Source: http://docplayer.net/20139134-Fresenius-medical-care.html
Timestamp: 2018-12-14 13:28:05
Document Index: 444282371

Matched Legal Cases: ['arts 800', 'art 802', 'art 801', 'art 820', 'art 801', 'art 21']

Fresenius Medical Care - PDF
Download "Fresenius Medical Care"
1 IO 20 20O 4 Fresenius Medical Care RAE Fresenius Optiflux F2ONRe, FI 8NRe and FI 6NR e Single Use Hemodkalyzers "Special" 510(k) Premarket Notification Summary of Safety and Effectiveness This 510O(k) Summary of Safety and Effectiveness is being submitted in accordance with the requirements of SMDA A. Submitter's Information: Name: Address: Fresenius Medical Care North America 95 Hayden Ave Two Ledgemont Center Lexington, MA Phone: Fax: (781) Contact Person: Date of Preparation: 19 November, 2004 B. Device Name: Arthur Eilinsfeld, Director of Regulatory Affairs Proprietary Name: Fresenius Optiflux F2ONRe, F18NR 0 and F16NRe Hemodialyzers Common Name: * Optiflux F2oNReandOptifiux F18NRC: Dialyzer, High Permeability with or without Sealed Dialysate System * Optiflux F16NRe: Dialyzer, Capillary, Hollow Fiber Product Code/Classification Panel: * Optiflux F2ONRe andaptiflux Fl8NR 0 : 78KDI/Gastroenterology-Urology * Optiflux F16NR 0 : 78FJI/Gastroenterology-Urology Classification: * Optiflux F20NR'andOptiflux F18NRe: Class II per * Optiflux F16NR 0 : Class II per
2 5AOE 2 cf-- Fresenius Medical Care Fresenius Optiflux F20NRe, F18NRe and F16NRe Single Use Hemodialyzers "Special" 510(k) Premarket Notification Summary of Safety and Effectiveness C. Predicate Devices The Fresenius Optiflux F20NR 0, F18NR 0, and F16NRe are modified versions of the Fresenius Optiflux 200NR and Fresenius Hemoflow F7NR 0, which were cleared under the following premarket notifications: Optiflux 200NR: * K (8/25/00) Hemoflow F7NR 0 * #K (12/04/00) D. Indications for Use/Intended Use Optiflux F20NRt, F18NRt, and F16NR 0 dialyzers are designed for single use acute and chronic hemodialysis. E. Substantial Equivalence: 1. Is the product a device? YES - The Fresenius Optiflux F20NR e, F18NRe, and F16NRe hemodialyzers are devices pursuant to 21 CFR 201 [321] (h). 2. Does the new device have the same intended use? YES - The intended use for the Optiflux F20NR e, F18NR 0, and F16NRe hemodialyzers is equivalent to that for the Fresenius Optiflux 200NR and is as follows: Optiflux F2ONRe. F18NRe, and F16NRe - Intended Use Optiflux F20NRF, F18NR, and F16NRt dialyzers are designed for single use acute and chronic hemodialysis. Fresenius Optiflux 200NR - Intended Use Optiflux 200NR dialyzers are designed for acute and chronic hemodialysis and are appropriate for single and multiple use
3 PA c2-4 4 V Fresenius Medical Care Fresenius Optiflux F20NR% F18NRe and F16NRe Single Use Hemodialyzers "Special" 510(k) Premarket Notification Summary of Safety and Effectiveness 3. Does the device have technological characteristics that raise new types of safety or effectiveness questions? NO - The Optifiux F20NRe, F18NRe, and F16NRe are e-beam sterilized, low-flux versions of the Optifiux 200NR. The technological characteristics of the Optiflux F20NR, F18NR 0, and F16NR e are equivalent to those of the Optiflux 200NR and Hemoflow F7NRe and raise no new types of safety or effectiveness questions. 4. Does descriptive or performance information demonstrate equivalence? YES - Fresenius Medical Care North America believes that the information provided in this submission clearly describes the Optiflux F20NR 0, F18NR e, and F16NR' and demonstrates that they are substantially equivalent to the Fresenius Optiflux 200NR and Hemoflow F7NR 0. F. Safety Summary The Optiflux F20NR 0, F18NR e, and F16NR' dialyzers are substantially equivalent in construction, design, materials, and intended use to the commercially available Fresenius F7NRe and Fresenius Optiflux 200NR dialyzers. In addition, testing of the Optiflux F20NR e, F18NRC, and F16NR 0 indicates that they are safe and effective for their intended use &
