 78 and knee center) can be determined from the height or other personal data of the particular patient using publicly available anthropometric data. From this data, a hip angle α between the femoral anatomic axis AN of the femoral bone 72 and the femoral mechanical axis FM can be determined. Similarly, a tibial angle β, i.e., an angle between the tibial mechanical axis TM of the tibial bone 74 and the femoral mechanical axis FM can be determined.
[0096] Referring to FIG. 16, a high resolution scan of the knee joint 70 of a patient 60 can be taken pre-operatively using an MRI scanner or CT scanner or other scanner 50. Scanning is focused on the knee joint 70 of the patient, as illustrated in FIG. 16. The high resolution scan of the knee joint obtained can be used for the methods illustrated in the flowcharts of FIGS. 18 and 20 as discussed below.
[0097] Referring to FIG. 18, various personal patient data are obtained preoperatively, at 1100. The personal patient data can include information for use with anthropometric databases and can include, for example, height, weight, body mass index (BMI), age, gender, race, ethnicity, daily activity, disability, etc. A high resolution scan of the knee joint 70 of the patient is also obtained pre-operatively at 1102, as illustrated in FIG. 16. Using the public anthropometric data, the hip height H1, the hip angle α, the tibial height H2 and tibial ankle β can be determined, at 1104 using the personal patient data. From this anthropometric data and the high resolution scan of the knee joint 70, the femoral mechanical axis FM and tibial mechanical axis TM can be determined, at 1106, and located in a two- or three-dimensional image model of the knee joint 70 as reconstructed from the knee joint scan, at 1108. Based on the image model, a pre-operative plan can be prepared, at 1110, as discussed above. Patient-specific alignment guides, other associated instruments and/or patient-specific, semi-custom, or non-custom implants can be made based on the pre-operative plan and the determination of the mechanical axis of the patient. The patient-specific alignment guides can include, for example, a three-dimensional bone engagement surface which is designed to be mate and be complementary to the three-dimensional surface of the image model of the knee joint, as discussed in detail in the patent applications cross-referenced above and incorporated herein by reference.
[0098] Referring to FIGS. 19 and 20, instead of using anthropometric data, a digital photographic image of the patient 60 in front of a grid surface 1120 or other reference surface can obtained pre-operatively using a digital photographic equipment 1130, as illustrated in FIG. 19. Using the digital photographic image, the hip center 76, knee center 75 and ankle center 78 can be identified on the grid surface 1120 and the corresponding hip height H1, tibial height H2 and various anatomic/mechanical axes (shown in FIG. 17) can be determined. It should be appreciated that other methods of reference for the digital photographic image can be used, including, for example size- or orientation indicative markers on the patient or in reference to the patient in the field of view of the camera.
[0099] Referring to flowchart of FIG. 20, a high resolution scan of the knee joint 70 of the patient can be obtained pre-operatively at 1150. A digital photographic image of the patient showing the leg anatomy in front of the grid surface 1120 is obtained, at 1152. The hip center 76 and ankle center 78 can be determined from the digital image, at 1154. The femoral mechanical axis FM and tibial mechanical axis TM can also be determined from the digital image, at 1156. Using this information and the using the high-resolution scan, a two-dimensional or a three-dimensional image model of the knee joint 70 showing the femoral and tibial mechanical axes FM and TM can be prepared at 1158. Based on the image model, a pre-operative plan can be prepared, at 1160, as discussed above. Patient-specific alignment guides, other associated instruments and/or patient-specific, semi-custom, or non-custom implants can be made based on the pre-operative plan and the determination of the mechanical axis.
[0100] Referring to FIGS. 21, 22A and 22B, an image-free method of representing a specific patient's anatomy for pre-operative planning is illustrated. A database that includes images and measurements of diseased/deformed bones and bone joints can be created from in-house patient data or other publicly available data, at 1170. Personal patient data can also be obtained preoperatively, at 1172. The personal patient data can include, for example, height, weight, body mass index (BMI), age, gender, race, ethnicity, daily activity, disability, etc. A best-fit bone model, illustrated at FIG. 22A, can be selected from the database based on the patient's personal data, at 1174. Optimization or iteration or visual selection can be used to select the best-fit bone model for the specific patient. The best-fit bone model may still have parameters that differ from the patient's personal data. After matching a patient with a best-fit bone model from the database that corresponds to one or more of gender, age, disability, race, etc, some other personal parameters may not quite match. For example, one or more of the hip height H1 or the tibial height H2, or the hip ankle α, or the tibial angle β may differ from corresponding known values H1′, H2′, α′ and β′ of the patient that are obtained pre-operatively for the specific patient, as discussed above. A transformation program/software can be used to transform, deform or morph the best-fit bone model of FIG. 22A to a patient customized model shown in FIG. 22B, in which all the known parameters of the patient are matched without altering other details of the best-fit bone model of FIG. 22A, at 1176. Various commercially available software programs can be used for the bone transformation/morphing, including those disclosed in U.S. Patent Application Publication 2004/006818, which is incorporated herein by reference. Based on the customized model, a pre-operative plan can be prepared, at 1178, as discussed above, and customized alignment guides, other associated instruments, and custom, semi-custom or non-custom implants can be made based on the preoperative plan.
[0101] The methods described above in reference with FIGS. 16-22B, provide different choices for controlling the extent of pre-operative image scanning of the patient for use in the pre-operative plan. The methods can be used to prepare corresponding patient-specific or patient-customized alignment guides and select implants, including patient-specific or customized implants, or semi-custom implants or non custom, off-the-shelf implants, as discussed above.
