Source: http://docplayer.net/114291-General-principles-of-software-validation-final-guidance-for-industry-and-fda-staff.html
Timestamp: 2017-06-25 07:20:50
Document Index: 761931049

Matched Legal Cases: ['art 820', 'art 11', 'art 11', 'art 2', 'art 11', 'art 11', 'art 11', 'art 11', 'art 11', 'art 11', 'art 11', 'art 11']

General Principles of Software Validation; Final Guidance for Industry and FDA Staff - PDF
Download "General Principles of Software Validation; Final Guidance for Industry and FDA Staff"
1 General Principles of Software Validation; Final Guidance for Industry and FDA Staff Document issued on: January 11, 2002 This document supersedes the draft document, "General Principles of Software Validation, Version 1.1, dated June 9, U.S. Department Of Health and Human Services Food and Drug Administration Center for Devices and Radiological Health Center for Biologics Evaluation and Research2 Preface Public Comment Comments and suggestions may be submitted at any time for Agency consideration to Dockets Management Branch, Division of Management Systems and Policy, Office of Human Resources and Management Services, Food and Drug Administration, 5630 Fishers Lane, Room 1061, (HFA-305), Rockville, MD, When submitting comments, please refer to the exact title of this guidance document. Comments may not be acted upon by the Agency until the document is next revised or updated. For questions regarding the use or interpretation of this guidance which involve the Center for Devices and Radiological Health (CDRH), contact John F. Murray at (301) or For questions regarding the use or interpretation of this guidance which involve the Center for Biologics Evaluation and Research (CBER) contact Jerome Davis at (301) or Additional Copies CDRH Additional copies are available from the Internet at: DeviceRegulationandGuidance/GuidanceDocuments/UCM htm. You may also send an request to to receive an electronic copy of the guidance or send a fax request to to receive a hard copy. Please use the document number (938) to identify the guidance you are requesting. CBER Additional copies are available from the Internet at: by writing to CBER, Office of Communication, Training, and Manufacturers' Assistance (HFM- 40), 1401 Rockville Pike, Rockville, Maryland , or by telephone request at or Page ii3 Table of Contents SECTION 1. PURPOSE... 1 SECTION 2. SCOPE Applicability Audience THE LEAST BURDENSOME APPROACH Regulatory Requirements for Software Validation Quality System Regulation vs Pre-Market Submissions... 4 SECTION 3. CONTEXT FOR SOFTWARE VALIDATION Definitions and Terminology Requirements and Specifications Verification and Validation IQ/OQ/PQ Software Development as Part of System Design Software is Different from Hardware Benefits of Software Validation Design Review... 9 SECTION 4. PRINCIPLES OF SOFTWARE VALIDATION Requirements Defect Prevention Time and Effort Software Life Cycle Plans Procedures Software Validation After a Change Validation Coverage Independence of Review Page iii4 4.10. Flexibility and Responsibility SECTION 5. ACTIVITIES AND TASKS Software Life Cycle Activities Typical Tasks Supporting Validation Quality Planning Requirements Design Construction or Coding Testing by the Software Developer User Site Testing Maintenance and Software Changes SECTION 6. VALIDATION OF AUTOMATED PROCESS EQUIPMENT AND QUALITY SYSTEM SOFTWARE How Much Validation Evidence Is Needed? Defined User Requirements Validation of Off-the-Shelf Software and Automated Equipment APPENDIX A - REFERENCES Food and Drug Administration References Other Government References International and National Consensus Standards Production Process Software References General Software Quality References APPENDIX B - DEVELOPMENT TEAM Page iv5 This document is intended to provide guidance. It represents the Agency s current thinking on this topic. It does not create or confer any rights for or on any person and does not operate to bind Food and Drug Administration (FDA) or the public. An alternative approach may be used if such approach satisfies the requirements of the applicable statutes and regulations. SECTION 1. PURPOSE This guidance outlines general validation principles that the Food and Drug Administration (FDA) considers to be applicable to the validation of medical device software or the validation of software used to design, develop, or manufacture medical devices. This final guidance document, Version 2.0, supersedes the draft document,, Version 1.1, dated June 9, SECTION 2. SCOPE This guidance describes how certain provisions of the medical device Quality System regulation apply to software and the agency s current approach to evaluating a software validation system. For example, this document lists elements that are acceptable to the FDA for the validation of software; however, it does not list all of the activities and tasks that must, in all instances, be used to comply with the law. The scope of this guidance is somewhat broader than the scope of validation in the strictest definition of that term. Planning, verification, testing, traceability, configuration management, and many other aspects of good software engineering discussed in this guidance are important activities that together help to support a final conclusion that software is validated. This guidance recommends an integration of software life cycle management and risk management activities. Based on the intended use and the safety risk associated with the software to be developed, the software developer should determine the specific approach, the combination of techniques to be used, and the level of effort to be applied. While this guidance does not recommend any specific life cycle model or any specific technique or method, it does recommend that software validation and verification activities be conducted throughout the entire software life cycle. Where the software is developed by someone other than the device manufacturer (e.g., off-the-shelf software) the software developer may not be directly responsible for compliance with FDA regulations. Page 16 In that case, the party with regulatory responsibility (i.e., the device manufacturer) needs to assess the adequacy of the off-the-shelf software developer s activities and determine what additional efforts are needed to establish that the software is validated for the device manufacturer s intended use APPLICABILITY This guidance applies to: Software used as a component, part, or accessory of a medical device; Software that is itself a medical device (e.g., blood establishment software); Software used in the production of a device (e.g., programmable logic controllers in manufacturing equipment); and Software used in implementation of the device manufacturer's quality system (e.g., software that records and maintains the device history record). This document is based on generally recognized software validation principles and, therefore, can be applied to any software. For FDA purposes, this guidance applies to any software related to a regulated medical device, as defined by Section 201(h) of the Federal Food, Drug, and Cosmetic Act (the Act) and by current FDA software and regulatory policy. This document does not specifically identify which software is or is not regulated AUDIENCE This guidance provides useful information and recommendations to the following individuals: Persons subject to the medical device Quality System regulation Persons responsible for the design, development, or production of medical device software Persons responsible for the design, development, production, or procurement of automated tools used for the design, development, or manufacture of medical devices or software tools used to implement the quality system itself FDA Investigators FDA Compliance Officers FDA Scientific Reviewers 2.3. THE LEAST BURDENSOME APPROACH We believe we should consider the least burdensome approach in all areas of medical device regulation. This guidance reflects our careful review of the relevant scientific and legal requirements and what we believe is the least burdensome way for you to comply with those requirements. However, if you believe that an alternative approach would be less burdensome, please contact us so we can consider Page 27 your point of view. You may send your written comments to the contact person listed in the preface to this guidance or to the CDRH Ombudsman. Comprehensive information on CDRH s Ombudsman, including ways to contact him, can be found on the Internet at: REGULATORY REQUIREMENTS FOR SOFTWARE VALIDATION The FDA s analysis of 3140 medical device recalls conducted between 1992 and 1998 reveals that 242 of them (7.7%) are attributable to software failures. Of those software related recalls, 192 (or 79%) were caused by software defects that were introduced when changes were made to the software after its initial production and distribution. Software validation and other related good software engineering practices discussed in this guidance are a principal means of avoiding such defects and resultant recalls. Software validation is a requirement of the Quality System regulation, which was published in the Federal Register on October 7, 1996 and took effect on June 1, (See Title 21 Code of Federal Regulations (CFR) Part 820, and 61 Federal Register (FR) 52602, respectively.) Validation requirements apply to software used as components in medical devices, to software that is itself a medical device, and to software used in production of the device or in implementation of the device manufacturer's quality system. Unless specifically exempted in a classification regulation, any medical device software product developed after June 1, 1997, regardless of its device class, is subject to applicable design control provisions. (See of 21 CFR ) This requirement includes the completion of current development projects, all new development projects, and all changes made to existing medical device