Patent Publication Number: US-2010114515-A1

Title: Calibration process management system and data structure

Description:
RELATED APPLICATION 
     This application is a divisional application of U.S. patent application Ser. No. 12/181,086, filed on Jul. 28, 2008, entitled “Calibration Process Management System and Data Structure,” which is a divisional application of U.S. patent application Ser. No. 10/527,627, filed on Mar. 10, 2005, entitled “Calibration Process Management System and Data Structure,” which issued as U.S. Pat. No. 7,406,388 granted on Jul. 29, 2008, which claims priority of PCT Patent application No. PCT/US03/28749 entitled “Calibration Process Management System and Method” filed on Sep. 11, 2003 which claims priority of U.S. Provisional Application Ser. No. 60/410,678, entitled “Calibration Process Management System and Data Structure” filed Sep. 13, 2002. The disclosures all of which are incorporated by reference herein. 
    
    
     COPYRIGHT STATEMENT 
     A portion of the disclosure of this patent document, namely, the screen captures, contain material which is subject to copyright protection under applicable conventions and national law. The copyright owner has no objection to the facsimile reproduction by anyone of the patent document or the patent disclosure, as it appears in official patent office records, but otherwise reserves all copyrights whatsoever. 
     APPENDIX DATA 
     Appendix A is a list of abbreviations used in this specification and drawings. 
     Appendix B is a list of the menu options that are available in the Calibration Process Manager application&#39;s main page for one version. 
     Appendix C describes each of the commands available in the Calibration Process Manager and their parameters. 
     Appendix D describes the interface features of the Calibration Process Manager main page (data sheet designer mode). 
     Appendix E contains tables listing interface features. 
     BACKGROUND OF INVENTION 
     As business grows more reliant on high-technology, the need for accurate and precise calibration and measurement becomes more imperative than ever. It is desirable to have standards and measurements that are directly traceable to NIST, and to have calibration and test program meet the highest standards of government and industry. Metrology laboratories may also be accredited by such programs as the National Voluntary Laboratory Accreditation Program (NVLAP). 
     A metrology laboratory might offer acceptance testing, calibration and repairs for instrumentation and standards in a wide array of discipline. Preferably, the metrology laboratory will issue certificates of calibration which can, when required, include as-found/as-left value and measurement uncertainties. A metrology laboratory may be able to calibrate equipment for measurement of temperature, humidity, mass, dimensional, pressure and vacuum, gas flow, force, torque, vibration, electrical, and time/frequency. 
     A metrology laboratory may also seek to provide complete calibration program management, including recall notification, interval management, reverse trace ability, procedure development, and records management. 
     A typical process for calibrating a unit under test (“UUT”) for a specific calibration function, is to connect it to a known pre-calibrated source having greater accuracy than the UUT, and then lead a putative measurement on the UUT. At the option of the metrologist, the UUT may then be left as is, merely recording the as-found measurement against the reference output, or optionally adjusted so that the measurement on the UUT matches within appropriate limits the reference. It is possible to independently measure each range of a UUT against a standard, and record the as-found/as-left measurement. 
     The heart of the metrology laboratory calibration process is the calibration report, generally referred to a data sheet, of a UUT. Typically a data sheet resembles a spreadsheet, with rows containing range data, and columns representing fields. Data sheets may be constructed for specific manufacturer&#39;s equipment, for example a Flukes® 787 process meter, which contain one row for every function of the UUT to be tested. Functions might include, for example, testing in different voltage ranges corresponding to different dial or switch settings on the UUT. Once a data sheet is determined for that particular make and model, e.g. the appropriate functions are identified and fixed, the data sheet may be saved as a template and can be reused for all UUTs of identical make and model. 
     In the prior art, a data sheet of calibration data for a particular UUT would typically be saved on an array data structure basis, with one or more identifier such as UUT serial number, ID number, test number, etc., serving as a database index field for the entire test of that UUT. This can be implemented in simple spreadsheet format that can be done by hand, or by using computerized spreadsheet programs such as Microsoft Excel®, or other available programs as will be known to those skilled in the art. 
     Data saved in an array structure, however, is difficult to maintain, and no easy solution exists for the identification of particular functions (e.g., individual rows in a data sheet) that may need re-calibration or different calibration intervals. Further, no easy solution exists for setting calibration intervals based upon drift analysis rather than failure based tolerance. Also, current data sheet based testing makes it difficult to tie standards to test points and calculate accuracy ratios for individual functions. 
     What is needed is a way to identify individual functions of a UUT, cross-link standards with these functions, and identify test points that may be problematic. Preferably, such a solution would optionally allow for simultaneous testing of like UUTs, and additionally optionally provide a comprehensive software tool that can augment a current system in place in an existing metrology laboratory, which simplifies and speeds up actual calibration process. Preferably this software tool would optionally perform multiple calibrations, reduce turnaround time, maintain a datasheet library, automate calibrations, calculate tolerances and analyze uncertainties. 
     SUMMARY OF INVENTION 
     The present invention meets this need by providing a way to identify individual functions of a UUT, cross-link standards with these functions, and identify test points that may be problematic. A method is disclosed for simultaneous testing of like UUTs, and a comprehensive software tool is described that can augment a current system in place in an existing metrology laboratory, which simplifies and speeds up actual calibration process. This software tool can perform multiple calibrations, reduce turnaround time, maintain a datasheet library, automate calibrations, calculate tolerances and analyze uncertainties. 
     A method for calibration process management of a calibration testing unit and a plurality of units under test is disclosed comprising configuring a user interface in communication with a calibration process management software system; configuring a communications link in communication with the software system, the software system capable of communicating with the calibration testing unit and the plurality of units under test; and wherein the software system manages the user interface and the communications link in a manner permitting an operator to calibrate the plurality of units under test. 
     Tracking of the calibrated units under test uses a permanent unique identifier and a dynamic unique identifier. Each permanent unique identifier is assigned when an object is created or altered in a manner that would make it unlinkable to its past usage, and each dynamic unique identifier is assigned any time the object is modified or saved. 
     A system for calibration process management of one or more than one unit under test (UUT), each UUT being an instance of a UUT equipment type is disclosed comprising a first computer readable medium for storing one or more first data objects representing calibration data from the one or more than one UUT; a second computer readable medium for storing a first globally unique identifier in an association relationship to one of the one or more first data objects; a third computer readable medium for storing one or more second data objects representing a reference standard for the UUT equipment class; a fourth computer readable medium for storing a second globally unique identifier in an association relationship to one of the one or more second data objects; a calibration management control program embodied on a fifth computer-readable medium for calibration of the one or more than one UUT, the calibration management control program in communication with the first data objects, second data objects, first globally unique identifier and second globally unique identifier, and an application program interface embodied on a sixth computer-readable medium for execution on a computer in conjunction with the calibration management control program. 
     The calibration management control program further optionally permits contemporaneous calibration of more than one UUT, and the application program interface is configured for contemporaneous calibration of more than one UUT, each UUT being an instance of the same UUT equipment type. 
     The application program interface can be configured to receive identification of the UUT equipment type, and a value as found for a specific calibration function, and display tolerance limits for the UUT based upon the UUT equipment type. 
     A method for calibrating a UUT for a specific calibration function is described comprising the steps of receiving a first identification attribute associated with a UUT; receiving a second identification attribute associated with a reference measuring unit; receiving a specific calibration function to be tested; maintaining in a first memory a reference database of one or more than one second identification attribute in holding relationship to one or more than one reference identification address, each reference identification address being associated with a single calibration function; looking up in the reference database, the reference identification address being held by the second identification attribute, and associated with the specific calibration function; selecting a unique record identification address; receiving a value as found for the specific calibration function for the UUT; storing in a second memory the value as found in a being-held relationship to the record identification address; and storing in a third memory the record identification address in a being-held relationship to the reference identification address. Further, either the first identification attribute or the second identification attribute, or both the first identification attribute and the second identification attribute, may be an asset number. 
     Optionally, the steps of receiving a value as left for the specific calibration function for the UUT and storing in the second memory the value as left in a being-held relationship to the record identification address may be added. A test accuracy ratio may be returned. An estimated measurement uncertainty may be returned. The step of automatically converting the units of the received value as found may be added. 
     Further, steps may be added of maintaining in a fourth memory an equipment database correlating for individual make and model of equipment, a nominal value to calibration function; receiving make and model of equipment attributes for the UUT; looking up in the equipment database, the nominal value for the specific calibration function for the received make and model of equipment; returning the nominal value; and calculating a calibration tolerance and returning the limits. Still further steps are receiving a value as left for the specific calibration function for the UUT and storing in a fifth memory the value as left in a being-held relationship to the record identification address. The value as found or the value as left is within the limits of the calibration tolerance may be indicated. 
    
    
     
       BRIEF DESCRIPTION OF DRAWINGS 
       These and other features, aspects and advantages of the present invention will become better understood with reference to the following description, appended claims, and accompanying drawings where: 
         FIGS. 1 to 79 ,  13 A and  17 A depict screen captures of exemplary pages and prompts displayed on a display device of a user interface of the present invention embodiments; 
         FIG. 80  is system diagram showing the calibration process management system of the present invention. 
         FIG. 81  is a flowchart showing the steps of the method for calibrating a UUT for a specific calibration function. 
         FIG. 82  is a block diagram of the component modules of the Calibration Process Management program. 
         FIG. 83  is a sample data sheet generated by the calibration process management system of  FIG. 80 . 
     
    
    