[0102] The foregoing discussion discloses and describes merely exemplary arrangements of the present teachings. Furthermore, the mixing and matching of features, elements and/or functions between various embodiments is expressly contemplated herein, so that one of ordinary skill in the art would appreciate from this disclosure that features, elements and/or functions of one embodiment may be incorporated into another embodiment as appropriate, unless described otherwise above. Moreover, many modifications may be made to adapt a particular situation or material to the teachings of the invention without departing from the essential scope thereof. One skilled in the art will readily recognize from such discussion, and from the accompanying drawings and claims, that various changes, modifications and variations can be made therein without departing from the spirit and scope of the present teachings as defined in the following claims.
Patent CitationsCited PatentFiling datePublication dateApplicantTitleUS1480285 *Dec 31, 1917Jan 8, 1924Moore Robert APortable sanding machineUS4246895 *May 29, 1979Jan 27, 1981Rehder GuentherApparatus for producing a conical surface on a boneUS4633862 *May 30, 1985Jan 6, 1987Petersen Thomas DPatellar resection sawguideUS4719907 *Mar 18, 1987Jan 19, 1988Orthospec, Inc.Orthopedic pin placement guideUS4721104 *Dec 2, 1985Jan 26, 1988Dow Corning Wright CorporationFemoral surface shaping apparatus for posterior-stabilized knee implantsUS4722330 *Apr 22, 1986Feb 2, 1988Dow Corning Wright CorporationFemoral surface shaping guide for knee implantsUS4800874 *Jul 14, 1987Jan 31, 1989Vereinigte Edelstahlwerke A.G.Anatomical bone plate and/or transfixion plateUS4893619 *Feb 4, 1988Jan 16, 1990Intermedics Orthopedics, Inc.Humeral osteotomy guideUS4896663 *Oct 14, 1988Jan 30, 1990Boehringer Mannheim CorporationSelf centering femoral drill jigUS4985037 *May 22, 1989Jan 15, 1991Petersen Thomas DUniversal modular prosthesis stem extensionUS5086401 *May 11, 1990Feb 4, 1992International Business Machines CorporationImage-directed robotic system for precise robotic surgery including redundant consistency checkingUS5176684 *Feb 20, 1992Jan 5, 1993Dow Corning WrightModular shaping and trial reduction guide for implantation of posterior-stabilized femoral prosthesis and method of using sameUS5282802 *Aug 19, 1991Feb 1, 1994Mahony Iii Thomas HMethod of securing a tendon graft with an interference fixation screwUS5490854 *Aug 17, 1993Feb 13, 1996Synvasive Technology, Inc.Surgical cutting block and method of useUS5593411 *Mar 13, 1995Jan 14, 1997Zimmer, Inc.Orthopaedic milling guide for milling intersecting planesUS5595703 *Mar 10, 1995Jan 21, 1997Materialise, Naamloze VennootschapMethod for supporting an object made by means of stereolithography or another rapid prototype production methodUS5704941 *Nov 3, 1995Jan 6, 1998Osteonics Corp.Tibial preparation apparatus and methodUS5709689 *Sep 25, 1995Jan 20, 1998Wright Medical Technology, Inc.Distal femur multiple resection guideUS5720752 *Feb 21, 1996Feb 24, 1998Smith & Nephew, Inc.Distal femoral cutting guide apparatus with anterior or posterior referencing for use in knee joint replacement surgeryUS5860981 *May 29, 1997Jan 19, 1999Dennis W. BurkeGuide for femoral milling instrumention for use in total knee arthroplastyUS5871018 *Jun 6, 1997Feb 16, 1999Delp; Scott L.Computer-assisted surgical methodUS6019767 *Jun 23, 1995Feb 1, 2000ArthrotekTibial guideUS6187010 *Sep 17, 1997Feb 13, 2001Medidea, LlcBone cutting guides for use in the implantation of prosthetic joint componentsUS6195615 *Feb 14, 1996Feb 27, 2001Pruftechnik Dieter Busch AgMethod of mutually aligning bodies and position-measuring sensor thereforUS6343987 *Nov 5, 1997Feb 5, 2002Kabushiki Kaisha Sega EnterprisesImage processing device, image processing method and recording mediumUS6503255 *Aug 24, 2000Jan 7, 2003Astra AktiebolagCutting guide instrumentUS6508980 *Sep 25, 1998Jan 21, 2003Massachusetts Institute Of TechnologyMetal and ceramic containing parts produced from powder using binders derived from saltUS6510334 *Nov 14, 2000Jan 21, 2003Luis SchusterMethod of producing an endoprosthesis as a joint substitute for a knee jointUS6514259 *Feb 2, 2001Feb 4, 2003Carnegie Mellon UniversityProbe and associated system and method for facilitating planar osteotomy during arthoplastyUS6517583 *Jan 30, 2000Feb 11, 2003Diamicron, Inc.Prosthetic hip joint having a polycrystalline diamond compact articulation surface and a counter bearing surfaceUS6519998 *Jan 2, 2002Feb 18, 2003Uniflex-Hydraulik GmbhRadial pressUS6520964 *May 1, 2001Feb 18, 2003Std Manufacturing, Inc.System and method for joint resurface repairUS6676892 *Jun 1, 2001Jan 13, 2004Board Of Regents, University Texas SystemDirect selective laser