software. Specific requirements for validation of device software are found in 21 CFR (g). Other design controls, such as planning, input, verification, and reviews, are required for medical device software. (See 21 CFR ) The corresponding documented results from these activities can provide additional support for a conclusion that medical device software is validated. Any software used to automate any part of the device production process or any part of the quality system must be validated for its intended use, as required by 21 CFR (i). This requirement applies to any software used to automate device design, testing, component acceptance, manufacturing, labeling, packaging, distribution, complaint handling, or to automate any other aspect of the quality system. In addition, computer systems used to create, modify, and maintain electronic records and to manage electronic signatures are also subject to the validation requirements. (See 21 CFR 11.10(a).) Such computer systems must be validated to ensure accuracy, reliability, consistent intended performance, and the ability to discern invalid or altered records. Page 38 Software for the above applications may be developed in-house or under contract. However, software is frequently purchased off-the-shelf for a particular intended use. All production and/or quality system software, even if purchased off-the-shelf, should have documented requirements that fully define its intended use, and information against which testing results and other evidence can be compared, to show that the software is validated for its intended use. The use of off-the-shelf software in automated medical devices and in automated manufacturing and quality system operations is increasing. Off-the-shelf software may have many capabilities, only a few of which are needed by the device manufacturer. Device manufacturers are responsible for the adequacy of the software used in their devices, and used to produce devices. When device manufacturers purchase "off-the-shelf'' software, they must ensure that it will perform as intended in their chosen application. For off-the-shelf software used in manufacturing or in the quality system, additional guidance is included in Section 6.3 of this document. For device software, additional useful information may be found in FDA s Guidance for Industry, FDA Reviewers, and Compliance on Off-The-Shelf Software Use in Medical Devices QUALITY SYSTEM REGULATION VS PRE-MARKET SUBMISSIONS This document addresses Quality System regulation issues that involve the implementation of software validation. It provides guidance for the management and control of the software validation process. The management and control of the software validation process should not be confused with any other validation requirements, such as process validation for an automated manufacturing process. Device manufacturers may use the same procedures and records for compliance with quality system and design control requirements, as well as for pre-market submissions to FDA. This document does not cover any specific safety or efficacy issues related to software validation. Design issues and documentation requirements for pre-market submissions of regulated software are not addressed by this document. Specific issues related to safety and efficacy, and the documentation required in pre-market submissions, should be addressed to the Office of Device Evaluation (ODE), Center for Devices and Radiological Health (CDRH) or to the Office of Blood Research and Review, Center for Biologics Evaluation and Research (CBER). See the references in Appendix A for applicable FDA guidance documents for pre-market submissions. Page 49 SECTION 3. CONTEXT FOR SOFTWARE VALIDATION Many people have asked for specific guidance on what FDA expects them to do to ensure compliance with the Quality System regulation with regard to software validation. Information on software validation presented in this document is not new. Validation of software, using the principles and tasks listed in Sections 4 and 5, has been conducted in many segments of the software industry for well over 20 years. Due to the great variety of medical devices, processes, and manufacturing facilities, it is not possible to state in one document all of the specific validation elements that are applicable. However, a general application of several broad concepts can be used successfully as guidance for software validation. These broad concepts provide an acceptable framework for building a comprehensive approach to software validation. Additional specific information is available from many of the references listed in Appendix A DEFINITIONS AND TERMINOLOGY Unless defined in the Quality System regulation, or otherwise specified below, all other terms used in this guidance are as defined in the current edition of the FDA Glossary of Computerized System and Software Development Terminology. The medical device Quality System regulation (21 CFR 820.3(k)) defines "establish" to mean "define, document, and implement." Where it appears in this guidance, the words "establish" and established should be interpreted to have this same meaning. Some definitions found in the medical device Quality System regulation can be confusing when compared to commonly used terminology in the software industry. Examples are requirements, specification, verification, and validation Requirements and Specifications While the Quality System regulation states that design input requirements must be documented, and that specified requirements must be verified, the regulation does not further clarify the distinction between the terms requirement and specification. A requirement can be any need or expectation for a system or for its software. Requirements reflect the stated or implied needs of the customer, and may be market-based, contractual, or statutory, as well as an organization's internal requirements. There can be many different kinds of requirements (e.g., design, functional, implementation, interface, performance, or physical requirements). Software requirements are typically derived from the system requirements for those aspects of system functionality that have been allocated to software. Software requirements are typically stated in functional terms and are defined, refined, and updated as a development project progresses. Success in accurately and completely documenting software requirements is a crucial factor in successful validation of the resulting software. Page 510 A specification is defined as a document that states requirements. (See 21 CFR 820.3(y).) It may refer to or include drawings, patterns, or other relevant documents and usually indicates the means and the criteria whereby conformity with the requirement can be checked. There are many different kinds of written specifications, e.g., system requirements specification, software requirements specification, software design specification, software test specification, software integration specification, etc. All of these documents establish specified requirements and are design outputs for which various forms of verification are necessary Verification and Validation The Quality System regulation is harmonized with ISO 8402:1994, which treats verification and validation as separate and distinct terms. On the other hand, many software engineering journal articles and textbooks use the terms "verification" and "validation" interchangeably, or in some cases refer to software "verification, validation, and testing (VV&T)" as if it is a single concept, with no distinction among the three terms. Software verification provides objective evidence that the design outputs of a particular phase of the software development life cycle meet all of the specified requirements for that phase. Software verification looks for consistency, completeness, and correctness of the software and its supporting documentation, as it is being developed, and provides support for a subsequent conclusion that software is validated. Software testing is one of many verification activities intended to confirm that software development output meets its input requirements. Other verification activities include various static and dynamic analyses, code and document inspections, walkthroughs, and other techniques. Software validation is a part of the design validation for a finished device, but is not separately defined in the Quality System regulation. For purposes of this guidance, FDA considers software validation to be confirmation by examination and provision of objective evidence that software specifications conform to user needs and intended uses, and that the particular requirements implemented through software can be consistently fulfilled. In practice, software validation activities may occur both during, as well as at the end of the software development life cycle to ensure that all requirements have been fulfilled. Since software is usually part of a larger hardware system, the validation of software typically includes evidence that all software requirements have been implemented correctly and completely and are traceable to system requirements. A conclusion that software is validated is highly dependent upon comprehensive software testing, inspections, analyses, and other verification tasks performed at each stage of the software development life cycle. Testing of device software functionality in a simulated use environment, and user site testing are typically included as components of an overall design validation program for a software automated device. Software verification and validation are difficult because a developer cannot test forever, and it is hard to know how much evidence is enough. In large measure, software validation is a matter of developing a level of confidence that the device meets all requirements and user expectations for the software automated functions and features of the device. Measures such as defects found in specifications