Source: http://www.google.com/patents/US7753856?dq=5,941,947
Timestamp: 2014-09-22 11:24:11
Document Index: 491218379

Matched Legal Cases: ['Application No. 60', 'Application No. 05077367', 'Application No. 60', 'Application No. 60', 'Application No. 60', 'Application No. 60', 'Application No. 60', 'Application No. 05077368']

Patent US7753856 - Method, device and system for cardio-acoustic signal analysis - Google PatentsSearch Images Maps Play YouTube News Gmail Drive More »Sign in<nobr>Advanced Patent Search</nobr>PatentsA method for cardio-acoustic signal analysis includes receiving a signal representative of heart sounds and displaying the signal in a time-perceptual frequency-perceptual loudness domain representation. The received signal is transformed to represent a time-perceptual frequency-amplitude domain. The...http://www.google.com/patents/US7753856?utm_source=gb-gplus-sharePatent US7753856 - Method, device and system for cardio-acoustic signal analysisAdvanced Patent SearchPublication numberUS7753856 B2Publication typeGrantApplication numberUS 12/102,470Publication dateJul 13, 2010Filing dateApr 14, 2008Priority dateOct 14, 2005Fee statusPaidAlso published asEP1952291A1, US20080260173, US20090171227, WO2007043903A1Publication number102470, 12102470, US 7753856 B2, US 7753856B2, US-B2-7753856, US7753856 B2, US7753856B2InventorsMarek DziubinskiOriginal AssigneeMedicalgorithmics Ltd.Export CitationBiBTeX, EndNote, RefManPatent Citations (102), Non-Patent Citations (11), Referenced by (2), Classifications (11), Legal Events (1) External Links: USPTO, USPTO Assignment, EspacenetMethod, device and system for cardio-acoustic signal analysisUS 7753856 B2Abstract A method for cardio-acoustic signal analysis includes receiving a signal representative of heart sounds and displaying the signal in a time-perceptual frequency-perceptual loudness domain representation. The received signal is transformed to represent a time-perceptual frequency-amplitude domain. The method further includes applying a human auditory modeling algorithm to the time-perceptual frequency-amplitude domain representation to generate the time-perceptual frequency-perceptual loudness domain representation.
1. A method for cardio-acoustic signal analysis, comprising:
receiving, via a signal receiver, a signal representative of heart sounds;
transforming, via a processing unit, the received signal to represent a time, perceptual-frequency, amplitude domain;
applying, via the processing unit, a human auditory system modeling algorithm to the time, perceptual frequency, amplitude domain representation to generate a time, perceptual-frequency, perceptual-loudness domain representation; and
displaying, via a visualization unit, the received signal in the time, perceptual-frequency, perceptual-loudness domain representation.
2. The method of claim 1, wherein the transformation is based on inner product calculations using a finite impulse response (FIR) filter bank representing a non-linear frequency resolution.
3. The method of claim 2, wherein the finite impulse response filter bank is complex numbered and represents properties of perceptual-frequency characteristics.
4. The method of claim 3, wherein the perceptual-frequency characteristics correspond to critical bands of hearing and represent a Bark frequency scale.
5. The method of claim 3, wherein the complex numbered impulse responses filter bank allow for calculating phase prediction and amplitude prediction of spectral components.
6. The method of claim 5, further comprising calculating, via the processing unit, tonality using an unpredictability measure parameter.
7. The method of claim 6, further comprising distinguishing, via the processing unit, between noisy and tonal components in the received signal.
8. The method of claim 7, further comprising simulating, via the processing unit, periodicity perception of the received signal.
9. The method of claim 1, further comprising interpreting, via the processing unit, the time, perceptual-frequency, perceptual-loudness domain representation for cardio-acoustic signal analysis.
10. The method of claim 9, wherein interpreting the time-perceptual frequency-perceptual loudness domain representation includes comparing, via the processing unit, the time, perceptual-frequency, perceptual-loudness domain representation to a perceived acoustical phenomena.
11. The method of claim 1, wherein the received signal is representative of at least cardiac auscultation and pulmonary auscultation.
12. The method of claim 1, further comprising applying, via the processing unit, periodicity analysis methods to the time, perceptual-frequency, perceptual-loudness domain representation.
13. The method of claim 12, further comprising detecting, via the processing unit, S1 and S2 of at least 2 cardiac cycles for calculating an averaged cardiac cycle representation.
14. The method of claim 13, further comprising limiting, via the processing unit, influence of noise and other non-repetitive parasite artifacts and disturbances to the received signal.
15. The method of claim 13, further comprising restoring, via the processing unit, the cardiac auscultation signal to remove noise, parasite clicks, or any combination thereof.
16. The method of claim 1, wherein the applying the human auditory system modeling algorithm simulates psychoacoustic masking phenomena.
17. A system for cardio-acoustic signal analysis, comprising:
an input sensor for receiving a signal representative of heart sounds;
a transformation module for transforming the receive signal to represent a time, perceptual-frequency, perceptual-loudness domain; and
a display device for displaying the received signal in the time, perceptual-frequency, perceptual-loudness domain.
18. A system for cardio-acoustic signal analysis, comprising:
means for receiving a signal representative of heart sounds;
means for transforming the received signal to represent a time, perceptual-frequency, perceptual-loudness domain; and
means for displaying the received signal in a time, perceptual-frequency, perceptual-loudness domain representation.
RELATED APPLICATIONS This application claims the benefit of U.S. Provisional Application No. 60/956,782, filed on Aug. 20, 2007 and is a continuation of International Application No. PCT/PL2006/000067 filed on Oct. 16, 2006, published in English, which claims priority under 35 U.S.C. �119 or 365 to EP Application No. 05077367.0, filed Oct. 14, 2005, the entire teachings of the above applications are incorporated herein by reference. This application is a continuation-in-part of U.S. application Ser. No. 12/090,183, filed on Apr. 14, 2008, which claims the benefit of U.S. Provisional Application No. 60/987,192, filed on Nov. 12, 2007, U.S. Provisional Application No. 60/987,180, filed on Nov. 12, 2007, U.S. Provisional Application No. 60/987,043, filed on Nov. 10, 2007, U.S. Provisional Application No. 60/986,761, filed on Nov. 9, 2007, Polish Patent Application No. P 383243, filed on Sep. 2, 2007, entitled �A system for remote cardiac rehabilitation� (English translation), U.S. Provisional Application No. 60/948,527, filed on Jul. 9, 2007, and is a continuation of International Application No. PCT/PL2006/000068, filed on Oct. 16, 2006, published in English, which claims priority under 35 U.S.C. �119 or 365 to EP Application No. 05077368.8. filed Oct. 14, 2005, the entire teachings of the above applications are incorporated herein by reference.