sintering of metalsUS6682566 *Jul 30, 2001Jan 27, 2004Klaus DraenertModular socket prosthesisUS6696073 *Aug 27, 2002Feb 24, 2004Osteotech, Inc.Shaped load-bearing osteoimplant and methods of making sameUS6697664 *Jun 18, 2001Feb 24, 2004Ge Medical Systems Global Technology Company, LlcComputer assisted targeting device for use in orthopaedic surgeryUS6772026 *Oct 5, 2001Aug 3, 2004Therics, Inc.System and method for rapidly customizing design, manufacture and/or selection of biomedical devicesUS6990220 *Jun 14, 2001Jan 24, 2006Igo Technologies Inc.Apparatuses and methods for surgical navigationUS6993406 *Apr 23, 2004Jan 31, 2006Sandia CorporationMethod for making a bio-compatible scaffoldUS7169185 *May 26, 2004Jan 30, 2007Impact Science And Technology, Inc.Canine acetabular cupUS7474223 *Apr 18, 2005Jan 6, 2009Warsaw Orthopedic, Inc.Method and apparatus for implant identificationUS7519540 *Aug 30, 2001Apr 14, 2009Christian MayaudComputerized prescription system for gathering and presenting information relating to pharmaceuticalsUS7646161 *Nov 15, 2006Jan 12, 2010Deutsches Zentrum Fuer Luft-Und Raumfahrt E.V.Method for controlling a robot arm, and robot for implementing the methodUS7651501 *Jan 26, 2010Wright Medical Technology, Inc.Instrument for use in minimally invasive hip surgeryUS8092465 *Jan 10, 2012Biomet Manufacturing Corp.Patient specific knee alignment guide and associated methodUS8105330 *Jun 9, 2008Jan 31, 2012Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS8444564 *May 21, 2013Jointvue, LlcNoninvasive diagnostic systemUS8473305 *Jun 12, 2009Jun 25, 2013Biomet Manufacturing Corp.Method and apparatus for manufacturing an implantUS8632547 *May 12, 2011Jan 21, 2014Biomet Sports Medicine, LlcPatient-specific osteotomy devices and methodsUS20020007294 *Apr 5, 2001Jan 17, 2002Bradbury Thomas J.System and method for rapidly customizing a design and remotely manufacturing biomedical devices using a computer systemUS20020059049 *Oct 5, 2001May 16, 2002Therics, IncSystem and method for rapidly customizing design, manufacture and/or selection of biomedical devicesUS20020128872 *Feb 5, 2002Sep 12, 2002Giammattei Charles P.Medical data recordation systemUS20020186818 *Feb 27, 2002Dec 12, 2002Osteonet, Inc.System and method for building and manipulating a centralized measurement value databaseUS20030009171 *Jul 8, 2002Jan 9, 2003Alain TornierAncillary tool for fitting a humeral component of an elbow prosthesisUS20030009234 *Jul 5, 2001Jan 9, 2003Treacy Patrick J.Pelvic prosthesis plus methods and tools for implantationUS20030011624 *Jul 13, 2001Jan 16, 2003Randy EllisDeformable transformations for interventional guidanceUS20030018338 *Dec 23, 2000Jan 23, 2003Axelson Stuart L.Methods and tools for femoral resection in primary knee surgeryUS20030039676 *Aug 27, 2002Feb 27, 2003Boyce Todd M.Shaped load-bearing osteoimplant and methods of making sameUS20040018144 *Jul 23, 2002Jan 29, 2004Briscoe Michael D.Hydrogen to steam reforming of natural gas to synthesis gasUS20040030245 *Apr 16, 2003Feb 12, 2004Noble Philip C.Computer-based training methods for surgical proceduresUS20040133276 *Oct 7, 2003Jul 8, 2004Imaging Therapeutics, Inc.Minimally invasive joint implant with 3-Dimensional geometry matching the articular surfacesUS20040243481 *Jul 1, 2004Dec 2, 2004Therics, Inc.System and method for rapidly customizing design, manufacture and/or selection of biomedical devicesUS20050008887 *Jul 9, 2004Jan 13, 2005Haymann Basil A.Mill blank library and computer-implemented method for efficient selection of blanks to satisfy given criteriaUS20050010300 *Jul 9, 2004Jan 13, 2005Disilvestro Mark R.Orthopaedic element with self-contained data storageUS20050015022 *Jul 15, 2003Jan 20, 2005Alain RichardMethod for locating the mechanical axis of a femurUS20050019664 *Aug 20, 2004Jan 27, 2005Isao MatsumotoNon-sintered type thin electrode for battery, battery using same and process for sameUS20050027303 *Jun 17, 2004Feb 3, 2005Lionberger David R.Pelvic waypoint clamp assembly and methodUS20050027361 *Jul 6, 2004Feb 3, 2005Reiley Mark A.Facet arthroplasty devices and methodsUS20050043806 *Aug 21, 2003Feb 24, 2005The Curators Of The University Of MissouriElbow arthroplasty systemUS20050043837 *Sep 27, 2004Feb 24, 2005Rudger RubbertInteractive orthodontic care system based on intra-oral scanning of teethUS20050059873 *Aug 26, 2003Mar 17, 2005Zeev GlozmanPre-operative medical planning system and method for use thereofUS20050216305 *Feb 7, 2003Sep 29, 2005Are FunderudSystem for preparation and transmission of specification for customized prosthesesUS20060004284 *Jun 30, 2005Jan 5, 2006Frank GrunschlagerMethod and system for generating three-dimensional model of part of a body from fluoroscopy image data and specific