Source: http://ecfr.io/Title-36/pt36.3.1195
Timestamp: 2018-12-17 04:30:15
Document Index: 336272312

Matched Legal Cases: ['art 1195', 'art 1195', 'art 1195', 'ART 1195', 'art 1195', 'art 1195']

[36 CFR 1195] Title 36 Part 1195 : Code of Federal Regulations ';
Title 36 Part 1195
Title 36 → Chapter XI → Part 1195
PART 1195—STANDARDS FOR ACCESSIBLE MEDICAL DIAGNOSTIC EQUIPMENT
Appendix to Part 1195—Standards for Accessible Medical Diagnostic Equipment
Source: 82 FR 2845, Jan. 9, 2017, unless otherwise noted.
The standards for accessible medical diagnostic equipment are set forth in the appendix to this part. Other agencies, referred to as an enforcing authority in the standards, may adopt the standards as mandatory requirements for entities subject to their jurisdiction. Advisory sections and figures that illustrate the technical requirements in the appendix to part 1195 are available on the Internet at: www.access-board.gov. These advisory materials provide guidance only and do not contain mandatory requirements.
M301.2.2 Sunset. The low transfer position height, Item A of M301.2.1, shall cease to have effect on January 10, 2022.
M302.2.2 Sunset. The low transfer position height, Item A of M302.2.1, shall cease to have effect on January 10, 2022.
Exception: Components shall be permitted to extend into the wheelchair space at a height of 11⁄2 inches (38 mm) maximum between 17 inches (430 mm) minimum and 25 inches (635 mm) maximum in depth measured from the leading edge of the breast platform. From 25 inches (635 mm) to 28 inches (710 mm) in depth the height of a component above 11⁄2 inches (38 mm) shall be beveled at a rate of 2.5:3 maximum.
M303.2.6 Edge Protection. Where wheelchair spaces are provided on a platform raised more than 11⁄2 inches (38 mm) in height, edge protection 2 inches (51 mm) high minimum measured from the surface of the platform shall be provided on each side not providing entry to or exit from the equipment.
M303.3.1 Vertical. Changes in level of 1⁄4 inch (6.4 mm) high maximum shall be permitted to be vertical.
M303.3.2 Beveled. Changes in level between 1⁄4 inch (6.4 mm) high and 1⁄2 inch (13 mm) high maximum shall be beveled with a slope not steeper than 1:2.
M303.3.3 Ramped. Changes in level greater than 1⁄2 inch (13 mm) high shall be ramped and shall comply with M303.3.3.
Exception: A running slope not steeper than 1:8 shall be permitted for ramp runs with a maximum height of 21⁄2 inches (64 mm).
M303.3.3.4 Edge Protection. Ramps with drop offs 1⁄2 inch (13 mm) or greater shall provide edge protection 2 inches (50 mm) high minimum on each side with a drop off.
M305.2.1 Location. Transfer supports shall comply with M305.2.1.1 or M305.2.1.2 and shall be located 11⁄2 inches (38 mm) maximum measured horizontally from the plane defined by the nearest edge of the transfer surface.
M305.2.2.1 End Transfer Supports. Transfer supports for transfer surfaces complying with M301.2.3.1 and M305.2.2.1 shall be 15 inches (380 mm) long minimum. Transfer supports shall be positioned along 131⁄2 inches (345 mm) minimum of the depth of the transfer surface.
M305.2.4.1 Circular Cross Section. Transfer supports with circular cross sections shall have an outside diameter of 11⁄4 inches (32 mm) minimum and 2 inches (51 mm) maximum.
M305.2.7 Clearance. Clearance between the transfer support gripping surface and adjacent surfaces or obstructions shall be 11⁄2 inches (38 mm) minimum.
M305.4 Leg Supports. Leg supports shall provide a method of supporting, positioning, and securing the patient's legs.