BACKGROUND Heart (cardiac) auscultation is a very old technique, which is still commonly used during medical examinations carried out by physicians. It is performed using a standard non-electronic stethoscope, an electronic analogue stethoscope or a modern digital stethoscope. Cardiac auscultation is based on auditory perception of sounds produced by the heart in certain areas of the human body. These sounds are often accompanied by special kind of maneuvers which enhance some significant sounds, clicks or murmurs produced by the heart muscle and the blood flow across heart valves. Cardiac auscultation is difficult to perform properly due to the limitations of human hearing.
A cardiac auscultation examination requires nearly perfect hearing and extensive experience of a physician in dealing with rare cases of heart abnormalities. However, while experience increases with age, hearing deteriorates. Furthermore, acoustic signals have very (low) small amplitude and a heart's activity is usually represented by very low frequencies which remain on the edge of human perception. Also, pathologies detectable via cardiac auscultation are rare, and therefore physicians are not able to compare various cases representative of a given pathology thereby making diagnosis even more difficult. That is, an ordinary physician usually does not have enough experience to determine the audible phenomena, e.g. whether the extra sounds, murmurs or clicks are pathological, or not.
Additionally, interpretation of the results of an auscultation exam is very subjective, because different doctors have different experience and hearing abilities. Cardiac auscultation also requires very good hearing, which is not very common among human population (while it is necessary to perform cardiac auscultation effectively). Since hearing and experience of each human being differ, result of the auscultation is often very subjective matter. Hence, in many cases Cardiac auscultation is not treated seriously among medical community despite its potential. For this reason, despite its potential, cardiac auscultation is treated as a preliminary overview of the patient's heart condition, rather than a source of reliable information for potential treatment.
SUMMARY Phonocardiography is the best known technology, which aids physicians in dealing with heart sounds. It is based on visualization of the time domain acoustic cardiac signal. The general idea of this technology has been adopted from the ECG analysis, where the time domain electrophysiological waveform is analyzed by a medical specialist. However such approach is not appropriate for acoustic signals, because listening to the signal differs from viewing the time domain waveform, especially since acoustic events may happen simultaneously at different frequencies. Physicians have tried to see in the waveform what should be heard (since the extensive cardiac auscultation knowledge, gathered over nearly 200 years, describes the acoustical phenomena), which turned out to be not very effective approach. For this reason phonocardiography was and still is rejected by the medical community.
Phonocardiography may be improved by multi-band analysis, where several waveforms related to specific sub-bands are filtered out and often processed in a non-linear fashion. Such improvements allow physicians to identify acoustic events related to different frequencies. However, this approach is still insufficient due to nonlinear complexity of sound perception.
Alternative methods of cardiac auscultation signal processing are based on simulating human auditory system. In those methods the time domain waveform is represented as a 3 dimensional surface (digital image�time-frequency-amplitude map) where changes of the amplitude over time, for certain frequencies can be observed. The human auditory system functions in a similar manner, i.e. a listener detects energy of certain acoustic events and is able to determine their frequency and the moment of their occurrence. For example, one can notice that at certain time high frequency click occurred and after some time low frequency-clatter was present, etc.
However, these methods do not accurately represent the nonlinear character of the human auditory system and therefore do not properly show what should be heard (but still are closer to simulating human hearing than phonocardiography). It is due to the fact that human auditory system operates with specific frequency resolution, which is very high for lower frequencies and decreases with the increase of the sounds frequency.
Additionally, some acoustic events are inaudible, when they occur in direct presence (in time and/or frequency) of louder events. This phenomenon is known as perceptual masking and is used for example in MPEG audio coding standard (i.e. in the popular MP3). Furthermore, loudness of certain acoustic events is not directly related to their amplitude (energy), i.e. some frequencies are heard better than others and consequently some cannot be heard at all (usually sounds below 20 Hz and above 20 kHz.
In general the methods described above are based on mathematical signal transformations, which convert the time domain waveform into frequency domain. They are usually based on methods such as DFT (Discrete Fourier Transform), DWT (Discrete Wavelet Transform�wavelet spectrum), CWT (Complex Wavelet Transform) and several other techniques with some nonlinear enhancements, which however, do not take into account complexity of the human auditory system.
An alternative technology (alternative examination) to cardiac auscultation is so called heart ECHO (USG or ultrasound of the heart). This method enables a physician to determine a valves performance and allows for detecting other pathologies of the heart muscle. This examination, however, is expensive and requires highly trained medical personnel. Therefore, providing a reliable alternative to standard cardiac auscultation could be beneficial for a patient and is economically justified since allows for avoiding unnecessary and expensive ECHO.
Further, an electrocardiogram (ECG) is a graphic produced by an electrocardiograph, which records the electrical activity of the heart over time. Analysis of the various waves and normal vectors of depolarization and repolarization yields important diagnostic information. The ECG does not allow a physician to detect heart valve pathologies because these pathologies do not correspond with electrical activity of the heart muscle.
There is provided a method and system for performing cardio-acoustic signal analysis. The method includes receiving a signal representative of heart sounds and displaying the signal in a time-perceptual frequency-perceptual loudness domain representation. The method further can include transforming the received signal to represent a time-perceptual frequency-amplitude domain.
In some embodiments, the method can further include applying a human auditory system modeling algorithm to the time-perceptual frequency-amplitude domain representation to generate a time-perceptual frequency-perceptual loudness domain representation.
In some embodiments, the transformation can be based on inner products calculations using at least complex numbered impulse response filter banks representing a non-linear frequency resolution character. The complex numbered impulse response can represent properties of perceptual frequency characteristics. The perceptual frequency characteristics can include at least Bark units and mel scale. The complex numbered impulse response can allow for calculating phase (and amplitude) prediction of the representation components.
In some embodiments, the method can include calculating tonality using an unpredictability measure parameter. The method can also include distinguishing between noisy and tonal components in the signal. The method can further simulate periodicity perception.
In some embodiments, the method can include interpreting the time-perceptual frequency-perceptual loudness domain representation. The interpreting the time-perceptual frequency-perceptual loudness domain representation can include comparing the time-perceptual frequency-perceptual loudness domain representation to a perceived acoustical phenomena.
