Source: http://www.google.com/patents/US7962298?dq=Richard+B.+Fuller
Timestamp: 2014-12-23 04:29:46
Document Index: 749157068

Matched Legal Cases: ['art 4', 'art 4', 'art 4', 'art 4', 'art 11', 'art 11', 'art 22', 'art 22', 'art 22', 'art 22']

Patent US7962298 - Revenue class power meter with frequency rejection - Google PatentsSearch Images Maps Play YouTube News Gmail Drive More »Sign inAdvanced Patent SearchPatentsA power meter for measuring power parameters on one or more electrical power lines/loads is configured to include a revenue class metering module and a power quality metering module. The revenue class metering module and the power quality metering module may use the same measured voltage and current...http://www.google.com/patents/US7962298?utm_source=gb-gplus-sharePatent US7962298 - Revenue class power meter with frequency rejectionAdvanced Patent SearchPublication numberUS7962298 B2Publication typeGrantApplication numberUS 11/504,269Publication dateJun 14, 2011Filing dateAug 15, 2006Priority dateSep 16, 2005Also published asEP1764618A2, EP1764618A3, EP1764618B1, US20070067119, US20070067121Publication number11504269, 504269, US 7962298 B2, US 7962298B2, US-B2-7962298, US7962298 B2, US7962298B2InventorsPiotr B. Przydatek, Michael D. Bandsmer, Erin C. McPhalen, Daniel N. Loewen, Darrin G. Marr, Robert M. LawOriginal AssigneePower Measurement Ltd.Export CitationBiBTeX, EndNote, RefManPatent Citations (12), Non-Patent Citations (27), Referenced by (11), Classifications (6), Legal Events (2) External Links: USPTO, USPTO Assignment, EspacenetRevenue class power meter with frequency rejectionUS 7962298 B2Abstract A power meter for measuring power parameters on one or more electrical power lines/loads is configured to include a revenue class metering module and a power quality metering module. The revenue class metering module and the power quality metering module may use the same measured voltage and current signals to generate revenue accurate power consumption information and power quality information, respectively. The power meter may also include a plurality of filters configured to attenuate high frequency signals included in the voltage and current signals. Each of the filters may be configured with a corner frequency, gain and signal propagation delay that is substantially equal. The filtered current and voltage signals may be received and processed by the revenue class metering module, and the power quality metering module to generate revenue accurate power consumption information and power quality information, respectively. Alternatively, the filtered current and voltage signals may be received and processed by the power quality metering module to generate power quality information, and unfiltered current and voltage signals may be received and processed by the revenue class metering module to generate revenue accurate power consumption information.
Patent CitationsCited PatentFiling datePublication dateApplicantTitleUS3641531Dec 30, 1970Feb 8, 1972Gen ElectricChangeable address display deviceUS5317250Jan 9, 1992May 31, 1994Zellweger Uster AgModular electronic meter having reduced widthUS6000034May 6, 1997Dec 7, 1999Power Measurement Ltd.Security system and method for revenue class electricity meterUS6008635 *Mar 4, 1998Dec 28, 1999Siemens AktiengesellschaftElectricity meterUS6016432Jun 6, 1995Jan 18, 2000Telefonaktiebolaget L/M Ericsson (Publ)Electronic metering equipment systemUS6614219Dec 21, 2001Sep 2, 2003Eaton CorporationMetering assemblyUS6615147Aug 9, 1999Sep 2, 2003Power Measurement Ltd.Revenue meter with power quality featuresUS6792364Sep 3, 2002Sep 14, 2004Power Measurement Ltd.Revenue meter with power quality featuresUS6871150Aug 15, 2001Mar 22, 2005Power Measurement Ltd.Expandable intelligent electronic deviceUS7191076Nov 18, 2004Mar 13, 2007Power Measurement Ltd.Expandable intelligent electronic deviceUS20070008171Jul 6, 2005Jan 11, 2007Bowman Eric LWireless meter-reading system and methods thereofUS20070067119May 31, 2006Mar 22, 2007Power Measurement Ltd.Rack-mounted power meter having removable metering options module* Cited by examinerNon-Patent CitationsReference1Alpha Z�hler, High Precision Rack Mounted Meter, alpha A2500, Elster Messtechnik GmbH, 2 pgs., published on or after Apr. 7, 2003.2American National Standard, ANSI C12.20-2002 (revision of ANSI C12.20-1998), For Electricity Meter-0.2 and 0.5 Accuracy Classes, National Electrical Manufacturers Association @ 2003.3American National Standard, ANSI C12.20-2002 (revision of ANSI C12.20-1998), For Electricity Meter�0.2 and 0.5 Accuracy Classes, National Electrical Manufacturers Association @ 2003.4ANSI C12.18-1996, Protocol Specification for ANSI Type 2 Optical Port, 39 pgs., National Electrical Manufacturers Association, Apr. 8, 1996.5CewePrometer, Product Guide, CEWE Instrument, 12 pgs., published on or after Oct. 22, 2004.6Einschubz�hler mit statischem Mebetawerk, DIN 43862, ICS 17.220.20, 4 pgs., Seite 2 bis 4, Ersatz F�r Ausgabe Jul. 1983; Deutsche Elektrotechnische Kommission im DIN and VDE (DKE).7Einschubz�hler mit statischem Meβwerk, DIN 43862, ICS 17.220.20, 4 pgs., Seite 2 bis 4, Ersatz F�r Ausgabe Jul. 1983; Deutsche Elektrotechnische Kommission im DIN and VDE (DKE).8Encore Series, 61000 System Manual/Specification, The Intelligent Instrument, 4 pgs., Dranetz BMI, Edison, NJ � 2005 Dranetz-BMI.9Enermet OY, Class 0.2S and 0.5S High Precision Meters Q402/405, 2 pgs., published on or after Apr. 28, 2003.10International Standard 61000-4-30, Electromagnetic compatibility (EMC)-Part 4-30: Testing and measurement techniques-Power quality measurements methods; First Ed., 90 pgs., Commission Electrotechnique Internationale, International Electrotechnical Commission, Geneva, Switzerland, Feb. 2003.11International Standard 61000-4-30, Electromagnetic compatibility (EMC)�Part 4-30: Testing and measurement techniques�Power quality measurements methods; First Ed., 90 pgs., Commission Electrotechnique Internationale, International Electrotechnical Commission, Geneva, Switzerland, Feb. 2003.12International Standard 61000-4-30, Electromagnetic compatibility (EMC)-Part 4-30: Testing and measurement techniques-Power quality measurements methods; Second Ed., Commission Electrotechnique Internationale, International Electrotechnical Commission, Geneva, Switzerland, Feb. 2003; 47 pgs. [Note: The complete document is 92 pages, however, only the pages in English (odd numbered pages) are submitted, the even numbered pages are in French, and are not submitted herewith.].13International Standard 61000-4-30, Electromagnetic compatibility (EMC)�Part 4-30: Testing and measurement techniques�Power quality measurements methods; Second Ed., Commission Electrotechnique Internationale, International Electrotechnical Commission, Geneva, Switzerland, Feb. 2003; 47 pgs. [Note: The complete document is 92 pages, however, only the pages in English (odd numbered pages) are submitted, the even numbered pages are in French, and are not submitted herewith.].14International Standard IEC1107, Data Exchange for Meter Reading, Tariff and Load Control-Direct Local Data Exchange, Commission Electrotechnique Internationale, International Electrotechnical Commission, 122 pgs., Geneva, Switzerland, Mar. 1996.15International Standard IEC1107, Data Exchange for Meter Reading, Tariff and Load Control�Direct Local Data Exchange, Commission Electrotechnique Internationale, International Electrotechnical Commission, 122 pgs., Geneva, Switzerland, Mar. 1996.16International Standard IEC60529, Degrees of Protection Provided by Enclosures (IP code), Commission Electrotechnique Internationale, International Electrotechnical Commission, 96 pgs., Geneva, Switzerland, Feb. 2001.17International Standard IEC62052-11, Electricity Metering Equipment (AC)-General Requirements-Part 11: Electricity metering equipment (AC)-General requirements, tests and test conditions,-First Ed., 92 pgs., Commission Electrotechnique Internationale, International Electrotechnical Commission, Geneva, Switzerland, Feb. 2003.18International Standard IEC62052-11, Electricity Metering Equipment (AC)�General Requirements�Part 11: Electricity metering equipment (AC)�General requirements, tests and test conditions,�First Ed., 92 pgs., Commission Electrotechnique Internationale, International Electrotechnical Commission, Geneva, Switzerland, Feb. 2003.19International Standard IEC62053-22, Electricity Metering Equipment (AC)-Particular Requirements-Part 22: Static meters for active energy (classes 0,2 S and 0,5 S), First Ed., 33 pgs., Commission Electrotechnique Internationale, International Electrotechnical Commission, Geneva, Switzerland, Jan. 2003.20International Standard IEC62053-22, Electricity Metering Equipment (AC)�Particular Requirements�Part 22: Static meters for active energy (classes 0,2 S and 0,5 S), First Ed., 33 pgs., Commission Electrotechnique Internationale, International Electrotechnical Commission, Geneva, Switzerland, Jan. 2003.21International Standard IEC62053-22, Electricity