Source: https://ru.scribd.com/document/102400061/As-NZS-CISPR-16-1-5-2004-Specification-for-Radio-Disturbance-and-Immunity-Measuring-Apparatus-and-Methods-Rad
Timestamp: 2019-09-18 17:56:40
Document Index: 291606196

Matched Legal Cases: ['art 1', 'art 1', 'art 1', 'art 1', 'art 1', 'art 1', 'art 1', 'art 1', 'art 1', 'art 1', 'art 1', 'art 4', 'art 4', 'art 4', 'art 4', 'art 1', 'art 1', 'art 1', 'art 1', 'art 4', 'art 4']

As NZS CISPR 16.1.5-2004 Specification for Radio Disturbance and Immunity Measuring Apparatus and Methods Rad | Calibration | Electromagnetic Compatibility
сохранитьСохранить «As NZS CISPR 16.1.5-2004 Specification for Radio D...» для последующего чтения
Hach HQ40-d
4225-14-36691-VER-B0001
Dse123 Datasheet
CorbettPA2Webinar
Perkin Elmer Lambda 20-40 Uv Vis Spectrometer Manual Eng
Quality Assuranae Plan Review Check List
D6347D6347M-Standard Test Method for Determination of Bulk Density of Coal Using Nuclear Backscatter Depth Density Methods
FCO560 Users Guide Issue 2a
Testing HiRes
Beyond Flowmeter Selection Consider Some of the Most Overlooked Criteria
AS/NZS CISPR 16.1.
CISPR 16-1-5:2003
AS/NZS CISPR 16.1.5:2004 This Joint Australian/New Zealand Standard was prepared by Joint Technical Committee TE-003, Electromagnetic Interference. It was approved on behalf of the Council of Standards Australia on 24 March 2004 and on behalf of the Council of Standards New Zealand on 16 April 2004. It was published on 2 June 2004.
This Standard was issued in draft form for comment as DR 04087.
Originated as part of AS 1052.1-1976. Previous edition AS/NZS CISPR 16.1:2002. Revised and redesignated in part as AS/NZS CISPR 16.1.5:2004.
ISBN 0 7337 5990 4
This Standard was prepared by the Joint Standards Australia/Standards New Zealand Committee TE-003, Electromagnetic Interference to supersede AS/NZS CISPR 16.1:2002. This Standard is identical with, and has been reproduced from, CISPR 16-1-5:2003, Specification for radio disturbance and immunity measuring apparatus and methods, Part 1-5: Radio disturbance and immunity measuring apparatusAntenna calibration test sites for 30 MHz to 1000 MHz. The objective of this Standard is to specify the requirements for calibration test sites, used to perform antenna calibrations, as well as the test antenna characteristics, calibration site verification procedure and site compliance criteria.
This Standard is Part 1.5 of AS/NZS CISPR 16.1, Specification for radio disturbance and immunity measuring apparatus and methods, which consists of the following: Part 1.1: Radio disturbance and immunity Part 1.2: Radio disturbance and immunity Part 1.3: Radio disturbance and immunity Part 1.4: Radio disturbance and immunity Part 1.5: Radio disturbance and immunity for 30 MHz to 1000 MHz (this standard) measuring measuring measuring measuring measuring apparatusMeasuring apparatus apparatusConducted disturbances apparatusDisturbance power apparatusRadiated disturbance apparatusAntenna calibration test sites
The terms normative and informative are used to define the application of the annex to which they apply. A normative annex is an integral part of a standard, whereas an informative annex is only for information and guidance. As this Standard is reproduced from an international standard, the following applies: (a) Its number appears on the cover and title page while the international standard number appears only on the cover. (b) In the source text this International Standard should read this Australian/New Zealand Standard. (c) A full point substitutes for a comma when referring to a decimal marker. References to International Standards should be replaced by references to Australian or Australian/New Zealand Standards, as follows: Reference to International Standard CISPR 16 Specification for radio disturbance and immunity measuring apparatus and methods Part 1-1: Radio disturbance and immunity measuring apparatus Measuring apparatus Part 1-4: Radio disturbance and immunity measuring apparatus Ancillary equipmentRadiated disturbances Australian/New Zealand Standard AS/NZS CISPR 16 Specification for radio disturbance and immunity measuring apparatus and methods Part 1.1: Radio disturbance and immunity measuring apparatus Measuring apparatus Part 1.4: Radio disturbance and immunity measuring apparatus Ancillary equipmentRadiated disturbances
Part 4-1: Uncertainties, statistics and limit modellingUncertainties in standardized EMC tests Part 4-2: Uncertainties, statistics and limit modellingMeasurement instrumentation uncertainty
Part 4.1: Uncertainties, statistics and limit modellingUncertainties in standardized EMC tests Part 4.2: Uncertainties, statistics and limit modellingMeasurement instrumentation uncertainty
Page 1 2 3 4 Scope .............................................................................................................................. 1 Normative references ....................................................................................................... 1 Definitions........................................................................................................................ 2 Specifications and validation procedures for a test site to be used to calibrate antennas in the frequency range of 30 MHz to 1 000 MHz ................................................ 3 4.1 4.2 Introduction............................................................................................................. 3 Antenna calibration test site (CALTS) specification ................................................. 3 4.2.1 Introduction ................................................................................................. 