In some embodiments, the signal can be representative of at least cardiac auscultation and pulmonary auscultation. The method can further include applying periodicity analysis methods to the time-perceptual frequency-perceptual loudness domain representation. The method can further detect S1 and S2 of at least 2 cardiac cycles for calculating an averaged representation. The method can also limit influence of noise and other non-repetitive parasite artifacts and disturbances to the signal. The method can restore the cardiac auscultation signal.
A system for cardio-acoustic signal analysis includes an input sensor for receiving a signal representative of heart sounds and a display device for displaying the signal in a time-perceptual frequency-perceptual loudness domain representation. The system can further include a transformation module for transforming the received signal to represent a time-perceptual frequency-amplitude domain.
The system and methods presented herein provide the following advantages. These include, but are not limited to, visualization of the signal exactly as it should be heard by an ideal listener; allows a physician to utilize knowledge directly related to cardiac auscultation gathered for last 200 years by thousands of physicians; objectified analysis, i.e., the same signal always will result in the same representation and interpretation, which is not always possible if the same signal is heard by different people (since they may have different hearing abilities and listening experience); and objectified analysis of the artificial valve's performance where ECHO performs poorly and is not very reliable.
BRIEF DESCRIPTION OF THE DRAWINGS The foregoing and other objects, features and advantages will be apparent from the following more particular description of the embodiments, as illustrated in the accompanying drawings in which like reference characters refer to the same parts throughout the different views. The drawings are not necessarily to scale, emphasis instead being placed upon illustrating the principles of the embodiments.
FIG. 1 shows graphical representations of acoustical phenomena that can be heard during cardiac auscultation;
FIG. 2 is a schematic illustration of a system for performing cardio-acoustic signal analysis;
FIG. 3 is a schematic illustration of a device for performing cardio-acoustic signal analysis;
FIG. 4 is a schematic illustration of another system embodiment for performing cardio-acoustic signal analysis;
FIG. 5 represents a standard spectrogram;
FIG. 6 represents a time-perceptual frequency�perceptual loudness image;
FIG. 7 represents a loudness contour of a chirp signal;
FIG. 8 represents a phonocardiographic representation (time domain waveform) of a cardiac auscultation signal with a holosystolic murmur;
FIG. 9 represents a psychoacoustic time-perceptual frequency-perceptual loudness of the same signal of FIG. 8;
FIG. 10 shows an example of a time-perceptual frequency-perceptual loudness domain representation of a cardio-acoustic signal;
FIG. 11 represents a stimulation of an ability of a human listener, with the use of the same signal of FIG. 10; and
FIG. 12 illustrates an example of auscultatory findings detected by a transformation algorithm.
DESCRIPTION In general, psychoacoustics (human auditory system modeling) is based on humans perceive sounds in terms of their amplitudes, frequencies, and time localization. Sounds of the same amplitudes, but different frequencies may be perceived as differently loud. Amplitude differences between sounds are frequency dependent, i.e. loudness relation between sounds of different frequencies varies with accordance to their energy. Quiet sounds may be masked (made inaudible) by louder sounds (maskers). This phenomenon depends on time and frequency distance between masked sounds and maskers. Perception of frequencies is non-linear, i.e. resolution of human auditory system is much higher for lower sounds, than for higher sounds. Noisy (chaotic) and periodic (tonal) components of the sounds are perceived differently.
Humans, while listening to the heart auscultation sounds, are able to focus on repetitive events, while some random events may be �separated� and disregarded. This innovation is based on a method of transforming the time domain waveform into the time�perceptual frequency�perceptual loudness domain. Such representation is obtained with regard to the mathematical model of the human auditory system.
Additionally, a set of algorithms responsible for simulating periodicity perception (pattern tracking) of acoustic events, i.e. humans are able to concentrate their attention on repetitive events (rhythmic events such as heart beat), rejecting incidental events, which usually are parasite disturbances.
Further, based on a periodicity analysis (pattern tracking and pattern tracking recovery); the auscultation signal may be significantly enhanced. Disturbing noise and parasite impulsive clicks of non-repetitive type may be removed, resulting in restored time domain representation of the auscultation sound.
FIG. 1 shows time domain graphical representations of acoustical phenomena that can be heard during cardiac auscultation. A standard cardiac stethoscope examination enables a doctor/physician to detect heart dysfunctions related to blood flow through the heart, such as heart valves dysfunctions. For example, valves leakiness (i.e. regurgitation) or improper (not complete) opening of the valve (i.e. stenosis). Cardiac auscultation is also commonly used to monitor the performance of artificial (implanted) mechanical valves, which tend to seal after some time and may require replacement. Also, presence of the extra sounds (S3 and S4) may be useful in detecting hemodynamic evidence of heart failure.
FIG. 2 is a schematic illustration of a system 100 for performing cardio-acoustic signal analysis. The system includes a digital input acoustic sensor 110, a psychoacoustic spectral transformation module 120, a Basiliar membrane excitation simulation module 130, and a psychoacoustic representation matrix module 140. Signal �S� represents a digital input acoustic sensor signal representative of heart sounds (time domain waveform), which is stored in an appropriate matrix and used as an input to the digital input acoustic sensor 110. The digital input acoustic sensor 100 signal is subject to segmentation, i.e. the digital input acoustic sensor signal is divided into shorter blocks of equal length. A matrix of the time domain short blocks becomes an input signal to the psychoacoustic spectral transformation module 120 where each time domain segment is transformed into a psychoacoustic spectral representation. Such representation is calculated with the use of inner products operations with regard to Bark units, i.e. frequency domain is represented by equally spaced spectral bins, however equal spacing refers to Bark units, thus such approach significantly differs from standard FFT, DWT, CWT, etc. The resulting representation vectors (each representing the input time domain blocks) is stored in an appropriate matrix and becomes input of to the Basiliar membrane excitation simulation module 130, which is responsible for simulating Basiliar membrane excitation and simultaneous masking phenomenon, i.e. total excitation is constructed of precise modeling of particular the membrane excitations.
A matrix constructed with the calculated Basiliar membrane deformation vectors becomes input to the psychoacoustic representation matrix module 140. The psychoacoustic representation matrix module 140 is responsible for simulating time domain Basiliar membrane behavior (time domain masking phenomenon), i.e. this method models the shape changes of the membrane over time. An output signal �P� from the psychoacoustic representation matrix module 140 is a matrix built from Basiliar membrane shapes, thus P is a psychoacoustic representation (in the form of the 3D map) in time-perceptual frequency-perceptual loudness domain.
