Source: https://www.shopulstandards.com/ProductDetail.aspx?UniqueKey=32287
Timestamp: 2019-11-21 06:47:58
Document Index: 360076135

Matched Legal Cases: ['art 031', 'art 031', 'art 031', 'art 031', 'art 031', 'art 031']

UL Standard | Standard 61010-031 | Edition 2
ulharm-61010-031.2
1 typical CONNECTORS
4 to equipment
2 PROBE TIP
5 SPRING-LOADED CLIP
3 probe body
6 PROBE WIRE
T61010.031.2-F1 Figure 1 - Examples of type A probe assemblies
4 PROTECTIVE FINGERGUARD
2 to equipment
5 hand-held area of probe body
3 reference CONNECTOR
T61010.031.2-F2 Figure 2 - Examples of type B probe assemblies
5 PROBE WIRE
2 probe body
6 examples of accessories
T61010.031.2-F3 Figure 3 - Examples of type C probe assemblies
3 hand-held area of SPRING-LOADED CLIP or clamp
4 PROBE WIRE
T61010.031.2-F4 Figure 4 - Examples of type D probe assemblies
This standard does not apply to current sensors within the scope of IEC 61010-2-032 ( Handheld and hand-manipulated current sensors), but may apply to their input measuring circuit leads and accessories.
This standard also specifies methods of verifying that the probe assembly meets the requirements of this standard, through inspection, TYPE TESTS, and ROUTINE TESTS.
This standard applies to probe assemblies designed to be safe at least under the following conditions:
a) altitude up to 2 000 m;
b) ambient temperature of 5 °C to 40 °C;
c) maximum relative humidity of 80 % for temperatures up to 31 °C decreasing linearly to 50 % relative humidity at 40 °C;
d) applicable POLLUTION DEGREE of the intended environment.
This standard applies to probe assemblies designed to be safe not only in the environmental conditions specified in 1.4.1, but also in any of the following conditions as RATED by the manufacturer of the probe assemblies:
b) altitudes above 2 000 m;
d) relative humidities above the levels specified in 1.4.1;
e) WET LOCATIONS.
1.4.3DV Addition of the following clause referencing the National Electrical Code and the Canadian Electrical Code:
b) Designed to comply with the general requirements of CAN/CSA-C22.2 No. 0 and to be installed in accordance with the Canadian Electrical Code (CE Code), Part I, CSA C22.1; or
Summary of Topics s61010-031_2_sum.html
Adoption of the Second Edition of UL 61010-031, Standard for Safety Requirements for Electrical Equipment for Measurement, Control and Laboratory Use - Part 031: Safety Requirements for Hand-Held Probe Assemblies for Electrical Measurement and Test
UL 61010-031 is an adoption of IEC 61010-031, Safety Requirements for Electrical Equipment for Measurement, Control and Laboratory Use - Part 031: Safety Requirements for Hand-Held Probe Assemblies for Electrical Measurement and Test, (second edition issued May 2015). Please note that the National Difference document incorporates all of the US National Differences for UL 61010-031.
As noted in the Commitment for Amendments statement located on the back side of the title page, UL and CSA are committed to updating this bi-national standard jointly.
Standard 61010-031, Edition 2 Edition Date: January 27, 2017 $440.00-$1,100.00
Summary of Topics b61010-031_2_20160930_sum.html
Subject 61010-031
The following changes in requirements to the Standard for Safety Requirements for Electrical Equipment for Measurement, Control and Laboratory Use - Part 031: Safety Requirements for Hand-Held Probe Assemblies for Electrical Measurement and Test, UL 61010-031, are being proposed:
1. Adoption of IEC 61010-031, Safety Requirements for Electrical Equipment for Measurement, Control and Laboratory Use - Part 031: Safety Requirements for Hand-Held Probe Assemblies for Electrical Measurement and Test (second edition, issued by IEC May 2015) as a new IEC-based UL standard, UL 61010-031 with US Differences.
Summary of Topics b61010-031_2_20160617_sum.html
The following changes in requirements to the Standard for Safety Requirements for Electrical Equipment for Measurement, Control and Laboratory Use - Part 031: Safety Requirements for Hand-Held Probe Assemblies for Electrical Measurement and Test, UL 61010-031, are being proposed for preliminary review and comment only:
1. For Preliminary Review Only: Adoption of IEC 61010-031, Safety Requirements for Electrical Equipment for Measurement, Control and Laboratory Use - Part 031: Safety Requirements for Hand-Held Probe Assemblies for Electrical Measurement and Test (second edition, issued by IEC May 2015) as a new IEC-based UL standard, UL 61010-031 with US Differences.