     DETAILED DESCRIPTION 
     Using the calibration process management system (“CPMS”) of the present invention, it is possible to create, manage and automate data sheets for the calibration of measuring and test equipment. The system has multiple tools to facilitate the translation of equipment specifications into electronic data sheets. These electronic data sheets can then be used to collect and store calibration data for a plurality of instruments simultaneously. Automation scripting can be added to these data sheets to configure instruments, capture data, perform mathematical operations, communicate with external software, and create instrument test points dynamically. While instruments are being tested, the invention can indicate the degree to which an instrument is in or out of tolerance. 
     A system using the present invention can be developed to satisfy ISO-17025, ANSI/NCSL 2-540-1, 10 CFR 50 Appendix B, ISO-9002, GMP, and other stringent quality requirements for calibration data collection. 
     In one preferred embodiment, the following assumptions are made regarding the system implementation and operation: (a) the software will run in a Windows 95 or higher environment, and (b) connection to a database server running Microsoft SQL Server version 7 or greater. Alternative environments and database servers are possible, as will be evident to those skilled in the art. To run automation scripts that use GPIB communications, a National Instruments GPIB controller card, or equivalent, may be required. 
     The CPMS of the present invention creates, manages, and automates data sheets, and analyses collected data. The system has multiple tools to facilitate the translation of equipment specifications into electronic data sheets. These electronic data sheets can then be used to collect and store calibration data for multiple instruments simultaneously. In one preferred embodiment, eight instruments can be calibrated simultaneously. 
     Optionally, automation scripting can be added to these data sheets to configure instruments, capture data, perform mathematical operations, communicate with external software, and even create instrument test points dynamically. While test point data is being collected, in a further embodiment, the CPMS according to the present invention will indicate the degree to which an instrument is in or out of tolerance. 
     System Description 
     One embodiment of the CPMS of the present invention will now be described. It will be understood by those skilled in the art that variations in the described embodiment are possible, and that specific aspects of this embodiment are not limitations on the claimed invention. This CPMS, with reference to  FIGS. 80 and 82 , comprises a calibration process management program  10  (one embodiment being the “Calibration Process Manager”) which is stored on a hard drive memory  54  of a computer  52  which is connected to a calibration testing unit  56 , and one or more units under test (UUT)  58 . It is possible to have multiple UUTs  58  connected and tested using the present invention. Optional printer  60  permits printing of data sheets  62 . An appropriate user interface on computer  52  is needed for operator interaction with the calibration process management program  10 . As described herein, an appropriate communications link is established between the calibration process management program  10 , the calibration testing unit  56 , and the one or more units under test (UUT)  58 . A series of pages and prompts as will now be described will appear on the user interface on computer  52 , guiding an operator to calibration of the one or more units under test. 
     The first time the Calibration Process Manager is run, a Database Server Page as shown in  FIG. 1  will appear to prompt the user for information that it will use to connect to the database server. The database server page can also be accessed by using the Administration, System Configuration, Database Server menu option. Changing database server settings requires the Calibration Process Manger to shut itself down. Calibration Process Manger will connect using the new settings the next time it is started. This page contains the interface features in Table 1. 
       FIG. 2  is a representation of a Master System Configuration Page for the CALIBRATION PROCESS MANAGER The Master System Configuration page is accessed by using the Administration, System Configuration, Master Setup menu option. This page contains the interface features shown in Table 2. 
     In a further embodiment a sound configuration page allows the user to set up sound clip options that are used with the Calibration Process Manager application.  FIG. 3  is a representation of what a sound configuration screen looks like. 
     The Calibration Process Manager provides a secure environment along with an audit trail to track changes to the system parameters as well as archiving of records. The security setup procedures of the Calibration Process Manager are performed by a user with System Administrator privileges. When the system database is initially installed a single user exists in the system, SysAdmin. The System Administrator performs the initial setup for the Calibration Process Manager, will login as SysAdmin and proceed to configure the system in preparation for use throughout the organization. System users must be setup prior to using the Calibration Process Manager. These users will be given the default set of permissions, once they are created. The System Administrator will adjust the user&#39;s permissions as required.  FIG. 4  is a representation of what the System Login Page looks like. 
     The System Login page is displayed when the Calibration Process Manager application is run. If the user fails to login, the application will terminate. This page contains the interface features shown in Table 3. 
       FIG. 5  is a representation of what the User Manager Page looks like. The User Manager page is accessed by using the Administration, User Manager menu option. System Administrator privileges are required to access the User Manager page. The User Manager page displays the master list of users for the Metrology suite of applications. All of the applications share a common user database. This page is the starting point for viewing, editing, and creating user accounts. The page also provides the gateway to user permissions, changing a user&#39;s password, and setting up the system&#39;s default login rules. This page contains the interface features in Table 4. 
       FIG. 6  is a representation of what the User Login Rules page looks like. The User Login Rules page is accessed by the Set Default Login Rules button on the User Manager page. This page is used to setup the system&#39;s basic security rules including user login rules. This page contains the interface features in Table 5. 
       FIG. 7  is a representation of what the Create a New User Page looks like. The Create a New User page is accessed by clicking the New User button on the User Manager page. The page is used to setup a new user account. This page contains the interface features in Table 6. 
       FIG. 8  is a representation of what the Edit User Login Page looks like. The Edit User Login page is accessed by clicking the Properties button on the User Manager page. The Edit User Login page displays and permits editing of the basic account information for the currently selected user. 
       FIG. 9  is a representation of what the User Permissions Page looks like. The User Permissions page is accessed by the permissions button on the user manager page. The User Permissions page is used to manage a user&#39;s privileges within the Calibration Process Manger. This page contains the interface features in Table 7. 
       FIG. 10  is a representation of what the Change Password Page looks like. The change password page can be accessed by two methods. The first is with the Options, Change Password menu option. The second is with the Change password button found in the User Manager page. However, with the Options, Change Password menu option, the user will have to input the current password. The Change Password Page contains the interface features in Table. Passwords cannot be less than six characters in length, or more than fourteen characters in length. They must be alphanumeric, and cannot match the login ID for the user. 
       FIG. 11  is a representation of what the Audit Report Page looks like. The Audit Report page is accessed by the main form&#39;s Administration, Audit Report menu option. The Audit Report page is used to setup, save, load, and execute Audit Report configurations. This page contains the interface features in Table 8. 
       FIG. 12  is a representation of what the main screen of the Calibration Process Manager application looks like. The main page is the screen that the Calibration Process Manager application opens to. The common thread of the Calibration Process Manager application is that it graphically represents, and is navigated as, a data sheet. It will display information as it is collected and auto format the test result columns to ensure each data sheet maintains a uniform look. 
     The main page has three master modes of operation: Data Sheet Designer, Data Collection, and statistics. When the application opens, the Master Mode is set to Data Collection. This page contains the interface features in Table 9. 
     Appendix B is a list of the menu options that are available in the Calibration Process Manager application&#39;s main page for one version. 
     A Master Mode selection drop down list allows the user to select what the current functionality of the application is to be. The choices are in Table 10. 
       FIG. 13  is a representation of what the Calibration Process Manger main page (Data Collection mode) looks like. The primary purpose of the Data Collection section is to facilitate the collection of measurement data and the design and execution of automation scripts. 
     Calibration Process Manager allows for the collection of data for more than one instrument simultaneously. This can be done, providing the instruments use the same data sheet. 
     In one embodiment, up to eight sets of measurement data can be collected. As is evident, a system could be constructed for more or fewer sets. Each instrument under test is a “channel” and will display the “As Found”, “As Left”, and “Out of Tolerance” data. All eight channels can be displayed and edited simultaneously. These channels are activated and configured independently. 
       FIG. 13A  is a representation of what a Calibration Process Manger Data Collection Channel looks like. A channel is container object that represents one of the units under test. When a channel is selected, its display is highlighted and measurement data is recorded into that channel. The channels display measurement data collected for the currently selected row of the data sheet grid. “As found”, “As left”, and “Out of Tolerance” information is displayed in relevant columns of the data sheet grid for the channel that is currently selected. These data sheet columns are not visible when editing automation script commands to preserve space. The script command list grid occupies part of the space normally reserved for the data sheet grid. The Tolerance column of the data sheet grid will display graphical status information for each of the active channels, as shown in  FIG. 13A . 
     The active tabs displayed at the top of the main page indicate which channels are designated for instruments and which are currently active for testing. When the active check box is selected, the UUT Boilerplate Setup page is displayed. The UUT Boilerplate Setup page is where the unit under test information is configured for the selected channel. 
     A data sheet must be loaded prior to activating a channel. The Calibration Process Manager uses the data sheet as a template for data collection. All determinations of the nominal test point definition, in tolerance status, and Specification Tracking standards used are derived from the active data sheet. The measurement data collected for each of the channels is stored separately in memory and is not stored in the active data sheet. When a calibration is saved, exported to a file, or printed, The Calibration Process Manager merges the channel and data sheet data into a single calibration file. If a calibration is to be reopened or imported for further data collection, the applicable data sheet must first be loaded. If the active data sheet is not the same data sheet or same version of the data sheet used to create the calibration file, a warning message will be displayed. 
     When a calibration is reopened into a channel, all measurement data is reevaluated, using the data sheet&#39;s information. If the data sheet&#39;s nominal test point definitions or tolerances have been modified, the calibrations in tolerance status will be adjusted accordingly. For this reason, historical calibration records should not be viewed or reprinted from within a channel. To view, reprint or export historical calibration files, set the Calibration Process Manager&#39;s Master Mode to Data Sheet Designer or Statistics, then use the File, Open, . . . menu options to load the calibration. These modes do not interpret the measurement data, they merely display it. 
     An active data collection channel contains the interface features in Table 11. 
     In one embodiment, when a channel is active the Show Standards button will be visible at the top of the main page. If no standards have been assigned to the data sheet the circle will flash red. Once standards have been assigned to the data sheet, the light will show green. When the button is clicked into the depressed position, the standards fields will be displayed at the bottom of the screen. However, the data collection channels will not be displayed again until the Show Standards button is de-selected (returned to the up position). 
       FIG. 14  is a representation of what the UUT Boilerplate Setup page looks like. If the channel in which the active check box is selected has not previously been configured, then the UUT Boilerplate Setup page will be opened. The UUT Boilerplate Setup page allows the user to designate asset numbers, test numbers, and other information to units under test for each channel. The UUT Boilerplate Setup page contains the interface features in Table 12. 
     The Function Field contains four indicators. The first is the Timer indicator. This will display a numeric value of a delay period specified by a script command. It will count down until the delay is completed. The remaining three indicators, GPIB, OLE, and RS-232 will show one of four colors: gray for not is use, green for reading data, yellow for writing data, and red for error. These colors will be visible during the corresponding command execution. Warnings and errors will cause a Calibration Process Manager Warning window to be displayed indicating the possible cause of the error or warning. The script will be halted, When the user has corrected the error, testing can resume. 
     The Measured Value Radio Selection field contains two radio buttons. The options are for “As Found” or “As Left” data collection. This selection represents the master data collection mode. Only one radio button can be selected at a time they will activate the corresponding field on the channel display. 
     The All Cal Files section has the following four buttons: to copy “as found” data to “as left”, print, view the master boilerplate data, and save calibrations. The copy “as found” to “as left” function will copy the numeric value from one field to the other within the channel display. These buttons will cause their action to propagate to all open channels. The operation of these buttons is identical to their operation when selected on the individual channel boxes, with the exception of the view boilerplate data button. 
     The “Show:” Selection panel contains two check boxes. The tolerance check box indicates if the tolerance value will be displayed in the “tolerance” column of the data sheet. The percentage of tolerance box indicates whether or not the percentage of tolerance consumed will be displayed in the “tolerance” column of the data sheet. These options only affect how data is displayed, not stored or printed. 
     The Data Sheet Master Grid contains the per line information for unit tests. The Tolerance field is updated as the tests are executed giving a real time indication of the test values. These results will not be user editable. 
     When a measured value is recorded for an instrument channel, the system will compare the value to the instrument&#39;s Tolerance as set in the data sheet, indicate the “out of tolerance” status, then display the percentage of the tolerance consumed. This operation is only performed on test points that contain calculable data in the Nominal Value and Tolerance cells. 
     Data can also be collected by use of the automated scripting system. This allows data collection from a GPIB, RS-232, Custom User Prompts, or the result of scripted mathematical operations. The user can design and execute automation scripting for the current data sheet. The Data Collection page contains commands for executing, halting, and when the Script Commands are visible, editing automation commands. 
     The script is executed by pressing the Command Menu key on the QuickData Keypad, clicking on the run button, or pressing the F12 key on the system keyboard. When an interface command group (command menu item) is selected from the pop-up Command Menu, the application will attempt to run the scripted commands, sequentially, for the interface command group being executed. The system will, as necessary, open other command based forms to assist the user in executing the test properly. 
     The Calibration Process Manager main page (Data Sheet Designer mode) option gives the user the means by which they may create data sheets. These data sheets may then be used as a template for automation scripting and the collection of calibration data. A data sheet is the starting point for all operations within the Calibration Process Manager application. This section describes the tools available to assist in the data sheet creation process. 
       FIG. 15  is a representation of what the Main Form&#39;s Data Sheet Designer mode looks like. The Calibration Process Manger main page (Data sheet designer mode) contains the interface features shown in Appendix D. 
       FIG. 16  is a representation of what the Calibration Process Manger main page (Statistics mode) looks like. The Statistics page allows the user to easily view past test results for a particular data sheet. The results can be displayed for variable date ranges, and can be specific to the data sheet or the asset. The statistics page contains the interface features in Table 13. The test point history is displayed as: As Found—Blue Diamonds; As Left—Red Diamonds; Nominal Value—Black Lines; Tolerance Value—Red Lines. 
       FIG. 17  is a representation of what the Drift Analyzer Interface page looks like. The Drift Analyzer Interface page is accessed by clicking on the Analyze Drift button, when the Calibration Process Manager&#39;s main form is in the Statistics mode. This page requires that an instrument history performance analysis (IHPA) application be installed on the computer. A suitable IHPA application is currently marketed by CRS Engineering Inc. The drift analyzer interface page allows the user to save drift analysis data as a spreadsheet file, such as such as Microsoft Excel format, for use with the IHPA application. Typically, the IHPA is a spreadsheet application. Once the file is created the IHPA spreadsheet application is executed. 
     The Drift Analyzer Interface page contains the interface features in Table 14. 
     Drift analysis predicts new values based on previously collected data using a least-squares linear regression of a range of the known data or known x-arrays and y-arrays. For example, given corporate earnings for each of the previous six quarters, a drift analysis function returns earnings expected for the next two quarters. This is the same methodology used in “forecasting” techniques. 
       FIG. 17A  is a representation of an interval analysis report. For calibration interval analysis, the term reliability refers to the probability that an item or parameter of measuring or test equipment (MTE) is in-tolerance. When items are periodically tested or calibrated, the observed reliability is the fraction of items observed to be in-tolerance at the end of the test or calibration interval. This is a binomial test, i.e., the item or parameter either passes or fails. 
     Testing the observed reliability R obs  involves comparing it against some established minimum acceptable in-tolerance confidence level, referred to as the reliability target R targ . With one method, the A3 Interval Method, the observed reliability is computed by simply dividing the Number of Tests into the Number In-Tolerance. For example, if the Number of Tests n for a grouping is 25 and the Number In-Tolerance x is 20, the observed reliability is 80%. 
     This test methodology works by comparing R obs  to R targ  to see if there is a significant difference between the two tests within the interval. The magnitude of the difference and the Test Confidence Level determine whether or not the difference is significant. If it is, then the interval corresponding to R obs  is said to have failed the test and a recommendation with respect to present interval is suggested (i.e., the interval will either be lengthened or shortened). 
     The algorithm used to test the interval evaluates whether upper and lower confidence limits for R obs  contain the reliability target. If so, then the interval passes the test. A preferred method for determining the confidence limits for R obs  uses a binomial distribution function. Given a Number of Tests n and a Number In-Tolerance x, the upper and lower confidence limits are obtained by solving a binomial confidence interval equation. If R targ  is within these limits, then the interval passes the test. If the interval is rejected, a new recommended interval is computed using a simple algorithm. The process takes place in two stages. In the first stage, the interval is adjusted to a trial value. In the second stage, the trial value is refined to ensure its feasibility. 
     Calibration Process Manager has two primary document types: data sheets and calibrations. A data sheet is essentially a template for a calibration. A calibration is a data sheet with actual measurement data recorded. Data sheets are actually much more than a simple template, as they may also contain complete automation scripting and detailed information regarding UUT and standard instrument specifications and uncertainties. 
     A data sheet is a pre-engineered template used to execute and record calibration data. Data sheets may also contain complete automation scripting and detailed information regarding UUT and standard instrument specifications and uncertainties. 
       FIG. 18  is a representation of what the Data Sheet Catalog Page looks like. The Data Sheet Catalog Page allows the user to locate and load a data sheet. The Data Sheet Catalog Page is displayed by selecting the File, Data Sheet, Open menu option. The catalog displays both data sheets and data sheet groups. A Data sheet group is a collection of data sheets established using the Data Sheet Group Wizard. 
     In one embodiment, the Calibration Process Manager does not have a database, consequentially it does not have a data sheet catalog. In that embodiment, it is therefore recommended that a shared network directory be used store and share data sheet files. Some degree of document control can be achieved by managing access rights to this directory. By default, the Calibration Process Manager will look for data sheets in the “DataSheets” directory of the path from which the application was run. If the application is stored in a “Read Only” directory on the local area network, the system administrator can create a “DataSheets” subdirectory. It is also recommended that a structured sub directory system be created inside the “ . . . \DataSheets\” directory to assist users in locating the appropriate data sheet. 
     The Data Sheet Catalog Page contains the interface features in Table 15. 
     An external data sheet file can be loaded into the Calibration Process Manager by selecting the File, Import, Data Sheet File menu option. The standard windows file location dialog is used to locate a file. Once loaded the data sheet is ready for editing, approval, or use in data collection. 
       FIG. 19  is a representation of what the Data Sheet Approval Page looks like. The Data Sheet Approval Page allows the user to approve a data sheet and enter it into The Calibration Process Manager&#39;s data sheet database. The Data Sheet Catalog Page is displayed by selecting the File, Data Sheet, Approve menu option. 
     The Data Sheet Approval Page also allows for other administrative functions including: Reactivating a previously inactivated data sheet, inactivating a data sheet, and deleting a data sheet. All of these administrative functions require the user to have System Administrator privileges. The data sheet must first be loaded into the Calibration Process Manger prior to activating the Data Sheet Approval Page. All of the actions taken on this page will require a special login of a user with data sheet approval or system administrator privileges. 
     The Data Sheet Approval Page contains the interface features in Table 16. 
     A Calibration Process Manager data sheet can be exported to an external data sheet file by selecting the File, Export, Data Sheet File menu option or the File, Data Sheet, (Save or Save As) menu option. The Standard windows file save dialog is used to set the file name and location. Working with files is the preferred method of saving a data sheet while it is being designed or edited. It should not be returned to the data base until it is ready for use in calibrations. 
     A Data sheet group is a collection of data sheets established using the Data Sheet Group Wizard. Data sheet groups are used to bind together a set of data sheets that are needed to calibrate an asset that may be comprised of two or more instruments. Once created, data sheet groups appear in the Data Sheet Catalog. Data sheet groups can be located in the Data Sheet Catalog using the same methods that would apply to a single data sheet. Once a data sheet group is selected using the Data Sheet Catalog, the user will be prompted to select a member data sheet using the Data Sheet Group Selector page. 
       FIG. 20  is a representation of what the Data Sheet Group Selector page looks like. The Data Sheet Group Selector page allows the user to select a data sheet from a data sheet group. The page is accessed from the Data Sheet Catalog Page. The Group Members button of the Data Sheet Catalog Page is enabled when a data sheet group is selected in the Data Sheet Catalog Page&#39;s data sheet grid. Click on this button to view the Data Sheet Group Selector page. 
     The Data Sheet Group Selector page contains the interface features in Table 17. 
       FIG. 21  is a representation of what the Data Sheet Group Wizard page looks like. 
     The Data Sheet Group Wizard page allows the user to create and edit data sheet groups. The page is accessed via the Tools, Wizards, Data Sheet Groups, New Group menu option to create a new group or the Tools, Wizards, Data Sheet Groups, Edit Existing Group menu option to edit an existing group. The features are in Table 18. 
       FIG. 22  is a representation of what the Data Sheet Group Approval page looks like. The Data Sheet Group Approval page allows the user to approve data sheet groups created or edited with the Data Sheet Group Wizard. The page is accessed by clicking on the Data Sheet Group Wizard page&#39;s Ok button upon completion of the data sheet group edit process. Features are in Table 19. 
     A calibration file is originated from a data sheet file for a unit whose measurement data has already been collected or is currently being collected. It retains and displays the measurement results and Specification Tracking standards used. This allows the user to save calibration information for units under test and cross-examine the results. The calibration file does not retain the automation or Specification Tracking default standard information of the data sheet. Calibrations can be stored in the system database for long term history or exported as a file for short term storage and transfer. In a more basic embodiment, calibration files do not contain any Specification Tracking data. 
       FIG. 23  is a representation of what the Calibration History Catalog Page looks like. The Calibration History Catalog Page allows the user to locate and load a Calibration. When in the Data Sheet Designer mode, the Calibration History Catalog Page is displayed by selecting the File, Calibration, Open menu option. When in the Data Collection mode, the Calibration History Catalog Page is displayed by clicking on the Open Cal button of the UUT Boilerplate Setup page. The UUT Boilerplate Setup page is accessed by clicking on the Set UUT Boilerplate button of one of the active UUT Channels. The data sheet used to create the calibration should be loaded into the Calibration Process Manager before loading a calibration. The data collection system uses the data sheet as a reference and the user will be warned if the applicable data sheet is not loaded first. 
     A basic embodiment of the Calibration Process Manager does not have a database, consequentially it does not have a calibration history catalog. It is therefore recommended that a shared network directory be used store and share calibration files. All users must have “Write” access to this directory. By default, the Calibration Process Manager will store calibration files in the subdirectories of the “Calibrations” directory of the path from which the application was run. If the application is stored in a directory on the local area network, the system administrator can create a “Calibrations” subdirectory. The Calibration History Catalog Page contains the interface features in Table 20. 
     An external calibration file can be loaded into the Calibration Process Manager. When in the Data Sheet Designer mode, the standard windows file locator dialog is displayed by selecting the File, Import, Calibration File menu option. Calibration files loaded in the Data Sheet Designer mode can not be stored in the database. The file must be imported in the Data Collection mode to be stored in the database. 
     When in the Data Collection mode, the standard windows file locator dialog is displayed by clicking on the Import Cal button of the UUT Boilerplate Setup page. The UUT Boilerplate Setup page is accessed by clicking on the Set UUT Boilerplate button of one of the active UUT Channels. The data sheet used to create the calibration should be loaded into the Calibration Process Manager before loading a calibration. The data collection system uses the data sheet as a reference and the user will be warned if the applicable data sheet is not loaded first. Once loaded the calibration is ready for editing or approval. 
       FIG. 24  is a representation of what the Search for File page looks like. This allows the user to look on the local system for a previously exported calibration file. This page will contain the information in Table 21. 
       FIG. 25  is a representation of what the Sign Calibration Page looks like. Calibrations can only be signed into the database when in the Data Collection mode and the calibration is in an active channel. When in the Data Collection mode, the Sign Calibration Page allows the user to sign a calibration and enter it into the Calibration Process Manager&#39;s calibration database. The user must have the calibration signature privilege assigned on the user permissions page. The Sign Calibration Page is displayed by clicking on the Save Cal button of the UUT Boilerplate Setup page. The UUT Boilerplate Setup page is accessed by clicking on the Set UUT Boilerplate button of one of the active UUT Channels. The Sign Calibration Page contains the interface features in Table 22. 
     A Calibration Process Manager Calibration and be exported to an external calibration file. When in the Data Sheet Designer mode, the standard windows file save dialog is displayed by selecting the File, Export, Calibration File menu option. When in the Data Collection mode, the standard windows file save dialog is displayed by clicking on the Export Cal button of the UUT Boilerplate Setup page. The UUT Boilerplate Setup page is accessed by clicking on the Set UUT Boilerplate button of one of the active UUT Channels. 
       FIG. 26  is a representation of what the Select Discipline Directory page looks like. The Select Discipline Directory page allows the user to export calibration files to a set of fixed discipline directories. 
       FIG. 27  is a representation of what the Data Sheet Print Shop page looks like. The Data Sheet Print Shop page allows the user to print data sheets or calibrations, and configure printers and page settings. The print button, in the Data Sheet Designer page accesses this page. The page can also be accessed by clicking on one of the Print buttons of an active calibration channel. The Data Sheet Print Shop page has the interface features in Table 23. 
       FIG. 28  is a representation of what the Document History Page looks like. The Document History Page allows the user to view the version history of data sheets and calibrations. To display the history of the document currently displayed in the Calibration Process Manager&#39;s primary data sheet display grid, select the Administration, Document History menu option. To display the history of a calibration currently displayed in one of the Calibration Process Manager&#39;s data collection channels, click on the Doc History button of the applicable UUT Boilerplate Setup page. The Document History Page contains the interface features in Table 24. 
     Data sheet design is performed when the Calibration Process Manager&#39;s main form is set to the Data Sheet Design mode. The electronic data sheets will be used to collect and store instrument calibration information. These data sheets will report the measurement data for instruments under test. Once approved for use, data sheets are stored in the system database. Data sheets can be saved as a flat file, with a .MDD extension, while in development or for transfer purposes. In one embodiment, a single data sheet will be able to collect and store calibration information for up to eight instruments simultaneously. It is possible to create a system for more or less units. The data sheets can be scripted to collect data from GPIB, RS-232, or custom user prompts. 
     The Calibration Process Manager application has different ways to create a data sheet, including manual entry (typing), the tolerance calculator, test point wizards, and data sheet generation wizards. 
     All Data Sheets will be automatically formatted to maintain a uniform look and feel from sheet to sheet. Calibration Process Manager will determine the necessary font size, spacing, and line splitting for each data column as entered. Each data sheet column has its own rules for formatting its content. The header row of each page also has formatting rules that differ from the data rows. Header rows are indicated in two ways: There cells are gray, and they are always on line number zero. The page and line numbers are displayed on the left edge of the data sheet grid. It is recommended that data sheet headers be inserted into data sheet last. As new rows are inserted into the data sheet, the rows containing the header information will be moved down. Once editing is completed, you will have to adjust your header rows. It is easiest to just delete header rows that are out of position and re-add add them to the first row of each page. The headers are inserted using the “Insert Headers” button. The “Insert Headers” button will insert the headers in the selected row and push the rest of the data sheet down one row. The “Insert Headers” function copies the header information from the first page of the data sheet. If you plan to use non-standard headers, change the first set of headers before using the function. While other data temporarily occupies the header rows, the text formatting will use the header row&#39;s formatting rules, but once the headers are inserted, the data will be pushed into normal data rows and assume proper formatting. 
     The Calibration Process Manager uses only ANSI standard characters within a data sheet. The main reason for this is universal data compatibility. Whether stored in the system database or exported to files, the information contained can be read and used by any application, on any operating system. Exported files are stored in a straight forward, tab delimited, text format so they may be used by other applications. If you decide to use something other than the Calibration Process Manager in the future, you can take your data with you. 
     There are only two special character flags used with the Calibration Process Manager&#39;s data sheet editor and they have the following meaning. The flag “˜” is a Forced Line Break. This character forces a line break wherever it appears. This overrides the internal automatic formatting of the data sheet editor. This is useful when it enhances readability of the data sheet. The character is not printed on data sheets or displayed in the data sheet grid. 
     The flag “˜@” is a Parametric Divider. These characters inform the Nominal Value decoder that what follows is another parametric value., i.e.: 190 mV˜@20 kHz˜@ 1 mA. When displayed the item will appear as: 190 mV@20 kHz@ 1 mA. 
     The decoder will interpret this as having three parametric values. The first parameter is used by to indicate the test point&#39;s nominal value. The Professional and Pro Remote versions will use all three parameters for the parametric matching of Specification Tracking Instrument Ranges. The “˜” character also serves its primary function of causing a forced line break. The decoder requires that each parameter have a numeric value followed by the units of measure. The decoder can resolve the parameters with or without spaces, but inserting the space between the value and unit of measure is easier for humans to read. There can be no space between the ˜ and @ characters. If the ˜ character appears by itself in the Nominal Value cell, it instructs the decoder to disregard it and anything that follows it. For printing and display purposes it will still serve its primary purpose of forcing a line break. Using the ˜ character by itself allows you to add additional text to the Nominal Value cell without confusing the decoder. i.e.: 190 mV˜@100 Hz˜(Slow Mode). When displayed the item will appear as: 190 mV@100 Hz (Slow Mode). The two parameters are recognized and the rest ignored. 
       FIG. 29  is a representation of what the Master Boiler Plate Data page looks like. The Calibration Process Manager Master Boiler Plate Data page allows the user to set header type information for the data sheet. This page also allows the user to configure system behaviors that apply when the data sheet is activated. The Master Boiler Plate Data page is activated by clicking the View/Edit Boilerplate button on the application&#39;s main page. The Boiler Plate Data page contains the interface features in Table 25. 
       FIG. 30  is a representation of what the Remarks Editor looks like. The Calibration Process Manager Remarks Editor allows the user to set global or per-line comments for the data sheet. The Remarks Editor contains the interface features in Table 26. 
       FIG. 31  is a representation of what the Data Sheet Operations Commands page looks like. 
     The Data Sheet Operations Commands page allows the user to set special data sheet directives that override normal data sheet behavior. 
       FIG. 32  is a representation of what the data sheet with the tolerance calculator looks like. 
     The integrated tolerance calculator allows a data sheet designer to generate test point row data by entering the nominal test point values and equipment specification data. The tolerance is calculated, formatted, and then the new test points are inserted into the data sheet. 
     The calculator uses a smart rounding system to appropriately round tolerance values. It will support both standard specifications, (% of I.V., ppm of I.V., % of F.S., ppm of F.S., ±n units of measure, and ±dB.) and custom specification calculators. The custom specifications contain more complex components (i.e. ±0.0025% of I.V., per volt above 750 volts). Custom specification calculators can be defined by the user, stored and used in conjunction with the standard specification components in the tolerance calculator. A basic version does not need to support the use of custom calculators. The Tolerance Calculator page contains the interface features in Table 27. 
       FIG. 33  is a representation of what the Automatic Line Generator page looks like. The Automatic Line Generator page is displayed when the tolerance calculator&#39;s Calculate button is clicked. When the new row is inserted, the user will be prompted to enter additional information to be placed in the new data sheet row. The Automatic Line Generator page contains the interface features in Table 28. 
     The tolerance calculator embeds the equipment specification data, used to calculate tolerances, into data sheet rows. A composite value for the standards requirement (0.25*Tolerance) is also embedded. This information is displayed for each data sheet row. 
     The Nominal Value and Tolerance columns of a data sheet play major roles in the Calibration Process Manager&#39;s overall function. The columns contain data that is used to assess in tolerance status during data collection and to interface with Specification Tracking instrument specifications. If the user were to directly edit the contents of these critical fields, the system could no longer be assured of their data integrity. It is therefore necessary to provide a means of modifying the content of the fields while preserving numerically critical content. A special editor page is provided for each of these fields to make this possible. These editors should be used to modify these fields. If a user directly edits the fields, all UUT specification and Specification Tracking Instrument references are dropped and the data sheet row looses all of its intelligence. 
       FIG. 34  is a representation of what the Nominal Value Wizard page looks like. The Nominal Value Wizard allows the user to configure the Nominal Value field of a data sheet row. All formatting is handled by the wizard. The user can edit the trailing text field without invalidating the field&#39;s data integrity, but any changes to field parametric values will invalidate any Specification Tracking Instrument range references. Changing the first parameter will invalidate all specification data for the data sheet row. Calibration Process Manger uses the first parameter of the nominal value field as its reference for the nominal value of calibration data collected. If data integrity is compromised, you must recalculate tolerances using the integrated tolerance calculator and reassign Specification Tracking instrument ranges. The Nominal Value Wizard page contains the interface features in Table 29. 
       FIG. 35  is a representation of what the Tolerance Column Wizard page looks like. The Tolerance Column Wizard allows the user to configure the Tolerance field of a data sheet row. All formatting is handled by the wizard. The user can edit the TAR, EMU, remark flag, and trailing text field without invalidating the field&#39;s data integrity. Calibration Process Manger uses the Tolerance field content to assess the in tolerance condition of calibration data collected. If data integrity is compromised, you must recalculate tolerances using the integrated tolerance calculator. The Tolerance Column Wizard page contains the interface features in Table 30. 
       FIG. 36  is a representation of what the Unit of Measure by Symbol page looks like, The Unit of Measure by Symbol page allows the user to search for a desired unit of measure by its symbol. The form is activated by clicking on the main form&#39;s Units button when the main form is in the Data Sheet Designer mode and the Lock Status&#39; Edit option is selected. The Unit of Measure by Symbol page contains the interface features in Table 31. 
     The Test Point Generation Wizard allows the user to predefine test point values, tolerance format, functions tested descriptions, and boilerplate information. The wizard can contain multiple test point definitions. The tolerance calculator will automatically be configured when the test point definition is selected, and setup an array of test points. 
       FIG. 37  is a representation of what the Wizard Loader page looks like. The Wizard Loader allows the user to load preset wizards for use in creating data sheets. This page is accessed through the Tools, Wizards, Test Point Wizards, Load menu option. The Wizard Loader page contains the interface features in Table 32. 
       FIG. 38  is a representation of what the Wizard Workshop page looks like. The Wizard Workshop allows the user to create test point wizards for use in creating data sheets. This page is accessed through the tools, wizards, test point generators, load menu item. The Wizard Loader page contains the interface features in Table 33. 
     The Data Sheet Generation Wizard allows the user to generate a complete data sheet by entering data into a module specifically designed for a specific instrument type. Data sheet wizards posses an advanced of knowledge of the rules applicable to the instruments for which they are designed. Once data sheets are created with the module, they can be edited and manipulated in the same manner as any other data sheet. 
       FIG. 39  is a representation of what the Pressure Gauge Data Sheet Generator page looks like. This is a Data Sheet generation wizard that will prompt the user for specific data relating to pressure instruments. Once the data has been inputted the entire data sheet is generated. This page is accessed by using the Tools, Data sheet Generators, Pressure option in the menu bar. The module supports the creation of multiple data sets within a single data sheet, i.e., Multiple sensor modules used with a single reader. Numeric fields on this page will turn red if an improper entry is made and an error message is then displayed. 
     The Pressure Gauge Data Sheet Generator uses specific information, embedded in the Suite&#39;s master units of measure table, to identify “absolute” and “gauge” pressure units of measure. The group names for these units must be “Pressure (absolute)” and “Pressure (gauge)”. These two groups contain the units that can be directly cross referenced to each other. Changing the group names or adding units that don&#39;t cross reference may cause errors in the Pressure Gauge Data Sheet Generator module. The Pressure Gauge Data Sheet Generator page contains the interface features in Table 34. 
       FIG. 40  is a representation of what the Test Point Uncertainty Wizard page looks like. The Test Point Uncertainty Wizard page is accessed from the Process Manger&#39;s main page when the main page is in the Data Sheet Designer mode. Click on the Measurement Uncertainty button (this is the button with the E symbol). The Test Point Uncertainty Wizard is used to view and edit the currently selected test point&#39;s measurement uncertainty model. The Calibration Process Manager will automatically manage the measurement uncertainty model when one or more Specification Tracking instrument ranges are assigned to the test point. The user can modify this model if the user has the necessary permissions. A fixed uncertainty value can also entered into the Tolerance Column Wizard page. If a value is entered directly into the Tolerance Column Wizard page, that value will override the uncertainty model. 
     The user can enter both type A and type B uncertainties. Some of the values used by the measurement uncertainty model can be set in the System Configuration page&#39;s “Measurement Uncty” configuration options. The measurement uncertainty model is also used in the calculation of the test accuracy ratio when directed to do so in the System Configuration page&#39;s “Measurement Uncty” configuration options. 
     The uncertainty model, test accuracy ratio, and applicable displayed results are always relevant to the currently selected Specification Tracking Interface mode. When in the Standards used mode, the Model is using the actual calibration standards&#39; specifications, when in the Default Standards mode, the model is using the default standards&#39; specifications, etc. The values displayed will vary as the Specification Tracking Interface mode is changed. Pay close attention to the Specification Tracking Interface mode indicators in the data sheet&#39;s Nominal Value column. The Test Point Uncertainty Wizard page contains the interface features in Table 35. 
       FIG. 41  is a representation of what the Uncertainty Element Editor Page looks like. The Uncertainty Element Editor Page allows the user to edit parameters of an uncertainty element. 
     The Uncertainty Element Editor Page contains the interface features in Table 36. 
     Specification Tracking 
     The Specification Tracking system records and manages instrument specifications. Specification Tracking was primarily designed for use with the Calibration Process Manager. The Calibration Process Manager has a built in interface to use the specification data stored within the Specification Tracking database system. 