FIG. 3 is a schematic illustration of a device 200 for performing cardio-acoustic signal analysis, where the device 200 may be for example incorporated into a microprocessor module 210. The microprocessor module includes a signal storage and pre-processing module 220, a psychoacoustic representation module 230, a detector 240, a cardiac auscultation module 250, and a cardiac events detection module 260.
The signal storage and pre-processing module 220 receives a digital input acoustic sensor signal �S� representative of heart sounds (in time domain waveform). The signal storage and pre-processing module 220 is responsible for selecting a fragment of the input signal of 10 seconds duration. The chosen fragment is the most periodic part of the input signal, while periodicity indicator is obtained with the use of cross correlation algorithm applied for RMS envelope of the digital input acoustic sensor signal �S�.
The psychoacoustic representation module 230 is responsible for calculating psychoacoustic representation (in time-perceptual frequency-perceptual loudness domain), as described with reference to FIG. 2. The output signal from the psychoacoustic representation module 230 is further processed in the cardiac auscultation module 250. The cardiac auscultation module 250 is a cardiac auscultation expert system (where parameterization of the examination information takes place).
Simultaneously, the periodicity detector 240 (simulating human perception of repetitive events versus incidental events), is responsible for processing of examination information (i.e. body position, auscultation area, inhalation/exhalation).
Such information is considered in determining significance of certain cardiac events, such as clicks and murmurs, which may be increased or decreased, depending on the body position, exhalation or inhalation or may not be present for certain auscultation areas. Output signals from the cardiac auscultation module 250 and the periodicity detector 240 are in the cardiac events detection module 260. After such processing, an output signal �V� of the cardiac events detection module 260 represents final results in the form of ascultatory findings.
FIG. 4 is a schematic illustration of another system embodiment 300 for performing cardio-acoustic signal analysis. The system 300 includes a microphone 310, a analog-to-digital (A/D) converter 320, a signal radio/wire transmitter 330, a processing unit 340, and a visualization unit VD 350.
The microphone 310 is responsible for converting acoustic pressure changes into voltage changes. The A/D converter 320 converts a analogue (voltage changes signal) into an equally spaced digital signal. The signal radio/wire transmitter 330 is responsible for sending the digital signal to the processing unit (e.g. a computer) 340. In some embodiments, transmission can be carried out via a USB cable, a wireless transmitter, or the like. In some embodiments, the processing unit 340 can be a personal computer, a personal digital assistant, a device based digital signal processor chip, or the like. In some embodiments, the visualization unit 350 can be a standard monitor, an LCD display, or the like and is responsible for enabling data presentation to a user.
The processing unit 340 includes a signal receiver 342, a data storage device 344, and a microprocessor 346. In some embodiments, the signal receiver 342 can be a USB adapter for the use with a USB connection, or a radio receiver for the use with a radio transmission. The microprocessor 346 is described in with reference to FIG. 2. In some embodiments, the data storage device 344 can be a hard drive, random access memory, or the like.
FIG. 5 represents a standard spectrogram, i.e. time-frequency-amplitude map based on a short term DFT, where the signal is divided into short segments and Fourier spectra are calculated for all of the segments, forming columns of the image (matrix). FIG. 5 shows a sinusoidal signal spectrogram with linearly increasing frequency and constant amplitude (a �chirp signal�). The signal frequency evolves over 5 seconds from 0 Hz at the beginning to 22050 Hz at the end (fifth second). The black color of the map (image) represents minimum energy, while the white color represents maximum energy. The chirp signal is the white line starting from the bottom left corner of the image. The constant amplitude over time (and frequency) of the chirp signal is indicated by a constant white color.
FIG. 6 represents a time-perceptual frequency-perceptual loudness image. The aforementioned embodiments utilize a transformation algorithm that transforms the time domain waveform of FIG. 5 into a time-perceptual frequency-loudness domain as shown in FIG. 6. It should be understood that the transformation algorithm is not based on the short-term DFT, DWT or any other known transformation technique, but utilizes inner products calculation of the segmented signal and complex FIR filter bank impulse responses (IRs), directly representing the Bark scale and critical bands properties. It should also be understood that any non-linear (e.g. �mel� scale) or linear scale can be utilized by the transformation algorithm during the filter-bank construction.
The Bark scale, particularly obtained with regard to Eq. 1 (shown below) has been found to be the closest to the human pitch perception and human auditory system frequency resolution, thus has been exploited in the implementation. The Bark scale is a psychoacoustic scale proposed by Eberhard Zwicker in 1961. It is named after Heinrich Barkhausen who introduced the first subjective measurements of loudness. The scale ranges from 1 to 24 and corresponds to the first 24 critical bands of hearing. The subsequent band edges are (in Hz) 20, 100, 200, 300, 400, 510, 630, 770, 920, 1080, 1270, 1480, 1720, 2000, 2320, 2700, 3150, 3700, 4400, 5300, 6400, 7700, 9500, 12000, 15500.
Typical modeling of Basilar membrane deformation, i.e., modeling psychoacoustic phenomena, is based on a discrete Fourier spectrum with equally spaced frequencies (in Hz) and energy represented by the spectrum bins. Such a model has several limitations, because the spectrum resolution does not represent the resolution of the human hearing. Therefore, it is necessary to chose an appropriate spectrum order (reflecting time vs. frequency resolution), which is achieved by manipulating signal block lengths.
Increasing signal block lengths results in increased frequency resolution (more spectrum bins represent the entire signal frequency range, limited by the Nyquist frequency), but it decreases time domain resolution. For the lower frequency bands, it is desirable to have high frequency domain resolution because humans perceive low frequency sounds with increased resolutions, while high frequency sounds should be represented by decreased frequency domain resolution. Also, the time resolution for different frequencies is not constant, i.e., lower sounds are perceived with lower time domain resolution, and higher pitched sounds are perceived with increased time domain resolution which corresponds with the time domain masking phenomena. It is clear, that using Fourier spectra has several limitations and does not allow for simulating the Basilar membrane deformation changes over time very accurately.