landmarksUS20060015120 *Apr 30, 2003Jan 19, 2006Alain RichardDetermining femoral cuts in knee surgeryUS20060038520 *Aug 4, 2005Feb 23, 2006Masanori NegoroVehicle control unit and vehicleUS20060052725 *Sep 2, 2005Mar 9, 2006Santilli Albert NSmall joint orthopedic implants and their manufactureUS20060142657 *Feb 21, 2006Jun 29, 2006Mako Surgical CorporationHaptic guidance system and methodUS20060287891 *Jun 16, 2005Dec 21, 2006Cerner Innovation, Inc.System and method in a computerized environment for charting pediatric growthUS20070015995 *Apr 25, 2006Jan 18, 2007Philipp LangJoint and cartilage diagnosis, assessment and modelingUS20070016008 *Nov 30, 2005Jan 18, 2007Ryan SchoenefeldSelective gesturing input to a surgical navigation systemUS20070016209 *Apr 3, 2006Jan 18, 2007Kelly AmmannMethod and apparatus for performing an open wedge, high tibial osteotomyUS20070027680 *Jul 27, 2005Feb 1, 2007Ashley James PMethod and apparatus for coding an information signal using pitch delay contour adjustmentUS20070039205 *Mar 2, 2006Feb 22, 2007Fila Luxembourg S.A.R.L.Method and system for identifying a kit of footwear components used to provide customized footwear to a consumerUS20070073133 *Sep 15, 2005Mar 29, 2007Schoenefeld Ryan JVirtual mouse for use in surgical navigationUS20070073136 *Mar 12, 2006Mar 29, 2007Robert MetzgerBone milling with image guided surgeryUS20070073137 *May 15, 2006Mar 29, 2007Ryan SchoenefeldVirtual mouse for use in surgical navigationUS20070272747 *May 25, 2006Nov 29, 2007Woods Sherrod AMethod and system for managing inventories of orthopaedic implantsUS20080009952 *Jul 2, 2007Jan 10, 2008Hodge W APrecision acetabular machining system and resurfacing acetabular implantUS20080015604 *Jun 30, 2006Jan 17, 2008Howmedica Osteonics Corp.Method for performing a high tibial osteotomyUS20080015605 *Jul 3, 2006Jan 17, 2008Howmedica Osteonics Corp.High tibial osteotomy guideUS20080021299 *Jul 18, 2006Jan 24, 2008Meulink Steven LMethod for selecting modular implant componentsUS20080021494 *Sep 24, 2007Jan 24, 2008Guenther Schmelzeisen-RedekerSystem for withdrawing body fluidUS20080021567 *Dec 27, 2006Jan 24, 2008Zimmer Technology, Inc.Modular orthopaedic component caseUS20080027563 *Oct 11, 2007Jan 31, 2008Zimmer Technology, Inc.Modular knee prosthesisUS20080094396 *Sep 2, 2005Apr 24, 2008Koninklijke Philips Electronics, N.V.System For The Three-Dimensional Imaging Of A Moving JointUS20080257363 *Apr 16, 2008Oct 23, 2008Biomet Manufacturing Corp.Method And Apparatus For Manufacturing An ImplantUS20090012526 *Feb 13, 2008Jan 8, 2009Fletcher Henry HDrill system for acetabular cup implantsUS20090012659 *Feb 20, 2008Jan 8, 2009Raytheon CompanyManual Human Interfaces to ElectronicsUS20090018546 *Jul 10, 2008Jan 15, 2009Daley Robert JMethods and apparatus for determining pin placement during hip surgeryUS20090018666 *Jul 9, 2008Jan 15, 2009Eska Implants Gmbh & Co. KgSet For Creating An Offset-Resurfacing Hip-Joint ImplantUS20090024131 *Sep 16, 2008Jan 22, 2009Biomet Manufacturing Corp.Patient specific guidesUS20090024169 *Sep 29, 2008Jan 22, 2009Facet Solutions, Inc.System and method for multiple level facet joint arthroplasty and fusionUS20090151736 *Feb 13, 2009Jun 18, 2009Biomet Manufacturing Corp.Method And Apparatus For Manufacturing An ImplantUS20090254367 *Jun 12, 2009Oct 8, 2009Biomet Manufacturing Corp.Method and Apparatus for Manufacturing an ImplantUS20100016984 *Jan 21, 2010Harutaro TrabishAcetabulum Surgical Resurfacing AidUS20100016986 *Jan 21, 2010Harutaro TrabishFemoral Head Surgical Resurfacing AidUS20100023015 *Jul 17, 2009Jan 28, 2010Otismed CorporationSystem and method for manufacturing arthroplasty jigs having improved mating accuracyUS20100152782 *Feb 26, 2010Jun 17, 2010Biomet Manufactring Corp.Patient Specific High Tibia OsteotomyUS20100198067 *Feb 2, 2009Aug 5, 2010Mahfouz Mohamed MNoninvasive Diagnostic SystemUS20100324692 *Aug 31, 2010Dec 23, 2010Biomet Manufacturing Corp.Method and Apparatus for Manufacturing an ImplantUS20110004317 *Dec 17, 2008Jan 6, 2011Hacking Adam SOrthopaedic implantsUS20110014081 *Jan 20, 2011Howmedica Osteonics Corp.Laser-produced porous structureUS20110015636 *Jan 20, 2011Biomet Manufacturing Corp.Patient-Specific Elbow Guides and Associated MethodsUS20110015639 *Sep 29, 2010Jan 20, 2011Biomet Manufacturing Corp.Femoral Acetabular Impingement GuideUS20110015752 *Jul 14, 2009Jan 20, 2011Biomet Manufacturing Corp.System and Method for Acetabular CupUS20110016690 *Dec 11, 2008Jan 27, 2011Universiti MalayaProcess to design and fabricate a custom-fit implantUS20110022174 *Jan 11, 2009Jan 27, 2011Technion - Research & Development Foundation LtdModeling micro-scaffold-based implants for bone tissue engineeringUS20110071802 *Mar 24, 2011Ray BojarskiPatient-adapted and