     Specification Tracking defines instruments using three hierarchical categories: Instruments, Functions and Ranges. The instrument is subdivided into instrument functions, which are subdivided into instrument ranges. Once an instrument is assigned to a data sheet, its ranges can then be assigned to individual data sheet rows. Calibration Process Manager has tools to automate the range selection process. 
     Specification Tracking Instruments make use of the underlying system of serialization in the CPMS that provides unique serial numbers to both the instrument and individual instrument ranges. As explained below, each instrument and range is assigned two serial numbers: A Permanent Unique Identifier (PUI) and a Dynamically Unique Identifier (DUI). The PUI is assigned when an object is created or altered in a manner that would make it unlinkable to its past usage. The DUI is assigned any time the object is modified/saved. These unique identifiers allow the Calibration Process Manager to track and relate linkages of these objects for historical and statistical purposes. Once assigned to a data sheet, these serial numbers form a permanent link between the data sheet/Calibration and the instrument&#39;s applicable ranges. 
     When instrument ranges are assigned to data sheet rows the Calibration Process Manager can use the data to determine Test Accuracy Ratios, Estimated Measurement Uncertainties, Instrument range utilization, and assist in the failure analysis of calibration standards. 
     The interface has three modes that correspond to each of the three instrument specification data applications used within the Calibration Process Manager. The Specification Tracking Interface mode indicators allow the user to determine which mode of the Specification Tracking Interface is active, even when the Specification Tracking Instruments Interface is not visible. 
     The Calibration Process Manager&#39;s test point measurement uncertainty model, test accuracy ratio, and applicable displayed results are always relevant to the currently selected Specification Tracking Interface mode. i.e.: When in the Standards used mode, the Model is using the actual calibration standards&#39; specifications, when in the Default Standards mode, the model is using the default standards&#39; specifications, etc. The values displayed will vary as the Specification Tracking Interface mode is changed. The following table describes the three modal applications of specification data. In the “Myself” mode the Specification Tracking instrument ranges of the instrument to be calibrated are assigned to applicable data sheet rows. This self definition information is used when the instrument to be calibrated is, itself, used as a calibration standard. When a calibration record is loaded into the Calibration Process Manger, the Statistics mode can be used to generate an Instrument Range Utilization Report. The report is generated for the currently selected data sheet row&#39;s applicable Specification Tracking instrument range. The report shows the instrument ranges&#39; use in other calibrations. This report is extremely useful in performing calibration standard failure analysis. 
     The “Def Stds” mode allows the data sheet designer to assign calibration standard instruments to the data sheet and assign instrument ranges to data sheet rows. These assignments represent the default standards, with range assignments, recommended for use when actual calibrations are performed. If default standards are assigned, the user has only to click on the “Stds Used” pages&#39; “Use ‘Default’ stds” button to transfer these predetermined selections to “Stds Used”. The data derived through these assignments will also permit the Calibration Process Manager to estimate accuracies, accuracy ratio&#39;s and estimated measurement uncertainties to assist the data sheet designer in determining the adequacy of test conditions. 
     The “Stds Used” mode is designed to assign calibration standard instruments to the data sheet and assign instrument ranges to data sheet rows at the time of calibration. The user will also assign the specific instrument asset numbers for the standards used. The data derived through these assignments will permit the Calibration Process Manager to calculate, record, and report test accuracy ratio&#39;s and estimated measurement uncertainties for data sheet test points. The modes are in table 37. 
     The Specification Tracking Instruments Interface opens in the lower portion of the screen. The data sheet grid is reduced in height by approximately a third, in order to accommodate the Specification Tracking Instruments Interface. The Specification Tracking Instruments Interface can be opened by selecting the View, Show Specification Tracking Instruments menu option. 
       FIG. 42  is a representation of what the Calibration Process Manger&#39;s main page (Data Sheet Designer mode) looks like with the Specification Tracking Instruments Interface displayed. 
       FIG. 43  is a representation of what the Specification Tracking Instruments Interface (Myself mode) looks like. In the Myself mode, specifications of the instrument being calibrated are assigned to the instrument itself. This self definition information is used to associate an instrument&#39;s calibration test points with its Specification Tracking instrument ranges. The process synchronizes Specification Tracking instrument and range serial numbers with the data sheet that calibrates the instrument. This identification process allows the Calibration Process Manger to link the usage of an instrument range to the data sheet on which it is calibrated. 
     This self awareness is especially important when the instrument is a calibration standard. When a calibration record is loaded into the Calibration Process Manger, the Statistics mode can be used to generate an Instrument Range Utilization Report. The report is generated for the currently selected data sheet row&#39;s applicable Specification Tracking instrument range. The report shows the instrument ranges&#39; use in other calibrations. This report is extremely useful in performing calibration standard failure analysis. 
     Another reason to perform this assignment is to compare data sheet specifications with Specification Tracking specifications. When performing an auto-assignment of instrument ranges to the data sheet, Calibration Process Manager will not auto assign an instrument range to a data sheet row if the calculated accuracies do not agree within 0.1% of each other. If the Calibration Process Manager is unable to make a range assignment, either the data sheet or instrument&#39;s specifications are in error. 
     The Specification Tracking Instruments Interface (Myself mode) contains the interface features in Table 38. 
       FIG. 44  is a representation of what the Specification Tracking Instruments Interface (Default Standards mode) looks like. In the Default Standards mode, specifications of the standards, planned for use during the execution of the data sheet (calibration), are assigned to the data sheet. This assignment of Specification Tracking instruments and instrument ranges to a data sheet&#39;s test points will enable the data sheet designer to pre-define a desired test point instrumentation configuration. When a user performs an actual calibration, with the data sheet, the user can select the Standards Used mode&#39;s “Use ‘Default’ Stds” button to copy the pre-defined configuration to the Standards Used configuration. The only other thing the user will have to do is assign asset numbers to the actual standards used. Another important purpose for assigning default standards is to provide the Calibration process Manger with the information necessary to calculate standards accuracies, test accuracy ratios, and estimated measurement uncertainty. This information is very useful in the calibration design process. The Specification Tracking Instruments Interface (Default Standards mode) contains the interface features in Table 39. 
       FIG. 45  is a representation of what the Specification Tracking Instruments Interface (Standards Used mode) looks like. In the Standards Used mode, specifications of the standards used during calibration are assigned. When a user performs a calibration, the user can click the “Use ‘Default’ Stds” button to copy the pre-defined default standards configuration to the Standards Used configuration. The only other thing the user will have to do is assign asset numbers to the actual standards used. If no default configuration is available or another configuration is used, the User can configure the standards at the time of calibration. 
     By associating the standards and standard ranges used during the calibration, to the data sheet test points, standard range utilization reporting can be achieved and greatly reduce the work required to perform standards failure analysis. The standard range data is also critical to the Calibration Process Manager&#39;s ability to calculate standards accuracies, test accuracy ratios and build an uncertainty model to calculate estimated measurement uncertainties. The Specification Tracking Instruments Interface (Standards Used mode) contains the interface features in Table 40. 
       FIG. 46  is a representation of what the Specification Tracking Interface mode indicators look like. The Specification Tracking Interface mode indicators allow the user to determine which mode of the Specification Tracking Interface is active, even when the Specification Tracking Instruments Interface is not visible. Each cell of the data sheet&#39;s Nominal Value column will have three small boxes located at the top of the cell. The indicators are only present in actual test point rows of the data sheet. The indicators appear in the same order as the mode selection buttons in the Specification Tracking Instruments Interface. A carrot symbol (▴) under one of the indicators indicates the present mode. The presence and color of the symbols convey meaning to the status of instrument ranges assigned, or not assigned, to the test point. The Specification Tracking Interface mode indicators convey the information in Table 41. 
       FIG. 47  is a representation of what the Instrument Specifications Catalog page looks like. 
     The Instrument Specifications Catalog page allows the user to choose an instrument to add to a data sheet. The Instrument Specifications Catalog page is accessed from the Calibration Process Manager&#39;s main page&#39;s Specification Tracking Instruments Interface by clicking on the Add button while in any of the three Specification Tracking Interface modes. The Instrument Specifications Catalog page contains the interface features in Table 42. 
       FIG. 48  is a representation of what the Auto Assign Instrument Ranges page looks like. The Auto Assign Instrument Ranges page is displayed when the user clicks the Auto Assign button, located on the Specification Tracking Instruments Interface&#39;s instrument section, while in any of the three Specification Tracking Interface modes. The Auto Assign Instrument Ranges page is displayed to inquire if the assignment is for the instrument currently selected in the Specification Tracking Instruments Interface&#39;s Instrument List grid or if the assignment is for all instruments assigned to the data sheet. The page also displays the following warning to the user describing the limitations of automatic instrument range selection: “Warning: This function will replace all previous assignments with the standard(s) most accurate range. After this operation, you should review my choices. After all, I am just a machine.” 
     Although auto selection is capable of choosing an inappropriate range, based on parametric matching and best accuracy, it is usually a good idea to let the Calibration Process Manager have a try at auto selecting the ranges. It is usually easier to change the few inappropriate selections than it is to do them all manually. Of course, the user&#39;s experience with the system and knowledge of the parametric matching probability of the instruments in question, is the best determination of the time saved by auto matching. This page contains the interface features in Table 43. 
       FIG. 49  is a representation of what the Manual Instrument Range Selector page looks like. 
     When the Calibration Process Manager is unable to auto select an instrument range or selects an inappropriate instrument range for a test point, the user must intervene. This intervention is performed using the Manual Instrument Range Selector. The Manual Instrument Range Selector page is accessed from the Calibration Process Manager&#39;s main page&#39;s Specification Tracking Instruments Interface by clicking on the Manual Assign to TP button while in any of the three Specification Tracking Interface modes. This page contains the interface features in Table 44. 
       FIG. 50  is a representation of what the Instrument Substitution page looks like. The Instrument Substitution page is displayed when clicking the “Replace selected with” button on the Calibration Process Manager&#39;s main page&#39;s Specification Tracking Instruments Interface. This button is available while in any of the three Specification Tracking Interface modes. 
     The Instrument Substitution page is used to remove the currently selected instrument&#39;s range assignments from all data sheet test point rows and replace those assignments with ranges from the instrument selected on the Instrument Substitution page. The Calibration Process manger will attempt to make range assignments for the newly selected instrument. Assignments will only be made for test point rows that had a range assigned for the instrument being replaced. All range assignments for the instrument being replaced will be removed, even if the newly selected instrument does not have an applicable range to assign in its place. This behavior is designed to reduce confusion as to why the removed instrument might still have ranges assigned to the data sheet. 
     The Instrument Substitution page also displays the following warning: “Warning: You should review all test points after this operation to ensure my selections are appropriate to the test point application. After all I will select ranges that meet my requirements, but the question is do they meet yours?” This page contains the interface features in Table 45. 
       FIG. 51  is a representation of what the Set Instrument Asset ID&#39;s page looks like. Asset ID&#39;s should be assigned to each standard used while performing a calibration. This option is only available when in the Specification Tracking Interface Standards Used mode (applicable only during an actual calibration). The Set Instrument Asset ID&#39;s page is accessed from the Calibration Process Manager&#39;s main page&#39;s Specification Tracking Instruments Interface by clicking on the Asset IDs button while in the Specification Tracking Interface “Stds Used” mode. This page contains the interface features in Table 46. 
     Calibration Process Manager is designed to automate data sheets for the calibration of measuring and test equipment. Once designed, Calibration Process Manager data sheets can then be used to collect and store calibration data for up to eight instruments simultaneously. Automation scripting can be added to these data sheets to configure instruments, capture data, perform mathematical operations, and create instrument test points dynamically. 
     The scripted data sheet may result in a completely unattended calibration, a series of user prompts and calculations, or only one row of a data sheet may have automation scripting. This flexibility is designed to maximize calibration productivity while minimizing the data sheet design process. 
     Whatever your choices in script design, the starting point is always a data sheet. The data sheet can be edited before, during, and after the automation scripting process. 
     The Calibration Process Manager will allow the user to add and configure scripting commands to any row of a data sheet. The system will allow the operator to cut and paste individual script commands, a series of script commands, or entire data sheet rows. Each row of a data sheet is an independent island of automation scripting. These islands can be bridged together by navigation script commands that produce a continuous execution of all or a portion of a data sheet. 
     Any data sheet can have scripting commands. The automation scripting editor manages the relationship of automation commands to data sheet rows. 
     The Calibration Process Manager&#39;s automation scripting system is based on the grouped and sequenced execution of parameterized script commands. Appendix C describes each of the commands available and their parameters. Note: Most command parameters allow the use of a user defined variable in place of a hard coded (fixed) value. 
     With the Master Mode is set to Data Collection, the View, Show Script menu option displays the command list box in the lower right corner of the screen. The data sheet grid is reduced in width by approximately half, in order to accommodate the command list box. The command list box contains the interface features in Table 47. 
       FIG. 52  is a representation of what the Automation Command Editor page looks like. This page allows the user to select and configure commands to be used in an automation script. This page also allows the user to assign the command to a command group to facilitate sequential execution of a collection of commands. As script commands represent the principle usage of user defined variables, this page is where the data sheet&#39;s user defined variables are managed. This page contains the interface features in Table 48. 
       FIG. 53  is a representation of what the Data Sheet Variables page looks like. The data sheet variables page allows the user to see the present content of all data sheet variables. The Automation scripting system uses these variables in many ways such as: Command parameter values, in equations, to hold user query values, etc. Any command parameter may have a user defined variable used in place of a direct value. 
       FIG. 54  is a representation of what the Data Sheet Variable Editor page looks like. The data sheet variable editor page allows the user to create and edit data sheet variables. This page contains the interface features in Table 49. 
       FIG. 55  is a representation of what the Command Group Editor page looks like. The Command Group Editor page is where the user can create command groups. Only one each, of command groups may be added into the list. Calibration Process Manager will display a warning screen if the user attempts to add a second iteration of a command group. Command groups are used to bind a set of commands into a command sequence. There are two types of command groups. 
     Interface Groups. These groups can be defined by the script author. When the user invokes the command menu, a unique list of all interface groups in the script, for the selected data sheet row, will appear on the menu. The Script author may assign a numeric “Key” value in the range of 0-9 to each interface group. The “Key” value will be displayed to the left of each item on the command menu. When the user presses the corresponding key, all of the commands assigned to the selected group are executed sequentially. Interface groups are displayed with a black font in the scripting system when created by the script author. 
     System Groups. These groups may not be edited by user or have “Key” values assigned to them. Their purpose is to bind script commands together for special behaviors that are programmed into the Calibration Process Manger&#39;s scripting system. 
     System and Interface groups may both be used within the same data sheet row to achieve a desired effect. While some script commands are designed to operate without being assigned to a command group, most commands will not execute unless they have a command group assigned to them. 
     This Command Group Editor page contains the interface features in Table 50. 
       FIG. 56  is a representation of what the User Interface Command Menu page looks like. The User Interface Command Menu page can be displayed when the Calibration Process Manger&#39;s main page is in the Data Collection Mode, a data sheet is loaded, and a data sheet row is selected that has at least one Interface Group assigned in its automation script command list. Once the above criteria is met the User Interface Command Menu page can be displayed by clicking the Run button, pressing the F12 key, or pressing the Command Menu button on the optional Quick Data Entry pad. 
     When the user invokes the command menu, a unique list of all interface groups in the script, for the selected data sheet row, will appear on the menu. When the user presses the corresponding key, all of the commands assigned to the selected group are executed sequentially. This page contains the interface features in Table 51. 
       FIG. 57  is a representation of what the Calibration Instruction Viewer page looks like. This page will display information about steps that need to be taken in order to perform a successful test. The information and instructions are set by the user in the Calibration Instruction Editor. It will prompt the user by opening a message window which contains information like: “ensure the unit is connected to the system”, and “test connections.” Issue warnings, such as: “High Voltage will be present during these tests”, and give information on how long a step will take before user intervention is required again. The calibration instructions viewer window has two buttons: O.K. move on, and Halt Script. The test will not proceed until the user presses the O.K. buttons. Pressing the Halt button will end the execution of the script. 
       FIG. 58  is a representation of what the Select a Calibration Instruction page looks like. This page allows the user to create instruction pages to assist during test execution. The page displays a list of the current instructions and allows for the addition of new instructions, removal of old instructions, and editing of all instructions. 
       FIG. 59  is a representation of what the Calibration Instruction Editor page looks like. This page allows the user to edit calibration instructions. 
       FIG. 60  is a representation of what the Graphical Instruction Viewer page looks like. This page allows the user to view instructions and graphical images to assist during test execution. The Instructions tab displays calibration instruction text. The text can be viewed by clicking on the tab. 
       FIG. 61  is a representation of what the Graphical Instruction Editor page looks like. This page allows the user to build a graphical instruction to assist during test execution. The page displays the name of the currently selected instruction and allows for the selection of other instructions, and the ability to edit instructions. The page also displays the currently selected images, their names, and figure numbers. Up to five images can be selected in this embodiment. The edit Binary List button provides the ability to add and remove images from the data sheet&#39;s Binary Resource List. This page contains the interface features in Table 52. 
       FIG. 62  is a representation of what the Binary Resource List page looks like. This page allows the user to manage binary resources contained within the data sheet. The page displays a list of the binary resources stored in memory and allows for the addition of new resources, removal of old resources, and editing of all resource names. As of the present version, the Calibration Process Manager only supports image files in the Binary Resource List, but this list will contain other types of files in the future. Resource files are stored in the database with the data sheet and are exported as files when the data sheet is exported to a file. This page contains the interface features in Table 53. 
       FIG. 63  is a representation of what the Binary Resource Editor page looks like. This page allows the user to add a binary resource from an external file and edit the name of the resource. This page contains the interface features in Table 54. 
       FIG. 64  is a representation of what the User Query Prompt page looks like. This page appears when the Sys_UserQuery command is executed. The Query User Prompt will assign values, entered by the user, into the assigned user defined variables. The Prompt is designed using the User Query Wizard. 
       FIG. 65  is a representation of what the User Query Wizard page looks like. This page allows the user to configure parameters for the Sys_UserQuery command&#39;s User Query Prompt. This page contains the interface features in Table 55. 
       FIG. 66  is a representation of what the GPIB Command Wizard page looks like. The GPIB command wizard allows the user to set the parameters for commands used to communicate with GIPB instruments. Clicking the Automation Command Editor&#39;s “Use the Command Wizard” button, with one of the GPIB commands selected, opens the GPIB Command Wizard page, described below. The GPIB Command Wizard page contains the interface features in Table 56. 
       FIG. 67  is a representation of what the RS-232 Command wizard page looks like. The RS-232 Command wizard allows the user to set the parameters for commands that communicate with instruments over a RS-232 connection. Clicking the Automation Command Editor&#39;s “Use the Command Wizard” button, with one of the RS-232 commands selected, opens the RS-232 Command Wizard page, described below. This page lists the current settings for the corn port. The settings can be edited using the Configure Port button. The RS-232 Command wizard page contains the interface features in Table 57. 
       FIG. 68  is a representation of what the Corn Port Setup Page looks like. This page allows the user to set the variables for the corn ports. This information is viewable at the bottom of the RS-232 Command wizard page. 
       FIG. 69  is a representation of what the ATE Clipboard Viewer page looks like. The ATE Clipboard Viewer allows the user to view the commands and associated objects that are currently held in memory. The ATE Clipboard Viewer contains the interface features in Table 58. 
     Calibration Process Manager includes several tools that can be used independently to assist the user in performing metrology work. All of the tools in this section are accessed by selecting one of the Tools menu options from the Calibration Process Manager&#39;s main page. 
       FIG. 70  is a representation of what the Equation Editor page looks like. The Equation Editor page contains the interface features in Table 59. 
       FIG. 71  is a representation of what the Data Sheet Variable Editor page looks like. The Equation Variable Editor Page allows the user to create and edit equation variables. 
       FIG. 72  is a representation of what the Custom Calculator Workshop page looks like. The Calibration Process Manager system allows the user to create calculators for use with data sheets and for standalone use. Custom calculators can be saved for re-use later. Custom Calculators can be configured to prompt the user for variable values and execute one or more equations. The equations are edited using the Equation Editor page. When a custom calculator is intended for use with the Calibration Process Manager&#39;s integrated tolerance calculator, the calculated result of all custom calculators should be in the same unit of measure as the “nominal value” of the test point. The value must also be absolute. If the value is negative, the tolerance calculator will declare it to be in error. The custom calculator workshop page contains the interface features in Table 60. 
     The Open Custom Calculator button allows the user to open previously saved Calibration Process Manager custom calculator files. These files will have a file extension such as .MCC (Calibration Process Manager Custom Calculator). This is a standard Windows open file dialogue box. 
     The Save a Custom Calculator button allows the user to save Calibration Process Manager custom calculator files. These files will have the file extension of .MCC (Calibration Process Manager Custom Calculator). This is a standard Windows open file dialogue box. 
       FIG. 73  is a representation of what the User Query Setup Page looks like. This page allows the user to input customized Calculator User Queries for use in the Custom Calculator Workshop. The features are in Table 61. 
       FIG. 74  is a representation of what the Custom Calculator Equation Variable Editor Page looks like. The Custom Calculator Equation Variable Editor Page allows the user to set variable values for equations in the custom calculator workshop. 
       FIG. 75  is a representation of what the GPIB Interactive Controller page looks like. The GPIB Interactive Controller page allows the user to set the parameters for data being collected over a GIPB connection. This interface allows for direct manipulation of attached units without requiring automation script commands. The GPIB Interactive Controller page contains the interface features in Table 62. 
       FIG. 76  is a representation of what the RS-232 Interactive Controller page looks like. The RS-232 Interactive Controller allows the user to set the parameters for data being collected over a RS-232 connection. This interface allows for direct manipulation of attached units without requiring automation script commands. This page also lists the current settings for the corn port. The settings can be edited using the Configure Corn Port button. The RS-232 Interactive Controller page contains the interface features in Table 63. 
       FIG. 77  is a representation of what the HTML Table Generator Page looks like. The HTML Table generator page allows the user to copy all, or part, of a data sheet to the Windows clipboard. The information will be in HTML format. 
       FIG. 78  is a representation of what the Tab Delimited Table Generator Page looks like. The Tab Delimited Table generator page allows the user to copy all, or part, of a data sheet to the Windows clipboard. The information will be in Tab Delimited format. 
       FIG. 79  is a representation of what the Units Converter Page looks like. The Units Converter page allows the user to convert values from one unit of measure to another. 
     Globally Unique Identifiers 
     Implementation of the Calibration Process Manager is facilitated by a data structure for recording calibration data from a UUT. Typically, with reference to  FIG. 80 , this data structure will be stored on a hard drive memory  54  of the computer  52  which is connected to the calibration testing unit  56 , and the UUT  58 . It is possible to have multiple UUTs  58  connected and tested using the present invention. 
     As will be known in the art with reference to this disclosure, it would be possible to store a data structure of the present invention on any computer readable medium, such as floppy drives, dismountable hard drives, DVDs, CD-ROM, CD-R, CD-RW, memory cards, and the like. 
     The structure centers on use of two guaranteed identification address globally unique identifiers (“GUID”). The generation of GUIDs is known in the art, and information relating to GUIDs may be found in the Microsoft Knowledge Base article Q176790 “HOWTO: Use CoCreateGUID API to Generate a GUID with VB.” 
     The first GUID used in the present invention is a permanent unique ID, or “PUI.” The second GUID used in the present invention is a dynamic unique ID, or “DUI.” 
     This underlying system of serialization provides the unique identifiers to both the data sheet and individual rows of the data sheet. Preferably, serial numbers are used as the identifiers. Each data sheet and data sheet row is assigned two serial numbers: A Permanent Unique Identifier (PUI) and a Dynamically Unique Identifier (DUI). The PUI is assigned when an object is created or altered in a manner that would make it unable to link to its past usage. A new DUI is assigned each time the object is modified/saved. 
     These unique identifiers allow the Calibration Process Manager to track and relate linkages of these objects for historical and statistical purposes. The relationship of the unique identifiers to data sheet and calibration history should be understood and considered any time a datasheet is modified. Calibration test point history is related to data sheet row by its PUI. Changing parametric data in the Nominal Value cell of a data sheet row will cause the Calibration Process Manger to issue the row a new PUI. When these parameters are changed the Calibration Process Manager must assume that the quantitative relationship to calibration history for the test point is no longer applicable. The result is that the data sheet row is no longer directly linkable to calibration test points that used the row in the past. The security of data preservation is not broken, but statistical analysis will be disrupted. The bottom line is that the user must consider these impacts in the decision to alter the data sheet. 
     The system permits a secure environment along with an audit trail to track changes to the system parameters. Archiving of records is also possible. 
     Using the PUI and DUI identifiers, for example, it will be possible to locate all UUTs that were measured with a particular calibration testing unit  56 . If it is later determined that the particular calibration testing unit  56  is itself out of calibration in a particular range, the PUI/DUI identifiers permit identifying each UUT  58 , and each range tested, that may be affected by the our of calibration testing unit. 
     Method for Calibration 
     With reference to  FIG. 81 , method  200  for calibrating a UUT for a specific calibration function is described, comprising the steps of receiving  202  a first identification attribute associated with a UUT; receiving  204  a second identification attribute associated with a reference measuring unit; receiving  206  a specific calibration function to be tested; maintaining  216  in a first memory a reference database of one or more than one second identification attribute in holding relationship to one or more than one reference identification address, each reference identification address being associated with a single calibration function; looking up  208  in the reference database, the reference identification address being held by the second identification attribute, and associated with the specific calibration function; selecting  210  a unique record identification address; receiving  212  a value as found for the specific calibration function for the UUT; storing  214  in a second memory the value as found in a being-held relationship to the record identification address; and storing  216  in a third memory the record identification address in a being-held relationship to the reference identification address. 
     In one preferred embodiment, either the first identification attribute (not numbered) or the second identification attribute  122 , or both the first identification attribute and the second identification attribute  122 , can be an asset number. 
     In a further embodiment, steps may be added of receiving (not shown) a value as left for the specific calibration function for the UUT and storing in the second memory the value as left in a being-held relationship to the record identification address. Additionally, optionally steps (not shown) may be added returning a test accuracy ratio, or returning an estimated measurement uncertainty, or automatically converting the units of the received value as found. 
     In a further embodiment, method  200  may further comprise maintaining in a fourth memory an equipment database correlating for individual make and model of equipment, a nominal value to calibration function; receiving make and model of equipment attributes for the UUT; looking up in the equipment database, the nominal value for the specific calibration function for the received make and model of equipment; returning the nominal value; and calculating a calibration tolerance and returning the limits. In a still further embodiment, additional steps may be receiving a value as left for the specific calibration function for the UUT and storing in a fifth memory the value as left in a being-held relationship to the record identification address. The method may still further comprise the step of indicating whether the value as found or the value as left is within the limits of the calibration tolerance. 
     Management System 
     As shown in  FIG. 82 , a Calibration Process Management Program  10  implementing method  200  is shown, of which the previously described Calibration Process Manager is one embodiment, comprising a set of modules including calibration tools  12 , engineering tools  14 , and management tools  16 . Multiple formerly independent modules can thus be integrated into a Calibration Process Management System (“CPMS”)  50  comprising program  10 , residing on memory  54  of an appropriate computer  52 , and an optional printer  60 . A data sheet  62  may then be printed. A sample of a data sheet  62  is shown in  FIG. 83 . 
     The calibration tools module  12  comprises calibration tools  121 , and optionally data sheet generation wizards  122 , and editors and calculators  123 . The engineering tools module  14  comprises measurement analysis tools  143  and optionally failure analysis tools  141 , drift analysis tools  142 , uncertainty tools  144 , and interval analysis tools  145 . The management tools module  16  comprises datasheet management tools  161 , and optionally security access  162 , and workload distribution  163 . These have been described in various embodiments in the Calibration Process Manager above. Other embodiments of these tools will be evident to those skilled in the art with reference to this disclosure. 
     It would be possible to implement the CPMS on any type of personal computer, such as a Pentium® class machine. Operating systems useable include those available from Microsoft®, including Windows® 98, ME, 2000, and XP. The computer memory can be any media useable for storage of computer data, including but not limited to hard drives, floppy drives, removable drives, RAM, ROM, DVDs, and CD-R. 
     It is possible to implement management program  10  as a 32 bit, Windows 9x, NT, or 2000 based program, designed for standalone or Client/Server SQL database applications. In one embodiment, it can be designed and developed to satisfy ISO-17025, ANSI Z-540, 10 CFR 50 Appendix B, ISO-9002, GMP, and other stringent quality requirements for calibration data collection. 
     Certain aspects of Calibration Process Management Program  10  are desirable, and will now be described. It will be understood that individually each of these are optional. 
     An integrated tolerance calculator allows a data sheet designer to generate test point row data by merely entering nominal test point values and equipment specification data. The tolerance is calculated, formatted, and then the new test points are inserted into the data sheet. Preferably, the calculator uses a smart rounding system to appropriately round tolerance values, and supports both standard specifications (% of I.V., ppm of I.V., % of F.S., ppm of F.S., ±n Units of measure, and ±db) and custom specifications. Custom specifications are more complex specification components such as: “±0.025% of I.V. per volt above 750 volts”. Custom specifications can be defined by the user, then used in combination with the standard specification components in the tolerance calculator. 
     A custom tolerance calculator is used to calculate tolerances, based on specification components that are more complex than the standard specification components (% of I.V., ppm of I.V., % of F.S., ppm of F.S.±n Units of measure, and ±db). These more complex specification components are things such as: “±0.025% of I.V. per volt above 750 volts”. CPMS 50 provides a special set of tools that allow you create your own library of custom specification statements and then use them in combination with the standard specification components in tolerance calculator. 
     An equation system permits equations to be constructed using a user-friendly equation editor. The system uses, standard operands, a function library, user defined variables, and conditional clauses in equations. The equation system can be used throughout CPMS 50 to create and execute equations in such operations as: custom calculators, units conversion, automation scripting math commands, etc. 
     When creating data sheets, development productivity and format continuity are important factors. This is where data sheet generation wizards fit the bill. These wizards allow for the predefinition of test point values, tolerance format, function tested descriptions, and data sheet boilerplate information. A single data sheet wizard can contain multiple test point definitions. 
     As an example, a wizard for digital multimeters with a range base of 200 (i.e. 200 mV Range) might contain the following test point definitions shown in Table 64. 
     Selecting one of these test point definitions will automatically configure the tolerance calculator with predefined numeric and tolerance formats, and setup an array of test points. When the new test points are inserted into the data sheet, the Step Number and Function Tested descriptions are taken from the wizard test point definition. Any of the predefined values can be overridden, if necessary. 
     Data sheets can be created using special data sheet wizards. These wizards prompt the user for specification data then generate an entire datasheet. An example of a user interface suitable for use in a data sheet generation wizard is shown in  FIG. 21 . 
     Multiple Channel Data Collection. For many types of instruments, it is much more efficient to collect data for more than one instrument at the same time. CPMS 50 allows you to do this, providing the instruments use the same data sheet. Up to eight simultaneous sets of measurement data can be collected within CPMS 50 in one embodiment. Each set of measurement data is referred to as a “Channel”. Each channel has a display module for “As Found”, “As Left”, and “Out of Tolerance” data. All eight channels can be displayed and edited simultaneously. 
     Automatic “Out of Tolerance” detection. When a measured value is recorded for an instrument channel, CPMS 50 will compare the value to the instrument&#39;s “Tolerance”, indicate the “Out of Tolerance” status, and display the percentage of the tolerance consumed. 
     Data collection via automation scripting. Instrument data can also be collected through the use of an automation scripting system. This allows data collection from GPIB, or RS-232, custom user prompts, or the result of scripted mathematical operations. 
     Storage and retrieval of collected data. Measurement data collected for each channel can be stored independently. This data can be recalled or reprinted at any time. The data can be either recalled, into any one of the eight instrument channels, or opened as an independent data sheet. 
     Automation Scripting. CPMS 50 allows you to add automation scripting commands to any row of a data sheet. Scripting data sheet rows independently has many advantages, but chief among them is the direct interaction between a row&#39;s script commands, test point data, and measurement data. With this direct correlation you can apply automation commands to a specific row or section of a data sheet, by simply selecting the applicable data sheet row. If only one section of a data sheet is to be automated, no action is required in any unaffected row. Even when scripting is available for a section/row of a data sheet, it can also be treated as just that, a simple data sheet row. The user can simply enter the measurement data or run the applicable part of the script. 
     Editing the automation script for a data sheet is simple. Script commands for a data sheet row are displayed whenever that row has focus in the data sheet editor. The script editor makes it simple to cut and paste a series of commands, from one data sheet row, to another data sheet row, or even another data sheet. When copying and pasting entire data sheet rows, the data sheet editor will also copy and paste all applicable commands. 
     Scripting of preexisting data sheets. Any data sheet can have scripting commands added to it. The result is incremental development. Whenever someone creates a data sheet, they are laying out the framework for an automation process. When scripting is undertaken, this work does not have to be repeated. As a result of the direct relationship of the script to the data sheet row, evaluation of measured values to the calibration tolerances is internally handled, with no scripting required. 
     Scripting commands are available for both GPIB and RS-232 operations. 
     CPMS 50 has the ability to use scripting commands to actually build a data sheet. Scripting commands are available to prompt the user for specification data, manipulate the tolerance calculator, perform math operations, and generate test points. This type of scripting is very useful as replacement for generic data sheets. Creating self generating data sheets for instruments such as torque wrenches, pressure gauges, micrometers, etc., is very easy. 
     The automation scripting system features commands can make full use of a CPMS 50 equation system described above. Interaction with external applications can be done by including in the scripting commands Object Linking and Embedding (OLE) commands. These commands allow the scripting system to share data with external applications, i.e., MATHCAD, WORD, EXCEL, etc. Calibration instructions can be created with a Rich Text Format (RTF) instruction editor or imported from another word processor. The instructions can be displayed using a script command. 
     Groups of commands can be saved to a file. These files are intended to serve as a command group library. Groups of commands can be inserted directly into a script from one of these files. Groups of commands can also be cut and pasted across rows or even data sheets. 
     An intelligent SI Units of Measure converter automatically interprets measurement units for the nominal values in the calibration data sheet. The Units of measure engine also automatically converts measurement units be the UT and applied standards. The Units of measure engine can also be used as a stand alone units converter for quick conversions. 
     Specification Tracking 
     Specification Tracking is an optional CPMS 50 module that contains a library of calibration standards accuracy specifications and/or uncertainties. Specification Tracking enables CPMS 50 to assign calibration standards, and their related function, range and accuracy specifications to data sheets and applicable UUT calibration data. Standards accuracy specifications are entered through a unique user interface which provides a systematic, logical architecture for building specification data. 
     Standards specifications can be divided by Function, Parameter, Range, Accuracy, Resolution, and other meteorological criteria to properly identify the specification. Specification data stored in Specification Tracking can be used throughout CPMS 50 in the calculation of Estimated Measurement Uncertainty (EMU), Test Accuracy Ratios (TAR) and failure analysis. 
     CPMS 50 can be used in the field to collect calibration data and merge data into the main system by adding optional Mobile PC support. 
     The CPMS can also optionally automatically calculate each test point&#39;s estimated measurement uncertainty (“EMU”), using principle sources of error from standards accuracy specifications, inverse normal distribution function, in-tolerance probability, and uncertainty contributed by the measurement system resolution. These sources of error are combined with a suitable coverage factor and expressed as EMU. This meets the Quality Standards and Practices requirement of ISO/IEC 170255.4.6 “Estimation of uncertainty of measurement” which states in section 5.4.6.2 “Nesting laboratories shall have and apply procedures for estimating uncertainty of measurement. In certain cases the nature of the test method may preclude rigorous, metrologically and statistically valid, calculation of uncertainty of measurement. In these cases the laboratory shall at least attempt to identify all the components of uncertainty and make a reasonable estimation.” ISO/IEC 17025, “General Requirements for the Competence of Testing and Calibration Laboratories.” 
     There are two distinct testing methods; Type A tolerance testing (method of evaluation of uncertainty by the statistical analysis of series of observations) and Type B uncertainty testing (method of evaluation of uncertainty by means other than the statistical analysis of series of observations). In tolerance testing measurement reliability is the objective, parametric data is not utilized, and is typically done in a secondary calibration laboratory. In uncertainty testing, typically done in a primary standards laboratory, measurement uncertainty is the objective, and parametric data is utilized. 
     Preferably, the CPMS is capable of Type B evaluation of uncertainty. It is sufficient to assume that a heuristic estimate is adequate, and that standards and UUT are the principle sources of error, where the standards contribution is based on in-tolerance probability, and the UUT contribution is based on measurement system resolution. 
     One way of defining the uncertainty contributed by the standard is shown in Equation (Eq. 1), where L S  is the standard accuracy specification, Φ −1  is the inverse normal distribution function, and P is the in tolerance probability. 
     