To overcome these limitations, the transformation algorithm is based on a complex numbered filter-bank impulse response directly related to human auditory system resolution. Time domain properties of the impulse responses depend on their center frequencies and bandwidth, i.e., the time domain resolution increases with the increase of the center frequencies of the filters and resembles the resolution of the human auditory system. In order to calculate center frequencies and bandwidth of the filters, the frequency scale is divided into N equally spaced (in Bark units) segments. Recalculation from Hz to Barks is based on the following empirical formula:
z = { 26.81 � f 1960 + f - 0.53 for ⁢ ⁢ z < 2 , z = z + 0.15 � ( 2 - z ) for ⁢ ⁢ z > 20.1 , z = z + 0.22 � ( z - 20.1 ) ( 1 ) Where:
z�critical band rate, f�frequency in Hz. The nth filter bandwidth ranges from Fz[n−1] to Fz[n+1], where Fz represents the frequency (in Hz) of the nth critical band frequency scale segment. The filter IRs can be obtained with any linear phase FIR filter design technique (e.g., with a window-based finite impulse response filter design method, as it is done in the implementation). In addition, each impulse response (k) can be weighted by a width controlled windowing function, such as Gaussian window. Such operation allows for additional control of the time domain performance of each transformation sub-band. The set of impulse responses forms matrix K:
K re = k n m � k n M ⋮ ⋱ ⋮ k N m � k N M ( 2 ) where:
M�is the IRs length, equal to processed signal segments length, N�number of sub-bands (frequency resolution of the representation) Kim�imaginary part of the matrix Kre is obtained with the use of Hilbert transform and both Kre and Kim are used to form complex impulse responses matrix K. The complex representation, with perceptually spaced frequency domain (i.e. equally spaced in Bark units), is calculated with the inner product operations:
P t =K�s t (3)
Pt�the invented complex representation at instance t (for tth processing block) K�complex filter impulse response matrix, st�tth input signal segment. Outer-middle ear frequency characteristics are simulated with accordance to the following formula and are used to weight each representation band:
A=−0.6�3.64�f −0.8+6.5�e −0.6�(f−3.3) 2 −10−3 �f 4 (4)
Based on the representation P (calculated with Eq. 3) it is possible to simulate the simultaneous masking phenomena and calculate Basilar membrane deformation curve. The simulation is performed similarly as with the use of complex Fourier spectrum. In the first step, tonality descriptors of the representation components are calculated. The tonality is represented by the Unpredictability Measure (UM) parameter and further used for calculating the masking offset. The offset for the excitation of bx Barks at frequency of bx Barks is given by the formula:
O k,x=αk t�(14.5+bark(x))+(1−αk t)�5.5 (5)
The tonality index αk t of excitation of bx Barks is represented by the UM parameter (αk t=ck t), where ck t is calculated in the following way:
c k t = ( r k t � cos ⁢ ⁢ Φ k t - r ^ k t � cos ⁢ ⁢ Φ ^ k t ) 2 + ( r k t � sin ⁢ ⁢ Φ k t - r ^ k t � sin ⁢ ⁢ Φ ^ k t ) 2 r k t -  r ^ k t  ( 6 ) For rk t denoting spectral magnitude and Φk t denoting phase, both at time t, while {circumflex over (r)}k t and {circumflex over (Φ)}k t represent the predicted values of Φk t, and are referred to the past information (calculated for two previous signal segments):
α k t = { r ^ k t = r k t - 1 + ( r k t - 1 - r k t - 2 ) Φ k t = Φ k t - 1 + ( Φ k t - 1 - Φ k t - 2 ) ⇒ { r ^ k t = 2 � r k t - 1 - r k t - 2 Φ k t = 2 � Φ k t - 1 - Φ k t - 2 ( 7 ) Phase prediction is possible due to the complex numbered character of the representation (P from Eq. 3). The masking threshold T of the Basilar membrane, stimulated by a single tone excitation of bx Barks of magnitude equal to Sx is calculated with regard to the formula:
{ T i , x = P i � 10 - P 1 � ( b x - b i ) / 10 - O i , x , b x ≤ b i T j , x = P j � 10 - P 2 � ( b j - b x ) / 10 - O ji , x , b x > b j ( 8 ) And the global masking threshold at bx Barks is:
T x = ∑ j ∈ ⁢ J ⁢ T j , x + ∑ i ∈ ⁢ I ⁢ T i , x ( 9 ) FIG. 6 shows a representation of the same signal of FIG. 5, however with the use of the transformation algorithm (Eq. 3) and perceptual masking modeling (Eq. 4-9). As shown in FIG. 6, the perceptual frequency of the chirp signal increases rapidly for lower frequencies and the increase slows down for higher perceptual frequency band. Also, it can be observed that the maximum loudness level is reached at 0.7 s (for 17 barks, i.e. around 2 kHz), which is indicated by white color, while below 1 Bark and above 24 Barks loudness is near zero. The loudness curve L (without the frequency domain properties) is achieved, by averaging energy (RMS) of each column of the psychoacoustic representation (matrix) T:
L m = 1 N ⁢ ∑ n = 1 N ⁢ T n , m 2 ( 10 ) Where N denotes number of subbands (see also Eq. 2).
FIG. 7 represents a loudness contour of the chirp signal. It can be observed that it reaches its maximum value at 0.7 seconds, which corresponds to image in of FIG. 6.
Additionally, the transformation algorithm contains a set of auxiliary algorithms for automated analysis of the psychoacoustic representation. The algorithms are responsible for simulating periodicity perception of the acoustic events, since humans have ability to concentrate their attention on repetitive events (rhythmic events such as heart beats, murmurs, etc.) and rejecting incidental events, which usually are parasite disturbances. The periodicity analysis can be utilized to detect cardiac cycle periods based on the T representation, however for reducing the computational complexity it is possible to utilize the loudness curve L only, or combine both (i.e. periodicity detection with L and periods averaging in T). The periods lengths are obtained by comparing segments of the representation with other segments offset by a trial period to find a match, similarly as it is done in the pitch detection algorithms operating in the time domain. Average magnitude difference function (AMDF) or the similar autocorrelation are used for this purpose:
α t ⁡ ( t ) = ∑ n = - N / 2 N / 2 - 1 ⁢ x ⁡ ( t + n ) � x ⁡ ( t + n - P ) ∑ n = - N / 2 N / 2 - 1 ⁢ x 2 ⁡ ( t + n ) � ∑ n = N / 2 N / 2 - 1 ⁢ x 2 ⁡ ( t + n + P ) ( 11 ) AMDF:
γ t ⁡ ( t ) = ∑ n = N / 2 N / 2 - 1 ⁢  x ⁡ ( n + m ) � w ⁡ ( m ) - x ⁡ ( n + m - t ) � w ⁡ ( m - t )  ( 12 ) where:
α�autocorrelation signal, γ�AMDF signal, w�weighting window function (e.g. Hann's, Kaiser's, etc.) In case of periodicity analysis of the time-perceptual frequency-loudness representation, above algorithms (Eq. 11-12) are applied for each row of the matrix T independently.