documents, estimates of defects remaining, testing coverage, and other techniques are all used to Page 611 develop an acceptable level of confidence before shipping the product. The level of confidence, and therefore the level of software validation, verification, and testing effort needed, will vary depending upon the safety risk (hazard) posed by the automated functions of the device. Additional guidance regarding safety risk management for software may be found in Section 4 of FDA s Guidance for the Content of Pre-market Submissions for Software Contained in Medical Devices, and in the international standards ISO/IEC and IEC referenced in Appendix A IQ/OQ/PQ For many years, both FDA and regulated industry have attempted to understand and define software validation within the context of process validation terminology. For example, industry documents and other FDA validation guidance sometimes describe user site software validation in terms of installation qualification (IQ), operational qualification (OQ) and performance qualification (PQ). Definitions of these terms and additional information regarding IQ/OQ/PQ may be found in FDA s Guideline on General Principles of Process Validation, dated May 11, 1987, and in FDA s Glossary of Computerized System and Software Development Terminology, dated August While IQ/OQ/PQ terminology has served its purpose well and is one of many legitimate ways to organize software validation tasks at the user site, this terminology may not be well understood among many software professionals, and it is not used elsewhere in this document. However, both FDA personnel and device manufacturers need to be aware of these differences in terminology as they ask for and provide information regarding software validation SOFTWARE DEVELOPMENT AS PART OF SYSTEM DESIGN The decision to implement system functionality using software is one that is typically made during system design. Software requirements are typically derived from the overall system requirements and design for those aspects in the system that are to be implemented using software. There are user needs and intended uses for a finished device, but users typically do not specify whether those requirements are to be met by hardware, software, or some combination of both. Therefore, software validation must be considered within the context of the overall design validation for the system. A documented requirements specification represents the user's needs and intended uses from which the product is developed. A primary goal of software validation is to then demonstrate that all completed software products comply with all documented software and system requirements. The correctness and completeness of both the system requirements and the software requirements should be addressed as part of the design validation process for the device. Software validation includes confirmation of conformance to all software specifications and confirmation that all software requirements are traceable to the system specifications. Confirmation is an important part of the overall design validation to ensure that all aspects of the medical device conform to user needs and intended uses. Page 712 3.3. SOFTWARE IS DIFFERENT FROM HARDWARE While software shares many of the same engineering tasks as hardware, it has some very important differences. For example: The vast majority of software problems are traceable to errors made during the design and development process. While the quality of a hardware product is highly dependent on design, development and manufacture, the quality of a software product is dependent primarily on design and development with a minimum concern for software manufacture. Software manufacturing consists of reproduction that can be easily verified. It is not difficult to manufacture thousands of program copies that function exactly the same as the original; the difficulty comes in getting the original program to meet all specifications. One of the most significant features of software is branching, i.e., the ability to execute alternative series of commands, based on differing inputs. This feature is a major contributing factor for another characteristic of software its complexity. Even short programs can be very complex and difficult to fully understand. Typically, testing alone cannot fully verify that software is complete and correct. In addition to testing, other verification techniques and a structured and documented development process should be combined to ensure a comprehensive validation approach. Unlike hardware, software is not a physical entity and does not wear out. In fact, software may improve with age, as latent defects are discovered and removed. However, as software is constantly updated and changed, such improvements are sometimes countered by new defects introduced into the software during the change. Unlike some hardware failures, software failures occur without advanced warning. The software s branching that allows it to follow differing paths during execution, may hide some latent defects until long after a software product has been introduced into the marketplace. Another related characteristic of software is the speed and ease with which it can be changed. This factor can cause both software and non-software professionals to believe that software problems can be corrected easily. Combined with a lack of understanding of software, it can lead managers to believe that tightly controlled engineering is not needed as much for software as it is for hardware. In fact, the opposite is true. Because of its complexity, the development process for software should be even more tightly controlled than for hardware, in order to prevent problems that cannot be easily detected later in the development process. Seemingly insignificant changes in software code can create unexpected and very significant problems elsewhere in the software program. The software development process should be sufficiently well planned, controlled, and documented to detect and correct unexpected results from software changes. Page 813 Given the high demand for software professionals and the highly mobile workforce, the software personnel who make maintenance changes to software may not have been involved in the original software development. Therefore, accurate and thorough documentation is essential. Historically, software components have not been as frequently standardized and interchangeable as hardware components. However, medical device software developers are beginning to use component-based development tools and techniques. Object-oriented methodologies and the use of off-the-shelf software components hold promise for faster and less expensive software development. However, component-based approaches require very careful attention during integration. Prior to integration, time is needed to fully define and develop reusable software code and to fully understand the behavior of off-the-shelf components. For these and other reasons, software engineering needs an even greater level of managerial scrutiny and control than does hardware engineering BENEFITS OF SOFTWARE VALIDATION Software validation is a critical tool used to assure the quality of device software and software automated operations. Software validation can increase the usability and reliability of the device, resulting in decreased failure rates, fewer recalls and corrective actions, less risk to patients and users, and reduced liability to device manufacturers. Software validation can also reduce long term costs by making it easier and less costly to reliably modify software and revalidate software changes. Software maintenance can represent a very large percentage of the total cost of software over its entire life cycle. An established comprehensive software validation process helps to reduce the long-term cost of software by reducing the cost of validation for each subsequent release of the software. 3.5 DESIGN REVIEW Design reviews are documented, comprehensive, and systematic examinations of a design to evaluate the adequacy of the design requirements, to evaluate the capability of the design to meet these requirements, and to identify problems. While there may be many informal technical reviews that occur within the development team during a software project, a formal design review is more structured and includes participation from others outside the development team. Formal design reviews may reference or include results from other formal and informal reviews. Design reviews may be conducted separately for the software, after the software is integrated with the hardware into the system, or both. Design reviews should include examination of development plans, requirements specifications, design specifications, testing plans and procedures, all other documents and activities associated with the project, verification results from each stage of the defined life cycle, and validation results for the overall device. Design review is a primary tool for managing and evaluating development projects. For example, formal design reviews allow management to confirm that all goals defined in the software validation plan have Page 914 been achieved. The Quality System regulation requires that at least one formal design review be conducted during the device design process. However, it is recommended that multiple design reviews be conducted (e.g., at the end of each software life cycle activity, in preparation for proceeding to the next activity). Formal design review is especially important at or near the end of the requirements activity, before major resources have been committed to specific design solutions. Problems found at this point can be resolved more easily, save time and money, and reduce the likelihood of missing a critical issue. Answers to some key questions should be documented during formal design reviews. These include: Have the appropriate tasks and expected results, outputs, or