       
         
           
             
               
                 
                   
                     u 
                     s 
                   
                   = 
                   
                     
                       L 
                       s 
                     
                     
                       
                         Φ 
                         
                           - 
                           1 
                         
                       
                        
                       
                         ( 
                         
                           
                             1 
                             + 
                             P 
                           
                           2 
                         
                         ) 
                       
                     
                   
                 
               
               
                 
                   ( 
                   
                     Eq 
                     . 
                     
                         
                     
                      
                     1 
                   
                   ) 
                 
               
             
           
         
       
     
     One way of defining the uncertainty contributed by the UUT is shown in Equation (Eq. 3), where X. R  is the measurement system resolution (e.g., ½ LSD) 
     
       
         
           
             
               
                 
                   
                     u 
                     R 
                   
                   = 
                   
                     
                       x 
                       R 
                     
                     
                       3 
                     
                   
                 
               
               
                 
                   ( 
                   
                     Eq 
                     . 
                     
                         
                     
                      
                     2 
                   
                   ) 
                 
               
             
             
               
                 
                   
                     u 
                     R 
                   
                   = 
                   
                     
                       X 
                       R 
                     
                     
                       3 
                     
                   
                 
               
               
                 
                   ( 
                   
                     Eq 
                     . 
                     
                         
                     
                      
                     3 
                   
                   ) 
                 
               
             
           
         
       
     
     The combined standard and UUT uncertainty is then given by Equation (Eq. 4), where u S  is the standard uncertainty and u R  is the UUT uncertainty. 
         u   C =√{square root over ( u   S   2   +u   R   2 )}  (Eq. 4) 
     The estimated measurement uncertainty may then be expressed by Equation (Eq. 5), where k is a coverage factor (e.g. k=2). “Estimates of uncertainty are derived form the statistical concept of the standard deviation. The standard deviation is a characteristic or number that describes the amount of variability of a distribution of measurements.” Jackson, D. and Castrup, H., “Uncertainty Propagation in Calibration Support.” 
         U=k·u   C   (Eq. 5) 
     Substituting Equation (Eq. 4) into Equation (Eq. 5), we can obtain an approximate value for U echelon I  in a Type B evaluation using Standard and UUT data from Equation (Eq. 6), where k factor  is the coverage factor. 
         U   echelon I   ≈k   factor   ·[u   standard   2   +u   resolution   2 ] 0.5   (Eq. 6) 
     Similarly, we can obtain an approximate value for U echelon II  in a Type A evaluation using parametric test data with Equation (Eq. 7). 
         U   echelon II   =k (ν eff   ,p )[Σ( c   i   u   standard     i   ) 2 +Σ( c   j   u   other ) 2 +u test data   2 ] 0.5   (Eq. 7) 
     Further, we can obtain an approximate value for U echelon III  in a Type A evaluation using parametric historical data with Equation (Eq. 8). 
         U   echelon III   =k (ν eff   ,p )[Σ( c   i   u   standard     i   ) 2 +Σ( c   j   u   other     j   ) 2   +u   historical data   2 ] 0.5   (Eq. 8) 
     A CPMS preferably automatically calculates each test point&#39;s test accuracy (“TAR”) ratio between the UUT and the applicable standard&#39;s function and range, preferably in accord with ANSI/NCSL Z540-1-1994 “Calibration Laboratories and Measuring and Test” section 10.2 b): “The laboratory shall ensure that the calibration uncertainties are sufficiently small so that the adequacy of the measurement is not affected . . . the collective uncertainty of the measurement standards shall not exceed 25% of the acceptable tolerance (e.g., manufacture&#39;s specification) for each characteristic of the measuring and test equipment being calibrated or verified.” 
     In one preferred embodiment, the CPMS uses an algorithm that combines the calibration tolerance of the UUT, and the collective uncertainty of the standard to derive the Test Accuracy Ratio or TAR. This method represents the model specified in ANSI/NCSL 2540-1-1994. Section 10.2, and is expressed in Equation (Eq. 9), where TAR is the Test Accuracy Ratio, Lx is the calibration tolerance of the UUT, and the radical denominator term is the collective uncertainty of standards, k=1. 
     
       
         
           
             
               
                 
                   TAR 
                   = 
                   
                     
                       
                         L 
                         X 
                       
                       / 
                       
                         3 
                       
                     
                     
                       
                         ∑ 
                         
                           u 
                           S 
                           2 
                         
                       
                     
                   
                 
               
               
                 
                   ( 
                   
                     Eq 
                     . 
                     
                         
                     
                      
                     9 
                   
                   ) 
                 
               
             
           
         
       
     
     Failure Analysis. This feature enables the user to automatically evaluate out-of-tolerance conditions by tracking the function, range and test point information between each standard and the unit under test. Failure analysis pertains to compliance with ISO 17025 paragraph 4.9 b) or Z540-1 paragraph 8.2. That is, the laboratory shall examine the effect of non-conforming work or equipment defects (out-of-tolerance conditions) on other work. Most labs keep track of what standards are used to calibrate what equipment. Therefore, the starting point for failure analysis is an out-of-tolerance condition on a standard and a list of items on which the standard was used. It is then up to the evaluator to re-trace each calibration to determine whether or not the individual range and test point that failed on the standard was actually used on the UUT, to quantitatively assess what effect the failure had, and to notify the client if the effect was significant. Imagine doing this if the standard is a multifunction meter calibrator and the UUT list contains 100 items, and across the laboratory&#39;s inventory of standards. 
     The CPMS enables this evaluation to be done automatically by keeping track of the function, range and test point information between each standard and UUT. By knowing the applicable specifications and the magnitude of the failure, it could determine whether or not the failure had an adverse effect on the calibration. It could produce a list of items that were actually affected, where and by how much. It could even prepare this information for transmittal to the client. This a huge feature because most labs are overwhelmed by this or do not comply with this provision of the standard. 
     Failure Report. Failure analysis reports can be generated by selecting the calibration standard, function and range within a specified date range. This report indicates which UUT where calibrated by the selected standard, grouped by individual Asset ID number, and the specific function, ranges, and test points affected. 
     Drift/Interval Analysis. Drift and Interval Analysis enables users to plot data and to analyze trends of measuring and test equipment, between calibrations, on a test point basis. This feature will allow users to determine optimum calibration intervals for their equipment. Drift analysis tracks and analyzes trends in laboratory standards and client M&amp;TE between calibrations on a test point by test point basis. The Drift analysis module allows you to plot data, look for trends and calculate basic statistics. Ultimately, this tool will enable the laboratory to establish the uncertainty of its standards and client&#39;s M&amp;TE in real time, to determine optimum calibration intervals for its equipment. 
     CPMS 50 may also incorporate advanced security features including; Comprehensive User Manager Detailed Audit Trail; and Selectable audit trail reporting. A User Manager allows the system Administrator to set user access, and permission levels within CPMS 50. Users&#39; activities are automatically tracked and logged in the audit trail for database trace ability and compliance to 21 CFR Part 11 audit requirements. 
     Revision Control. Calibration data sheets and subsequent revisions can be maintained and controlled through CPMS 50 exclusive data sheet approval process. Data sheets are created in an ASCII file based format out side of the SQL database. Data sheets can only be inserted into a database by individuals authorized for data sheet approval. Upon review and approval, data sheets are e-signed, and time and date logged as new or a revision, as selected by the authorized individual. When a data sheet is revised, the old data sheet is inactivated, but maintained in the system, thus preventing the use of retired data sheets, and only allowing the currently approved data sheet for use. 
     Storage options. Typically, a Data sheet file contains all of the test point data, specifications, automation script commands, and calibration instructions for a single data sheet. A calibration file is a simplified version of a Data sheet file. No automation scripting or calibration instructions are stored. The calibration file also stores the UUT&#39;s measurement data. The calibration file can be either recalled into one of the UUT channels or reopened independently as a data sheet, with UUT measurement data. 
     CPMS 50 may also feature an automated cataloging system of all data sheets and calibration files. The catalog can be used to locate a data sheet or a calibration file. The data sheet catalog can be searched by model, manufacturer, equipment description or calibration procedure. CPMS 50 can also support barcode data entry for M&amp;TE ID and Test Report numbers. 
     Data sheets can be created by any person with the prerequisite technical knowledge of the unit under test. The actual process of creating data sheets requires minimal training. Data sheets should be approved by the appropriate authorized individual before they can be inserted into the database for use. Data sheets are initially created in file format, and must be approved, and e-signed prior to submittal into the database. This process assures proper data sheet validation and approval in accordance with industry quality standards, and users quality procedures. Data sheet approvals are controlled by the customer, adequate controls should be in place for the review, acceptance and approval of data sheets prior to use. 
     Data sheet revision control. In further embodiments, Data sheets must not only be approved, their revision are also tracked and controlled. Upon approval of a new data sheet, CPMS 50 tracks previously created data sheets of the same number. If previous data sheets exist, the new data sheet can be added as a revision, or a new data sheet. CPMS 50 tracks old revisions and only allows the current revision for use. 
     The usual starting place for an automation script is a previously created data sheet. Having said that, scripting commands can be added to a blank data sheet if desired. The training requirements for script authors will vary with the nature of the scripts being written. If, for example, a mathematically intensive script is being written, the author must have the requisite math skills. As far as the actual scripting process is concerned, the direct relationship of data sheet rows and script commands keeps the process relatively easy. 
     All features disclosed in the specification, including the claims, abstract, and drawings, and all the steps in any method or process disclosed, may be combined in any combination, except combinations where at least some of such features and/or steps are mutually exclusive. Each feature disclosed in the specification, including the claims, abstract, and drawings, can be replaced by alternative features serving the same, equivalent or similar purpose, unless expressly stated otherwise. Thus, unless expressly stated otherwise, each feature disclosed is one example only of a generic series of equivalent or similar features. 
     Also, any element in a claim that does not explicitly state “means for” performing a specified function or “step for” performing a specified function, should not be interpreted as a “means” or “step” clause as specified in 35 U.S.C. §112. 
     APPENDIX A 
     
         
         
           
             AF As Found 
             AL As Left 
             A/N Alpha-numeric 
             Bar Barometer 
             Cal Calibration 
             CFR Code of Federal Regulations 
             Com Port Communications Port 
             CR Carriage Return 
             CRLF Carriage Return Line Feed 
             dB Decibels 
             OLE Object Linking and Embedding 
             Esc Escape 
             F.S. Full Scale 
             GPIB General Purpose Interface Bus 
             Hg Mercury 
             inHg In Mercury 
             HW Hardware 
             IEEE Institute of Electrical and Electronic Engineers 
             I.V. Indicated Value 
             LF Line Feed 
             Max Maximum 
             Mhz Mega Hertz 
             mm Millimeter 
             mV millivolt 
             N/A Not Applicable 
             Nav Navigation 
             No Number 
             Pa Pascal 
             ppm Parts Per Million 
             Psig Pounds per Square Inch Gauge 
             Psia Pounds Per Square Inch Absolute 
             RAM Random Access Memory 
             Rcv Receive 
             Ref Reference 
             RS-232 Recommended Standard 232 (computer interface cable) 
             RTSCTS Request to send clear to send 
             SDD Software Design Description 
             SOPs Standard Operating Procedures 
             SQL Structured Query Language 
             SW Software 
             Tol Tolerance 
             Tx Transmit 
             UUT Unit Under Test 
             Val Value 
           
         
       
    
     
       
         
           
               
               
               
               
             
               
                 APPENDIX B 
               
               
                   
               
             
            
               
                 File 
                 Data Sheet 
                   
                 This option allows the following: 
               
               
                   
                   
                 Open 
                 Loads a previously saved data sheet. 
               
               
                   
                   
                   
                 This option opens the Open Data Sheet catalog. 
               
               
                   
                   
                 Save 
                 Saves edits made to the currently opened data 
               
               
                   
                   
                   
                 sheet. This option exports the Data Sheet to 
               
               
                   
                   
                   
                 a File. 
               
               
                   
                   
                 Save As 
                 Saves edits made to the currently opened data 
               
               
                   
                   
                   
                 sheet, also allows the operator to rename the 
               
               
                   
                   
                   
                 data sheet. This option exports the Data Sheet 
               
               
                   
                   
                   
                 to a File. 
               
               
                   
                   
                 New 
                 Creates a blank data sheet. 
               
               
                   
                   
                 Approve 
                 This option allows users with data sheet 
               
               
                   
                   
                   
                 approval privileges to approve a data sheet for 
               
               
                   
                   
                   
                 use in the system database. 
               
               
                   
                 Calibration File 
                   
                 This option allows the following: 
               
               
                   
                   
                 Open 
                 This option allows the user to open a 
               
               
                   
                   
                   
                 previously saved Calibration. 
               
               
                   
                   
                 Save 
                 This option allows the user to Export the 
               
               
                   
                   
                   
                 current calibration to a file. 
               
               
                   
                 Import 
                   
                 This option allows the following: 
               
               
                   
                   
                 Data Sheet File 
                 This option imports data sheet files into the 
               
               
                   
                   
                   
                 editor for review/edit/approval. 
               
               
                   
                   
                 Calibration File 
                 This option imports a calibration file into the 
               
               
                   
                   
                   
                 editor for review/edit/approval. 
               
               
                   
                 Export 
                   
                 This option allows the following: 
               
               
                   
                   
                 Data Sheet File 
                 This option exports data sheets to a file 
               
               
                   
                   
                 Calibration File 
                 This option imports calibrations to a file. 
               
               
                   
                 Find 
                   
                 This option allows the following: 
               
               
                   
                   
                 Data Sheet 
                 This option opens the Data Sheet Database 
               
               
                   
                   
                   
                 Catalog. 
               
               
                   
                   
                 Calibration 
                 This option allows users to search the system 
               
               
                   
                   
                   
                 database for calibrations. 
               
               
                   
                   
                 Calibration 
                 This allows the user to search the system for 
               
               
                   
                   
                 File&gt;(*.mdc) 
                 files with the .mdc extension to open and 
               
               
                   
                   
                   
                 calibrate. 
               
               
                   
                   
                 Calibration File&gt;All 
                 This allows the user to search the system for 
               
               
                   
                   
                 Files 
                 files to pen. 
               
               
                   
                 Exit 
                   
                 This option terminates the application. 
               
               
                 View 
                 Channel 
                   
                 This option allows the following: 
               
               
                   
                 Matrix 
               
               
                   
                   
                 2 × 4 
                 Displays 2 rows of 4 Channel boxes. 
               
               
                   
                   
                 1 × 4 
                 Displays 1row of 4 Channel boxes. 
               
               
                   
                   
                 1 × 3 
                 Displays 1row of 3Channel boxes. 
               
               
                   
                 Show Script 
                 Commands 
                 This displays the Command List in the lower right 
               
               
                   
                   
                   
                 of the main screen. The data sheet grid section of 
               
               
                   
                   
                   
                 the screen will be resized to approximately half the 
               
               
                   
                   
                   
                 original size while the command list is visible. 
               
               
                   
                   
                   
                 “As found,” “As left,” and “out of tolerance” 
               
               
                   
                   
                   
                 columns are not displayed in this mode. 
               
               
                   
                 Show Specification 
                 Instruments 
                 This option opens the Specification Tracking 
               
               
                   
                 Tracking 
                   
                 Interface page. 
               
               
                 Tools 
                 Wizards 
                 Test Point 
                 This option opens the Wizard Loader page 
               
               
                   
                   
                 Generators&gt;Load 
               
               
                   
                   
                 Test Point 
                 This option opens the Wizard Workshop 
               
               
                   
                   
                 Generators&gt;Workshop 
               
               
                   
                   
                 Data Sheet 
                 This opens the Pressure Gauge Data Sheet Generator 
               
               
                   
                   
                 Generators&gt;Pressure 
                 page. 
               
               
                   
                   
                 Data Sheet 
                 Allows the user to create a new group of data Sheets. 
               
               
                   
                   
                 Group&gt;New Group 
               
               
                   
                   
                 Data Sheet 
                 Allows the user to edit members of an existing group 
               
               
                   
                   
                 Group&gt;Edit Existing 
               
               
                   
                 Calculators 
                 Units Conversion 
                 This option opens the Specification Tracking 
               
               
                   
                   
                   
                 units conversion tool details. 
               
               
                   
                   
                 Custom 
                 This opens the Custom Calculator Workshop page. 
               
               
                   
                   
                 Equation Editor 
                 This option opens the Equation Editor page. 
               
               
                   
                 Instrument 
                 GPIB (IEEE-488) 
                 This opens the GPIB Interactive page. 
               
               
                   
                 Controllers 
                 Com Ports (RS-232) 
                 This opens the RS-232 Interactive page. 
               
               
                   
                 Clipboard 
                 HTML Table 
                 This opens the HTML Table Generator page. 
               
               
                   
                 Special Copy 
               
               
                   
                   
                 Tab Delimited 
                 This opens the Tab Delimited Table Generator page. 
               
               
                   
                 Modules 
                 Specification 
                 This option activates Specification Tracking. 
               
               
                   
                   
                 Tracking 
               
               
                   
                 External 
                 Microsoft Notepad 
                 This opens the standard Windows notepad. 
               
               
                   
                 Applications 
                   
               
               
                   
                   
                 Microsoft Explorer 
                 This opens Windows Explorer. 
               
               
                   
                   
                 Microsoft Calculator 
                 This opens the Windows calculator. 
               
               
                 Options 
                 Preferences 
                 Sound Settings 
                 This option allows the operator to set the audio 
               
               
                   
                   
                   
                 properties for the Calibration Process Manager 
               
               
                   
                   
                   
                 application, if implemented. 
               
               
                   
                 Change Password 
                   
                 This option allows the user to change their personal 
               
               
                   
                   
                   
                 password. 
               
               
                   
                 Show Hints 
                   
                 Turns on or off the hint comments to assist users. 
               
               
                 Administration 
                 Document History 
                   
                 This option opens the Document History page. 
               
               
                   
                 Audit Report 
                   
                 This option opens the Audit File Report Selection 
               
               
                   
                   
                   
                 Criteria page. 
               