The transformation algorithm allows for calculating psychoacoustic auscultation sound representation, which is visualization of the basilar membrane deformation changes over time (tth column of matrix T, representing tth signal block is the basilar membrane deformation at time instant t). Further, the auscultation signal findings are described in terms of perception of the acoustical phenomena. Therefore, audible events can be observed in a psychoacoustic representation image, and inaudible events cannot (which is not always the case in phonocardiographic, or other non-psychoacoustic representations), thereby allowing for utilizing the representation for supporting physicians during the auscultation procedure. The transformation algorithm further allows physicians to utilize the existing descriptive auscultation knowledge, gathered over nearly 200 years. Furthermore, the transformation algorithm allows for objectification of the examination because the sense of sight is far better developed among humans, than the sense of hearing. Such a representation also allows for automated analysis of the acoustic signal, for the data (images) exchange and consultations.
FIG. 8 represents a phonocardiographic representation (time domain waveform) of a cardiac auscultation signal with a holosystolic murmur (lasting from S1 to S2�see FIG. 1) with an additional S3 sound. When carefully listening to the signal, S1, S2 and S3 are audible. The holosystolic murmur is loud and easy to detect. The recording is poor quality and very noisy, thus S3 is obscured by noise and not visible in the phonocardiographic representation. Moreover, S1 and S2 are obscured by the holosystolic murmur, and cannot be observed in the waveform either.
FIG. 9 represents a psychoacoustic time-perceptual frequency-loudness of the same signal of FIG. 8 determined by the above-mentioned transformation algorithm. It can be observed that S1 and S2 are easily distinguishable from the holosystolic murmur because they have significantly lower frequencies. Further, S3 is visible in the image, and is distinguishable from the background high frequency noise.
FIG. 10 shows an example of the time-perceptual frequency-perceptual loudness domain representation, i.e., basilar membrane shape changes over time. In the representation of an certain cardio-acoustic signal, impulsive events (such as heart beats or clicks) are visible as vertical shapes of short duration, while long lasting murmurs are visible as frequency limited with relatively high center frequency shapes of longer duration.
FIG. 11 represents a simulation of the pattern tracking ability of a human listener, with the use of the same cardio-acoustic signal of FIG. 10. It can be observed that the representation is much smoother than the one showed in FIG. 10 with repetitive events emphasized.
FIG. 12 illustrates an example of the auscultatory findings detected in the signal of FIG. 11. In this example, the cardio-acoustic signal may be interpreted as indicative hypertrophic cardiomyopathy.
The transformation algorithm provides the use of human auditory system modeling application for analysis of auscultation acoustic signal, which could be implemented on various computers or other devices, with regard to specific requirements and knowledge of the users. The innovation is based on the algorithm (could be viewed as a DSP hardware device with dedicated DSP chip), which transforms the time domain waveform into the time�perceptual frequency�perceptual loudness domain.
The transmitting device can include, for example, a computer, a computer with a browser device, a telephone, an IP phone, a mobile device (e.g., cellular phone, personal digital assistant (PDA) device, laptop computer, electronic mail device), and/or other communication devices. The browser device includes, for example, a computer (e.g., desktop computer, laptop computer) with a world wide web browser (e.g., Microsoft� Internet Explorer� available from Microsoft Corporation, Mozilla� Firefox available from Mozilla Corporation). The mobile computing device includes, for example, a Blackberry�.
Patent CitationsCited PatentFiling datePublication dateApplicantTitleUS4356486Mar 24, 1980Oct 26, 1982Medimetric CompanyTelemetering system for operating room and the likeUS4407288Mar 16, 1981Oct 4, 1983Mieczyslaw MirowskiImplantable heart stimulator and stimulation methodUS4572182Dec 27, 1983Feb 25, 1986Instromedix, Inc.Notched pressure pad for an artery clampUS4622979Mar 2, 1984Nov 18, 1986Cardiac Monitoring, Inc.User-worn apparatus for monitoring and recording electrocardiographic data and method of operationUS4763663Oct 28, 1986Aug 16, 1988Vitacomm, Ltd.Esophageal stethoscope and vital signs monitor systemUS4803625Jun 30, 1986Feb 7, 1989Buddy Systems, Inc.Personal health monitorUS4889131Dec 20, 1988Dec 26, 1989American Health Products, Inc.Portable belt monitor of physiological functions and sensors thereforUS4916441Sep 19, 1988Apr 10, 1990Clinicom IncorporatedPortable handheld terminalUS4938228Feb 15, 1989Jul 3, 1990Righter William HWrist worn heart rate monitorUS4958641Mar 10, 1989Sep 25, 1990Instromedix, Inc.Heart data monitoring method and apparatusUS4967756Jun 15, 1988Nov 6, 1990Instromedix, Inc.Blood pressure and heart rate monitoring method and apparatusUS4977899Nov 24, 1989Dec 18, 1990Instromedix, Inc.Portable ECG monitoring apparatusUS5003984Jan 31, 1990Apr 2, 1991Fukuda Densky Co, Ltd.Apparatus for alerting a doctor by radio to provide an alarm for a patientUS5012814Nov 9, 1989May 7, 1991Instromedix, Inc.Implantable-defibrillator pulse detection-triggered ECG monitoring method and apparatusUS5012815 *Jun 5, 1990May 7, 1991Yale UniversityDynamic spectral phonocardiographUS5024225Jan 5, 1990Jun 18, 1991William FangPersonal health monitor enclosureUS5033474Jan 5, 1990Jul 23, 1991Buddy Systems, Inc.ECG