products been established for each software life cycle activity? Do the tasks and expected results, outputs, or products of each software life cycle activity: Comply with the requirements of other software life cycle activities in terms of correctness, completeness, consistency, and accuracy? Satisfy the standards, practices, and conventions of that activity? Establish a proper basis for initiating tasks for the next software life cycle activity? Page 1015 SECTION 4. PRINCIPLES OF SOFTWARE VALIDATION This section lists the general principles that should be considered for the validation of software REQUIREMENTS A documented software requirements specification provides a baseline for both validation and verification. The software validation process cannot be completed without an established software requirements specification (Ref: 21 CFR 820.3(z) and (aa) and (f) and (g)) DEFECT PREVENTION Software quality assurance needs to focus on preventing the introduction of defects into the software development process and not on trying to test quality into the software code after it is written. Software testing is very limited in its ability to surface all latent defects in software code. For example, the complexity of most software prevents it from being exhaustively tested. Software testing is a necessary activity. However, in most cases software testing by itself is not sufficient to establish confidence that the software is fit for its intended use. In order to establish that confidence, software developers should use a mixture of methods and techniques to prevent software errors and to detect software errors that do occur. The best mix of methods depends on many factors including the development environment, application, size of project, language, and risk TIME AND EFFORT To build a case that the software is validated requires time and effort. Preparation for software validation should begin early, i.e., during design and development planning and design input. The final conclusion that the software is validated should be based on evidence collected from planned efforts conducted throughout the software lifecycle SOFTWARE LIFE CYCLE Software validation takes place within the environment of an established software life cycle. The software life cycle contains software engineering tasks and documentation necessary to support the software validation effort. In addition, the software life cycle contains specific verification and validation tasks that are appropriate for the intended use of the software. This guidance does not recommend any particular life cycle models only that they should be selected and used for a software development project. Page 1116 4.5. PLANS The software validation process is defined and controlled through the use of a plan. The software validation plan defines what is to be accomplished through the software validation effort. Software validation plans are a significant quality system tool. Software validation plans specify areas such as scope, approach, resources, schedules and the types and extent of activities, tasks, and work items PROCEDURES The software validation process is executed through the use of procedures. These procedures establish how to conduct the software validation effort. The procedures should identify the specific actions or sequence of actions that must be taken to complete individual validation activities, tasks, and work items SOFTWARE VALIDATION AFTER A CHANGE Due to the complexity of software, a seemingly small local change may have a significant global system impact. When any change (even a small change) is made to the software, the validation status of the software needs to be re-established. Whenever software is changed, a validation analysis should be conducted not just for validation of the individual change, but also to determine the extent and impact of that change on the entire software system. Based on this analysis, the software developer should then conduct an appropriate level of software regression testing to show that unchanged but vulnerable portions of the system have not been adversely affected. Design controls and appropriate regression testing provide the confidence that the software is validated after a software change VALIDATION COVERAGE Validation coverage should be based on the software s complexity and safety risk not on firm size or resource constraints. The selection of validation activities, tasks, and work items should be commensurate with the complexity of the software design and the risk associated with the use of the software for the specified intended use. For lower risk devices, only baseline validation activities may be conducted. As the risk increases additional validation activities should be added to cover the additional risk. Validation documentation should be sufficient to demonstrate that all software validation plans and procedures have been completed successfully INDEPENDENCE OF REVIEW Validation activities should be conducted using the basic quality assurance precept of independence of review. Self-validation is extremely difficult. When possible, an independent evaluation is always better, especially for higher risk applications. Some firms contract out for a third-party independent Page 1217 verification and validation, but this solution may not always be feasible. Another approach is to assign internal staff members that are not involved in a particular design or its implementation, but who have sufficient knowledge to evaluate the project and conduct the verification and validation activities. Smaller firms may need to be creative in how tasks are organized and assigned in order to maintain internal independence of review FLEXIBILITY AND RESPONSIBILITY Specific implementation of these software validation principles may be quite different from one application to another. The device manufacturer has flexibility in choosing how to apply these validation principles, but retains ultimate responsibility for demonstrating that the software has been validated. Software is designed, developed, validated, and regulated in a wide spectrum of environments, and for a wide variety of devices with varying levels of risk. FDA regulated medical device applications include software that: Is a component, part, or accessory of a medical device; Is itself a medical device; or Is used in manufacturing, design and development, or other parts of the quality system. In each environment, software components from many sources may be used to create the application (e.g., in-house developed software, off-the-shelf software, contract software, shareware). In addition, software components come in many different forms (e.g., application software, operating systems, compilers, debuggers, configuration management tools, and many more). The validation of software in these environments can be a complex undertaking; therefore, it is appropriate that all of these software validation principles be considered when designing the software validation process. The resultant software validation process should be commensurate with the safety risk associated with the system, device, or process. Software validation activities and tasks may be dispersed, occurring at different locations and being conducted by different organizations. However, regardless of the distribution of tasks, contractual relations, source of components, or the development environment, the device manufacturer or specification developer retains ultimate responsibility for ensuring that the software is validated. Page 1318 SECTION 5. ACTIVITIES AND TASKS Software validation is accomplished through a series of activities and tasks that are planned and executed at various stages of the software development life cycle. These tasks may be one time occurrences or may be iterated many times, depending on the life cycle model used and the scope of changes made as the software project progresses SOFTWARE LIFE CYCLE ACTIVITIES This guidance does not recommend the use of any specific software life cycle model. Software developers should establish a software life cycle model that is appropriate for their product and organization. The software life cycle model that is selected should cover the software from its birth to its retirement. Activities in a typical software life cycle model include the following: Quality Planning System Requirements Definition Detailed Software Requirements Specification Software Design Specification Construction or Coding Testing Installation Operation and Support Maintenance Retirement Verification, testing, and other tasks that support software validation occur during each of these activities. A life cycle model organizes these software development activities in various ways and provides a framework for monitoring and controlling the software development project. Several software life cycle models (e.g., waterfall, spiral, rapid prototyping, incremental development, etc.) are defined in FDA s Glossary of Computerized System and Software Development Terminology, dated August These and many other life cycle models are described in various references listed in Appendix A TYPICAL TASKS SUPPORTING VALIDATION For each of the software life cycle activities, there are certain typical tasks that support a conclusion that the software is validated. However, the specific tasks to be performed, their order of performance, and the iteration and timing of their performance will be dictated by the specific software life cycle model that is selected and the safety risk associated with the software application. For very low risk applications, certain tasks may not be needed at all. However, the software developer should at least consider each of these tasks and should define and document which tasks are or are not appropriate for Page 1419 their specific application. The following discussion is generic and is not intended to prescribe any particular software life cycle model or any particular order