improved articular implants, designs and related guide toolsUS20110172672 *Jul 14, 2011Biomet Manufacturing Corp.Instrument with transparent portion for use with patient-specific alignment guideUS20120010619 *Jan 12, 2012The Cleveland Clinic FoundationMethod and apparatus for providing a relative location indication during a surgical procedureUS20120010710 *Jan 12, 2012Robert FriggAdvanced Bone Marker and Custom ImplantsUS20120010711 *Jan 12, 2012Antonyshyn OlehMethod of forming patient-specific implantUS20120029345 *Feb 2, 2012Joint Vue, LLCNoninvasive diagnostic systemUS20120303004 *Jun 20, 2012Nov 29, 2012Biomet Manufacturing Corp.Backup surgical instrument system and methodUS20120316564 *Jun 11, 2012Dec 13, 2012Biomet Manufacturing Corp.Drill guides for confirming alignment of patient-specific alignment guidesUS20130001121 *Jul 1, 2011Jan 3, 2013Biomet Manufacturing Corp.Backup kit for a patient-specific arthroplasty kit assemblyUS20130018483 *Jun 1, 2012Jan 17, 2013Zimmer, Inc.Rapid manufacturing of porous metal prosthesesUS20140005672 *Jun 30, 2012Jan 2, 2014Jon M. EdwardsCutting block including modular mounting systemsUS20140018934 *Sep 16, 2013Jan 16, 2014Biomet Manufacturing, LlcPatient-Specific Tools And ImplantsUS20140018948 *Sep 6, 2013Jan 16, 2014Biomet Manufacturing, LlcPre-Operative Planning And Manufacturing Method For Orthopedic ProcedureUS20140066937 *Sep 3, 2013Mar 6, 2014Smith & Nephew, Inc.Patient specific implant positioningUS20140081659 *Mar 14, 2013Mar 20, 2014Depuy Orthopaedics, Inc.Systems and methods for surgical and interventional planning, support, post-operative follow-up, and functional recovery trackingUS20140222157 *Jan 30, 2014Aug 7, 2014Conformis, Inc.Acquiring and Utilizing Kinematic Information for Patient-Adapted Implants, Tools and Surgical ProceduresUS20140257508 *Oct 12, 2012Sep 11, 2014Conformis, Inc.Methods and Systems for Identification, Assessment, Modeling and Repair of Anatomical Disparities in Joint ReplacementUS20140303938 *Jun 5, 2013Oct 9, 2014Biomet Manufacturing Corp.Integrated orthopedic planning and management processUS20140303990 *Oct 14, 2013Oct 9, 2014Biomet Manufacturing Corp.Integrated orthopedic planning and management processUS20140324058 *Jul 9, 2014Oct 30, 2014Biomet Manufacturing, LlcPatient-Specific Knee Alignment Guide And Associated Method* Cited by examinerNon-Patent CitationsReference1 *Mahfouz, Noninvasive Diagnostic System, Published 8/5/10, WO 2010/088696A1* Cited by examinerReferenced byCiting PatentFiling datePublication dateApplicantTitleUS8070752Jan 9, 2008Dec 6, 2011Biomet Manufacturing Corp.Patient specific alignment guide and inter-operative adjustmentUS8092465Jan 10, 2012Biomet Manufacturing Corp.Patient specific knee alignment guide and associated methodUS8133234Feb 20, 2009Mar 13, 2012Biomet Manufacturing Corp.Patient specific acetabular guide and methodUS8170641May 1, 2012Biomet Manufacturing Corp.Method of imaging an extremity of a patientUS8241293Feb 26, 2010Aug 14, 2012Biomet Manufacturing Corp.Patient specific high tibia osteotomyUS8265949Sep 11, 2012Depuy Products, Inc.Customized patient surgical planUS8282646Oct 9, 2012Biomet Manufacturing Corp.Patient specific knee alignment guide and associated methodUS8298237Oct 30, 2012Biomet Manufacturing Corp.Patient-specific alignment guide for multiple incisionsUS8343159Jan 1, 2013Depuy Products, Inc.Orthopaedic bone saw and method of use thereofUS8357111Jan 22, 2013Depuy Products, Inc.Method and system for designing patient-specific orthopaedic surgical instrumentsUS8357166Sep 29, 2008Jan 22, 2013Depuy Products, Inc.Customized patient-specific instrumentation and method for performing a bone re-cutUS8361076Jan 29, 2013Depuy Products, Inc.Patient-customizable device and system for performing an orthopaedic surgical procedureUS8377066Feb 19, 2013Biomet Manufacturing Corp.Patient-specific elbow guides and associated methodsUS8377068Sep 29, 2008Feb 19, 2013DePuy Synthes Products, LLC.Customized patient-specific instrumentation for use in orthopaedic surgical proceduresUS8398645Mar 19, 2013DePuy Synthes Products, LLCFemoral tibial customized patient-specific orthopaedic surgical instrumentationUS8398646Mar 19, 2013Biomet Manufacturing Corp.Patient-specific knee alignment guide and associated methodUS8407067Aug 31, 2010Mar 26, 2013Biomet Manufacturing Corp.Method and apparatus for manufacturing an implantUS8425523Apr 23, 2013DePuy Synthes Products, LLCCustomized patient-specific instrumentation for use in orthopaedic surgical proceduresUS8425524Apr 23, 2013DePuy Synthes Products, LLCCustomized patient-specific multi-cutting blocksUS8460302Jun 11, 2013Otismed CorporationArthroplasty devices and related methodsUS8460303Oct 25, 2007Jun 11, 2013Otismed CorporationArthroplasty systems