cable storage means in a personal health monitorUS5111396Nov 9, 1989May 5, 1992Instromedix, Inc.Portable ecg data-storage apparatusUS5128552Jan 5, 1990Jul 7, 1992Buddy Systems, Inc.System and method for power supply preservation in a personal health monitorUS5191891Sep 10, 1991Mar 9, 1993Ralin, Inc.Portable ECG monitor/recorderUS5226425Nov 30, 1992Jul 13, 1993Ralin, Inc.Portable ECG monitor/recorderUS5238001Nov 12, 1991Aug 24, 1993Stuart Medical Inc.Ambulatory patient monitoring system having multiple monitoring units and optical communications therebetweenUS5289824Dec 26, 1991Mar 1, 1994Instromedix, Inc.Wrist-worn ECG monitorUS5304186Jun 23, 1992Apr 19, 1994Semler Herbert JArtery clampUS5307818Jul 7, 1992May 3, 1994Jacob SegalowitzWireless electrocardiographic and monitoring system and wireless electrode assemblies for sameUS5317269Jan 8, 1993May 31, 1994Instromedix, Inc.Wrist-worn ECG monitor with battery end of life predictionUS5319363Aug 20, 1992Jun 7, 1994The General Hospital CorporationNetwork for portable patient monitoring devicesUS5333616Jan 7, 1993Aug 2, 1994Instromedix, Inc.Wrist-worn ECG monitorUS5351695Jan 7, 1993Oct 4, 1994Instromedix, Inc.Wrist worn ECG monitorUS5365935Jul 12, 1993Nov 22, 1994Ralin, Inc.Portable, multi-channel ECG data monitor/recorderUS5381804Oct 14, 1993Jan 17, 1995Aspect Medical Systems, Inc.Monitor and method for acquiring and processing electrical signals relating to bodily functionsUS5417222Jan 21, 1994May 23, 1995Hewlett-Packard CompanyPatient monitoring systemUS5441047May 25, 1993Aug 15, 1995David; DanielAmbulatory patient health monitoring techniques utilizing interactive visual communicationUS5467773May 21, 1993Nov 21, 1995Paceart Associates, L.P.Cardiac patient remote monitoring using multiple tone frequencies from central station to control functions of local instrument at patient's homeUS5573506Nov 25, 1994Nov 12, 1996Block Medical, Inc.Remotely programmable infusion systemUS5581369Sep 25, 1992Dec 3, 1996Ralin, Inc.Apparatus and method for communicating electrocardiographic data to a facsimile machineUS5630710Mar 9, 1994May 20, 1997Baxter International Inc.Ambulatory infusion pumpUS5631495Nov 29, 1994May 20, 1997International Business Machines CorporationHigh performance bipolar devices with plurality of base contact regions formed around the emitter layerUS5687717Aug 6, 1996Nov 18, 1997Tremont Medical, Inc.Patient monitoring system with chassis mounted or remotely operable modules and portable computerUS5704364Nov 8, 1995Jan 6, 1998Instromedix, Inc.Concurrent medical patient data and voice communication method and apparatusUS5730143May 3, 1996Mar 24, 1998Ralin Medical, Inc.For attachment to a patientUS5749367Sep 5, 1996May 12, 1998Cardionetics LimitedHeart monitoring apparatus and methodUS5752976Jun 23, 1995May 19, 1998Medtronic, Inc.World wide patient location and data telemetry system for implantable medical devicesUS5819007Mar 15, 1996Oct 6, 1998Siemens Medical Systems, Inc.Feature-based expert system classifierUS5885245Oct 16, 1997Mar 23, 1999Sabratek CorporationMedical apparatus with remote virtual input deviceUS5931791Nov 5, 1997Aug 3, 1999Instromedix, Inc.Medical patient vital signs-monitoring apparatusUS5941829Oct 24, 1997Aug 24, 1999Instromedix, Inc.Concurrent medical patient data and voice communication method and apparatusUS5959529Mar 7, 1997Sep 28, 1999Kail, Iv; Karl A.Reprogrammable remote sensor monitoring systemUS6171237Mar 30, 1998Jan 9, 2001Boaz AvitallRemote health monitoring systemUS6225901Jul 20, 1999May 1, 2001Cardionet, Inc.Reprogrammable remote sensor monitoring systemUS6366871Mar 3, 1999Apr 2, 2002Card Guard Scientific Survival Ltd.Personal ambulatory cellular health monitor for mobile patientUS6466806May 17, 2000Oct 15, 2002Card Guard Scientific Survival Ltd.Photoacoustic material analysisUS6486779Feb 22, 1999Nov 26, 2002Shl Telemedicine International Ltd.Emergency signaling or diagnostic deviceUS6537233 *Nov 6, 2000Mar 25, 2003University Technologies International Inc.Auditory display of knee joint vibration signalsUS6569095Apr 23, 2001May 27, 2003Cardionet, Inc.Adaptive selection of a warning limit in patient monitoringUS6664893Apr 23, 2001Dec 16, 2003Cardionet, Inc.Method for controlling access to medical monitoring device serviceUS6665385Apr 23, 2001Dec 16, 2003Cardionet, Inc.Medical monitoring system having multipath communications capabilityUS6694177Apr 23, 2001Feb 17, 2004Cardionet, Inc.Control of data transmission between a remote monitoring unit and a central unitUS6801137Apr 23, 2001Oct 5, 2004Cardionet, Inc.Bidirectional communication between a sensor unit and a monitor unit in patient monitoringUS6871089Jun 5, 2002Mar 22, 2005Card Guard Technologies, Inc.Portable ECG monitor and method for atrial fibrillation detectionUS6940403Nov 12, 2002Sep 6, 2005Cardionet, Inc.Reprogrammable remote sensor monitoring systemUS6957107Mar 13, 2002Oct 18, 2005Cardionet, Inc.Method and apparatus for monitoring and communicating with an implanted medical deviceUS7002468Dec 5, 2003Feb 21, 2006Cardionet, Inc.Controlling access to a medical monitoring systemUS7099715Feb 17, 2004Aug 29, 2006Cardionet, Inc.Distributed cardiac activity monitoring with selective filteringUS7130396Dec 15, 2003Oct 31, 2006Cardionet, Inc.Medical monitoring system having multiple communications channelsUS7194300Jan 21, 2004Mar 20, 2007Cardionet, Inc.Cardiac monitoringUS7212850Jan 16, 2004May 1, 2007Cardionet, Inc.System and method for processing and presenting arrhythmia information to facilitate heart arrhythmia identification and treatmentUS7222054Mar 4, 2002May 22, 2007Card Guard Scientific Survival Ltd.Personal ambulatory wireless health monitorUS7266405Aug 23, 2000Sep 4, 2007Shl Telemedicine International Ltd.Compact electrode assembly for a portable ECG signaling deviceUS7299159Aug 26, 2004Nov 20, 2007Reuven