in which tasks are to be performed Quality Planning Design and development planning should culminate in a plan that identifies necessary tasks, procedures for anomaly reporting and resolution, necessary resources, and management review requirements, including formal design reviews. A software life cycle model and associated activities should be identified, as well as those tasks necessary for each software life cycle activity. The plan should include: The specific tasks for each life cycle activity; Enumeration of important quality factors (e.g., reliability, maintainability, and usability); Methods and procedures for each task; Task acceptance criteria; Criteria for defining and documenting outputs in terms that will allow evaluation of their conformance to input requirements; Inputs for each task; Outputs from each task; Roles, resources, and responsibilities for each task; Risks and assumptions; and Documentation of user needs. Management must identify and provide the appropriate software development environment and resources. (See 21 CFR (b)(1) and (2).) Typically, each task requires personnel as well as physical resources. The plan should identify the personnel, the facility and equipment resources for each task, and the role that risk (hazard) management will play. A configuration management plan should be developed that will guide and control multiple parallel development activities and ensure proper communications and documentation. Controls are necessary to ensure positive and correct correspondence among all approved versions of the specifications documents, source code, object code, and test suites that comprise a software system. The controls also should ensure accurate identification of, and access to, the currently approved versions. Procedures should be created for reporting and resolving software anomalies found through validation or other activities. Management should identify the reports and specify the contents, format, and responsible organizational elements for each report. Procedures also are necessary for the review and approval of software development results, including the responsible organizational elements for such reviews and approvals. Typical Tasks Quality Planning Risk (Hazard) Management Plan Configuration Management Plan Page 1520 Software Quality Assurance Plan Software Verification and Validation Plan Verification and Validation Tasks, and Acceptance Criteria Schedule and Resource Allocation (for software verification and validation activities) Reporting Requirements Formal Design Review Requirements Other Technical Review Requirements Problem Reporting and Resolution Procedures Other Support Activities Requirements Requirements development includes the identification, analysis, and documentation of information about the device and its intended use. Areas of special importance include allocation of system functions to hardware/software, operating conditions, user characteristics, potential hazards, and anticipated tasks. In addition, the requirements should state clearly the intended use of the software. The software requirements specification document should contain a written definition of the software functions. It is not possible to validate software without predetermined and documented software requirements. Typical software requirements specify the following: All software system inputs; All software system outputs; All functions that the software system will perform; All performance requirements that the software will meet, (e.g., data throughput, reliability, and timing); The definition of all external and user interfaces, as well as any internal software-to-system interfaces; How users will interact with the system; What constitutes an error and how errors should be handled; Required response times; The intended operating environment for the software, if this is a design constraint (e.g., hardware platform, operating system); All ranges, limits, defaults, and specific values that the software will accept; and All safety related requirements, specifications, features, or functions that will be implemented in software. Software safety requirements are derived from a technical risk management process that is closely integrated with the system requirements development process. Software requirement specifications should identify clearly the potential hazards that can result from a software failure in the system as well as any safety requirements to be implemented in software. The consequences of software failure should be evaluated, along with means of mitigating such failures (e.g., hardware mitigation, defensive programming, etc.). From this analysis, it should be possible to identify the most appropriate measures necessary to prevent harm. Page 16 View more
General Principles of Software Validation; Final Guidance for Industry and FDA Staff Document issued on: January 11, 2002 This document supersedes the draft document, "General Principles of Software Validation, More information Medical Device Software - Software Life Cycle Processes
1 Medical Device Software - Software Life Cycle Processes IEC 62304 2 Credits John F. Murray Software Compliance Expert U.S. Food and Drug Administration Marcie R. Williams Medical Device Fellow Ph.D. More information Using TechExcel s DevSuite to Achieve FDA Software Validation Compliance For Medical Software Device Development
Using TechExcel s DevSuite to Achieve FDA Software Validation Compliance For Medical Software Device Development The FDA requires medical software development teams to comply with its standards for software More information INTRODUCTION. This book offers a systematic, ten-step approach, from the decision to validate to
INTRODUCTION This book offers a systematic, ten-step approach, from the decision to validate to the assessment of the validation outcome, for validating configurable off-the-shelf (COTS) computer software More information Document issued on: May 11, 2005
Guidance for Industry and FDA Staff Guidance for the Content of Premarket Submissions for Software Contained in Medical Devices Document issued on: May 11, 2005 This document supersedes Guidance for the More information Contains Nonbinding Recommendations
Guidance for Industry and FDA Staff Laser Products Conformance with IEC 60825-1 and IEC 60601-2-22; Guidance for Industry and FDA Staff (Laser Notice No. 50) Document issued on June 24, 2007 This document More information Effective Software Verification for Medical Devices
STERLINGTECH AND KLOCWORK WHITE PAPER NOVEMBER 2009 Effective Software Verification for Medical Devices Achieving compliance and meeting productivity goals with static analysis In addition to producing More information 8. Master Test Plan (MTP)
8. Master Test Plan (MTP) The purpose of the Master Test Plan (MTP) is to provide an overall test planning and test management document for multiple levels of test (either within one project or across More information Overview of Medical Device Design Controls in the US. By Nandini Murthy, MS, RAC
Overview of Medical Device Controls in the US By Nandini Murthy, MS, RAC 18 controls are a regulatory requirement for medical devices. In the US, compliance with the design controls section of 21 Code More information Quality System: Design Control Procedure - Appendix
Quality System: Design Control Procedure - Appendix Page 1 of 10 Quality System: Design Control Procedure - Appendix CORP Medical Products Various details have been removed, indicated by [ ] 1. Overview More information R214 SPECIFIC REQUIREMENTS: INFORMATION TECHNOLOGY TESTING LABORATORY ACCREDITATION PROGRAM
The American Association for Laboratory Accreditation Document Revised: R214: Specific Requirements: Information Technology Testing Laboratory Accreditation July 13, 2010 Program Page 1 of 26 R214 SPECIFIC More information Medical Device Data Systems, Medical Image Storage Devices, and Medical Image Communications Devices
Medical Device Data Systems, Medical Image Storage Devices, and Medical Image Communications Devices Draft Guidance for Industry and Food and Drug Administration Staff DRAFT GUIDANCE This guidance document More information DRAFT REGULATORY GUIDE
U.S. NUCLEAR REGULATORY COMMISSION August 2012 OFFICE OF NUCLEAR REGULATORY RESEARCH Division 1 DRAFT REGULATORY GUIDE Contact: K. Sturzebecher (301) 251-7494 DRAFT REGULATORY GUIDE DG-1206 (Proposed Revision More information SOFTWARE DEVELOPMENT STANDARD FOR SPACECRAFT
SOFTWARE DEVELOPMENT STANDARD FOR SPACECRAFT Mar 31, 2014 Japan Aerospace Exploration Agency This is an English translation of JERG-2-610. Whenever there is anything ambiguous in this document, the original More information Risk Assessment for Medical Devices. Linda Braddon, Ph.D. Bring your medical device to market faster 1
Risk Assessment for Medical Devices Linda Braddon, Ph.D. Bring your medical device to market faster 1 My Perspective Work with start up medical device companies Goal: Making great ideas into profitable More information Medical Device Software Verification, Validation, and Compliance
Medical Device Software Verification, Validation, and Compliance David A. Vogel ARTECH HOUSE BOSTON LONDON artechhouse.com Contents Preface The Author's Background and Perspective of Validation Acknowledgments More information Guidance for Industry
Guidance for Industry Electronic Source Data in Clinical Investigations DRAFT GUIDANCE This guidance document is being distributed for comment purposes only. Comments and suggestions regarding this draft More information Developing a Mobile Medical App? How to determine if it is a medical device and get it cleared by the US FDA
Developing a Mobile Medical App? How to determine if it is a medical device and get it cleared by the US FDA In this presentation: App stats: Explosive growth Examples already cleared by the US FDA Is More information Your Software Quality is Our Business. INDEPENDENT VERIFICATION AND VALIDATION (IV&V) WHITE PAPER Prepared by Adnet, Inc.