4 '7 ~~~DEPARTMENT OF HEALTH & HUMAN SERVICES Public Health Service Food and Drug Administration 9200 Corporate Boulevard DEC ~~~~Rockville MD Ms. Nichole Riek Regulatory Affairs Supervisor Fresenius Medical Care North America 95 Hayden Avenue LEXINGTON MA Re: K Trade/Device Name: Fresenius Optiflux 2ONRe, I8NRe and I6NRe Hemodialyzers Regulation Number: 21 CFR Regulation Name: High permeability hemodialysis system Product Code: 78 KDI Regulation Number: 21 CFR Regulation Name: Hemodialysis system and accessories Product Code: 78 FJI Regulatory Class: 11 Dated: November 19, 2004 Received: November 2S, 2004 Dear Ms. Riek: We have reviewed your Section 5 10(k) premarket notification of intent to market the device referenced above and have determined the device is substantially equivalent (for the indications for use stated in the enclosure) to legally marketed predicate devices marketed in interstate commerce prior to May 28, 1976, the enactment date of the Medical Device Amendments, or to devices that have been reclassified in accordance with the provisions of the Federal Food, Drug, and Cosmetic Act (Act) that do not require approval of a premarket approval application (PMA). You may, therefore, market the device, subject to the general controls provisions of the Act. The general controls provisions of the Act include requirements for annual registration, listing of devices, good manufacturing practice, labeling, and prohibitions against misbranding and adulteration. If your device is classified (see above) into either class LI (Special Controls) or class III (PMA), it may be subject to additional controls. Existing major regulations affecting your device can be found in the Code of Federal Regulations, Title 21, Parts 800 to 898. In addition, FDA may publish further announcements concerning your device in the Federal eise. Please be advised that FDA's issuance of a substantial equivalence determination does not mean that FDA has made a determination that your device complies with other requirements of the Act or any Federal statutes and regulations administered by other Federal agencies. You must comply with all the Act's requirements, including, but not limited to: registration and listing (21 CFR Part 802); labeling (21 CFR Part 801); good manufacturing practice requirements as set forth in the quality systems (QS) regulation (21 CFR Part 820); and if applicable, the electronic product radiation control provisions (sections of the Act); 21 CFR
5 This letter will allow you to begin marketing your device as described in your Section 510(k) premarket notification. The FDA finding of substantial equivalence of your device to a legally marketed predicate device results in a classification for your device and thus, permits your device to proceed to the market. If you desire specific advice for your device on our labeling regulation (21 CFR Part 801), please contact the Office of Compliance at one of the following numbers, based on the regulation number at the top of this letter: 21 CFR 876.xxxx (Gastroenterology/Renal/Urology) CFR 884.xxxx (Obstetrics/Gynecology) CFR 892.xxxx (Radiology) Other Also, please note the regulation entitled, "Misbranding by reference to premarket notification" (21 CFR ). You may obtain other general information on your responsibilities under the Act from the Division of Small Manufacturers, International and Consumer Assistance at its toll-free number (800) or (301) or at its Internet address Sincerely yours, Enclosure Nan4cyC.rogdoon Director, Division of Reproductive, Abdominal, and Radiological Devices Office of Device Evaluation Center for Devices and Radiological Health
6 Indications for Use 510(k) Number (if known): Device Name: K Fresenius Optiflux F20NR e, F18NRe, and F16NR e Hremodialyzers Indications For Use: Optiflux F20NRe, F1 8NR e, and F1 6NRe dialyzers are designed for single use acute and chronic hemodialysis. Prescription Use [/ -AND/OR Over-The-Counter Use (Part 21 CFR 801 Subpart D) (21 CFR 801 Subpart C) (PLEASE DO NOT WRITE BELOW THIS LINE-CONTINUE ON ANOTHER PAGE IF NEEDED) Concurrence of CDRH, Office of Device Evaluation (ODE) (Division Sign-0ff) Division of Reproductive, Divisi Abdominal,Pae1o ~~~~~~~Page 1 of1 1 and Radiological Devices.2 510(k) Number.