Source: http://docplayer.fr/55891021-International-standard-norme-internationale.html
Timestamp: 2018-10-23 19:23:05+00:00
Document Index: 311043090

Matched Legal Cases: ['art 3', 'art 3', 'art 2', 'art 3', 'art 1', 'art 1']

1 INTERNATIONAL STANDARD NORME INTERNATIONALE IEC Edition Thermocouples Part 3: Extension and compensating cables Tolerances and identification system Couples thermoélectriques Partie 3: Câbles d extension et de compensation Tolérances et système d identification INTERNATIONAL ELECTROTECHNICAL COMMISSION COMMISSION ELECTROTECHNIQUE INTERNATIONALE PRICE CODE CODE PRIX J ICS ISBN
2 IEC:2007 INTERNATIONAL ELECTROTECHNICAL COMMISSION THERMOCOUPLES Part 3: Extension and compensating cables Tolerances and identification system FOREWORD 1) The International Electrotechnical Commission (IEC) is a worldwide organization for standardization comprising all national electrotechnical committees (IEC National Committees). The object of IEC is to promote international co-operation on all questions concerning standardization in the electrical and electronic fields. To this end and in addition to other activities, IEC publishes International Standards, Technical Specifications, Technical Reports, Publicly Available Specifications (PAS) and Guides (hereafter referred to as IEC Publication(s) ). Their preparation is entrusted to technical committees; any IEC National Committee interested in the subject dealt with may participate in this preparatory work. International, governmental and non-governmental organizations liaising with the IEC also participate in this preparation. IEC collaborates closely with the International Organization for Standardization (ISO) in accordance with conditions determined by agreement between the two organizations. 2) The formal decisions or agreements of IEC on technical matters express, as nearly as possible, an international consensus of opinion on the relevant subjects since each technical committee has representation from all interested IEC National Committees. 3) IEC Publications have the form of recommendations for international use and are accepted by IEC National Committees in that sense. While all reasonable efforts are made to ensure that the technical content of IEC Publications is accurate, IEC cannot be held responsible for the way in which they are used or for any misinterpretation by any end user. 4) In order to promote international uniformity, IEC National Committees undertake to apply IEC Publications transparently to the maximum extent possible in their national and regional publications. Any divergence between any IEC Publication and the corresponding national or regional publication shall be clearly indicated in the latter. 5) IEC provides no marking procedure to indicate its approval and cannot be rendered responsible for any equipment declared to be in conformity with an IEC Publication. 6) All users should ensure that they have the latest edition of this publication. 7) No liability shall attach to IEC or its directors, employees, servants or agents including individual experts and members of its technical committees and IEC National Committees for any personal injury, property damage or other damage of any nature whatsoever, whether direct or indirect, or for costs (including legal fees) and expenses arising out of the publication, use of, or reliance upon, this IEC Publication or any other IEC Publications. 8) Attention is drawn to the Normative references cited in this publication. Use of the referenced publications is indispensable for the correct application of this publication. 9) Attention is drawn to the possibility that some of the elements of this IEC Publication may be the subject of patent rights. IEC shall not be held responsible for identifying any or all such patent rights. International Standard IEC has been prepared by subcommittee 65B: Devices and process analysis, of IEC Technical Committee 65: Industrial-process measurement, control and automation. This second edition cancels and replaces the first edition of IEC issued in It constitutes a technical revision. The main technical changes with regard to the previous edition are as follows: Addition of Subclause 5.4 Connectors. Addition of Clauses 7 Dimensions and 8 Requirements.
3 IEC: The text of this standard is based on the following documents: FDIS 65B/642/FDIS Report on voting 65B/646/RVD Full information on the voting for the approval of this standard can be found in the report on voting indicated in the above table. This publication has been drafted in accordance with the ISO/IEC Directives, Part 2. A list of all parts of the IEC series, under the general title Thermocouples, can be found on the IEC website. The committee has decided that the contents of this publication will remain unchanged until the maintenance result date indicated on the IEC web site under " in the data related to the specific publication. At this date, the publication will be reconfirmed, withdrawn, replaced by a revised edition, or amended.