INDEPENDENT VERIFICATION AND VALIDATION (IV&V) WHITE PAPER Prepared by Adnet, Inc. February 2013 1 Executive Summary Adnet is pleased to provide this white paper, describing our approach to performing More information Medical Device Software Standards for Safety and Regulatory Compliance
Medical Device Software Standards for Safety and Regulatory Compliance Sherman Eagles +1 612-865-0107 seagles@softwarecpr.com www.softwarecpr.com Assuring safe software SAFE All hazards have been addressed More information Contains Nonbinding Recommendations. Document issued on October 20, 2006
Guidance for Industry and FDA Staff Exemption from Certain Reporting and Recordkeeping Requirements for Television Receivers and Computer Monitors with Cathode Ray Tubes Document issued on October 20, More information Software Engineering. Software Processes. Based on Software Engineering, 7 th Edition by Ian Sommerville
Software Engineering Software Processes Based on Software Engineering, 7 th Edition by Ian Sommerville Objectives To introduce software process models To describe three generic process models and when More information Guidance for Industry Part 11, Electronic Records; Electronic Signatures Scope and Application
Guidance for Industry Part 11, Electronic Records; Electronic Signatures Scope and Application U.S. Department of Health and Human Services Food and Drug Administration Center for Drug Evaluation and Research More information (Refer Slide Time: 01:52)
Software Engineering Prof. N. L. Sarda Computer Science & Engineering Indian Institute of Technology, Bombay Lecture - 2 Introduction to Software Engineering Challenges, Process Models etc (Part 2) This More information The FDA Perspective on Human Factors in Medical Device Software Development
The FDA Perspective on in Medical Device Development Molly Follette Story, PhD FDA /CDRH / ODE 2012 IQPC Design for Medical Devices Europe Munich, Germany Februar y1, 2012 Overview Guidance for FDA premarket More information Use of Electronic Health Record Data in Clinical Investigations
Use of Electronic Health Record Data in Clinical Investigations Guidance for Industry DRAFT GUIDANCE This guidance document is being distributed for comment purposes only. Comments and suggestions regarding More information Montana Department of Transportation Information Services Division. System Development Life Cycle (SDLC) Guide
Montana Department of Transportation Information Services Division System Development Life Cycle (SDLC) Guide Version 2 August 2, 2007 \mdt_sdlc_process\mdt_sdlc_v02.doc Table of Contents 1 Business Analysis...3 More information Computer System Validation - It s More Than Just Testing
Computer System Validation - It s More Than Just Testing Introduction Computer System Validation is the technical discipline that Life Science companies use to ensure that each Information Technology application More information Guidance for Industry. 21 CFR Part 11; Electronic. Records; Electronic Signatures. Time Stamps
Guidance for Industry 21 CFR Part 11; Electronic Records; Electronic Signatures Time Stamps Draft Guidance This guidance document is being distributed for comment purposes only. Comments and suggestions More information Fundamentals of Measurements
Objective Software Project Measurements Slide 1 Fundamentals of Measurements Educational Objective: To review the fundamentals of software measurement, to illustrate that measurement plays a central role More information Medical Product Software Development and FDA Regulations Software Development Practices and FDA Compliance
Medical Product Development and FDA Regulations IEEE Orange County Computer Society March 27, 2006 Carl R. Wyrwa Medical Product Development and FDA Regulations Introduction Regulated FDA Overview Medical More information ORACLE CONSULTING GROUP
ORACLE CONSULTING GROUP An Official United States Agent Firm for Foreign Establishments CONSULTING MEMORANDUM: DEALING WITH A MEDICAL DEVICE IN THE U.S. 5398 Golder Ranch Rd., Ste. 1 Tucson, Arizona 85739 More information To introduce software process models To describe three generic process models and when they may be used
Software Processes Objectives To introduce software process models To describe three generic process models and when they may be used To describe outline process models for requirements engineering, software More information AC 20-148 REUSABLE SOFTWARE COMPONENTS
AC 20-148 REUSABLE SOFTWARE COMPONENTS December 7, 2004 12/7/04 AC 20-148 CONTENTS Paragraph Title Page 1. Purpose....1 2. Motivation for this Guidance....1 3. Document Overview...1 4. General Guidelines More information How to Use the Design Process to Manage Risk: Elements of Design Controls and Why It Matters
environmental failure analysis & prevention health technology development How to Use the Design Process to Manage Risk: Elements of Design Controls and Why It Matters Kevin L. Ong, Ph.D., P.E. Managing More information Guidance for Industry
Guidance for Industry Electronic Source Data in Clinical Investigations U.S. Department of Health and Human Services Food and Drug Administration Center for Drug Evaluation and Research (CDER) Center for More information Clinical Risk Management: Agile Development Implementation Guidance
Document filename: Directorate / Programme Document Reference NPFIT-FNT-TO-TOCLNSA-1306.02 CRM Agile Development Implementation Guidance v1.0 Solution Design Standards and Assurance Project Clinical Risk More information CORPORATE QUALITY MANUAL
Corporate Quality Manual Preface The following Corporate Quality Manual is written within the framework of ISO 9001:2008 Quality System by the employees of CyberOptics. CyberOptics recognizes the importance More information Software Project Management Plan
The document in this file is adapted from the IEEE standards for Software Project Management Plans, 1058-1998, which conforms to the requirements of ISO standard 12207 Software Life Cycle Processes. Tailor More information Considerations When Validating Your Analyst Software Per GAMP 5
WHITE PAPER Analyst Software Validation Service Considerations When Validating Your Analyst Software Per GAMP 5 Blair C. James, Stacy D. Nelson Introduction The purpose of this white paper is to assist More information Software Test Plan (STP) Template
(STP) Template Items that are intended to stay in as part of your document are in bold; explanatory comments are in italic text. Plain text is used where you might insert wording about your project. This More information Regulatory Guide 1.169 Configuration Management Plans for Digital Computer Software Used in Safety Systems of Nuclear Power Plants
Regulatory Guide 1.169Configuration Managemen... Page 1 of 10 September 1997 Regulatory Guide 1.169 Configuration Management Plans for Digital Computer Software Used in Safety Systems of Nuclear Power More information Haulsey Engineering, Inc. Quality Management System (QMS) Table of Contents
Haulsey Engineering, Inc. Quality Management System (QMS) Table of Contents 1.0 Introduction 1.1 Quality Management Policy and Practices 2.0 Quality System Components 2.1 Quality Management Plans 2.2 Quality More information System Development Life Cycle Guide
TEXAS DEPARTMENT OF INFORMATION RESOURCES System Development Life Cycle Guide Version 1.1 30 MAY 2008 Version History This and other Framework Extension tools are available on Framework Web site. Release More information Overview of the System Engineering Process. Prepared by
Overview of the System Engineering Process Prepared by Ed Ryen, PE Maintenance ITS March 2008 Introduction This document provides a high level look at the Systems Engineering Process for ITS projects. More information Guidance for Industry and FDA Staff Tonometers - Premarket Notification [510(k)] Submissions
Guidance for Industry and FDA Staff Tonometers - Premarket Notification [510(k)] Submissions Document issued on: March 27, 2006 For questions regarding this document, contact Everette Beers, Ph.D. at 240-276-4200 More information Guidance for Industry COMPUTERIZED SYSTEMS USED IN CLINICAL TRIALS
Guidance for Industry COMPUTERIZED SYSTEMS USED IN CLINICAL TRIALS U.S. Department of Health and Human Services Food and Drug Administration Center for Biologic Evaluation and Research (CBER) Center for More information Example Software Development Process.