               
                   
                 User Manager 
                   
                 This option opens the User Manager page. This option 
               
               
                   
                   
                   
                 is only available to users with administrator 
               
               
                   
                   
                   
                 privileges. 
               
               
                   
                 System 
               
               
                   
                 Configuration 
               
               
                   
                   
                 Database Server 
                 This option opens the Database Server Settings page. 
               
               
                   
                   
                 Master Setup 
                 This option opens the System Configuration page. 
               
               
                 Help 
                 Cal Manager Help 
                   
                 This provides help on the calibration manager. 
               
               
                   
                 Web site 
                   
                 This accesses the web site for the suite of applications. 
               
               
                   
                 About 
                   
                 Help About will provide the operator with the program 
               
               
                   
                   
                   
                 name, current version number, and copyright 
               
               
                   
                   
                   
                 information for the application currently installed on 
               
               
                   
                   
                   
                 the user&#39;s system. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 APPENDIX C 
               
               
                   
               
               
                 Command 
                 Parameters 
                 Description 
               
               
                   
               
             
            
               
                 DataSht_StepGroupParent 
                   
                 Indicates that the current data sheet row is a parent 
               
               
                   
                   
                 to child rows that will follow. A Step Group 
               
               
                   
                   
                 Parent row normally contains configuration 
               
               
                   
                   
                 commands, user queries, or instructions that must 
               
               
                   
                   
                 be executed prior to executing subsequent rows. 
               
               
                   
                   
                 Children of the parent will not be allowed to 
               
               
                   
                   
                 execute until the parent is executed. See 
               
               
                   
                   
                 “DataSht_StepGroupChild” 
               
               
                   
                 StepGrpName 
                 The user defined text that represents the name of 
               
               
                   
                   
                 the group. This parameter must exactly match the 
               
               
                   
                   
                 value entered for the other applicable step group 
               
               
                   
                   
                 commands to which it is associated. 
               
               
                 DataSht_StepGroupChild 
                   
                 Indicates that the current data sheet row has a 
               
               
                   
                   
                 subordinate relationship to its Step Group Parent. 
               
               
                   
                   
                 Step Group Children will not be allowed to 
               
               
                   
                   
                 execute until the parent is executed. See 
               
               
                   
                   
                 “DataSht_StepGroupParent”. 
               
               
                   
                 StepGrpName 
                 The user defined text that represents the name of 
               
               
                   
                   
                 the group. This parameter must exactly match the 
               
               
                   
                   
                 value entered for the other applicable step group 
               
               
                   
                   
                 commands to which it is associated. 
               
               
                 DataSht_GoToStepGroupParent 
                   
                 Send the execution pointer to the Parent Step 
               
               
                   
                   
                 indicated by the “StepGrpName” parameter and 
               
               
                   
                   
                 Halt the Automation script. See 
               
               
                   
                   
                 “DataSht_StepGroupParent” 
               
               
                   
                 StepGrpName 
                 The user defined text that represents the name of 
               
               
                   
                   
                 the group. This parameter must exactly match the 
               
               
                   
                   
                 value entered for the other applicable step group 
               
               
                   
                   
                 commands to which it is associated. 
               
               
                 DataSht_NormalizeMeasure 
                   
                 Compensate the measured value for the deviation 
               
               
                   
                   
                 between the reference (true) and test point 
               
               
                   
                   
                 nominal values. (ie. Ohms nominal 190 but 
               
               
                   
                   
                 reference value is 189.99982). The measured 
               
               
                   
                   
                 value will be adjusted by the deviation quantity of 
               
               
                   
                   
                 the nominal vs. reference values. This 
               
               
                   
                   
                 compensation allows the test point to retain the 
               
               
                   
                   
                 fixed nominal value, as opposed to replacing it 
               
               
                   
                   
                 with the reference value. 
               
               
                   
                 NominalValue 
                 The nominal value of the test point. The value 
               
               
                   
                   
                 must be numeric. This parameter should match 
               
               
                   
                   
                 the test point&#39;s nominal value. 
               
               
                   
                 ReferenceVal 
                 The Reference (True) Value of the measurement. 
               
               
                   
                   
                 The value must be numeric. 
               
               
                   
                 MeasuredVal 
                 The Unknown (indicated) Measured Value. The 
               
               
                   
                   
                 value must be numeric. 
               
               
                   
                 ExponentLetter 
                 A single letter that represents the desired 
               
               
                   
                   
                 exponential format of the measurement result. 
               
               
                   
                 NumDecimals 
                 The desired number of decimal places the 
               
               
                   
                   
                 measured value should have. 
               
               
                 DataSht_PassFailToMeasure 
                   
                 Invokes a popup form containing Pass/Fail 
               
               
                   
                   
                 criteria. The popup form has buttons to select 
               
               
                   
                   
                 Pass or Fail. 
               
               
                   
                 Criteria 
                 The name of a calibration instruction to display in 
               
               
                   
                   
                 the popup form 
               
               
                 DataSht_ResetScript 
                 [None] 
                 Resets all running automation status flags, 
               
               
                   
                   
                 including Parent Step flags. The user will be 
               
               
                   
                   
                 required to re-run a parent step prior to any 
               
               
                   
                   
                 children. This is usually invoked at the end of a 
               
               
                   
                   
                 scripted test series. 
               
               
                 DataSht_ShowBoilerPlate 
                 [None] 
                 The master Boilerplate form will be displayed to 
               
               
                   
                   
                 allow the user to interact with it. 
               
               
                 DataSht_ToCalc(H/L) 
                   
                 Calculates the test point tolerance using the 
               
               
                   
                   
                 parametric values indicated above. The formatted 
               
               
                   
                   
                 nominal value, calibration tolerance, and test point 
               
               
                   
                   
                 specifications are then inserted into the data sheet 
               
               
                   
                   
                 row. Once inserted the row behaves as though the 
               
               
                   
                   
                 values were placed there when the data sheet was 
               
               
                   
                   
                 designed. Unused parameter values can be set to 
               
               
                   
                   
                 zero, but none can be left blank. 
               
               
                   
                 NominalVal 
                 The test point&#39;s nominal value 
               
               
                   
                 FullScaleVal 
                 The full scale value of the applicable instrument 
               
               
                   
                   
                 range. 
               
               
                   
                 ErrPctIV 
                 The % I.V. specification 
               
               
                   
                 ErrPctFS 
                 The % F.S. specification, If this parameter is not 
               
               
                   
                   
                 zero, then the FullScaleVal parameter is relevant. 
               
               
                   
                 ErrFloor 
                 The floor error specification relative to the 
               
               
                   
                   
                 NominalVal parameters unit of measure. This 
               
               
                   
                   
                 value will be used in its absolute form (not 
               
               
                   
                   
                 negative). 
               
               
                   
                 NumDecimals 
                 The desired number of decimal places. 
               
               
                   
                 Units 
                 The test point&#39;s unit of measure 
               
               
                 DataSht_VarToMeasure 
                   
                 The contents of a user defined variable are 
               
               
                   
                   
                 formatted and inserted into the data sheet row as 
               
               
                   
                   
                 the measured value. The variable must contain a 
               
               
                   
                   
                 numeric value. 
               
               
                   
                 UserVar 
                 A user defined variable. 
               
               
                   
                 NumDecimals 
                 The desired number of decimal places the 
               
               
                   
                   
                 measured value should have. 
               
               
                   
                 ExponentLetter 
                 A single letter that represents the desired 
               
               
                   
                   
                 exponential format of the measurement result. 
               
               
                 GPIB_ReadToMeasure 
                   
                 The result of a GPIB query is formatted and 
               
               
                   
                   
                 inserted into the data sheet row as the measured 
               
               
                   
                   
                 value. The GPIB query result must be convertible 
               
               
                   
                   
                 to a numeric value. 
               
               
                   
                 Addr 
                 The GPIB address of the instrument to receive the 
               
               
                   
                   
                 communication 
               
               
                   
                 Cmnd 
                 The command to be issued to the instrument. 
               
               
                   
                 BufferSize 
                 The maximum number of characters to receive. 
               
               
                   
                   
                 This number should be set higher than the number 
               
               
                   
                   
                 of characters actually expected. When in doubt, 
               
               
                   
                   
                 set the number to 100. 
               
               
                   
                 CutLeft 
                 The number of characters to remove from the 
               
               
                   
                   
                 response string. An entry of −1 indicates no 
               
               
                   
                   
                 characters are removed. 
               
               
                   
                 KeepLeft 
                 The number of characters to retain after the “cut 
               
               
                   
                   
                 left”. An entry of −1 indicates all characters are 
               
               
                   
                   
                 retained. 
               
               
                   
                 ExponentLetter 
                 A single letter that represents the desired 
               
               
                   
                   
                 exponential format of the measurement result. 
               
               
                 GPIB_ReadToMeasure 
                   
                 The result of a GPIB query is formatted and 
               
               
                   
                   
                 inserted into the data sheet row as the measured 
               
               
                   
                   
                 value. The GPIB query result must be convertible 
               
               
                   
                   
                 to a numeric value. 
               
               
                 GPIB_ReadToVar 
                   
                 The result of a GPIB query is trimmed and 
               
               
                   
                   
                 numerically formatted, then inserted into the user 
               
               
                   
                   
                 defined variable specified by the “UserVar” 
               
               
                   
                   
                 parameter. The GPIB query result must be 
               
               
                   
                   
                 convertible to a numeric value. 
               
               
                   
                 Addr 
                 The GPIB address of the instrument to receive the 
               
               
                   
                   
                 communication 
               
               
                   
                 Cmnd 
                 The command to be issued to the instrument. 
               
               
                   
                 UserVar 
                 A user defined variable. 
               
               
                   
                 BufferSize 
                 The maximum number of characters to receive. 
               
               
                   
                   
                 This number should be set higher than the number 
               
               
                   
                   
                 of characters actually expected. When in doubt, 
               
               
                   
                   
                 set the number to 100. 
               
               
                   
                 CutLeft 
                 The number of characters to remove from the 
               
               
                   
                   
                 response string. An entry of −1 indicates no 
               
               
                   
                   
                 characters are removed. 
               
               
                   
                 KeepLeft 
                 The number of characters to retain after the “cut 
               
               
                   
                   
                 left”. An entry of −1 indicates all characters are 
               
               
                   
                   
                 retained. 
               
               
                   
                 NumDecimals 
                 The desired number of decimal places the 
               
               
                   
                   
                 resulting value should have. 
               
               
                   
                 ExponentLetter 
                 A single letter that represents the desired 
               
               
                   
                   
                 exponential format of the result. 
               
               
                 GPIB_Write 
                   
                 A command is issued to a GPIB device at the 
               
               
                   
                   
                 address specified. 
               
               
                   
                 Addr 
                 The GPIB address of the instrument to receive the 
               
               
                   
                   
                 communication 
               
               
                   
                 Cmnd 
                 The command to be issued to the instrument. 
               
               
                 Math_CalcToVar 
                   
                 The specified equation is executed and the result 
               
               
                   
                   
                 is placed into the indicated user defined variable. 
               
               
                   
                 UserVar 
                 A user defined variable. 
               
               
                   
                 Equation 
                 The equation to be executed. Once the equation is 
               
               
                   
                   
                 executed the result is placed in the User Defined 
               
               
                   
                   
                 Variable specified in the “UserVar” parameter. 
               
               
                   
                   
                 The equation must be capable of being executed 
               
               
                   
                   
                 by the Calibration Process Manager&#39;s internal 
               
               
                   
                   
                 equation system. The best way to ensure this is to 
               
               
                   
                   
                 design and test the equation using The Calibration 
               
               
                   
                   
                 Process Manager&#39;s Equation Editor. The equation 
               
               
                   
                   
                 editor is the command wizard for the 
               
               
                   
                   
                 “Math_CalcToVar” command. 
               
               
                 RS232_Config 
                   
                 This command configures the specified com port 
               
               
                   
                   
                 for RS-232 communications. 
               
               
                   
                 ComPort 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 ComPortID 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 BaudRate 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 Parity 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 DataBits 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 StopBits 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 HwHandshake 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 SwHandshake 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 TxPrefix 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 TxTermination 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 RxTermination 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 RxTimeOut 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                 RS232_ReadToMeasure 
                   
                 The result of an RS-232 query is formatted and 
               
               
                   
                   
                 inserted into the data sheet row as the measured 
               
               
                   
                   
                 value. The RS-232 query result must be 
               
               
                   
                   
                 convertible to a numeric value. 
               
               
                   
                 ComPort 
                 The com port number to be used. The allowable 
               
               
                   
                   
                 range is 1-9. 
               
               
                   
                 Cmnd 
                 The command to be issued to the instrument. 
               
               
                   
                 CutLeft 
                 The number of characters to remove from the 
               
               
                   
                   
                 response string. An entry of −1 indicates no 
               
               
                   
                   
                 characters are removed. 
               
               
                   
                 KeepLeft 
                 The number of characters to retain after the “cut 
               
               
                   
                   
                 left”. An entry of −1 indicates all characters are 
               
               
                   
                   
                 retained. 
               
               
                   
                 NumDecimals 
                 The desired number of decimal places the 
               
               
                   
                   
                 measured value should have. 
               
               
                   
                 ExponentLetter 
                 A single letter that represents the desired 
               
               
                   
                   
                 exponential format of the measurement result. 
               
               
                 RS232_ReadToVar 
                   
                 The result of an RS-232 query is trimmed and 
               
               
                   
                   
                 numerically formatted then inserted into the user 
               
               
                   
                   
                 defined variable specified by the “UserVar” 
               
               
                   
                   
                 parameter. The RS-232 query result must be 
               
               
                   
                   
                 convertible to a numeric value. 
               
               
                   
                 ComPort 
                 The com port number to be used. The allowable 
               
               
                   
                   
                 range is 1-9. 
               
               
                   
                 Cmnd 
                 The command to be issued to the instrument. 
               
               
                   
                 UserVar 
                 A user defined variable. 
               
               
                   
                 CutLeft 
                 The number of characters to remove from the 
               
               
                   
                   
                 response string. An entry of −1 indicates no 
               
               
                   
                   
                 characters are removed. 
               
               
                   
                 KeepLeft 
                 The number of characters to retain after the “cut 
               
               
                   
                   
                 left”. An entry of −1 indicates all characters are 
               
               
                   
                   
                 retained. 
               
               
                   
                 NumDecimals 
                 The desired number of decimal places the 
               
               
                   
                   
                 resulting value should have. 
               
               
                   
                 ExponentLetter 
                 A single letter that represents the desired 
               
               
                   
                   
                 exponential format of the result. 
               
               
                 RS232_Write 
                   
                 A command is issued to an RS-232 device at the 
               
               
                   
                   
                 address specified. 
               
               
                   
                 ComPort 
                 The com port number to be used. The allowable 
               
               
                   
                   
                 range is 1-9. 
               
               
                   
                 Cmnd 
                 The command to be issued to the instrument. 
               
               
                 Sys_AssignToVariable 
                   
                 Copies the Contents of one user defined variable 
               
               
                   
                   
                 into another. 
               
               
                   
                 UserVar 
                 A user defined variable. 
               
               
                   
                 NewValue 
                 The value to assign to the user defined variable. 
               
               
                 Sys_AutoExecNextRow 
                 [None] 
                 Instructs the script engine to proceed to the next 
               
               
                   
                   
                 data sheet row and continue script execution. 
               
               
                 Sys_Instruction 
                   
                 Displays a pop-up form containing 
               
               
                   
                   
                 instructions/notes/warnings for the user to read.. 
               
               
                   
                 Name 
                 The name of the Calibration Instruction to be 
               
               
                   
                   
                 displayed. 
               
               
                 Sys_ImageInstruct 
                   
                 Displays a pop-up form containing 
               
               
                   
                   
                 instructions/notes/warnings for the user to read. 
               
               
                   
                   
                 and images for the user to view. 
               
               
                   
                 InstName 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 ImageNames 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 FigNums 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                 Sys_QueryUser 
                   
                 Displays a user query prompt for the user to enter 
               
               
                   
                   
                 values. The values are then assigned to user 
               
               
                   
                   
                 defined variables. 
               
               
                   
                 Title 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 Captions 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 Variables 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 DefValues 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                   
                 Tails 
                 This parameter can only be set using the command 
               
               
                   
                   
                 wizard. 
               
               
                 Sys_Wait 
                   
                 Causes the scripting system to pause for a fixed 
               
               
                   
                   
                 amount of time. 
               
               
                   
                 Seconds 
                 The number of seconds to wait. This value must 
               
               
                   
                   
                 be numeric. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 APPENDIX D 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Primary Objects 
                 Panel 
                 This panel contains the primary interface objects for the data sheet 
               
               
                   
                   
                 and is located just above the data sheet grid. 
               
               
                 Edit Cell Field 
                 A/N 
                 Text box 
               
               
                 . . . 
                 Button 
                 This button is only visible when a Nominal Value or Tolerance cell is 
               
               
                 (Nominal Value/ 
                   
                 selected in the data sheet grid. Selecting the button causes the 
               
               
                 Tolerance Cell 
                   
                 applicable wizard to be displayed. The purpose of these wizards is to 
               
               
                 wizard) 
                   
                 allow some content of the cell to be edited without invalidating 
               
               
                   
                   
                 calculable aspects of the cell&#39;s content. (The Standard version does 
               
               
                   
                   
                 not have this feature) 
               
               
                 View/Edit Master 
                 Button 
                 Opens the Master Boilerplate Setup page. 
               
               
                 Boilerplate 
               
               
                 Save Files 
                 Button 
                 Saves the current file. 
               
               
                 Print Data Sheets 
                 Button 
                 Opens the Data Sheet Print Shop page. 
               
               
                 Data Sheet Grid 
                 Grid 
                 This grid is a representation of the printed data sheet rows. The data 
               
               
                   
                   
                 sheet grid is used as the primary method of navigation within the 
               
               
                   
                   
                 Calibration Process Manager. The currently selected data sheet row is 
               
               
                   
                   
                 the target of all data sheet operations. 
               
               
                 Step number 
                 Column 
                 The applicable step number of a calibration procedure document. 
               
               
                 Function Tested 
                 Column 
                 Description of the test. 
               
               
                 Nominal Value 
                 Column 
                 The “nominal” value of the test point. These cells also have three 
               
               
                   
                   
                 indicators to represent the status of the Specification Tracking 
               
               
                   
                   
                 Interface and instrumentation. The cells also display a □ symbol 
               
               
                   
                   
                 when a test point&#39;s measurement uncertainty model has been modified 
               
               
                   
                   
                 by a user. 
               
               
                 As Found 
                 Column 
                 The “as found” value for the unit under test. 
               
               
                 As Left 
                 Column 
                 The “as left” value for the unit under test. 
               
               
                 Out of Tolerance 
                 Column 
                 “X” indicates “out of tolerance.” 
               
               
                 Calibration 
                 Column 
                 The measurement “tolerance.” 
               
               
                 Tolerance 
               
               
                 Specifications 
                 Column 
                 Displays the specifications entered into the tolerance calculator when 
               
               
                   
                   
                 the tolerance calculator is used to create the test point. 
               
               
                 Lock Status 
                 Drop 
                 This option is used to lock or unlock the data sheet for editing. 
               
               
                   
                 Down List 
               
               
                 View Only 
                 Selection 
                 This selection only allows users to view data sheets. No edits are 
               
               
                   
                   
                 allowed. 
               
               
                 Edit mode 
                 Selection 
                 This selection allows the user access to all edit tools within the page. 
               
               
                 Cell Clipboard 
                 panel 
                 This clipboard operates on a single data sheet cell and is independent 
               
               
                   
                   
                 of the MS Windows clipboard. 
               
               
                 View (Cell) 
                 Button 
                 View “Cell” clipboard. 
               
               
                 Copy (Cell) 
                 Button 
                 Copies currently selected cell to the “cell” clipboard. 
               
               
                 Paste (Cell) 
                 Button 
                 Pastes information from the “cell” clipboard to the currently selected 
               
               
                   
                   
                 cell. 
               
               
                 Clear (Cell) 
                 Button 
                 Clears the currently selected “cell” of information. 
               
               
                 Line Clipboard 
                 panel 
                 This clipboard operates on one or more selected data sheet rows 
               
               
                   
                   
                 (lines) and is independent of the MS Windows clipboard. 
               
               
                 View (Line) 
                 Button 
                 View “Line” clipboard. 
               
               
                 Copy (Line) 
                 Button 
                 Copies currently selected line/s to the “line” clipboard. 
               
               
                   
                   
                 Note: This clipboard is independent of the MS Windows clipboard. 
               
               
                 Paste (Line) 
                 Button 
                 Pastes information from the “line” clipboard before the currently 
               
               
                   
                   
                 selected line. 
               
               
                 Clear (Line) 
                 Button 
                 Clears the currently selected “line”/s of information. 
               
               
                 Insert (Line) 
                 Button 
                 Inserts a new “line” before the currently selected line of the data 
               
               
                   
                   
                 sheet. 
               
               
                 Delete (Line) 
                 Button 
                 Deletes the currently selected “line”/s. 
               
               
                 Add (5) (Line) 
                 Button 
                 Inserts five new “lines” at the bottom of the data sheet. 
               
               
                 Add Page (Line) 
                 Button 
                 Inserts a new page of “lines” at the bottom of the data sheet. 
               
               
                 Group Clipboard 
                 panel 
                 This clipboard operates on one or mores selected data sheet cells 
               
               
                   
                   
                 (group) and is independent of the MS Windows clipboard. 
               
               
                 View (Group) 
                 Button 
                 View “group” clipboard. 
               
               
                 Copy (Group) 
                 Button 
                 Copies the “group” of the currently selected cells to the clipboard. 
               
               
                   
                   
                 Note: This clipboard is independent of the MS Windows clipboard. 
               
               
                 Paste (Group) 
                 Button 
                 Pastes information from the clipboard to the currently selected cells. 
               
               
                   
                   
                 Note: The selected cells area for pasting must be a single or multiple, 
               
               
                   
                   
                 two dimensional, match of the cells area selected when the group of 
               
               
                   
                   
                 cells was copied. 
               
               
                 Clear (Group) 
                 Button 
                 Clears the contents of the currently selected “group” of cells. 
               
               
                 Tools 
                 panel 
                 This panel contains buttons that invoke editor tool modules. 
               
               
                 Tolerance Calculator 
                 Button 
                 Opens the Tolerance Calculator page at the bottom of the screen. The 
               
               
                   
                   
                 application resizes the main screen by approximately ⅓. 
               
               
                 Test Point 
                 Button 
                 Opens the Measurement Uncertainty Wizard page. 
               
               
                 Measurement 
               
               
                 Uncertainty Wizard 
               
               
                 Specification 
                 Button 
                 Opens the Configure Instrumentation with Specification Tracking 
               
               
                 Tracking 
                   
                 Data page at the bottom of the screen. The application resizes the 
               
               
                 Instruments 
                   
                 main screen by approximately ⅓. 
               
               
                 Command Count 
                 Label 
                 Displays the number of scripted commands contained within the 
               
               
                 Label 
                   
                 selected data sheet row. 
               
               
                 Warnings 
                 panel 
                 This panel contains a button that resets editor warnings. 
               
               
                 Reset All 
                 Button 
                 Resets all warnings to default. When editing the Nominal Value or 
               
               
                   
                   
                 Tolerance cells, a warning is display to indicate the ramifications of 
               
               
                   
                   
                 manually editing these cells. The user can choose to ignore the 
               
               
                   
                   
                 warning, have the system stop issuing the warning, or cancel the 
               
               
                   
                   
                 operation. If one or more of these warning was directed to stop 
               
               
                   
                   
                 issuing the warning, this option directs the system to re-assert the 
               
               
                   
                   
                 issuance of all warnings. 
               
               
                 Special Ops 
                 panel 
                 This panel contains buttons that invoke special editor operations. 
               
               
                 “Same” 
                 Button 
                 Copies the word “Same” into all “As Left” data cells that presently 
               
               
                   
                   
                 have no content. 
               
               
                 Insert Header 
                 Button 
                 Inserts a header line into the Data Sheet before the currently selected 
               
               
                   
                   
                 cell. The header line is a duplicate of the first row of the data sheet. 
               
               
                 Add Page Break 
                 Button 
                 Inserts a page break into the Data Sheet. A page break is a 
               
               
                   
                   
                 mechanism that directs the system to restart row numbering on the 
               
               
                   
                   
                 next line of the data sheet. When the data sheet is printed, the first 
               
               
                   
                   
                 row following a page break is the first row of a new page. 
               
               
                 Copy symbol to 
                 panel 
                 This panel contains buttons that copy a symbol onto the Ms Windows 
               
               
                 clipboard 
                   
                 clipboard. 
               
               
                 ± 
                 Button 
                 Copies the ± symbol to the clipboard. 
               
               
                 ° 
                 Button 
                 Copies the ° symbol to the clipboard. 
               
               
                 μ 
                 Button 
                 Copies the μ symbol to the clipboard. 
               
               
                 
                   1 
                 
                 Button 
                 Copies the  1  symbol to the clipboard. 
               
               
                 
                   2 
                 
                 Button 
                 Copies the  2  symbol to the clipboard. 
               
               
                 
                   3 
                 
                 Button 
                 Copies the  3  symbol to the clipboard. 
               
               
                 Units 
                 Button 
                 Opens the Units of Measure Selection form. 
               
               
                 Row Level Remark 
                 panel 
                 This panel contains a drop down list that manages row level remarks. 
               
