Source: http://www.readbag.com/accessdata-fda-cdrh-docs-pdf3-k030594
Timestamp: 2019-08-17 17:30:13
Document Index: 772255621

Matched Legal Cases: ['arts 800', 'art 807', 'art 801', 'art 820', 'art 801', 'art 807']

Read K030594.pdf
Read K030594.pdf text version
K$3$57
The following 510(k) Summary of Safety and Effectiveness is prepared and provided in accordance with the requirements of 21 CFR 807.92 as amended under the Safe Medical Devices Act of 1990 (SMDA).
Company Name: Address: Contact Name: Telephone Number: Summary Prepared Date: Medi-Physics, Inc DBA Amersham Health 101 Carnegie Center Princeton, NJ 08540-623 1 Subhash Pate1 Associate Director, Regulatory Affairs (609) 514-6846 February 19,2003
Subiect Device Information
Trade Name: Common Name: Class: Classification: RAPID StrandTM Model: 7000
Radionuclide Brachytherapy Source I1 21 CFR 892.5730 Product Code: 90-IWI
Legally marketed devices to which equivalence is claimed. RAPID StrandTM Model: 7000 Radionuclide Brachytherapy Source I1 21 CFR 892.5730 Product Code: 90-IWI K940632 Medi-Physics, Inc., Arlington Heights, I L OncoSeedTM Model: 6711 Radionuclide Brachytherapy Source I1 21 CFR 892.5730 Product Code: 90-KXK K914281 Medi-Physics, Inc., Arlington Heights, IL Coated VICRYL (Polyglactin 910) Synthetic Absorbable Suture Absorbable Poly (glycolideL-lactide) surgical suture I1 21 CFR 892.4493 Product Code: GAM KO22269 (September 13,2002) Ethicon, Inc., Somerville, NJ
Trade Name: Common Name: Class: Classification: Cleared 510(k) No.: Submitted by: Trade Name: Common Name: Class: Classification: Cleared 5 1O(k) No.: Submitted by: Trade Name: Common Name: Class: Classification: Cleared 510(k) No.: Submitted by:
Trade Name: Common Name: Class: Classification: Cleared 5 10(k) No.: Submitted by:
Absorbable Seeding Spacers Accessory to applicator and accessory to radionuclide Brachytherapy Sources I1 21 CFR 892.5730 Product Code: 90 KXK KO13964 (January 24,2002) CP Medical, Portland, OR
RAPID StrandTMconsists of 11 absorbable seeding spacers and 10 Model 671 1 OncoSeedTM seeds (welded titanium capsule containing 1-125 adsorbed onto a silver rod) spaced at a fixed distance within absorbable braided carrier. The absorbable braided carrier containing OncoSeeds and absorbable seeding spacer is stiffened and then sterilized by Ethylene Oxide. The seeds are housed in a plastic spacing jig within a stainless steel shielding tube which attenuates &gt;99% of the I- 125 photons. RAPID StrandTM STERILE when shipped. is The Customized RAPID Strand allows the variable seed spacing to reflect individual patient prescriptions as specified by the treating physician. This Customized RAPID Strand will have less than ten (10) OncoSeedTM seeds and a variable number of Absorbable Seeding Spacers. The treating physician will determine the required dose and total number of active seeds for each Customized RAPID Strand. Since it will contain less than 10 Brachytherapy seeds, it does not raise any issues of safety. Please refer to Seed Spacing Diagram in Section-E
RAPID StrandTM consists of absorbable seeding spacers and Model 671 1 OncoSeedTM seeds, embedded within Polyglactin 910 Synthetic Absorbable Braided Carrier and is intended for medical purposes to be placed into a body cavity or tissue as a source of nuclear radiation for therapy.
RAPID Strand TM is indicated for permanent interstitial implantation of selected localized tumors which are low to moderate radiosensitivity. They may be used either as primary treatment (such as prostate cancer or unresectable tumors) or for treatment of residual disease after excision of the primary tumor. RAPID StrandTM may be indicated for use concurrent with or at the completion of other treatment modalities, such as external beam radiation therapy or chemotherapy. When compared with the Intended Use and Indications for Use of the predicate model 7000 RAPID StrandTM, the Intended Use and Indications for Use of the subject model 7000 RAPID StrandTMhave not changed.