Metering Equipment (AC)-Particular Requirements-Part 22: Static meters for active energy (classes 0,2 S and 0,5 S), Second Ed., 33 pgs., Commission Electrotechnique Internationale, International Electrotechnical Commission, Geneva, Switzerland, Jan. 2003; [Note: The complete document is 33 pages, however, only the pages in English (odd numbered pages) are submitted, the even numbered pages are in French, and are not submitted herewith.].22International Standard IEC62053-22, Electricity Metering Equipment (AC)�Particular Requirements�Part 22: Static meters for active energy (classes 0,2 S and 0,5 S), Second Ed., 33 pgs., Commission Electrotechnique Internationale, International Electrotechnical Commission, Geneva, Switzerland, Jan. 2003; [Note: The complete document is 33 pages, however, only the pages in English (odd numbered pages) are submitted, the even numbered pages are in French, and are not submitted herewith.].23ION� 8600 User Guide, Power Measurement, 219 pgs., Jun. 2005.24Iskraemeco, Energy Measurement and Management, TE340, DE340 Product Guide, 4 pgs., published on or after Dec. 14, 1999.25Linear Technology, LTC1563-2/LTC1563-3, Active RC, E Order Lowpass Filter Family, 20 pgs, Dec. 5, 2005.26Power Logic� Series 4000 Circuit Monitors, Document # 3020H00601, 4 pgs., Jan. 2006; � 2006 Schneider Electric.27Quantum � Q1000, Electronic Multimeasurement Meter, 4 pgs., Schlumberger, Jan. 2001.Referenced byCiting PatentFiling datePublication dateApplicantTitleUS8160824Apr 3, 2008Apr 17, 2012Electro Industries/Gauge TechIntelligent electronic device with enhanced power quality monitoring and communication capabilitiesUS8190381Mar 13, 2008May 29, 2012Electro Industries/Gauge TechIntelligent electronic device with enhanced power quality monitoring and communications capabilitiesUS8587949 *Mar 26, 2008Nov 19, 2013Electro Industries/Gauge TechElectronic meter having user-interface and central processing functionality on a single printed circuit boardUS8666688Mar 13, 2008Mar 4, 2014Electro Industries/Gauge TechHigh speed digital transient waveform detection system and method for use in an intelligent electronic deviceUS8700347May 24, 2012Apr 15, 2014Electro Industries/Gauge TechIntelligent electronic device with enhanced power quality monitoring and communications capabilityUS8717007Oct 13, 2009May 6, 2014Electro Industries/Gauge TechIntelligent electronic device having a terminal assembly for coupling to a meter mounting socketUS8773108 *Nov 10, 2010Jul 8, 2014Power Monitors, Inc.System, method, and apparatus for a safe powerline communications instrumentation front-endUS8797202Mar 29, 2011Aug 5, 2014Electro Industries/Gauge TechIntelligent electronic device having circuitry for highly accurate voltage sensingUS20080238713 *Mar 26, 2008Oct 2, 2008Electro Industries/Gauge Tech.Electronic meter having user-interface and central processing functionality on a single printed circuit boardUS20110109320 *Nov 10, 2010May 12, 2011Power Monitors, Inc.System, method, and apparatus for a safe powerline communications instrumentation front-endUS20120029850 *Sep 19, 2011Feb 2, 2012Electro Industries/Gauge TechIntelligent Electronic Device for Providing Broadband Internet Access* Cited by examinerClassifications U.S. Classification702/61International ClassificationG06F19/00Cooperative ClassificationG01R22/06, G01R22/065European ClassificationG01R22/06, G01R22/06D2Legal EventsDateCodeEventDescriptionAug 29, 2006ASAssignmentOwner name: POWER MEASUREMENT LTD., CANADAFree format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:PRZYDATEK, PIOTR B.;BANDSMER, MICHAEL D.;MCPHALEN, ERIN C.;AND OTHERS;REEL/FRAME:018259/0375;SIGNING DATES FROM 20060811 TO 20060814Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:PRZYDATEK, PIOTR B.;BANDSMER, MICHAEL D.;MCPHALEN, ERIN C.;AND OTHERS;SIGNING DATES FROM 20060811 TO 20060814;REEL/FRAME:018259/0375Aug 15, 2006ASAssignmentOwner name: POWER MEASUREMENT LTD., CANADAFree format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:PRZYDATEK, PIOTR B.;BANDSMER, MICHAEL D.;MCPHALEN, ERIN C.;AND OTHERS;REEL/FRAME:018204/0584;SIGNING DATES FROM 20060811 TO 20060814Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:PRZYDATEK, PIOTR B.;BANDSMER, MICHAEL D.;MCPHALEN, ERIN C.;AND OTHERS;SIGNING DATES FROM 20060811 TO 20060814;REEL/FRAME:018204/0584RotateOriginal ImageGoogle Home - Sitemap - USPTO Bulk Downloads - Privacy Policy - Terms of Service - About Google Patents - Send FeedbackData provided by IFI CLAIMS Patent Services©2012 Google