and devices, and related methodsUS8473305Jun 12, 2009Jun 25, 2013Biomet Manufacturing Corp.Method and apparatus for manufacturing an implantUS8480679Apr 29, 2008Jul 9, 2013Otismed CorporationGeneration of a computerized bone model representative of a pre-degenerated state and useable in the design and manufacture of arthroplasty devicesUS8483469Oct 2, 2012Jul 9, 2013Otismed CorporationSystem and method for image segmentation in generating computer models of a joint to undergo arthroplastyUS8486150Apr 7, 2011Jul 16, 2013Biomet Manufacturing Corp.Patient-modified implantUS8500451Jan 13, 2008Aug 6, 2013Simbionix Ltd.Preoperative surgical simulationUS8532361Jan 25, 2012Sep 10, 2013Otismed CorporationSystem and method for image segmentation in generating computer models of a joint to undergo arthroplastyUS8532807Jun 6, 2011Sep 10, 2013Biomet Manufacturing, LlcPre-operative planning and manufacturing method for orthopedic procedureUS8535387Mar 7, 2011Sep 17, 2013Biomet Manufacturing, LlcPatient-specific tools and implantsUS8543338Mar 17, 2009Sep 24, 2013Simbionix Ltd.System and method for performing computerized simulations for image-guided procedures using a patient specific modelUS8568487Dec 23, 2010Oct 29, 2013Biomet Manufacturing, LlcPatient-specific hip joint devicesUS8591516Nov 29, 2010Nov 26, 2013Biomet Manufacturing, LlcPatient-specific orthopedic instrumentsUS8594395Sep 29, 2008Nov 26, 2013DePuy Synthes Products, LLCSystem and method for fabricating a customized patient-specific surgical instrumentUS8597365Aug 4, 2011Dec 3, 2013Biomet Manufacturing, LlcPatient-specific pelvic implants for acetabular reconstructionUS8603180May 19, 2011Dec 10, 2013Biomet Manufacturing, LlcPatient-specific acetabular alignment guidesUS8608748Sep 16, 2008Dec 17, 2013Biomet Manufacturing, LlcPatient specific guidesUS8608749Mar 7, 2011Dec 17, 2013Biomet Manufacturing, LlcPatient-specific acetabular guides and associated instrumentsUS8617171Apr 13, 2011Dec 31, 2013Otismed CorporationPreoperatively planning an arthroplasty procedure and generating a corresponding patient specific arthroplasty resection guideUS8617175Dec 14, 2009Dec 31, 2013Otismed CorporationUnicompartmental customized arthroplasty cutting jigs and methods of making the sameUS8632547May 12, 2011Jan 21, 2014Biomet Sports Medicine, LlcPatient-specific osteotomy devices and methodsUS8641721Jun 30, 2011Feb 4, 2014DePuy Synthes Products, LLCCustomized patient-specific orthopaedic pin guidesUS8668700Apr 29, 2011Mar 11, 2014Biomet Manufacturing, LlcPatient-specific convertible guidesUS8715289Apr 15, 2011May 6, 2014Biomet Manufacturing, LlcPatient-specific numerically controlled instrumentUS8715291Aug 24, 2009May 6, 2014Otismed CorporationArthroplasty system and related methodsUS8734455Feb 23, 2009May 27, 2014Otismed CorporationHip resurfacing surgical guide toolUS8735773Jun 10, 2011May 27, 2014Conformis, Inc.Implant device and method for manufactureUS8737700Apr 14, 2010May 27, 2014Otismed CorporationPreoperatively planning an arthroplasty procedure and generating a corresponding patient specific arthroplasty resection guideUS8764760Jul 1, 2011Jul 1, 2014Biomet Manufacturing, LlcPatient-specific bone-cutting guidance instruments and methodsUS8771365Jun 23, 2010Jul 8, 2014Conformis, Inc.Patient-adapted and improved orthopedic implants, designs, and related toolsUS8777875Jul 17, 2009Jul 15, 2014Otismed CorporationSystem and method for manufacturing arthroplasty jigs having improved mating accuracyUS8828087Aug 13, 2012Sep 9, 2014Biomet Manufacturing, LlcPatient-specific high tibia osteotomyUS8858561Jun 18, 2009Oct 14, 2014Blomet Manufacturing, LLCPatient-specific alignment guideUS8864769Mar 7, 2011Oct 21, 2014Biomet Manufacturing, LlcAlignment guides with patient-specific anchoring elementsUS8900244Jan 5, 2012Dec 2, 2014Biomet Manufacturing, LlcPatient-specific acetabular guide and methodUS8903530Sep 6, 2013Dec 2, 2014Biomet Manufacturing, LlcPre-operative planning and manufacturing method for orthopedic procedureUS8926706Nov 11, 2011Jan 6, 2015Conformis, Inc.Patient-adapted and improved articular implants, designs and related guide toolsUS8956364Aug 29, 2012Feb 17, 2015Biomet Manufacturing, LlcPatient-specific partial knee guides and other instrumentsUS8968320Jun 5, 2012Mar 3, 2015Otismed CorporationSystem and method for manufacturing arthroplasty jigsUS8974539Nov 11, 2011Mar 10, 2015Conformis, Inc.Patient-adapted and improved articular implants, designs and related guide toolsUS8979855Feb 23, 2011Mar 17, 2015DePuy Synthes Products, Inc.Customized patient-specific bone cutting blocksUS8979936Jun 21, 2013Mar 17, 2015Biomet Manufacturing, LlcPatient-modified implantUS9005297Jan 17, 2013Apr 14, 2015Biomet Manufacturing, LlcPatient-specific elbow guides and associated