3 4.2.2 Normative specification ............................................................................... 4 Test antenna specification ...................................................................................... 4 4.3.1 Introduction ................................................................................................. 4 4.3.2 Normative specifications ............................................................................. 4 Antenna calibration test site validation procedure ................................................... 6 4.4.1 Introduction ................................................................................................. 6 4.4.2 Test set-up.................................................................................................. 7 4.4.3 Test frequencies and receiving antenna heights .......................................... 8 4.4.4 Site attenuation measurements ................................................................... 9 4.4.5 Antenna-height scan measurements ......................................................... 11 4.4.6 Frequency scan measurements ................................................................. 11 Antenna calibration test site compliance criteria.................................................... 11 4.5.1 Introduction ............................................................................................... 11 4.5.2 Tolerances and measurement uncertainties .............................................. 12 4.5.3 Compliance criteria ................................................................................... 13 The validation report ............................................................................................. 15 4.6.1 Introduction ............................................................................................... 15 4.6.2 Validation report requirements .................................................................. 15 Validation of the CALTS for vertical polarization ................................................... 17 (informative) CALTS requirements......................................................................... 18
4.7 Annex A
Annex B (informative) Test antenna considerations ............................................................. 21 Annex C (informative) Antenna and site attenuation theory................................................. 27 Annex D (informative) Application of a fixed length dipole (30 MHz f 80 MHz) ............... 38 Annex E (informative) Pascal Program used in C.1.3........................................................... 39 Annex F (informative) Checklist validation procedure .......................................................... 43
1 STANDARDS AUSTRALIA/STANDARDS NEW ZEALAND Australian/New Zealand Standard Specification for radio disturbance and immunity measuring apparatus and methods Part 1.5: Radio disturbance and immunity measuring apparatusAntenna calibration test sites for 30 MHz to 1000 MHz
This part of CISPR 16 is designated a basic standard which specifies the requirements for calibration test sites, used to perform antenna calibrations, as well as the test antenna characteristics, calibration site verification procedure and site compliance criteria. Further information on calibration site requirements, test antenna considerations and the theory of antennas and site attenuation is provided in informative annexes. Measurement instrumentation specifications are given in CISPR 16-1-1 and CISPR 16-1-4. Further information and background on uncertainties in general is given in CISPR 16-4-1, which may be helpful in establishing uncertainty estimates for the calibration processes of antennas.
The following referenced documents are indispensable for the application of this document. For dated references, only the edition cited applies. For undated references, the latest edition of the referenced document (including any amendments) applies. CISPR 14-1:2000, Electromagnetic compatibility Requirements for household appliances, electric tools and similar apparatus Part 1: Emission CISPR 16-1-1:2003, Specification for radio disturbance and immunity measuring apparatus and methods Part 1-1: Radio disturbance and immunity measuring apparatus Measuring apparatus CISPR 16-1-4:2003, Specification for radio disturbance and immunity measuring apparatus and methods Part 1-4: Radio disturbance and immunity measuring apparatus Ancillary equipment - Radiated disturbances CISPR 16-4-1:2003, Specification for radio disturbance and immunity measuring apparatus and methods Part 4-1: Uncertainties, statistics and limit modelling Uncertainties in standardized EMC tests CISPR 16-4-2:2003, Specification for radio disturbance and immunity measuring apparatus and methods Part 4-2: Uncertainties, statistics and limit modelling Measurement instrumentation uncertainty IEC 60050(161):1990, International Electrotechnical Vocabulary (IEV) Chapter 161: Electromagnetic compatibility International Vocabulary of Basic and General Terms in Metrology, International Organization for Standardization, Geneva, 2nd edition, 1993
AS/NZS CISPR 16.1.5:2004, Specification for radio disturbance and immunity measuring apparatus and methods Radio disturbance and immunity measuring apparatus - Antenna calibration test sites for 30 MHz to 1000 MHz
Документы, похожие на «As NZS CISPR 16.1.5-2004 Specification for Radio Disturbance and Immunity Measuring Apparatus and Methods Rad»
VasilisPapantinas
QRB Photoresistive Flame Detectors
Quick Reference-Precise 100 Series
certificado luxometro
70k06-e HW Connection
Farid Dafa