               
                 Flags 
               
               
                 Level Remark 
                 Drop 
                 This inserts a notation into the Tolerance column of the selected row. 
               
               
                   
                 Down Box 
                 The notations listed in the drop down box are: [None], *, **, ***, 
               
               
                   
                   
                 ****, *****, *{circumflex over ( )}, *{circumflex over ( )}{circumflex over ( )}, and *{circumflex over ( )}{circumflex over ( )}{circumflex over ( )}. These symbols direct the data sheet 
               
               
                   
                   
                 printing system to print the remark, associated with the selected 
               
               
                   
                   
                 symbol, in the remarks section of the data sheet page that used the 
               
               
                   
                   
                 symbol. The symbols are automatically added to the end of the 
               
               
                   
                   
                 Tolerance cell&#39;s content. When a data sheet row is selected, the drop 
               
               
                   
                   
                 down box indicates the current symbol assignment. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 1 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts all changes and closes the page. 
               
               
                 Cancel 
                 Button 
                 Cancels all changes and closes the page. 
               
               
                 Server Name 
                 Field 
                 The name of the computer running the SQL 
               
               
                   
                   
                 Server. 
               
               
                 Database name 
                 Field 
                 This is the name of the Database on the 
               
               
                   
                   
                 Server computer. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 2 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts all changes and closes the page. 
               
               
                 Cancel 
                 Button 
                 Cancels all changes and closes the page. 
               
               
                 Configuration 
                 Drop 
                 This drop down box contains the following selection 
               
               
                 Group 
                 Down 
                 options: 
               
               
                   
                 Selection 
               
               
                 All 
                 Selection 
                 This selection contains all of the configuration options listed 
               
               
                   
                   
                 in the specific sections below. 
               
               
                 General 
                 Selection 
                 This selection contains following general information 
               
               
                   
                   
                 options: 
               
               
                   
                   
                 Company Name 
               
               
                 Files 
                 Selection 
                 This selection contains the following options for default file 
               
               
                   
                   
                 search path locations: 
               
               
                   
                   
                 Master Data Sheet Path 
               
               
                   
                   
                 Master Calibration File Path 
               
               
                 Field Labels 
                 Selection 
                 This selection contains the following fields that the system 
               
               
                   
                   
                 administrator can change the display label for: 
               
               
                   
                   
                 Asset Number 
               
               
                   
                   
                 Test Number 
               
               
                   
                   
                 Cal Procedure 
               
               
                 Hints 
                 Selection 
                 This selection contains the following fields that the system 
               
               
                   
                   
                 administrator can change the “fly over” hint label for: 
               
               
                   
                   
                 Set As Left Data to “Same” 
               
               
                   
                   
                 View Boilerplate 
               
               
                 Pressure 
                 Selection 
                 This selection contains the following fields that the system 
               
               
                 Wizard 
                   
                 administrator can use to configure the Pressure Gauge Data 
               
               
                   
                   
                 Sheet Generator wizard&#39;s maximum vacuum values: 
               
               
                   
                   
                 Psia in Hg 
               
               
                 Datasheet 
                 Selection 
                 This selection option contains the following data sheet 
               
               
                   
                   
                 configuration options: 
               
               
                   
                   
                 Tolerance: Include TAR 
               
               
                   
                   
                 Tolerance: Include Uncertainty 
               
               
                   
                   
                 Datasheet Title 
               
               
                   
                   
                 Custom Datasheet title 
               
               
                   
                   
                 Document Page Nums 
               
               
                   
                   
                 Datasheet Page Nums 
               
               
                   
                   
                 Datasheet.Use Attachment 
               
               
                   
                   
                 Def Headers: Step Num 
               
               
                   
                   
                 Def Headers: Func.Tested 
               
               
                   
                   
                 Def Headers: Nominal Val 
               
               
                   
                   
                 Def Headers: As Found 
               
               
                   
                   
                 Def Headers: As Left 
               
               
                   
                   
                 Def Headers: Out of Tol 
               
               
                   
                   
                 Def Headers: Tolerance 
               
               
                 Configurations 
                 Grid 
                 Lists all of the available configuration options. Filtered by 
               
               
                 List 
                   
                 the Configuration Group drop down list selection. 
               
               
                 . . . 
                 Button 
                 This button is located in the Value cell of the Configurations 
               
               
                 (Set Directory) 
                   
                 List grid when a file path type of configuration option is 
               
               
                   
                   
                 visible in the row with the button. This button will open the 
               
               
                   
                   
                 standard Windows open file window. 
               
               
                 Set Value 
                 Button 
                 Sets the value in the change value field in the permissions 
               
               
                   
                   
                 group section. 
               
               
                 Default Item 
                 Button 
                 Sets the selected permission to the default value. 
               
               
                 Default All 
                 Buttons 
                 Sets all permissions to the default value. Filtered by the 
               
               
                   
                   
                 Configuration Group drop down list selection. 
               
               
                 Change Value 
                 Field 
                 This field is used to input the values for the Configurations. 
               
               
                   
                   
                 The value ranges are listed in the configuration List grid in 
               
               
                   
                   
                 the Range column. 
               
               
                 Information 
                 Field 
                 This field will display a short explanation of the 
               
               
                   
                   
                 configuration item selected in the Configurations List grid. 
               
               
                   
               
            
           
         
       
     
     APPENDIX E 
       
     
       
         
           
               
               
               
             
               
                 TABLE 3 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 OK 
                 Button 
                 Attempts the login. 
               
               
                 Cancel 
                 Button 
                 Cancels the login attempt and closes the page. 
               
               
                 Login ID 
                 Field 
                 Sets the login ID for the login attempt. 
               
               
                 Password 
                 Field 
                 Sets the password for the login attempt. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 4 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Exit 
                 Button 
                 Closes the User Manager Page. 
               
               
                 User List 
                 Grid 
                 Displays the following fields: 
               
               
                 Name 
                 Field 
                 Displays full name of the user. 
               
               
                 Login ID 
                 Field 
                 Displays the login ID of the user. 
               
               
                 Can Login 
                 Field 
                 Indicates if the user&#39;s account has been disabled. 
               
               
                 Sys Admin 
                 Field 
                 Indicates if the user has System Administrator privileges. 
               
               
                 Custom ID 
                 Field 
                 Displays the custom ID of the user, if there is one. This field 
               
               
                   
                   
                 is to allow the system administrator to enter an employee ID 
               
               
                   
                   
                 from another application. The purpose is to permit “joins”, 
               
               
                   
                   
                 on employee ID, between the application&#39;s data. 
               
               
                 New user 
                 Button 
                 Opens the Create a New User page 
               
               
                 Properties 
                 Button 
                 Opens the Edit User Login page 
               
               
                 Permissions 
                 Button 
                 Opens the User Permissions page 
               
               
                 Change 
                 Button 
                 Opens the Change Your Password page 
               
               
                 Password 
               
               
                 Set Default 
                 Button 
                 Opens the User Login Rules page. 
               
               
                 Login Rules 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 5 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Password 
                 Check 
                 This check box will set the password to automatically 
               
               
                 Expires 
                 Box 
                 expire, and force the user to create a new password. 
               
               
                 User Can Not 
                 Check 
                 This check box will set the application to not allow 
               
               
                 Reuse Password 
                 Box 
                 passwords to be reused. 
               
               
                 Logout After 
                 Check 
                 This check box sets the application to automatically logoff 
               
               
                 Idle Time Period 
                 Box 
                 the user after a period of idle time. A user will be required 
               
               
                   
                   
                 log back in or terminate the application. 
               
               
                 Terminate 
                 Check 
                 This check box sets the application to terminate after a set 
               
               
                 Application 
                 Box 
                 number of failed login attempts. 
               
               
                 Suspend 
                 Check 
                 This check box sets the application to suspend the account 
               
               
                 Account on 
                 Box 
                 of the user who has surpassed the number of failed login 
               
               
                 Termination 
                   
                 attempts. 
               
               
                 Log Login 
                 Check 
                 This check box sets the application to track logins via the 
               
               
                   
                 Box 
                 audit report. 
               
               
                 Log Logout 
                 Check 
                 This check box sets the application to track logouts via the 
               
               
                   
                 Box 
                 audit report. 
               
               
                 Log Each Failed 
                 Check 
                 This check box sets the application to track failed login 
               
               
                 Attempt 
                 Box 
                 attempts via the audit report. 
               
               
                 Log Password 
                 Check 
                 This check box sets the application to track password 
               
               
                 Change 
                 Box 
                 changes via the audit report. 
               
               
                 After “x” Days 
                 Numeric 
                 Sets the number of days prior to requiring the user to change 
               
               
                   
                 Field 
                 their password. 
               
               
                 Max Idle “x” 
                 Numeric 
                 Sets the number of minutes until the system auto logs off 
               
               
                 Min 
                 Field 
                 the user. 
               
               
                 After “x” Login 
                 Numeric 
                 Sets the number of failed logon attempts prior to 
               
               
                 Attempts 
                 Field 
                 terminating the application. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 6 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts all changes and closes the page. 
               
               
                 Cancel 
                 Button 
                 Cancels all changes and closes the page. 
               
               
                 Employee ID 
                 Field 
                 Displays the unique ID number the system gives each user. 
               
               
                   
                   
                 This number is viewable only in the User properties page. 
               
               
                   
                   
                 The field is not user editable. 
               
               
                 Login ID 
                 Field 
                 The login name the user is to be assigned. 
               
               
                 Custom User ID 
                 Field 
                 This field is to allow the system administrator to enter an 
               
               
                   
                   
                 employee ID from another application. The purpose is to 
               
               
                   
                   
                 permit “joins”, on employee ID, between the applications&#39; 
               
               
                   
                   
                 data. 
               
               
                 First Name 
                 Field 
                 The first name of the user. 
               
               
                 Middle Initial 
                 Field 
                 The middle initial of the user. 
               
               
                 Last Name 
                 Field 
                 The last name of the user. 
               
               
                 Composite 
                 Field 
                 This will display the user&#39;s full name: “Last Name, First 
               
               
                 Name 
                   
                 Name MI”. This field is not user editable. 
               
               
                 Password 
                 Field 
                 This field will display the entries as *. The password will 
               
               
                   
                   
                 have to be re-entered. 
               
               
                 Re-enter 
                 Field 
                 This field also displays entries a *. If the two password 
               
               
                 Password 
                   
                 entries do not match, an error message will be displayed. 
               
               
                   
                   
                 The user cannot be saved until the passwords match. 
               
               
                 System 
                 Check 
                 This check box will grant system administrator access. If 
               
               
                 Administrator 
                 Box 
                 the check box is checked, the user will have system admin 
               
               
                   
                   
                 access. 
               
               
                 Account 
                 Check 
                 This check box will disable the users account. If the box is 
               
               
                 Disabled 
                 Box 
                 checked the user will not be able to log into the application. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 7 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts all changes and closes the page. 
               
               
                 Cancel 
                 Button 
                 Cancels all changes and closes the page. 
               
               
                 Permissions 
                 Drop Down 
                 This drop down box contains the following selection options: 
               
               
                 Group 
                 Selection 
               
               
                 All 
                 Selection 
                 This selection option displays all of the permission 
               
               
                   
                   
                 described in the sections below: 
               
               
                 Calibrations 
                 Selection 
                 This selection option contains the following options: 
               
               
                   
                   
                 Can Sign 
               
               
                 Data Sheet 
                 Selection 
                 This selection option contains the following options: 
               
               
                   
                   
                 Can Approve 
               
               
                 DS Design 
                 Selection 
                 This selection option contains the following options: 
               
               
                   
                   
                 Can Edit 
               
               
                 Engineering 
                 Selection 
                 This selection option contains the following options: 
               
               
                   
                   
                 Edit MU Wizard 
               
               
                 Change 
                 Field 
                 This field is used to input the values for the Permissions. 
               
               
                 Value 
                   
                 The value ranges are listed in the Permissions Group 
               
               
                   
                   
                 section under the Range column. 
               
               
                 Set Value 
                 Button 
                 Sets the value in the change value field to the value field 
               
               
                   
                   
                 in the permissions group section. 
               
               
                 Default Item 
                 Button 
                 Sets the selected permission to the default value. 
               
               
                 Default All 
                 Buttons 
                 Sets all permissions to the default value. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 8 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                   
               
            
           
           
               
               
               
            
               
                 Date Range 
                 Drop Down Box 
                 This has the following information: 
               
            
           
           
               
               
               
            
               
                   
                 This Month 
                 Last Month 
               
               
                   
                 This Week 
                 Last Week 
               
               
                   
                 This Quarter 
                 Last Quarter 
               
               
                   
                 This Year 
                 Last Year 
               
               
                   
                 Today 
                 Yesterday 
               
            
           
           
               
               
               
            
               
                 From 
                 Drop Down Box 
                 This drop down box opens the windows calendar box. 
               
               
                 To 
                 Drop Down Box 
                 This drop down box opens the windows calendar box. 
               
               
                 Sort 1 
                 Drop Down Box 
                 The Sort boxes designate the sorting order of the 
               
               
                   
                   
                 report. The Sort 1 drop down box contains the 
               
               
                   
                   
                 following options: 
               
            
           
           
               
               
               
            
               
                   
                 [None] 
                 File Name 
               
               
                   
                 Login ID 
                 Application Name 
               
               
                   
                 Trans Type 
               
            
           
           
               
               
               
            
               
                 Sort 2 
                 Drop Down Box 
                 The Sort boxes designate the sorting order of the 
               
               
                   
                   
                 report. The Sort 2 drop down box contains the 
               
               
                   
                   
                 following options: 
               
            
           
           
               
               
               
            
               
                   
                 [None] 
                 File Name 
               
               
                   
                 Login ID 
                 Application Name 
               
               
                   
                 Trans Type 
               
            
           
           
               
               
               
            
               
                 Sort 3 
                 Drop Down Box 
                 The Sort boxes designate the sorting order of the 
               
               
                   
                   
                 report. The Sort 3 drop down box contains the 
               
               
                   
                   
                 following options: 
               
            
           
           
               
               
               
            
               
                   
                 [None] 
                 File Name 
               
               
                   
                 Login ID 
                 Application Name 
               
               
                   
                 Trans Type 
               
            
           
           
               
               
               
            
               
                 Sort 4 
                 Drop Down Box 
                 The Sort boxes designate the sorting order of the 
               
               
                   
                   
                 report. The Sort 4 drop down box contains the 
               
               
                   
                   
                 following options: 
               
            
           
           
               
               
               
            
               
                   
                 [None] 
                 File Name 
               
               
                   
                 Login ID 
                 Application Name 
               
               
                   
                 Trans Type 
               
            
           
           
               
               
               
            
               
                 Show 
                 Check Box 
                 This option allows the user to preview the report prior 
               
               
                 Preview 
                   
                 to printing. 
               
               
                 Print 
                 Check Box 
                 This option allows the user to print the transaction 
               
               
                 Transaction 
                   
                 ID&#39;s on the reports. 
               
               
                 ID 
               
               
                 Selection 
                 Drop Down Box 
                 This drop down box contains the interface features in 
               
               
                 Fields 
                   
                 Table. 
               
            
           
           
               
               
               
            
               
                   
                 Application 
                 File Name 
               
               
                   
                 Login ID 
                 Trans Type 
               
            
           
           
               
               
               
            
               
                 Application 
                 Selection 
                 This selection option contains a dynamically 
               
               
                   
                   
                 generated list of applications, for filtering the reports. 
               
               
                 Login IDs 
                 Selection 
                 This selection option a dynamically generated list of 
               
               
                   
                   
                 login IDs, for filtering the reports. 
               
               
                 Files 
                 Selection 
                 This selection option contains a dynamically 
               
               
                   
                   
                 generated list of files, for filtering the reports. 
               
               
                 Trans Type 
                 Selection 
                 This selection option contains a dynamically 
               
               
                   
                   
                 generated list of transaction types, for filtering the 
               
               
                   
                   
                 reports. 
               
               
                 Clear 
                 Button 
                 This button clears all filter selections. 
               
               
                 Selections 
               
               
                 Retrieve 
                 Button 
                 This button allows the user to load pre-saved settings. 
               
               
                 Settings 
               
               
                 Save 
                 Button 
                 This button allows the user to save setting to be 
               
               
                 Settings 
                   
                 loaded at a later date. 
               
               
                 Print 
                 Button 
                 This button prints the report or displays the print 
               
               
                   
                   
                 preview. 
               
               
                 Exit 
                 Button 
                 This button clears all changes and closes the page. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 9 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Copy “A/L” 
                 Button 
                 Copies the “As found” data to the “As left” field for all rows 
               
               
                 Fields 
                   
                 of all channels. 
               
               
                 Print Cal 
                 Button 
                 Opens the Data Sheet Print Shop page to print all channels. 
               
               
                 Files 
               
               
                 Save Cal 
                 Button 
                 Saves calibration data for all active channels 
               
               
                 Files 
               
               
                 Set UUT 
                 Button 
                 Opens the UUT Boilerplate Setup page. 
               
               
                 Boilerplate 
               
               
                 As Found 
                 Radio 
                 Sets the system to the collect “As found” data navigation 
               
               
                   
                 Button 
                 mode. This option sets up navigation and data collection 
               
               
                   
                   
                 behaviors in the Calibration Process Manger that will 
               
               
                   
                   
                 automatically direct measurement data to the “As found” field 
               
               
                   
                   
                 of a data sheet row. This automatic navigation removes a lot 
               
               
                   
                   
                 of the tedium from data collection and automation scripting. 
               
               
                 As Left 
                 Radio 
                 Sets the system to the collect “As left” data navigation mode. 
               
               
                   
                 Button 
                 This option sets up navigation and data collection behaviors in 
               
               
                   
                   
                 the Calibration Process Manger that will automatically direct 
               
               
                   
                   
                 measurement data to the “As left” field of a data sheet row. 
               
               
                   
                   
                 This automatic navigation removes a lot of the tedium from 
               
               
                   
                   
                 data collection and automation scripting. 
               
               
                 Active 
                 Check 
                 Opens the Channel display window. 
               
               
                   
                 Box 
               
               
                 Show 
                 Check 
                 Displays the tolerance values in the data sheet tolerance 
               
               
                 Tolerance 
                 Box 
                 column. 
               
               
                 Show % of 
                 Check 
                 Displays the percentage of tolerance consumed value in the 
               
               
                 Tolerance 
                 Box 
                 data sheet tolerance column. 
               
               
                 Timer 
                 Indicator 
                 The timer displays a count down for a wait period specified by 
               
               
                   
                   
                 a script command. 
               
               
                 OLE 
                 Indicator 
                 Indicates when an OLE communication is occurring. 
               
               
                 GPIB 
                 Indicator 
                 Indicates when a GPIB communication is occurring. 
               
               
                 RS-232 
                 Indicator 
                 Indicates when a RS-232 communication is occurring. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 10 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Data 
                 Selection 
                 This displays the main screen for calibration 
               
               
                 Collection 
                   
                 data collection. This is the primary mode used in 
               
               
                   
                   
                 the execution of calibrations and development 
               
               
                   
                   
                 of automation scripts for data sheets. 
               
               
                 Data Sheet 
                 Selection 
                 This mode is used to create, edit, and manage 
               
               
                 Designer 
                   
                 data sheets. It is also used to view and export 
               
               
                   
                   
                 historical calibration records 
               
               
                 Statistics 
                 Selection 
                 This opens the Statistics page. This mode is used 
               
               
                   
                   
                 to view and analyze historical data collected 
               
               
                   
                   
                 with the Calibration Process Manager. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 11 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Active 
                 Check 
                 Activates or deactivates the channel. If selected on an 
               
               
                   
                 Box 
                 empty channel the UUT Boilerplate Setup page opens. 
               
               
                 Unit Text Field 
                 A/N 
                 Identifies the Unit being tested on that particular channel. 
               
               
                 As Found Text 
                 A/N 
                 Displays the values found during testing. 
               
               
                 Field 
               
               
                 As Left Text Field 
                 A/N 
                 Displays the values left after testing. 
               
               
                 Set UUT 
                 Button 
                 Opens the UUT Boilerplate Setup page. 
               
               
                 Boilerplate 
               
               
                 Copy “A/L” Fields 
                 Button 
                 Copies the “As found” vales to the “As left” field. 
               
               
                 Print Cal Files 
                 Button 
                 Opens the Data sheet Print Shop page. 
               
               
                 Save Cal Files 
                 Button 
                 Saves the calibration. 
               
               
                 Release the Channel 
                 Button 
                 This clears all information within the channel and closes 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 12 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 This button accepts the current criteria, and returns to the main 
               
               
                   
                   
                 page of the Calibration Process Manager. 
               
               
                 Cancel 
                 Button 
                 This option cancels any changes made, or aborts opening of a 
               
               
                   
                   
                 new channel. Returns to the main page of the Calibration 
               
               
                   
                   
                 Process Manager. 
               
               
                 Doc History 
                 Button 
                 This selection accesses the Document History page. (Button 
               
               
                   
                   
                 not present in Pro Remote or Standard versions.) 
               
               
                 New Cal 
                 Button 
                 This clears the calibration information for the UUT. 
               
               
                 Release 
                 Button 
                 This clears all information within the channel and closes it. 
               
               
                 Channel 
               
               
                 Reset 
                 Button 
                 Resets Boilerplate information to the Data Sheet default. 
               
               
                 Boilerplate 
               
               
                 Open Cal 
                 Button 
                 Opens the Calibration History Catalog page to load a 
               
               
                   
                   
                 previously saved calibration. Optionally, displays the standard 
               
               
                   
                   
                 MS Windows file open dialog to load a previously saved 
               
               
                   
                   
                 calibration 
               
               
                 Save Cal 
                 Button 
                 This option opens the Sign Calibration Password window to 
               
               
                   
                   
                 save the calibration to the system database. Optionally, this 
               
               
                   
                   
                 option opens the discipline directory selection page to save the 
               
               
                   
                   
                 calibration as a file 
               
               
                 Import Cal 
                 Button 
                 This option opens the Search page. 
               
               
                 Export Cal 
                 Button 
                 This option allows the user to export the current calibration to a 
               
               
                   
                   
                 file. 
               
               
                 Find Other 
                 Button 
                 Opens the Calibration History Catalog page to view a listing of 
               
               
                 Test Reports 
                   
                 previously saved calibrations with the same Test Number. 
               
               
                   
                   
                 Optionally, opens the Calibration file search page to view a 
               
               
                   
                   
                 listing of previously saved calibrations with the same Test 
               
               
                   
                   
                 Number. 
               
               
                 Asset No. 
                 Field 
                 This field is alphanumeric and allows the user to set a unique 
               
               
                   
                   
                 identifier for the UUT. 
               
               
                 Test No. 
                 Field 
                 This field is numeric and allows the user to set the test number 
               
               
                   
                   
                 for the UUT. 
               
               
                 Model: 
                 Field 
                 This field is alphanumeric and allows the user to set a unique 
               
               
                   
                   
                 model identifier for the UUT. 
               
               
                 Manufacturer 
                 Field 
                 This field is alphanumeric and allows the user to set a unique 
               
               
                   
                   
                 manufacturer identifier for the UUT. 
               
               
                 Description 
                 Field 
                 This field is alphanumeric and allows the user to set a unique 
               
               
                   
                   
                 description for the UUT. 
               
               
                 Part 
                 Field 
                 This field is numeric. The user can use this field to denote 
               
               
                   
                   
                 which part of a multipart test report the channel is to be used 
               
               
                   
                   
                 for. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 13 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Date 
                 Group 
                 This group consists of the following items: 
               
               
                 Range 
               
               
                 Start Date 
                 Field 
                 This field allows the user to input the starting date 
               
               
                   
                   
                 for test point history retrieval. 
               
               
                 End Date 
                 Field 
                 This field allows the user to input the ending date 
               
               
                   
                   
                 for test point history retrieval. 
               
               
                 Test Point 
                 Group 
                 This group contains the following items: 
               
               
                 History 
               
               
                 Load 
                 Button 
                 Loads the test point history for the asset or 
               
               
                   
                   
                 data sheet. 
               
               
                 Analyze 
                 Button 
                 This button opens the Drift Analyzer Interface page. 
               
               
                 Drift 
               
               
                 Do Report 
                 Button 
                 Creates a report showing a selected test point&#39;s 
               
               
                   
                   
                 Specification Tracking Instrument range usage as a 
               
               
                   
                   
                 “standard” in other calibrations. The report&#39;s 
               
               
                   
                   
                 primary purpose is to assist the “standards failure 
               
               
                   
                   
                 analysis” process. 
               
               
                 This Asset 
                 Radial 
                 This option will allow the application to load 
               
               
                   
                 Button 
                 only the test point history for the selected asset. 
               
               
                 This Data 
                 Radial 
                 This option will allow the application to load the 
               
               
                 Sheet 
                 Button 
                 test point history for all calibrations that utilized the 
               
               
                   
                   
                 data sheet. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 14 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Go 
                 Button 
                 Accepts parameters, and creates file. 
               
               
                 Cancel 
                 Button 
                 Aborts the file creation, and closes page. 
               
               
                 File Path 
                 Field 
                 Shows the file path to the directory, where the 
               
               
                   
                   
                 file will be created. 
               
               
                 File Name 
                 Field 
                 This alphanumeric field allows the users to input 
               
               
                   
                   
                 a desired file name. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 15 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 OK 
                 Button 
                 Closes the page and loads the selected data sheet. 
               