methodsUS9017336Jan 19, 2007Apr 28, 2015Otismed CorporationArthroplasty devices and related methodsUS9020788Feb 15, 2012Apr 28, 2015Conformis, Inc.Patient-adapted and improved articular implants, designs and related guide toolsUS9060788Dec 11, 2012Jun 23, 2015Biomet Manufacturing, LlcPatient-specific acetabular guide for anterior approachUS9066727Mar 3, 2011Jun 30, 2015Materialise NvPatient-specific computed tomography guidesUS9066734Aug 31, 2011Jun 30, 2015Biomet Manufacturing, LlcPatient-specific sacroiliac guides and associated methodsUS9084618Jun 11, 2012Jul 21, 2015Biomet Manufacturing, LlcDrill guides for confirming alignment of patient-specific alignment guidesUS9095355Jan 15, 2014Aug 4, 2015DePuy Synthes Products, Inc.Customized patient-specific orthopaedic pin guidesUS9113921Jan 25, 2011Aug 25, 2015Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9113971Sep 29, 2010Aug 25, 2015Biomet Manufacturing, LlcFemoral acetabular impingement guideUS9138239Feb 23, 2011Sep 22, 2015DePuy Synthes Products, Inc.Customized patient-specific tibial cutting blocksUS9173661Oct 1, 2009Nov 3, 2015Biomet Manufacturing, LlcPatient specific alignment guide with cutting surface and laser indicatorUS9173662Feb 23, 2011Nov 3, 2015DePuy Synthes Products, Inc.Customized patient-specific tibial cutting blocksUS9173666Jun 27, 2014Nov 3, 2015Biomet Manufacturing, LlcPatient-specific-bone-cutting guidance instruments and methodsUS9180015Mar 24, 2014Nov 10, 2015Conformis, Inc.Implants for altering wear patterns of articular surfacesUS9204977Mar 8, 2013Dec 8, 2015Biomet Manufacturing, LlcPatient-specific acetabular guide for anterior approachUS9208263Dec 31, 2012Dec 8, 2015Howmedica Osteonics CorporationSystem and method for image segmentation in generating computer models of a joint to undergo arthroplastyUS9220516Jan 20, 2011Dec 29, 2015Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9220517Jan 20, 2011Dec 29, 2015Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9237950Jan 31, 2013Jan 19, 2016Biomet Manufacturing, LlcImplant with patient-specific porous structureUS9241724Jan 25, 2011Jan 26, 2016Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9241725Jan 25, 2011Jan 26, 2016Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9241745Dec 13, 2012Jan 26, 2016Biomet Manufacturing, LlcPatient-specific femoral version guideUS9271744Apr 18, 2011Mar 1, 2016Biomet Manufacturing, LlcPatient-specific guide for partial acetabular socket replacementUS9289253Nov 3, 2010Mar 22, 2016Biomet Manufacturing, LlcPatient-specific shoulder guideUS9295481Jan 25, 2011Mar 29, 2016Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9295482Jun 17, 2011Mar 29, 2016Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9295497Dec 18, 2012Mar 29, 2016Biomet Manufacturing, LlcPatient-specific sacroiliac and pedicle guidesUS9301812Oct 17, 2012Apr 5, 2016Biomet Manufacturing, LlcMethods for patient-specific shoulder arthroplastyUS9308005Jan 26, 2011Apr 12, 2016Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9308053Jul 30, 2013Apr 12, 2016Conformis, Inc.Patient-specific joint arthroplasty devices for ligament repairUS9308091May 12, 2009Apr 12, 2016Conformis, Inc.Devices and methods for treatment of facet and other jointsUS9314251Mar 17, 2015Apr 19, 2016DePuy Synthes Products, Inc.Customized patient-specific bone cutting blocksUS9314256Jan 26, 2011Apr 19, 2016Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9317631Mar 11, 2013Apr 19, 2016DePuy Synthes Products, Inc.Surgical guide with cut resistant insertsUS9317634May 23, 2013Apr 19, 2016DePuy Synthes Products, Inc.Surgical guide with cut resistant insertsUS9320620Jul 9, 2013Apr 26, 2016Conformis, Inc.Patient-adapted and improved articular implants, designs and related guide toolsUS9339278Feb 21, 2012May 17, 2016Biomet Manufacturing, LlcPatient-specific acetabular guides and associated instrumentsUS9345548 *Dec 20, 2010May 24, 2016Biomet Manufacturing, LlcPatient-specific pre-operative planningUS9351743Oct 17, 2012May 31, 2016Biomet Manufacturing, LlcPatient-specific glenoid guidesUS9361410Mar 11, 2013Jun 7, 2016DePuy Synthes Products, Inc.Surgical guides from scanned implant dataUS9375222Jan 25, 2011Jun 28, 2016Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9381025Jan 26, 2011Jul 5, 2016Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9386993Sep 26, 2012Jul 12, 2016Biomet Manufacturing, LlcPatient-specific femoroacetabular impingement instruments and methodsUS9387079Oct 10, 2013Jul 12, 2016Conformis, Inc.Patient-adapted and improved articular implants, designs and related guide toolsUS9387083Jan 30, 2014Jul 12, 2016Conformis, Inc.Acquiring and utilizing kinematic information for patient-adapted implants, tools and surgical proceduresUS9393028Aug 10, 2010Jul 19, 2016Biomet