Example Software Development Process. The example software development process is shown in Figure A. The boxes represent the software development process kernels. The Software Unit Testing, Software Component More information How to Develop an Effective Design Control Program: A Life-Cycle Multi-Phase Approach
GMP How to Develop an Effective Design Control Program: A Life-Cycle Multi-Phase Approach Jackelyn Rodriguez This article provides a life-cycle methodology in a multi-phase approach that includes an effective More information Certified Professional in Configuration Management Glossary of Terms
Certified Professional in Configuration Management Glossary of terms used in Configuration Management Issue 2007.07 Association of the International Certified Configuration Manager e.v. Copyright 2007, More information Guidance for Industry. Q10 Pharmaceutical Quality System
Guidance for Industry Q10 Pharmaceutical Quality System U.S. Department of Health and Human Services Food and Drug Administration Center for Drug Evaluation and Research (CDER) Center for Biologics Evaluation More information Guidance for Industry Quality Systems Approach to Pharmaceutical CGMP Regulations
Guidance for Industry Quality Systems Approach to Pharmaceutical CGMP Regulations U.S. Department of Health and Human Services Food and Drug Administration Center for Drug Evaluation and Research (CDER) More information Procedure for Assessment of System and Software
Doc. No: STQC IT/ Assessment/ 01, Version 1.0 Procedure for Assessment of System and Software May, 2014 STQC - IT Services STQC Directorate, Department of Electronics and Information Technology, Ministry More information Software testing. Objectives
Software testing cmsc435-1 Objectives To discuss the distinctions between validation testing and defect testing To describe the principles of system and component testing To describe strategies for generating More information Lecture 17: Requirements Specifications
Lecture 17: Requirements Specifications Why we need to write specifications Purpose and audience Choosing an appropriate size and formality Desiderata for Specifications Properties of good specifications More information FDA Software Validation-Answers to the Top Five Software Validation Questions
Whitepaper FDA Software Validation-Answers to the Top Five Software Validation Questions Author: Penny Goss, Penny Goss Technical Solutions The FDA (Food and Drug Administration) and IEC (International More information DNV GL Assessment Checklist ISO 9001:2015
DNV GL Assessment Checklist ISO 9001:2015 Rev 0 - December 2015 4 Context of the Organization No. Question Proc. Ref. Comments 4.1 Understanding the Organization and its context 1 Has the organization More information Basic Testing Concepts and Terminology
T-76.5613 Software Testing and Quality Assurance Lecture 2, 13.9.2006 Basic Testing Concepts and Terminology Juha Itkonen SoberIT Contents Realities and principles of Testing terminology and basic concepts More information Intland s Medical Template
Intland s Medical Template Traceability Browser Risk Management & FMEA Medical Wiki Supports compliance with IEC 62304, FDA Title 21 CFR Part 11, ISO 14971, IEC 60601 and more INTLAND codebeamer ALM is More information Guidance for Industry Computerized Systems Used in Clinical Investigations
Guidance for Industry Computerized Systems Used in Clinical Investigations U.S. Department of Health and Human Services Food and Drug Administration (FDA) Office of the Commissioner (OC) May 2007 Guidance More information BPA. Disclaimer. Applying Critical Thinking to the. Applications, Purchased Applications, and Web-Hosted Applications. RiskVal Life Cycle
BPA Applying Critical Thinking to the Validation of Spreadsheets, Access Applications, Purchased Applications, and Web-Hosted Applications IVT 11 th Annual Computer and Software Validation Conference April More information Software Project Models
INTERNATIONAL JOURNAL OF TECHNOLOGY ENHANCEMENTS AND EMERGING ENGINEERING RESEARCH, VOL 1, ISSUE 4 135 Software Project Models Abhimanyu Chopra, Abhinav Prashar, Chandresh Saini Email-abhinav.prashar@gmail.com, More information Introduction into IEC 62304 Software life cycle for medical devices
Introduction into IEC 62304 Software life cycle for medical devices Christoph Gerber 4. September 2008 SPIQ 9/5/2008 1 Agenda Current Picture Regulatory requirements for medical device software IEC 62304 More information Guidance for Industry. 21 CFR Part 11; Electronic Records; Electronic Signatures. Maintenance of Electronic Records
Draft Guidance for Industry -- Not For Implementation Guidance for Industry 21 CFR Part 11; Electronic Records; Electronic Signatures Maintenance of Electronic Records Draft Guidance This guidance document More information IVD Regulation Overview. Requirements to Assure Quality & Effectiveness
IVD Regulation Overview Requirements to Assure Quality & Effectiveness CLIAC Jan. 2002 Statutory and Regulatory Requirements Statute: Food, Drug, and Cosmetic Act Food and Drugs Act of 1906 Food and Drug More information Guidance for Industry
Guidance for Industry Providing Regulatory Submissions in Electronic Format Content of Labeling U.S. Department of Health and Human Services Food and Drug Administration Center for Drug Evaluation and More information ELECTROTECHNIQUE IEC INTERNATIONALE 61508-3 INTERNATIONAL ELECTROTECHNICAL
61508-3 ª IEC: 1997 1 Version 12.0 05/12/97 COMMISSION CEI ELECTROTECHNIQUE IEC INTERNATIONALE 61508-3 INTERNATIONAL ELECTROTECHNICAL COMMISSION Functional safety of electrical/electronic/ programmable More information The purpose of this Supplier Quality Standard is to communicate the expectations and requirements of Baxter Healthcare Corporation to its suppliers.
Supplier Quality Standard 1.0 Purpose The purpose of this Supplier Quality Standard is to communicate the expectations and requirements of Baxter Healthcare Corporation to its suppliers. These expectations More information Eclipsys Sunrise Clinical Manager Enterprise Electronic Medical Record (SCM) and Title 21 Code of Federal Regulations Part 11 (21CFR11)
Eclipsys Sunrise Clinical Manager Enterprise Electronic Medical Record (SCM) and Title 21 Code of Federal Regulations Part 11 (21CFR11) The title 21 code of federal regulations part 11 deals with an institutions More information ISCT Cell Therapy Liaison Meeting AABB Headquarters in Bethesda, MD. Regulatory Considerations for the Use of Software for Manufacturing HCT/P
ISCT Cell Therapy Liaison Meeting AABB Headquarters in Bethesda, MD September 10, 2009 David Doleski, Team Leader, Branch 2 Division of Manufacturing and Product Quality (DMPQ) Office of Compliance and More information Mobile Medical Application Development: FDA Regulation
Mobile Medical Application Development: FDA Regulation Mobile Medical Applications: Current Environment Currently over 100,000 mobile health related applications are available for download; there is an More information REGULATORY GUIDE 1.170 (Draft was issued as DG-1207, dated August 2012)
Purpose U.S. NUCLEAR REGULATORY COMMISSION July 2013 Revision 1 REGULATORY GUIDE OFFICE OF NUCLEAR REGULATORY RESEARCH REGULATORY GUIDE 1.170 (Draft was issued as DG-1207, dated August 2012) Technical More information OMCL Network of the Council of Europe QUALITY ASSURANCE DOCUMENT
OMCL Network of the Council of Europe QUALITY ASSURANCE DOCUMENT PA/PH/OMCL (08) 69 3R Full document title and reference Document type VALIDATION OF COMPUTERISED SYSTEMS Legislative basis - CORE DOCUMENT More information 6500m HOV Project Stage 1: A-4500 HOV
6500m HOV Project Stage 1: A-4500 HOV Systems Engineering, Integration & Testing Plan Document Control No.: 0000000 01-November-2009 WOODS HOLE OCEANOGRAPHIC INSTITUTION WOODS HOLE, MA 02543 Document Control More information SOFTWARE CONFIGURATION MANAGEMENT GUIDEBOOK
Office of Safety and Mission Assurance NASA-GB-9503 SOFTWARE CONFIGURATION MANAGEMENT GUIDEBOOK AUGUST 1995 National Aeronautics and Space Administration Washington, D.C. 20546 PREFACE The growth in cost More information U.S. Department of Energy
U.S. Department of Energy Washington, D.C. NOTICE DOE N 203.1 Approved: Expires: 06-02-01 SUBJECT: SOFTWARE QUALITY ASSURANCE 1. OBJECTIVES. To define requirements and responsibilities for software quality More information Guidance for Industry Classifying Resubmissions in Response to Action Letters
Guidance for Industry Classifying Resubmissions in Response to Action Letters U.S. Department of Health and Human Services Food and Drug Administration Center for Drug Evaluation and Research (CDER) Center More information Implementation of ANSI/AAMI/IEC 62304 Medical Device Software Lifecycle Processes.