4 IEC:2007 THERMOCOUPLES Part 3: Extension and compensating cables Tolerances and identification system 1 Scope This part of IEC specifies manufacturing tolerances for extension and compensating cables (other than mineral insulated cables) provided directly to users of industrial processes. These tolerances are determined with respect to the e.m.f.-temperature relationship of Part 1 of the standard. The method for identification of insulated thermocouple extension and compensating cables other than mineral insulated cables is described. Furthermore, requirements for extension and compensating cables for use in industrial process control are specified. 2 Normative references The following referenced documents are indispensable for the application of this document. For dated references, only the edition cited applies. For undated references, the latest edition of the referenced document (including any amendments) applies. IEC :1995, Thermocouples Part 1: Reference tables
5 CEI:2007 COMMISSION ÉLECTROTECHNIQUE INTERNATIONALE COUPLES THERMOÉLECTRIQUES Partie 3: Câbles d extension et de compensation Tolérances et système d identification AVANT-PROPOS 1) La Commission Electrotechnique Internationale (CEI) est une organisation mondiale de normalisation composée de l'ensemble des comités électrotechniques nationaux (Comités nationaux de la CEI). La CEI a pour objet de favoriser la coopération internationale pour toutes les questions de normalisation dans les domaines de l'électricité et de l'électronique. A cet effet, la CEI entre autres activités publie des Normes internationales, des Spécifications techniques, des Rapports techniques, des Spécifications accessibles au public (PAS) et des Guides (ci-après dénommés "Publication(s) de la CEI"). Leur élaboration est confiée à des comités d'études, aux travaux desquels tout Comité national intéressé par le sujet traité peut participer. Les organisations internationales, gouvernementales et non gouvernementales, en liaison avec la CEI, participent également aux travaux. La CEI collabore étroitement avec l'organisation Internationale de Normalisation (ISO), selon des conditions fixées par accord entre les deux organisations. 2) Les décisions ou accords officiels de la CEI concernant les questions techniques représentent, dans la mesure du possible, un accord international sur les sujets étudiés, étant donné que les Comités nationaux de la CEI intéressés sont représentés dans chaque comité d études. 3) Les Publications de la CEI se présentent sous la forme de recommandations internationales et sont agréées comme telles par les Comités nationaux de la CEI. Tous les efforts raisonnables sont entrepris afin que la CEI s'assure de l'exactitude du contenu technique de ses publications; la CEI ne peut pas être tenue responsable de l'éventuelle mauvaise utilisation ou interprétation qui en est faite par un quelconque utilisateur final. 4) Dans le but d'encourager l'uniformité internationale, les Comités nationaux de la CEI s'engagent, dans toute la mesure possible, à appliquer de façon transparente les Publications de la CEI dans leurs publications nationales et régionales. Toutes divergences entre toutes Publications de la CEI et toutes publications nationales ou régionales correspondantes doivent être indiquées en termes clairs dans ces dernières. 5) La CEI n a prévu aucune procédure de marquage valant indication d approbation et n'engage pas sa responsabilité pour les équipements déclarés conformes à une de ses Publications. 6) Tous les utilisateurs doivent s'assurer qu'ils sont en possession de la dernière édition de cette publication. 7) Aucune responsabilité ne doit être imputée à la CEI, à ses administrateurs, employés, auxiliaires ou mandataires, y compris ses experts particuliers et les membres de ses comités d'études et des Comités nationaux de la CEI, pour tout préjudice causé en cas de dommages corporels et matériels, ou de tout autre dommage de quelque nature que ce soit, directe ou indirecte, ou pour supporter les coûts (y compris les frais de justice) et les dépenses découlant de la publication ou de l'utilisation de cette Publication de la CEI ou de toute autre Publication de la CEI, ou au crédit qui lui est accordé. 8) L'attention est attirée sur les références normatives citées dans cette publication. L'utilisation de publications référencées est obligatoire pour une application correcte de la présente publication. 9) L attention est attirée sur le fait que certains des éléments de la présente Publication de la CEI peuvent faire l objet de droits de propriété intellectuelle ou de droits analogues. La CEI ne saurait être tenue pour responsable de ne pas avoir identifié de tels droits de propriété et de ne pas avoir signalé leur existence. La Norme internationale CEI a été établie par le sous-comité d'études 65B: Dispositifs, du comité d'études 65 de la CEI: Mesure, commande et automation dans les processus industriels. Cette deuxième édition annule et remplace la première édition de la CEI , publiée en 1989, dont elle constitue une révision technique. Les principaux changements techniques par rapport à l édition précédente sont les suivants : Ajout du Paragraphe 5.4 Connecteurs. Ajout des Articles 7 Dimensions et 8 Exigences.
6 CEI: Le texte de la présente norme est issu des documents suivants: FDIS 65B/642/FDIS Rapport de vote 65B/646/RVD Le rapport de vote indiqué dans le tableau ci-dessus donne toute information sur le vote ayant abouti à l'approbation de la présente Norme. Cette publication a été rédigée selon les Directives ISO/CEI, Partie 2. Une liste de toutes les parties de la série CEI 60584, présentées sous le titre général Couples thermoélectriques, peut être consultée sur le site web de la CEI. Le comité a décidé que le contenu de cette publication ne sera pas modifié avant la date de maintenance indiquée sur le site web de la CEI sous " dans les données relatives à la publication recherchée. A cette date, la publication sera reconduite, supprimée, remplacée par une édition révisée, ou amendée.
7 CEI:2007 COUPLES THERMOÉLECTRIQUES Partie 3: Câbles d extension et de compensation Tolérances et système d identification 1 Domaine d application La présente partie de la CEI spécifie les tolérances de fabrication pour les câbles d extension et de compensation (autres que ceux à isolation minérale) fournis directement aux utilisateurs pour des procédés industriels. Ces tolérances sont déterminées par rapport à la relation fem-température de la Partie 1 de cette norme. La méthode d identification des câbles d extension et de compensation des couples thermoélectriques isolés autres que les câbles à isolation minérale est décrite. De plus, des exigences sont spécifiées pour les câbles d extension et de compensation pour utilisation dans des procédés industriels. 2 Références normatives Les documents de référence suivants sont indispensables pour l'application du présent document. Pour les références datées, seule l'édition citée s'applique. Pour les références non datées, la dernière édition du document de référence s'applique (y compris les éventuels amendements). CEI :1995, Couples thermoélectriques Partie 1: Tables de référence
This is a preview - click here to buy the full publication NORME INTERNATIONALE INTERNATIONAL STANDARD. Connecteurs pour fibres et câbles optiques
NORME INTERNATIONALE INTERNATIONAL STANDARD CEI IEC 60874-1-1 QC 910001 to QC 910006 QC 910001 à QC 910006 et/and QC 910099 Deuxième édition Second edition 2006-10 Connecteurs pour fibres et câbles optiques