Manufacturing, LlcDevice for the resection of bones, method for producing such a device, endoprosthesis suited for this purpose and method for producing such an endoprosthesisUS9402637Jan 24, 2013Aug 2, 2016Howmedica Osteonics CorporationCustomized arthroplasty cutting guides and surgical methods using the sameUS9408615Jan 25, 2011Aug 9, 2016Conformis, Inc.Patient selectable joint arthroplasty devices and surgical toolsUS9408616May 12, 2014Aug 9, 2016Biomet Manufacturing, LlcHumeral cut guideUS9408618Feb 23, 2009Aug 9, 2016Howmedica Osteonics CorporationTotal hip replacement surgical guide toolUS9408686Jan 22, 2013Aug 9, 2016Conformis, Inc.Devices, systems and methods for manufacturing orthopedic implantsUS9411939Mar 13, 2013Aug 9, 2016DePuy Synthes Products, Inc.Method for producing patient-specific plateUS9427320Nov 27, 2013Aug 30, 2016Biomet Manufacturing, LlcPatient-specific pelvic implants for acetabular reconstructionUS9439659Jun 29, 2015Sep 13, 2016Biomet Manufacturing, LlcPatient-specific sacroiliac guides and associated methodsUS9439767Oct 10, 2013Sep 13, 2016Conformis, Inc.Patient-adapted and improved articular implants, designs and related guide toolsUS20080147072 *Dec 18, 2006Jun 19, 2008Ilwhan ParkArthroplasty devices and related methodsUS20090177454 *Jul 9, 2009Ran BronsteinSystem and method for performing computerized simulations for image-guided procedures using a patient specific modelUS20100324692 *Aug 31, 2010Dec 23, 2010Biomet Manufacturing Corp.Method and Apparatus for Manufacturing an ImplantUS20110214279 *Sep 8, 2011Otismed CorporationPreoperatively planning an arthroplasty procedure and generating a corresponding patient specific arthroplasty resection guideUS20110224674 *Sep 15, 2011Biomet Manufacturing Corp.Patient-specific acetabular alignment guidesUS20110236868 *Mar 24, 2010Sep 29, 2011Ran BronsteinSystem and method for performing a computerized simulation of a medical procedureUS20120197619 *Jan 27, 2011Aug 2, 2012Einav Namer YelinSystem and method for generating a patient-specific digital image-based model of an anatomical structureUS20130066319 *Feb 23, 2011Mar 14, 2013Luke J. AramMethod of fabricating customized patient-specific bone cutting blocksUS20140189508 *Dec 31, 2013Jul 3, 2014Mako Surgical Corp.Systems and methods for guiding a user during surgical planningUS20140259629 *Mar 17, 2014Sep 18, 2014Conformis, Inc.Devices, methods and systems for forming implant componentsUS20140303938 *Jun 5, 2013Oct 9, 2014Biomet Manufacturing Corp.Integrated orthopedic planning and management processUS20140303990 *Oct 14, 2013Oct 9, 2014Biomet Manufacturing Corp.Integrated orthopedic planning and management processUS20150223941 *Aug 27, 2013Aug 13, 2015Conformis, Inc.Methods, Devices and Techniques for Improved Placement and Fixation of Shoulder Implant ComponentsUSD691719Jun 22, 2011Oct 15, 2013Otismed CorporationArthroplasty jig blankWO2013013170A1 *Jul 20, 2012Jan 24, 2013Smith & Nephew, Inc.Systems and methods for optimizing fit of an implant to anatomyWO2013033205A1Aug 29, 2012Mar 7, 2013Biomet Manufacturing Corp.Patient-specific sacroiliac guides and associated methodsWO2013165558A1 *Mar 11, 2013Nov 7, 2013DePuy Synthes Products, LLCSurgical guides from scanned implant dataWO2013165883A1 *Apr 29, 2013Nov 7, 2013Siemens Product Lifecycle Management Software Inc.Feature-driven rule-based framework for orthopedic surgical planningWO2013177334A1 *May 22, 2013Nov 28, 2013Mako Surgical Corp.Soft tissue cutting instrument and method of useWO2014070889A1Oct 30, 2013May 8, 2014Biomet Manufacturing, LlcPatient-specific sacroiliac and pedicle guidesWO2015102964A1 *Dec 19, 2014Jul 9, 2015Mako Surgical Corp.Systems and methods for generating customized haptic boundaries* Cited by examinerClassifications U.S. Classification600/416, 600/425, 382/131International ClassificationA61B5/055, G06K9/00, A61B6/03Cooperative ClassificationA61F2002/4687, A61B2034/108, A61F2/4607, A61B34/10, A61B2034/105, A61B2034/252, A61F2250/0097, A61F2002/4668, A61F2002/4662, A61F2002/4633, A61F2002/30617, A61B17/152, A61B17/151European ClassificationA61B17/15C, A61B17/15C2, A61B19/50Legal EventsDateCodeEventDescriptionDec 18, 2013ASAssignmentOwner name: BIOMET MANUFACTURING CORPORATION, INDIANAFree format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:SCHOENEFELD, RYAN;UTHGENANNT, BRIAN;MORRISON, BRYAN;AND OTHERS;SIGNING DATES FROM 20101202 TO 20101221;REEL/FRAME:031804/0729Owner name: BIOMET MANUFACTURING, LLC, INDIANAFree format text: CHANGE OF NAME;ASSIGNOR:BIOMET MANUFACTURING CORPORATION;REEL/FRAME:031805/0378Effective date: 20130603RotateOriginal ImageGoogle Home - Sitemap - USPTO Bulk Downloads - Privacy Policy - Terms of Service - About Google Patents - Send FeedbackData provided by IFI CLAIMS Patent Services