The following modifications have been incorporated in the subject device - Model 7000 RAPID StrandTM when compared to predicate device:
7000 RAPID Strand (Subject Device)
7000 RAPID Strand (Predicate Device) Cleared in 5 10(k)## W40632
Regulatory Status Number of Spacers Base Material Color
Not applicable Cleared in 510(k)# KO13964 None 11 (Eleven) 90% Glycolide + 10% L-lactide None D&amp;C Violet No.2 None I (21 CFR 74.3602) Coated VICRYL (Polyglactin 910) Braided Carrier Material Cleared in 510(k) # KO22269 Ethicon JB947 - Coated VICRYL, (Polyglactin 910) Synthetic Absorbable Suture Polyglactin 910 [90% Glycolide + 10% L-lactide coated with Polyglactin 370 and calcium stearate] Undyed (Natural) Sterile (Et0 sterilization performed on finished device) Cleared in 5 10(k) # K940632 Ethicon 5617 - Coated VICRYL (Polyglactin 910) Synthetic Absorbable Suture Polyglactin 910 [90% Glycolide + 10% L-lactide coated with Polyglactin 370 and calcium stearate] Coated Vicryl Violet (D&amp;C Violet #2) Sterile (Et0 sterilization performed on finished device)
Regulatory Status Base Material Formulation
Appearance Sterilization
All other technological characteristics of the subject 7000 RAPID StrandTM remain unchanged compared to the predicate RAPID StrandTM was cleared in the 5 10(k) # K940632. that
Nonclinical Test Data
The copies of the following Nonclinical Test Data reports have been provided in the Section-G of this submission. 1. Report of Analysis: Infrared Spectroscopy - FTIR Spectra of the predicate RAPID Strand (finished device) by Northview Laboratories, Inc., Northbrook, IL 2. Report of Analysis: Infrared Spectroscopy - FTIR Spectra of the subject RAPID Strand 1.5 (finished device) by Northview Laboratories, Inc., Northbrook, IL
Upon reviewing the safety and effectiveness information provided in this submission and comparing the intended use, indications for use, method of use and other technological characteristics, it can be concluded that the subject RAPID StrandTM substantially equivalent to is the predicate RAPID StrandTM, which was cleared under 5 10(k)# K940632.
Food and Drug Administration 9200 CorporateBoulevard Rockville MD 20850
Mr. Subhash Patel Associate Director Regulatory Affairs Amersham Health 101 Carnegie Center PRINCETON NJ 08540
Re: KO30594 Trade/Device Name: 1-125 RAPID Strand Model 7000 Regulation Number: 21 CFR 892.5730 Regulation Name: Radionuclide brachytherapy source Regulatory Class: I1 Product Code: 90 KXK Dated: March 25,2003 Received: March 27, 2003
Dear Mr. Patel: We have reviewed your Section 5 10(k) premarket notification of intent to market the device referenced above and have determined the device is substantially equivalent (for the indications for use stated in the enclosure) to legally marketed predicate devices marketed in interstate commerce prior to May 28, 1976, the enactment date of the Medical Device Amendments, or to devices that have been reclassified in accordance with the provisions of the Federal Food, Drug, and Cosmetic Act (Act) that do not require approval of a premarket approval application (PMA). You may, therefore, market the device, subject to the general controls provisions of the Act, The general controls provisions of the Act include requirements for annual registration, listing of devices, good manufacturing practice, labeling, and prohibitions against misbranding and adulteration. If your device is classified (see above) into either class I1 (Special Controls) or class 111 (PMA), it may be subject to additional controls. Existing major regulations affecting your device can be found in the Code of Federal Regulations, Title 21, Parts 800 to 898. In addition, FDA may publish further announcements concerning your device in the Federal Register. Please be advised that FDA's issuance of a substantial equivalence determination does not mean that FDA has made a determination that your device complies with other requirements of the Act or any Federal statutes and regulations administered by other Federal agencies. You must comply with all the Act's requirements, including, but not limited to: registration and listing (21 CFR Part 807); labeling (21 CFR Part 801); good manufacturing practice requirements as set forth in the quality systems (QS) regulation (2 1 CFR Part 820); and if applicable, the electronic product radiation control provisions (sections 53 1-542 of the Act); 2 1 CFR 1000-1050.
This letter will allow you to begin marketing your device as described in your Section 5 lO(k) premarket notification. The FDA finding of substantial equivalence of your device to a legally marketed predicate device results in a classification for your device and thus, permits your device to proceed to the market. If you desire specific advice for your device on our labeling regulation (21 CFR Part 801), please contact the Office of Compliance at one of the following numbers, based on the regulation number at the top of the letter:
8xx. lxxx 34Sxxx 876.2884.2~ X X X 4xr0c,5, 6xx~ 4xxx, 5xxx 892.2xxx, 3Other
Additionally, for questions on the promotion and advertising of your device, please contact the Office of Compliance at (301) 594-4639. Also, please note the regulation entitled, &quot;Misbranding by reference to premarket notification&quot; (21CFR Part 807.97) you may obtain. Other general information on your responsibilities under the Act may be obtained from the Division of Small Manufacturers, International and Consumer Assistance at its toll-free number (800) 638-204 1 or (301) 443-6597 or at its Internet address http://www.fda.gov/cdrh/dsma/dsmamain. html.
Nancy C. Brogdon Director, Division of Reproductive, Abdominal and Radiological Devices Office of Device Evaluation Center for Devices and Radiological Health Enclosure
KjGp5.r ~ f Model: 7000 RAPID StrandTM
RAPID Strand TM is indicated for permanent interstitial implantation of selected localized tumors which are low to moderate radiosensitivity. They may be used either as primary treatment (such as prostate cancer or unresectable tumors) or for treatment of residual disease after excision of the primary tumor. RAPID StrandTM may be indicated for use concurrent with or at the completion of other treatment modalities, such as external beam radiation therapy or chemotherapy.
PLEASE DO NOT WRITE BELOW THIS LINE - CONTINUE ON ANOTHER PAGE IF NEEDED
Concurrence of CDRH Office of Device Evaluation (ODE)
(Division Sign-off) Division o Reproductive,Awominsl, f 51Qk) Number and Radiological Oevices
D ,oS'w 3
Prescription Use (Per 21 CFR 801.109)