Implementation of ANSI/AAMI/IEC 62304 Medical Device Software Lifecycle Processes.. www.pharmout.net Page 1 of 15 Version-02 1. Scope 1.1. Purpose This paper reviews the implementation of the ANSI/AAMI/IEC More information Guidance for Industry and FDA Staff: Current Good Manufacturing Practice Requirements for Combination Products
Guidance for Industry and FDA Staff: Current Good Manufacturing Practice Requirements for Combination Products DRAFT GUIDANCE This guidance document is being distributed for comment purposes only. Comments More information CDRH Regulated Software
CDRH Regulated Software An Introduction John F. Murray Jr. CDRH Software Compliance Expert CDRH Regulates Software in the following areas Medical Devices Automation of Production Systems Automation of More information JOURNAL OF MEDICAL INFORMATICS & TECHNOLOGIES Vol. 21/2012, ISSN 1642-6037
JOURNAL OF MEDICAL INFORMATICS & TECHNOLOGIES Vol. 21/2012, ISSN 1642-6037 FDA, medical software, recall, safety of medical devices. Leszek DREWNIOK 1, Ewelina PIEKAR 1, Mirosław STASIAK 1, Remigiusz MANIURA More information Quality Management System Manual
Effective Date: 03/08/2011 Page: 1 of 17 Quality Management System Manual Thomas C. West Eric Weagle Stephen Oliver President ISO Management General Manager Representative Effective Date: 03/08/2011 Page: More information 2015. All rights reserved.
DOCUMENT: Future AAMI/IEC 62304:2006/AMD1, 18-August-2015 Final Draft International Standard for Vote, Amendment 1 to IEC 62304: Medical device software Software life cycle processes. Public Review Draft More information WHITEPAPER: SOFTWARE APPS AS MEDICAL DEVICES THE REGULATORY LANDSCAPE
WHITEPAPER: SOFTWARE APPS AS MEDICAL DEVICES THE REGULATORY LANDSCAPE White paper produced by Maetrics For more information, please contact global sales +1 610 458 9312 +1 877 623 8742 globalsales@maetrics.com More information Revision History Revision Date 3.0 14.02.10. Changes Initial version published to http://www.isasecure.org
SDLA-312 ISA Security Compliance Institute Security Development Lifecycle Assurance - Security Development Lifecycle Assessment v3.0 Lifecycle Phases Number Phase Name Description PH1 Security Management More information Refuse to Accept Policy for 510(k)s. Guidance for Industry and Food and Drug Administration Staff
Refuse to Accept Policy for 510(k)s Guidance for Industry and Food and Drug Administration Staff Document issued on: December 31, 2012 This document supersedes Center for Devices and Radiological Health More information CHAPTER 7 Software Configuration Management
CHAPTER 7 Software Configuration Management ACRONYMS CCB CM FCA MTBF PCA SCCB SCI SCM SCMP SCR SCSA SEI/CMMI SQA SRS USNRC INTRODUCTION Configuration Control Board Configuration Management Functional Configuration More information ISO/IEC 17025 QUALITY MANUAL
1800 NW 169 th Pl, Beaverton, OR 97006 Revision F Date: 9/18/06 PAGE 1 OF 18 TABLE OF CONTENTS Quality Manual Section Applicable ISO/IEC 17025:2005 clause(s) Page Quality Policy 4.2.2 3 Introduction 4 More information SOFTWARE ASSURANCE STANDARD
NOT MEASUREMENT SENSITIVE National Aeronautics and NASA-STD-8739.8 w/change 1 Space Administration July 28, 2004 SOFTWARE ASSURANCE STANDARD NASA TECHNICAL STANDARD REPLACES NASA-STD-2201-93 DATED NOVEMBER More information Medical Device Data Systems, Medical Image Storage Devices, and Medical Image
4164-01-P DEPARTMENT OF HEALTH AND HUMAN SERVICES Food and Drug Administration [Docket No. FDA-2014-D-0798] Medical Device Data Systems, Medical Image Storage Devices, and Medical Image Communication Devices; More information JOURNAL OF OBJECT TECHNOLOGY
JOURNAL OF OBJECT TECHNOLOGY Online at www.jot.fm. Published by ETH Zurich, Chair of Software Engineering JOT, 2006 Vol. 5. No. 8, November-December 2006 Requirements Engineering Tasks Donald Firesmith, More information Acknowledgement. Software Engineering. CS 3141: Team Software Project Introduction
CS 3141: Team Software Project Introduction Ali Ebnenasir Department of Computer Science Michigan Technological University Acknowledgement Betty H.C. Cheng Software Engineering Systematic approach for More information Off-the-Shelf Software: A Broader Picture By Bryan Chojnowski, Reglera Director of Quality
Off-the-Shelf Software: A Broader Picture By Bryan Chojnowski, Reglera Director of Quality In the past decade, there has been a sea change in the business software domain. Many companies are no longer More information Guidance for Industry
Guidance for Industry Q8, Q9, and Q10 Questions and Answers(R4) U.S. Department of Health and Human Services Food and Drug Administration Center for Drug Evaluation and Research (CDER) Center for Biologics More information Quality Management System Manual
Quality Management System Manual Assurance ISO / AS Manual Quality Management System EXCEEDING ALL EXPECTATIONS Since our inception in 1965, Swiss-Tech has supplied the medical, aerospace, hydraulic, electronic More information 074-8432-552 Page 1 of 7 Effective Date: 12/18/03 Software Supplier Process Requirements
Page 1 of 7 Software Supplier Process Requirements 1.0 QUALITY SYSTEM FRAMEWORK 1.1 QUALITY POLICY The Seller shall document and implement a quality program in the form of Quality manual or detailed Quality More information 3SL. Requirements Definition and Management Using Cradle
3SL Requirements Definition and Management Using Cradle November 2014 1 1 Introduction This white paper describes Requirements Definition and Management activities for system/product development and modification More information SA Tool Kit release life cycle
Release management Release management process is a software engineering process intended to oversee the development, testing, deployment and support of software releases. A release is usually a named collection More information Develop Project Charter. Develop Project Management Plan
Develop Charter Develop Charter is the process of developing documentation that formally authorizes a project or a phase. The documentation includes initial requirements that satisfy stakeholder needs More information 2017 © DocPlayer.net Privacy Policy | Terms of Service | Feedback