               
                 Cancel 
                 Button 
                 Closes the page. 
               
               
                 Inactivate 
                   
                 Data Sheet Approval privileges required. Inactivates the 
               
               
                 selected 
                   
                 currently selected version of the data sheet. This is often 
               
               
                   
                   
                 used when a data sheet was created in error. Inactivating 
               
               
                   
                   
                 the last active version of a data sheet will prevent the data 
               
               
                   
                   
                 sheet from appearing in the data sheet catalog. A system 
               
               
                   
                   
                 or database administrator will be required to reactivate the 
               
               
                   
                   
                 data sheet. 
               
               
                 Delete 
                   
                 System Administrator privileges required. Delete the 
               
               
                 selected 
                   
                 selected version of the data sheet. This is often used when 
               
               
                   
                   
                 a data sheet was approved in error. Deleting the last active 
               
               
                   
                   
                 version of a data sheet will prevent the data sheet from 
               
               
                   
                   
                 appearing in the data sheet catalog. A system database 
               
               
                   
                   
                 administrator will be required to reactivate the data sheet. 
               
               
                 Search for 
                 Field 
                 Searches the database using the criteria entered and the 
               
               
                   
                   
                 search index selected. 
               
               
                 Search 
                 Drop Down 
                 Lists the available search criteria (Index). The selected 
               
               
                 Criteria 
                 List 
                 item dictates the search type. 
               
               
                 Data Sheet 
                 Grid 
                 Lists data sheets and data sheet groups that match the 
               
               
                 List 
                   
                 search criteria 
               
               
                 Group 
                 Button 
                 Enabled when a data sheet group is selected in the Data 
               
               
                 Members 
                   
                 Sheet List grid. Displays the Data Sheet Group Selector 
               
               
                   
                   
                 page to enable the user to select one of the member data 
               
               
                   
                   
                 sheets. 
               
               
                 Specification 
                 Field 
                 Displays the specification reference 
               
               
                 Reference 
               
               
                 Catalog 
                 Field 
                 Displays the catalog notes 
               
               
                 Notes 
               
               
                 Cal Proc. 
                 Field 
                 Displays the calibration procedure 
               
               
                 Att # 
                 Field 
                 Displays the calibration procedure attachment number 
               
               
                 Author 
                 Field 
                 Displays the name of the author 
               
               
                 Revision 
                 Field 
                 Displays the name of the revision editor 
               
               
                 Editor 
               
               
                 Approved 
                 Field 
                 Displays the name of the approver 
               
               
                 by 
               
               
                 Approved 
                 Field 
                 Displays the date approved 
               
               
                 Date 
               
               
                 Model 
                 Field 
                 Displays the model number 
               
               
                 Manufacturer 
                 Field 
                 Displays the manufacturer 
               
               
                 Description 
                 Field 
                 Displays the equipment description 
               
               
                 Permanent 
                 Field 
                 Displays the data sheet&#39;s Permanent Unique ID. This is a 
               
               
                 Unique 
                   
                 Guaranteed Unique ID (GUID). It is assigned to all 
               
               
                 Identifier 
                   
                 versions of a document. 
               
               
                 (PUI) 
               
               
                 Dynamic 
                 Field 
                 Displays the data sheet&#39;s Dynamic Unique ID. This is a 
               
               
                 Unique 
                   
                 Guaranteed Unique ID (GUID). It identifies a specific 
               
               
                 Identifier 
                   
                 version of a document. 
               
               
                 (DUI) 
               
               
                 Custom 
                 Check Box 
                 Read only. Indicates if the data sheet is based on custom 
               
               
                 Spec 
                   
                 specifications (other than the manufactures specifications). 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 16 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Close 
                 Button 
                 Closes the page. 
               
               
                 Approve as 
                 Button 
                 Data sheet approval privileges required. Approves the 
               
               
                 revision 
                   
                 data sheet and saves it to the database as a revision of 
               
               
                   
                   
                 the active data sheet. All previous versions of the data sheet are 
               
               
                   
                   
                 marked inactive. 
               
               
                 Approve as new 
                 Button 
                 Data sheet approval privileges required. Approves the 
               
               
                 data sheet 
                   
                 data sheet and saves it to the database as a new data 
               
               
                   
                   
                 sheet. Use this option when an existing data sheet has 
               
               
                   
                   
                 been used as a template for another data sheet, 
               
               
                   
                   
                 importing a data sheet file, or no previous version 
               
               
                   
                   
                 exists. 
               
               
                 Revision 
                 Grid 
                 Displays all of the previous versions of the data sheet. 
               
               
                 History 
               
               
                 Specification 
                 Field 
               
               
                 Reference 
               
               
                 Catalog Notes 
                 Field 
                 Displays the catalog notes 
               
               
                 procedure 
                 Field 
                 Displays the calibration procedure 
               
               
                 Att # 
                 Field 
                 Displays the calibration procedure attachment number 
               
               
                 Author 
                 Field 
                 Displays the name of the author 
               
               
                 Revision Editor 
                 Field 
                 Displays the name of the revision editor 
               
               
                 Reactivate 
                 Button 
                 System Administrator privileges required. Reactivate 
               
               
                 Selected 
                   
                 the selected version of the data sheet. Will cause all 
               
               
                   
                   
                 other versions to be inactivated. 
               
               
                 Inactivate all 
                   
                 System Administrator privileges required. Inactivate 
               
               
                   
                   
                 all versions of the data sheet. This is often used when 
               
               
                   
                   
                 a data sheet is removed from service. Inactivating the 
               
               
                   
                   
                 last active version of a data sheet will prevent the data 
               
               
                   
                   
                 sheet from appearing in the data sheet catalog. A 
               
               
                   
                   
                 database administrator will be required to reactivate 
               
               
                   
                   
                 the data sheet. 
               
               
                 Delete selected 
                   
                 System Administrator privileges required. Delete the 
               
               
                   
                   
                 selected version of the data sheet. This is often used 
               
               
                   
                   
                 when a data sheet was approved in error. The System 
               
               
                   
                   
                 Administrator must take great care not to delete a data 
               
               
                   
                   
                 sheet once used. Deleting the last active version of a 
               
               
                   
                   
                 data sheet will prevent the data sheet from appearing in 
               
               
                   
                   
                 the data sheet catalog. A database administrator will 
               
               
                   
                   
                 be required to reactivate the data sheet. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 17 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 OK 
                 Button 
                 Use the selected data sheet and closes the page. 
               
               
                 Cancel 
                 Button 
                 Cancels data sheet selection and closes the page. 
               
               
                 Group Header 
                 Section 
               
               
                 Author 
                 Field 
                 Author of the New Group. 
               
               
                 Manufacturer 
                 Field 
                 The Equipment manufacturer 
               
               
                 Revision Editor 
                 Field 
                 User Revising Existing Group. 
               
               
                 Model Number 
                 Field 
                 Model of the UUT. 
               
               
                 Description 
                 Field 
                 Description of the Group. 
               
               
                 Procedure 
                 Field 
                 Calibration procedure associated with the group. 
               
               
                 Cal Interval 
                 Field 
                 The calibration for the unit. 
               
               
                 Spec Reference 
                 Field 
                 The specifications reference for the unit. 
               
               
                 Catalog Notes 
                 Field 
                 Notes to assist the user in identifying the 
               
               
                   
                   
                 particular group or data sheet. 
               
               
                 Group Members 
                 Section 
                 Applies to the selected member and displays 
               
               
                   
                   
                 information from the member data sheet&#39;s header. 
               
               
                 Member List 
                 Grid 
                 Lists the data sheet members of the group 
               
               
                 Cal Interval 
                 Field 
                 Displays the calibration interval 
               
               
                 Specifications 
                 Field 
                 Displays the Specification Reference 
               
               
                 Reference 
               
               
                 Catalog Notes 
                 Field 
                 Notes to assist the user in identifying the 
               
               
                   
                   
                 particular group or data sheet in the data sheet 
               
               
                   
                   
                 catalog. 
               
               
                 Procedure 
                 Field 
                 Calibration procedure. 
               
               
                 Att # 
                 Field 
                 Calibration procedure attachment number. 
               
               
                 Author 
                 Field 
                 Data sheet author. 
               
               
                 Revision Editor 
                 Field 
                 Data sheet reviser. 
               
               
                 Approved by 
                 Field 
                 Approver. 
               
               
                 Approved on 
                 Field 
                 Date approved. 
               
               
                 Model 
                 Field 
                 Model number. 
               
               
                 Manufacturer 
                 Field 
                 Equipment Manufacturer. 
               
               
                 Description 
                 Field 
                 Equipment description. 
               
               
                 Custom Specs 
                 Check Box 
                 Checked if Custom Specifications were used in the creation 
               
               
                   
                   
                 of the data sheet. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 18 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Approve Changes 
                 Button 
                 Accepts all changes and displays the Data Sheet 
               
               
                   
                   
                 Group Approval page. 
               
               
                 Cancel 
                 Button 
                 Cancels all changes and closes the page. 
               
               
                 Group Header 
                 Section 
               
               
                 Author 
                 Field 
                 Author of the New Group. 
               
               
                 Manufacturer 
                 Field 
                 The Equipment manufacturer 
               
               
                 Revision Editor 
                 Field 
                 User Revising Existing Group. 
               
               
                 Model Number 
                 Field 
                 Model of the UUT. 
               
               
                 Description 
                 Field 
                 Description of the Group. 
               
               
                 Procedure 
                 Field 
                 Calibration procedure for the Group. 
               
               
                 Cal Interval 
                 Field 
                 The calibration for the Group. 
               
               
                 Spec Reference 
                 Field 
                 The specifications reference for the Group. 
               
               
                 Catalog Notes 
                 Field 
                 Notes to assist the user in identifying the 
               
               
                   
                   
                 particular group or data sheet in the data sheet 
               
               
                   
                   
                 catalog. 
               
               
                 Group Members 
                 Section 
                 Applies to the selected member and displays 
               
               
                   
                   
                 information from the member data sheet&#39;s header. 
               
               
                 Member List 
                 Grid 
                 Lists the data sheet members of the group 
               
               
                 Insert member 
                 Button 
                 Inserts a new member into the member list. 
               
               
                 Above Selected 
               
               
                 Row 
               
               
                 Delete Selected 
                 Button 
                 Deletes the selected member. 
               
               
                 member 
               
               
                 Move member 
                 Button 
                 Moves the selected member up in the member 
               
               
                 Up 
                   
                 list. 
               
               
                 Move member 
                 Button 
                 Moves the selected member down in the member list. 
               
               
                 Down 
               
               
                 Cal Interval 
                 Field 
                 Displays the calibration interval 
               
               
                 Specifications 
                 Field 
                 Displays the Specification Reference 
               
               
                 Reference 
               
               
                 Catalog Notes 
                 Field 
                 Catalog notes. 
               
               
                 Procedure 
                 Field 
                 Calibration procedure. 
               
               
                 Att # 
                 Field 
                 Calibration procedure attachment number. 
               
               
                 Author 
                 Field 
                 Data sheet author. 
               
               
                 Revision Editor 
                 Field 
                 Data sheet reviser. 
               
               
                 Approved by 
                 Field 
                 Approver. 
               
               
                 Approved on 
                 Field 
                 Date approved. 
               
               
                 Model 
                 Field 
                 Model number. 
               
               
                 Manufacturer 
                 Field 
                 Equipment Manufacturer. 
               
               
                 Description 
                 Field 
                 Equipment description. 
               
               
                 Custom Specs 
                 Check Box 
                 Checked if Custom Specifications were used in the creation 
               
               
                   
                   
                 of the data sheet. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 19 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Close 
                 Button 
                 Closes the page. 
               
               
                 Approve as 
                 Button 
                 Data sheet approval privileges required. Approves the 
               
               
                 revision 
                   
                 data sheet group and saves it to the database as a 
               
               
                   
                   
                 revision of the active data sheet group. All previous 
               
               
                   
                   
                 versions of the data sheet group are marked inactive. 
               
               
                 Approve as new 
                 Button 
                 Data sheet approval privileges required. Approves the 
               
               
                 data sheet 
                   
                 data sheet group and saves it to the database as a new 
               
               
                   
                   
                 data sheet group. Use this option when an existing 
               
               
                   
                   
                 data sheet group has been used as a template for 
               
               
                   
                   
                 another data sheet group or no previous version exists. 
               
               
                 Revision 
                 Grid 
                 Displays all of the previous versions of the data sheet 
               
               
                 History 
                   
                 group. 
               
               
                 Specification 
                 Field 
                 Displays the specification reference for the data sheet 
               
               
                 Reference 
                   
                 group. 
               
               
                 Catalog Notes 
                 Field 
                 Displays the catalog notes for the data sheet group. 
               
               
                 procedure 
                 Field 
                 Displays the calibration procedure for the data sheet 
               
               
                   
                   
                 group. 
               
               
                 Att # 
                 Field 
                 Displays the calibration procedure attachment number 
               
               
                   
                   
                 for the data sheet group. 
               
               
                 Author 
                 Field 
                 Displays the name of the author of the data sheet 
               
               
                   
                   
                 group. 
               
               
                 Revision Editor 
                 Field 
                 Displays the name of the revision editor of the data 
               
               
                   
                   
                 sheet group. 
               
               
                 Reactivate 
                 Button 
                 System Administrator privileges required. Reactivate 
               
               
                 Selected 
                   
                 the selected version of the data sheet group. Will 
               
               
                   
                   
                 cause all other versions to be inactivated. 
               
               
                 Inactivate all 
                   
                 System Administrator privileges required. Inactivate 
               
               
                   
                   
                 all versions of the data sheet group. This is often used 
               
               
                   
                   
                 when a data sheet group is removed from service. 
               
               
                   
                   
                 Inactivating the last active version of a data sheet 
               
               
                   
                   
                 group will prevent the data sheet group from appearing 
               
               
                   
                   
                 in the data sheet catalog. A database administrator will 
               
               
                   
                   
                 be required to reactivate the data sheet group. 
               
               
                 Delete selected 
                   
                 System Administrator privileges required. Delete the 
               
               
                   
                   
                 selected version of the data sheet group. This is often 
               
               
                   
                   
                 used when a data sheet group was approved in error. 
               
               
                   
                   
                 Deleting the last active version of a data sheet group 
               
               
                   
                   
                 will prevent the data sheet group from appearing in the 
               
               
                   
                   
                 data sheet catalog. A database administrator will be required 
               
               
                   
                   
                 to reactivate the data sheet group. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 20 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 OK 
                 Button 
                 Closes the page and loads the selected calibration. 
               
               
                 Cancel 
                 Button 
                 Closes the page. 
               
               
                 Inactivate 
                   
                 Data Sheet Approval privileges required. Inactivates 
               
               
                 selected 
                   
                 the currently selected version of the calibration. This 
               
               
                   
                   
                 is often used when a calibration was created in error. 
               
               
                   
                   
                 Inactivating the last active version of a calibration will 
               
               
                   
                   
                 prevent the calibration from appearing in the 
               
               
                   
                   
                 calibration history catalog. A system or database 
               
               
                   
                   
                 administrator will be required to reactivate the 
               
               
                   
                   
                 calibration. 
               
               
                 Delete selected 
                   
                 System Administrator privileges required. Delete the 
               
               
                   
                   
                 selected version of the calibration. This is often used 
               
               
                   
                   
                 when a calibration was approved in error. Deleting the 
               
               
                   
                   
                 last active version of a calibration will prevent the 
               
               
                   
                   
                 calibration from appearing in the calibration history 
               
               
                   
                   
                 catalog. A system database administrator will be 
               
               
                   
                   
                 required to reactivate the calibration. 
               
               
                 Search for 
                 Field 
                 Searches the data base using the criteria entered and 
               
               
                   
                   
                 the search index selected. 
               
               
                 Search Criteria 
                 Drop 
                 Lists the available search criteria (Index). The selected 
               
               
                   
                 Down List 
                 item dictates the search type. 
               
               
                 Specification 
                 Field 
                 Displays the specification reference 
               
               
                 Reference 
               
               
                 Catalog Notes 
                 Field 
                 Displays the catalog notes 
               
               
                 Cal Proc. 
                 Field 
                 Displays the calibration procedure 
               
               
                 Att # 
                 Field 
                 Displays the calibration procedure attachment number 
               
               
                 Author 
                 Field 
                 Displays the name of the author 
               
               
                 Revision Editor 
                 Field 
                 Displays the name of the revision editor 
               
               
                 Approved by 
                 Field 
                 Displays the name of the approver 
               
               
                 Approved Date 
                 Field 
                 Displays the date approved 
               
               
                 Model 
                 Field 
                 Displays the model number 
               
               
                 Manufacturer 
                 Field 
                 Displays the manufacturer 
               
               
                 Description 
                 Field 
                 Displays the equipment description 
               
               
                 Permanent 
                 Field 
                 Displays the calibration&#39;s Permanent Unique ID. This 
               
               
                 Unique 
                   
                 is a Guaranteed Unique ID (GUID). It is assigned to 
               
               
                 Identifier (PUI) 
                   
                 all versions of a document. 
               
               
                 Dynamic 
                 Field 
                 Displays the calibration&#39;s Dynamic Unique ID. This is 
               
               
                 Unique 
                   
                 a Guaranteed Unique ID (GUID). It identifies a 
               
               
                 Identifier (DUI) 
                   
                 specific version of a document. 
               
               
                 Custom Spec 
                 Check Box 
                 Read only. Indicates if the data sheet is based on 
               
               
                   
                   
                 custom specifications (other than the manufactures 
               
               
                   
                   
                 specifications). 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 21 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Search 
                 Button 
                 This will search for the test number indicated in 
               
               
                   
                   
                 the Test Number Field. 
               
               
                 Manual Search 
                 Button 
                 This will open the standard Windows Open File 
               
               
                   
                   
                 page. 
               
               
                 Accept 
                 Button 
                 Accepts inputted values. 
               
               
                 Cancel 
                 Button 
                 Clears and closes Enter the Following Values 
               
               
                   
                   
                 Page. 
               
               
                 Test Number 
                 A/N 
                 Allows the user to search for specific test 
               
               
                 field 
                   
                 numbers. 
               
               
                 Select File 
                 A/N 
                 Displays the list of files found. 
               
               
                 Field 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 22 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Close 
                 Button 
                 Closes the page. 
               
               
                 OK 
                 Button 
                 Calibration signature privileges required. 
               
               
                   
                   
                 Signs the calibration, saves it to the database, 
               
               
                   
                   
                 and closes the page. 
               
               
                 Cancel 
                 Button 
                 Cancels the operation and closes the page. 
               
               
                 Enter your login 
                 Field 
                 Sets the user login. 
               
               
                 ID 
               
               
                 Enter your 
                 Field 
                 Sets the user password. 
               
               
                 password 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 23 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Print Now 
                 Button 
                 Prints the data sheet with the selected options. 
               
               
                 Printer Options 
                 Button 
                 Opens the standard Windows Print Setup page. 
               
               
                 Cancel 
                 Button 
                 Clears and closes the page. 
               
               
                 Data Sheet 
                 Check Box 
                 If this box is not selected, no print options are 
               
               
                   
                   
                 available and the only active button is Cancel. 
               
               
                 Specifications 
                 Check Box 
                 Includes the specifications in the data sheet 
               
               
                   
                   
                 printout. (not enabled in the current version of software). 
               
               
                   
                   
                 This feature will be available in a future version. 
               
               
                 All Pages 
                 Radio Button 
                 Allows all pages to be printed. 
               
               
                 Pages 
                 Radio Button 
                 Allows a user specified page range to be printed. 
               
               
                   
                   
                 This button activates the “Pages From” and the 
               
               
                   
                   
                 “Pages To” numeric fields. 
               
               
                 Automatic 
                 Radio Button 
                 Adds pages to the start and end of the numbering 
               
               
                   
                   
                 sequence base on the Boiler Plate options. 
               
               
                 Leave Blank 
                 Radio Button 
                 Leaves the page numbers off of the printouts. 
               
               
                 Modify Now 
                 Radio Button 
                 Allows the user to over ride the Boiler Plate 
               
               
                   
                   
                 options. This button activates the “Add n Pages 
               
               
                   
                   
                 to Front”, and the “Add n Pages to Back” 
               
               
                   
                   
                 numeric fields. 
               
               
                 Pages From 
                 Numeric 
                 Determines the number to start printing at. This 
               
               
                   
                   
                 is only available when the Pages radio box is selected. 
               
               
                 Pages To 
                 Numeric 
                 Determines the number to stop printing at. This 
               
               
                   
                   
                 is only available when the Pages radio box is selected. 
               
               
                 Add n Pages to 
                 Numeric 
                 Adds the selected amount of pages at the 
               
               
                 Front 
                   
                 beginning of the printout. This is only available 
               
               
                   
                   
                 when the Modify Now radio box is selected. 
               
               
                 Add n Pages to 
                 Numeric 
                 Adds the selected amount of pages at the end of 
               
               
                 Back 
                   
                 the printout. This is only available when the Modify 
               
               
                   
                   
                 Now radio box is selected. 
               
               
                 Print n Copies 
                 Numeric 
                 Sets the number of copies to be printed. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 24 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Close 
                 Button 
                 Closes the page. 
               
               
                 Revision 
                 Grid 
                 Displays all of the previous versions of the data 
               
               
                 History 
                   
                 sheet. 
               
               
                 Specification 
                 Field 
                 Displays the specification reference for the 
               
               
                 Reference 
                   
                 instrument. 
               
               
                 Catalog Notes 
                 Field 
                 Displays the catalog notes 
               
               
                 procedure 
                 Field 
                 Displays the calibration procedure 
               
               
                 Att # 
                 Field 
                 Displays the calibration procedure attachment 
               
               
                   
                   
                 number 
               
               
                 Author 
                 Field 
                 Displays the name of the author 
               
               
                 Revision Editor 
                 Field 
                 Displays the name of the revision editor 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 25 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 O.K. 
                 Button 
                 Accepts inputted values. 
               
               
                 Cancel 
                 Button 
                 Clears and closes the page. 
               
               
                 Search 
                 Button 
                 Opens the standard windows search dialogue. 
               
               
                 Edit Remarks 
                 Button 
                 Opens the Remarks Editor. 
               
               
                 Configure 
                 Button 
                 Allows for the selection of special data sheet 
               
               
                   
                   
                 operation commands (directives). 
               
               
                 TAR 
                 Check Box 
                 This includes the TAR information in the 
               
               
                   
                   
                 Tolerance column. 
               
               
                 Measurement 
                 Check Box 
                 This includes the Measurement Uncertainty 
               
               
                 Uncertainty 
                   
                 information in the Tolerance column. 
               
               
                 Custom 
                 Check Box 
                 Indicates that the specifications for the unit are 
               
               
                 Specifications 
                   
                 non-standard specs. 
               
               
                 Asset No. 
                 A/N 
                 The ID number for the unit under test. 
               
               
                 Test Number 
                 A/N 
                 The test identification number 
               
               
                 Manufacturer 
                 A/N 
                 The manufacturer of the unit under test. 
               
               
                 Model Number 
                 A/N 
                 The model number of the unit under test. 
               
               
                 Description 
                 A/N 
                 A brief description of the unit under test. 
               
               
                 Attachment No. 
                 A/N 
                 The number of attachments to the data sheet. 
               
               
                 Cal Proc 
                 A/N 
                 Name of calibration procedure. 
               
               
                 Cal Interval 
                 A/N 
                 Displays the calibration interval for the unit under 
               
               
                   
                   
                 test. 
               
               
                 Spec Reference 
                 A/N 
                 Displays a list of the specifications used. 
               
               
                 Catalog Notes 
                 A/N 
                 Additional information to aid in the selection of an 
               
               
                   
                   
                 appropriate data sheet. 
               
               
                 Add to Front 
                 Numeric 
                 Adds the number of pages to the front of the data 
               
               
                   
                   
                 sheet printout. 
               
               
                 Add to Back 
                 Numeric 
                 Adds the number of pages to the end of the data 
               
               
                   
                   
                 sheet printout. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 26 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts inputted values. 
               
               
                 Cancel 
                 Button 
                 Clears and closes the page. 
               
               
                 Global 
                 Button 
                 Setting a remark to Global will cause it to print on 
               
               
                   
                   
                 each page of the printed data sheet. If a remark is 
               
               
                   
                   
                 not global, it will only be printed if it&#39;s 
               
               
                   
                   
                 corresponding Remark Marker appears 
               
               
                   
                   
                 in the tolerance column of the printed page. 
               
               
                 Remarks 
                 A/N 
                 This field is where the user inputs their 
               
               
                   
                   
                 comments. 
               
               
                 Remark 
                 Symbol 
                 The symbol next to the global button indicates the 
               
               
                 Marker 
                   
                 marker used to indicate the comment. (i.e. *, **, 
               
               
                   
                   
                 ***). This symbol is placed in tolerance column 
               
               
                   
                   
                 of a data sheet row. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 27 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 % of I.V. 
                 Check Box 
                 Sets specification to include % of I.V. 
               
               
                 % of F.S. 
                 Check Box 
                 Sets specification to include % of F.S. 
               
               
                 Floor (Fixed 
                 Check Box 
                 Sets specifications to include a Floor value. 
               
               
                 Value relative 
               
               
                 to the units of 
               
               
                 measure) 
               
               
                 ±dB 
                 Check Box 
                 Sets specifications to ±dB. This check box excludes 
               
               
                   
                   
                 all other check boxes. 
               