NanikashviliHealth monitor system and method for health monitoringUS7374545Jul 23, 2002May 20, 2008Shl Telemedicine International, Ltd.Device for sampling blood droplets under vacuum conditionsUS20020032387May 8, 1997Mar 14, 2002Jacob GevaApparatus and method for remote spirometryUS20020067256Apr 9, 2001Jun 6, 2002Kail, Karl A.Reprogrammable remote sensor monitoring systemUS20030028442Aug 6, 2001Feb 6, 2003Craig WagstaffMethod of supplying heart screening services directly to a customerUS20030176795Dec 2, 2002Sep 18, 2003Harris Thomas JohnBlood pressure measurement apparatus and methodUS20030187363Mar 20, 2001Oct 2, 2003Yoram AlroyPortable egg signaling deviceUS20040260189Feb 13, 2004Dec 23, 2004Cardionet, Inc., A California Corporation.Control of data transmission between a remote monitoring unit and a central unitUS20050119580Dec 2, 2004Jun 2, 2005Eveland Doug C.Controlling access to a medical monitoring systemUS20050171448Feb 2, 2004Aug 4, 2005Lev KorzinovBiological signal managementUS20050203349Feb 16, 2005Sep 15, 2005Reuven NanikashviliPersonal health monitor and a method for health monitoringUS20050222515Feb 23, 2005Oct 6, 2005Biosignetics CorporationCardiovascular sound signature: method, process and formatUS20060084881Mar 15, 2005Apr 20, 2006Lev KorzinovMonitoring physiological activity using partial state space reconstructionUS20060155203 *Oct 9, 2003Jul 13, 2006Flemming MunkMethod for arbitrary two-dimensional scaling of phonocardiographic signalsUS20070015973Jun 3, 2005Jan 18, 2007Reuven NanikashviliCommunication terminal, medical telemetry system and method for monitoring physiological dataUS20070129642Feb 12, 2007Jun 7, 2007Cardionet, Inc.Cardiac MonitoringUS20070130657Oct 31, 2006Jun 7, 2007Cardionet, Inc., A California CorporationMedical monitoring system having multiple communications channelsUS20070156054Jul 31, 2006Jul 5, 2007Cardionet, Inc.Distributed cardiac activity monitoring with selective filteringUS20070219453Mar 14, 2006Sep 20, 2007Michael KremliovskyAutomated analysis of a cardiac signal based on dynamical characteristics of the cardiac signalUS20070288067Apr 12, 2007Dec 13, 2007Cardionet, Inc.Controlling Access to a Medical Monitoring SystemUS20070293776Aug 20, 2007Dec 20, 2007Cardionet, Inc.Biological signal managementUSD341659Jan 9, 1992Nov 23, 1993Instromedix, Inc.Wrist-worn cardiac monitorUSD372785Jan 12, 1995Aug 13, 1996Instromedix, Inc.External lead adaptor for wrist-worn cardiac monitorUSD377983Sep 13, 1995Feb 11, 1997 Cardiac monitorUSD414870Jan 2, 1998Oct 5, 1999Instromedix, Inc.Vital signs monitorUSD427315Jan 2, 1998Jun 27, 2000Instromedix, Inc.Vital signs monitoring unitUSD429336Apr 1, 1999Aug 8, 2000Cardionetics LimitedHeart monitoring deviceEP1473667A2Apr 28, 2004Nov 3, 2004Topvine Medical Microsystems LimitedPortable device for classification of medical dataWO1993022970A1May 17, 1993Nov 25, 1993Mcg International IncProducing spectral patterns of heart soundsWO2001093758A1May 15, 2001Dec 13, 2001Cardionetics LtdA blood pressure measurement apparatus and methodWO2004032742A1Oct 9, 2003Apr 22, 2004Bang & Olufsen Medicom AsA method for arbitrary two-dimensional scaling of phonocardiographic signalsWO2005006209A1Jul 12, 2004Jan 20, 2005Univ DublinMethod and apparatus for detecting paroxysmal atria fibrillationWO2007043902A2Oct 16, 2006Apr 19, 2007Medicalgorithmics Sp Z O OMethod, device and system for cardio-acoustic signal analysis* Cited by examinerNon-Patent CitationsReference1E. Zwicker et al., "Audio Engineering and Psychoacoustics: Matching Signals to the Final Receiver, the Human auditory System," Journal of the Audio Engineering Society, vol. 39, No. 3, Mar. 1, 1991, pp. 115-126.2European Search Report-(EP 08 15 9995) Date of Mailing Oct. 7, 2008.3European Search Report�(EP 08 15 9995) Date of Mailing Oct. 7, 2008.4Krishna Prasad G et al: "Classification of ECG Arrhythmias using Multi-Resolution Analysis and Neural Networks" IEEE TENCON 2003-Conference on Technologies for the Asia-Pacific Region-vol. IV, Oct. 15-17, 2003 Bangalore, India: pp. 227-231.5Krishna Prasad G et al: "Classification of ECG Arrhythmias using Multi-Resolution Analysis and Neural Networks" IEEE TENCON 2003�Conference on Technologies for the Asia-Pacific Region�vol. IV, Oct. 15-17, 2003 Bangalore, India: pp. 227-231.6M. Brusco et al., "Digital phonocardiography: a PDA-based approach," 26th Annual International Conference of the IEEE Engineering in Medicine and Biology Society, vol. 3, 2004, pp. 229-2302.7PCT International Search Report-PCT/PL2006 /000068 Date of Mailing Feb. 26, 2007.8PCT International Search Report�PCT/PL2006 /000068 Date of Mailing Feb. 26, 2007.9PCT International Search Report-PCT/PL2006/000067 Dated of Mailing Jun. 21, 2007.10PCT International Search Report�PCT/PL2006/000067 Dated of Mailing Jun. 21, 2007.11R. Erickson et al., "In-vitro study of mechanical heart valve sound loudness as measured by ISO-532/B," 1994 Proceeding of IEEE Seventh Symposium on Computer-Based Medical Systems, 1994, pp. 53-54.Referenced byCiting PatentFiling datePublication dateApplicantTitleUS8301236May 6, 2010Oct 30, 2012Biomedical Systems CorporationSystem and method for high resolution wireless full disclosure ECG episode monitoring and analysisUS8475396Feb 10, 2012Jul 2, 2013AventuSoft, LLCMethod and system of an acoustic scene analyzer for body soundsClassifications U.S. Classification600/528, 600/480, 600/450, 600/463, 600/514International ClassificationA61B7/00Cooperative ClassificationA61B5/0464, A61B5/7203, A61B5/7264European ClassificationA61B5/72K12, A61B5/0464Legal EventsDateCodeEventDescriptionJan 10, 2014FPAYFee paymentYear of fee payment: 4RotateOriginal ImageGoogle Home - Sitemap - USPTO Bulk Downloads - Privacy Policy - Terms of Service - About Google Patents - Send FeedbackData provided by IFI CLAIMS Patent Services©2012 Google