               
                 Custom 
                 Check Box 
                 Sets specifications to include a custom specification 
               
               
                   
                   
                 calculator. (Professional and Pro Remote versions 
               
               
                   
                   
                 only) 
               
               
                 Nom Value 
                 Numeric 
                 Nominal value for specification. 
               
               
                 Units of 
                 A/N 
                 Units of measure and additional parameters. 
               
               
                 Measure 
               
               
                 Nominal Value 
                 Button 
                 Displays the Nominal Value Wizard to assist in the 
               
               
                 Wizard 
                   
                 selection of appropriate units of measure and 
               
               
                   
                   
                 formatting. 
               
               
                 x % of I.V. 
                 Numeric 
                 When the ppm check box is selected this becomes I.V. 
               
               
                   
                   
                 ppm. This is only available when the % of I.V. check 
               
               
                   
                   
                 box is selected. 
               
               
                 F.S. Val 
                 Numeric 
                 This is only available when the % of F.S. check box is 
               
               
                   
                   
                 selected. 
               
               
                 % of F.S. 
                 Numeric 
                 When the ppm check box is selected this becomes F.S. 
               
               
                   
                   
                 ppm. This is only available when the % of F.S. check 
               
               
                   
                   
                 box is selected. 
               
               
                 Floor Val 
                 Numeric 
                 This is only available when the Floor (Fixed Value) 
               
               
                   
                   
                 check box is selected. 
               
               
                 ±dB of I.V. 
                 Numeric 
                 This only available when the ±dB check box is 
               
               
                   
                   
                 selected. 
               
               
                 Db Type 
                 Drop 
                 This only available when the ±dB check box is 
               
               
                   
                 Down Box 
                 selected. 
               
               
                   
                   
                 Contains the following options: 
               
               
                   
                   
                 Volt/Amp: Log20 based calculation 
               
               
                   
                   
                 Power: Log10 based calculation 
               
               
                 Open 
                 Button 
                 This allows the use of a custom calculator. The 
               
               
                 Calculation 
                   
                 Custom Calculator List is displayed to allow the user 
               
               
                   
                   
                 to select a custom calculator. (Professional and Pro 
               
               
                   
                   
                 Remote versions only) 
               
               
                 Custom 
                 Field 
                 Displays the name of the Custom Calculator in use. 
               
               
                 Calculation 
                   
                 (Professional and Pro Remote versions only) 
               
               
                 Asymmetrical 
                 Check Box 
                 Limits the Selection Specification Types to: % of I.V., 
               
               
                 Tol 
                   
                 % of F.S., and Floor (Fixed Value). Also allows entry 
               
               
                   
                   
                 of separate specifications for the high and low 
               
               
                   
                   
                 tolerances. 
               
               
                 Multiple Test 
                 Check Box 
                 Allows for the creation of a series of test points based 
               
               
                 Point 
                   
                 on the currently entered specifications. It opens the 
               
               
                   
                   
                 Test Point List. 
               
               
                 As % of 
                 Numeric 
                 This list is only available when the Multiple Test Point 
               
               
                   
                   
                 check box is selected. This value is the base value for 
               
               
                   
                   
                 percentages entered in the Test Point List. 
               
               
                 Test Point List 
                 Grid 
                 This list is only available when the Multiple Test Point 
               
               
                   
                   
                 check box is selected. This list contains percentages of 
               
               
                   
                   
                 the base value entered in the “As % of” field. The 
               
               
                   
                   
                 result of each is the creation of a data sheet row for 
               
               
                   
                   
                 each value entered. The nominal test point value is the 
               
               
                   
                   
                 derived as the percentage of the “As % of” field. 
               
               
                 Lo to High 
                 Radio 
                 This sets the tolerance format to indicate low and high 
               
               
                   
                 Button 
                 tolerance values. (No operand selection is available 
               
               
                   
                   
                 with this choice.) 
               
               
                 n Units 
                 Radio 
                 This sets the tolerance format to indicate [operand] n 
               
               
                   
                 Button 
                 [units of measure]. 
               
               
                 n % 
                 Radio 
                 This sets the tolerance format to indicate [operand] n 
               
               
                   
                 Button 
                 [%]. 
               
               
                 n ppm 
                 Radio 
                 This sets the tolerance format to indicate [operand] n 
               
               
                   
                 Button 
                 [ppm]. 
               
               
                 Opperand 
                 Drop 
                 This drop down box contains the following symbols: ±, 
               
               
                   
                 Down Box 
                 &gt;, &gt;=, &lt;, &lt;=, and =. 
               
               
                 Decimals 
                 Numeric 
                 This numerical box only accepts entries of one to nine 
               
               
                   
                   
                 decimal digits. 
               
               
                 Calculate 
                 Button 
                 This button will prompt the user for any required 
               
               
                   
                   
                 custom calculator values, calculate the tolerance, 
               
               
                   
                   
                 format numeric values, prompt the user for a “function 
               
               
                   
                   
                 tested” and “step number”. Then insert a new data 
               
               
                   
                   
                 sheet row. Prior to the currently selected row. 
               
               
                 Recalc 
                 Button 
                 This button will prompt the user for any required 
               
               
                   
                   
                 custom calculator values, calculate the tolerance, 
               
               
                   
                   
                 format numeric values, prompt the user for a “function 
               
               
                   
                   
                 tested” and “step number”. Then update the currently 
               
               
                   
                   
                 selected data sheet row. This button is disabled when 
               
               
                   
                   
                 the Multiple Test Point check box is selected. 
               
               
                 Test Point 
                 Drop 
                 This is a drop down box contains a list of test point 
               
               
                 Definitions 
                 Down Box 
                 wizards. This list is created by selecting the menu 
               
               
                   
                   
                 option Tools, Wizards, Test Point Definitions, Load. 
               
               
                   
                   
                 Selecting one of these wizards will configure the 
               
               
                   
                   
                 calculator. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 28 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 OK 
                 Button 
                 Accepts inputted values and closes the page. 
               
               
                 Cancel 
                 Button 
                 Closes the page and aborts the new line creation. 
               
               
                 Procedure 
                 Field 
                 Sets the text to be displayed in the data sheet row&#39;s 
               
               
                 Step No 
                   
                 Step Num cell. 
               
               
                 Function 
                 Field 
                 Sets the text to be displayed in the data sheet row&#39;s 
               
               
                 Tested 
                   
                 Function Tested cell. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 29 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts edited values. 
               
               
                 Cancel 
                 Button 
                 Cancels edits and closes the page. 
               
               
                 Units by symbol 
                 Button 
                 Displays the Units of Measure by Symbol page to 
               
               
                   
                   
                 allow the user to select a unit using a symbol search. 
               
               
                 Units Groups 
                 Grid 
                 Lists all available units of measure groups. 
               
               
                 Units of measure 
                 Grid 
                 Lists all available units of measure for the group 
               
               
                   
                   
                 currently selected in the Units Groups grid. 
               
               
                 Parameters 
                 Grid 
                 Lists parametric data used in the Nominal Value. The 
               
               
                   
                   
                 first item listed is considered the primary parameter 
               
               
                   
                   
                 and is the quantitative reference for the nominal value. 
               
               
                   
                   
                 Other parameters listed are used as additional selection 
               
               
                   
                   
                 criteria used to assign Specification Tracking 
               
               
                   
                   
                 instrument ranges. 
               
               
                 Update from 
                 Button 
                 Replace the unit of measure in the selected row of the 
               
               
                 Unit 
                   
                 Parameters grid with the unit from the selected row of 
               
               
                   
                   
                 the Units of Measure grid. 
               
               
                 Add Param 
                 Button 
                 Appends a new parameter into the Parameters grid 
               
               
                   
                   
                 using the unit from the selected row of the Units of 
               
               
                   
                   
                 Measure grid. 
               
               
                 Delete 
                 Button 
                 Deletes the selected parameter from the Parameters 
               
               
                   
                   
                 grid. 
               
               
                 Set Nominal 
                 Field 
                 Sets the value of the currently selected parameter. 
               
               
                 Value 
               
               
                 Source Nominal 
                 A/N 
                 Shows the original content of the data sheet row&#39;s 
               
               
                   
                   
                 nominal value cell. 
               
               
                 Data Component 
                 A/N 
                 Shows the parametric data component of the nominal 
               
               
                   
                   
                 value text. 
               
               
                 Trailing Text 
                 Field 
                 Sets the text that will be displayed at the end of the 
               
               
                   
                   
                 nominal value text. Trailing text is used for 
               
               
                   
                   
                 documentation purposes only. It is has no parametric role. 
               
               
                 Composite 
                 A/N 
                 Shows the fully constructed content for the data sheet 
               
               
                 Result 
                   
                 row&#39;s nominal value text. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 30 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts edited values. 
               
               
                 Cancel 
                 Button 
                 Cancels edits and closes the page. 
               
               
                 TAR 
                 Section 
                 Sets the status of Test Accuracy Ration (TAR) printing in the 
               
               
                   
                   
                 data sheet column and the value to be used. 
               
               
                 TAR 
                 Drop 
                 Sets the status of TAR printing in the data sheet column. Has 
               
               
                   
                 Down 
                 the following options: Use Boiler Plate (Inclusion is 
               
               
                   
                 List 
                 determined by the TAR check box setting in the Master 
               
               
                   
                   
                 Boiler Plate.) Included (The TAR will be printed.) Not 
               
               
                   
                   
                 Included (The TAR will not be printed.) 
               
               
                 Boilerplate 
                 Field 
                 Indicates the setting of the TAR check box in the Master 
               
               
                   
                   
                 Boiler Plate. 
               
               
                 Auto . . . : 1 
                 Field 
                 Indicates the automatically determined TAR. This field will 
               
               
                   
                   
                 be left blank if the application is unable to calculate a TAR. 
               
               
                 User Declared 
                 Field 
                 Sets the TAR to be printed. While this value is stored with 
               
               
                   
                   
                 the data sheet, it is used for printing purposes only. The 
               
               
                   
                   
                 value will not be used by statistical analysis software. Only 
               
               
                   
                   
                 the internally calculated value can occupy the requisite 
               
               
                   
                   
                 database field. 
               
               
                 EMU 
                 Section 
                 Sets the status of Estimated Measurement Uncertainty 
               
               
                   
                   
                 (EMU) printing in the data sheet column and the value to be 
               
               
                   
                   
                 used. 
               
               
                 EMU 
                 Drop 
                 Sets the status of EMU printing in the data sheet column. 
               
               
                   
                 Down 
                 Has the following options: Use Boiler Plate (Inclusion is 
               
               
                   
                 List 
                 determined by the EMU check box setting in the Master 
               
               
                   
                   
                 Boiler Plate). Included (The TAR will be printed). Not 
               
               
                   
                   
                 Included (The TAR will not be printed). 
               
               
                 Boilerplate 
                 Field 
                 Indicates the setting of the EMU check box in the Master 
               
               
                   
                   
                 Boiler Plate. 
               
               
                 Auto . . . % 
                 Field 
                 Indicates the automatically determined EMU as a percentage 
               
               
                   
                   
                 of the nominal value. This field will be left blank if the 
               
               
                   
                   
                 application is unable to calculate an EMU %. 
               
               
                 Auto . . . 
                 Field 
                 Indicates the automatically determined EMU in units of 
               
               
                 U of M 
                   
                 measure. This field will be left blank if the application is 
               
               
                   
                   
                 unable to calculate an EMU in units of measure. 
               
               
                 User Declared 
                 Field 
                 Sets the EMU to be printed. When entering data into this 
               
               
                   
                   
                 field include any applicable symbols, such as “%”, “mV”, 
               
               
                   
                   
                 “m”, etc. While this value is stored with the data sheet, it is 
               
               
                   
                   
                 used for printing purposes only. The value will not be used 
               
               
                   
                   
                 by statistical analysis software. Only the internally 
               
               
                   
                   
                 calculated value can occupy the requisite database field. 
               
               
                 Remark Flag 
                 Drop 
                 Select a symbol that corresponds to the desired remark from 
               
               
                   
                 Down 
                 the Master Boiler Plate. 
               
               
                   
                 List 
               
               
                 Tolerance Info 
                 Field 
                 Displays the actual tolerance component of the data sheet 
               
               
                   
                   
                 row&#39;s Tolerance cell. 
               
               
                 Auto Data 
                 Field 
                 Displays the automatically generated component of the data 
               
               
                   
                   
                 sheet row&#39;s Tolerance cell. 
               
               
                 Trailing Text 
                 Field 
                 Sets the user inserted text that will be displayed within the 
               
               
                   
                   
                 Tolerance cell text. Trailing text is used for documentation 
               
               
                   
                   
                 purposes only. It is has no systemic role. 
               
               
                 Composite 
                 Field 
                 Shows the fully constructed content for the data sheet row&#39;s 
               
               
                 Result 
                   
                 Tolerance cell text. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 31 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Places the selected symbol on the windows 
               
               
                   
                   
                 clipboard and close the form. 
               
               
                 Cancel 
                 Button 
                 Closes the page. 
               
               
                 Unit of measure 
                 Grid 
                 Lists the available units of measure. 
               
               
                 list 
               
               
                 Find Symbol 
                 Field 
                 Searches the Unit of Measure list for 
               
               
                   
                   
                 symbol values entered. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 32 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 OK 
                 Button 
                 Accepts inputted values. 
               
               
                 Cancel 
                 Button 
                 Clears and closes the page. 
               
               
                 Fetch Wizards 
                 Button 
                 Displays the wizards contained in the listed 
               
               
                   
                   
                 files, from the selected folder. 
               
               
                 Drive 
                 Drop 
                 Lists the drives and network drives available 
               
               
                   
                 Down Box 
                 to the system. 
               
               
                 Folder 
                 A/N 
                 Shows the folder tree for the selected folder. 
               
               
                 Files 
                 A/N 
                 Shows the wizard files in the selected folder. 
               
               
                 Select a 
                 Grid 
                 Shows wizards in the selected directory&#39;s 
               
               
                 Wizard 
                   
                 files. This grid is updated when the user 
               
               
                 to Load 
                   
                 clicks on the Fetch Wizard button. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 33 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Create new 
                 Button 
                 Creates a new list, clearing all unsaved data. This 
               
               
                 macro list 
                   
                 button will be followed by a confirmation page 
               
               
                   
                   
                 requiring the user to choose yes or no to continue. 
               
               
                 Open Macro 
                 Button 
                 This button opens the standard Windows open file 
               
               
                 List 
                   
                 dialogue box. 
               
               
                 Save This 
                 Button 
                 This button opens the standard Windows save file 
               
               
                 Macro List 
                   
                 dialogue box. 
               
               
                 Boiler Plate 
                 Button 
                 This button opens the Calibration Process Manager 
               
               
                   
                   
                 Boiler Plate Data page. 
               
               
                 Insert a Test 
                 Button 
                 Inserts a new test point into the list before the 
               
               
                 Point Before 
                   
                 selected test point. 
               
               
                 This One 
               
               
                 Add Test Point 
                 Button 
                 Adds a new test point to the end of the list. 
               
               
                 To The End 
               
               
                 Delete This Test 
                 Button 
                 Deletes the currently selected test point. 
               
               
                 Point 
               
               
                 Clear All Test 
                 Button 
                 Clears all test points. 
               
               
                 Points 
               
               
                 Accept Changes 
                 Button 
                 Accepts all inputted data. 
               
               
                 to This Test 
               
               
                 Point 
               
               
                 Undo the 
                 Button 
                 Clears all edits. 
               
               
                 Changes to the 
               
               
                 Test Point 
               
               
                 Exit 
                 Button 
                 Closes the page. 
               
               
                 Lo to High 
                 Radio 
                 Sets the test point calculations to a “lo to high” 
               
               
                   
                 Button 
                 value calculation. 
               
               
                 +/−n Units 
                 Radio 
                 Sets the test point calculations +/−n units where n 
               
               
                   
                 Button 
                 is the variable number. 
               
               
                 +/−n % 
                 Radio 
                 Sets the test point calculations +/−n percentage 
               
               
                   
                 Button 
                 where n is the variable number. 
               
               
                 +/−n ppm 
                 Radio 
                 Sets the test point calculations +/−n parts per 
               
               
                   
                 Button 
                 million where n is the variable number. 
               
               
                 Multiple Test 
                 Check Box 
                 This opens the Test Point List frame. 
               
               
                 Points 
               
               
                 ppm 
                 Check Box 
                 Sets the appropriate “% of” values to a ppm 
               
               
                   
                   
                 value. 
               
               
                 ±(% IV) 
                 Tab 
                 This selection allows for data entry in the 
               
               
                   
                   
                 following fields: 
               
               
                   
                   
                 Nom. Val. Units of Measure % of I.V. 
               
               
                 ±(% IV + % 
                 Tab 
                 This selection allows for data entry in the 
               
               
                 FS) 
                   
                 following fields: 
               
               
                   
                   
                 Nom. Val Units of Measure % of I.V. 
               
               
                   
                   
                 FS Val. % of F.S. 
               
               
                 ±(% IV + 
                 Tab 
                 This selection allows for data entry in the 
               
               
                 Floor) 
                   
                 following fields: 
               
               
                   
                   
                 Nom. Val. Units of Measure % of I.V. 
               
               
                   
                   
                 Floor 
               
               
                 ±(% IV + % FS + 
                 Tab 
                 This selection allows for data entry in the 
               
               
                 Floor)) 
                   
                 following fields: 
               
               
                   
                   
                 Nom. Val. Units of Measure % of I.V. 
               
               
                   
                   
                 FS Val. % of F.S. Floor Val 
               
               
                 ±(% FS) 
                 Tab 
                 This selection allows for data entry in the 
               
               
                   
                   
                 following fields: 
               
               
                   
                   
                 Nom. Val. Units of Measure FS Val. % 
               
               
                   
                   
                 of F.S. 
               
               
                 ±(% FS + 
                 Tab 
                 This selection allows for data entry in the 
               
               
                 Floor) 
                   
                 following fields: 
               
               
                   
                   
                 Nom. Val. Units of Measure FS Val. 
               
               
                   
                   
                 % of F.S. Floor 
               
               
                 ±(Floor) 
                 Tab 
                 This selection allows for data entry in the 
               
               
                   
                   
                 following fields: 
               
               
                   
                   
                 Nom. Val. Units of Measure Floor 
               
               
                 Wizard Title 
                 A/N 
                 Title of wizard as set by the user. 
               
               
                 Created By 
                 A/N 
                 Name of user creating wizard. 
               
               
                 Description 
                 A/N 
                 Description of wizard and function. 
               
               
                 Test point Title 
                 A/N 
                 Definition of test point. 
               
               
                 Step Num 
                 A/N 
                 Content for Step Number cell of test points 
               
               
                   
                   
                 created with the wizard. 
               
               
                 Function Tested 
                 A/N 
                 Content for Function Tested cell of test points 
               
               
                   
                   
                 created with the wizard. 
               
               
                 Nom. Val 
                 Numeric 
                 The nominal value for the test point. 
               
               
                 Units of 
                 Numeric 
                 The units of measure for the test point. 
               
               
                 Measure 
               
               
                 % of I.V. 
                 Numeric 
                 The acceptable percentage of the indicated value. 
               
               
                 FS Val 
                 Numeric 
                 The full scale value. 
               
               
                 % of F.S. 
                 Numeric 
                 The acceptable percentage range of the full scale 
               
               
                   
                   
                 value. 
               
               
                 Floor Val 
                 Numeric 
                 The floor error value. 
               
               
                 Fixed Decimal 
                 Numeric 
                 The number of decimal places. 
               
               
                 As % of 
                 A/N 
                 Sets the variable for the multiple test points. 
               
               
                 Test Point List 
                 A/N 
                 Shows a list of the multiple test points. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 34 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 O.K. create the 
                 Button 
                 Creates the data set specified and asks the user if 
               
               
                 data sheet 
                   
                 another data set is to be added. If another data set is to 
               
               
                   
                   
                 be added, the user is returned to the module to specify 
               
               
                   
                   
                 the next data set. 
               
               
                 No more data 
                 Button 
                 If at least one data set was created: Opens the 
               
               
                 sets 
                   
                 Calibration Process Manager Boiler Plate Data page. 
               
               
                   
                   
                 If no data sets were created: Clears information and 
               
               
                   
                   
                 closes the page. 
               
               
                 Gauge Type 
                 Drop 
                 This contains the following gauge selections: Simple 
               
               
                   
                 Down Box 
                 Gauge, Compound Gauge, Vacuum Gauge, 
               
               
                   
                   
                 Atmospheric Ref. Gauge, and Barometer. 
               
               
                 Analog 
                 Button 
                 Applies rules relative to analog gauges. Only one of 
               
               
                   
                   
                 the Analog or Digital buttons may be selected at one 
               
               
                   
                   
                 time. 
               
               
                 Digital 
                 Button 
                 Applies rules relative to digital gauges. Only one of 
               
               
                   
                   
                 the Analog or Digital buttons may be selected at one 
               
               
                   
                   
                 time. 
               
               
                 View Test 
                 Button 
                 Creates and displays the test point information in the 
               
               
                 Points 
                   
                 text field to the right. 
               
               
                 Top Scale Units 
                 Drop 
                 This drop down box contains units of measure options 
               
               
                   
                 Down Box 
                 to the selected gauge type. In Professional and Pro 
               
               
                   
                   
                 Remote versions the units list is dynamically generated 
               
               
                   
                   
                 from the system&#39;s units of measure list. In Standard 
               
               
                   
                   
                 Version the units list is fixed and may contain any of 
               
               
                   
                   
                 the following units, as applicable: Psig, mmHg, kPa, 
               
               
                   
                   
                 psid, Mbar, psia, InH20, bar, mHga, InHg, Pa, inHga 
               
               
                 Scale Bottom 
                 Numeric 
                 The lowest scale value. 
               
               
                 Full Scale 
                 Numeric 
                 The highest scale value. 
               
               
                 % Full Scale 
                 Numeric 
                 The percentage of full value. 
               
               
                 % I.V. 
                 Numeric 
                 The percentage of input value. 
               
               
                 Floor Value 
                 Numeric 
                 The unit in the parentheses reflects the chosen Top 
               
               
                 (unit) 
                   
                 Scale Unit. 
               
               
                 Ref Temp 
                 Numeric 
                 Only available when inH20, inHg, or mmHg units are 
               
               
                   
                   
                 selected. 
               
               
                 Max Test Point 
                 Numeric 
                 Only available for vacuum gauge. 
               
               
                 Value 
               
               
                 Use % Span 
                 Check Box 
                 This option sets the module to use a percentage of the 
               
               
                   
                   
                 gauges span (top and bottom scales combined) in place 
               
               
                   
                   
                 of the normal percentage of full scale for each scale. 
               
               
                   
                   
                 Applies when the Compound gauge option is selected. 
               
               
                   
                   
                 This option requires the module to cross reference the 
               
               
                   
                   
                 scales&#39; units of measure in real time to perform span 
               
               
                   
                   
                 and scale calculations. Unfortunately, this makes the 
               
               
                   
                   
                 feature unavailable to the Standard version 
               
               
                 Calculate Span 
                 Button 
                 Calculates the span of the gauge using the Full Scale 
               
               
                   
                   
                 value of each scale and their associated units of 
               
               
                   
                   
                 measure. The span is resolved to the top scale&#39;s units 
               
               
                   
                   
                 of measure. The button is only visible when the Use % 
               
               
                   
                   
                 Span option is Checked. (Unavailable in the Standard 
               
               
                   
                   
                 version) 
               
               
                 Span 
                 Field 
                 The span of a compound gauge, resolved to the top 
               
               
                   
                   
                 scale&#39;s units of measure. The field is only visible 
               
               
                   
                   
                 when the Use % Span option is Checked. (Unavailable 
               
               
                   
                   
                 in the Standard version) 
               
               
                 % of Span 
                 Field 
                 The percentage of span to be applied to the tolerance 
               
               
                   
                   
                 calculation, of each scale, of a compound gauge. Once 
               
               
                   
                   
                 the percentage of Span is calculated for the top scale 
               
               
                   
                   
                 units of measure, the value will be converted to each 
               
               
                   
                   
                 scale&#39;s units of measure. The field is only visible 
               
               
                   
                   
                 when the Use % Span option is Checked. (Unavailable 
               
               
                   
                   
                 in the Standard version) 
               
               
                 No. of Decimals 
                 Numeric 
                 Displays the number of decimal places. 
               
               
                 Calculator 
                 Button 
                 Calculates the number of points 
               
               
                 Step No. 
                 A/N 
                 Displays the identification number of the step. 
               
               
                 Increasing 
                 Check Box 
                 Creates increasing results values. 
               
               
                 Decreasing 
                 Check Box 
                 Creates decreasing result values 
               
               
                 Function tested 
                 A/N 
                 Identification for the tested function. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 35 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts edits to the uncertainty model and closes the 
               
               
                   
                   
                 page. 
               
               
                 Cancel 
                 Button 
                 Ignores edits to the uncertainty model and closes the 
               
               
                   
                   
                 page. 
               
               
                 Use system 
                 Check Box 
                 When checked, the Calibration Process Manager will 
               
               
                 Default 
                   
                 completely manage the uncertainty model. When 
               
               
                   
                   
                 unchecked, the user assumes responsibility for the 
               
               
                   
                   
                 uncertainty model. 
               
               
                 Copy Wizard 
                 Button 
                 Places the contents of the measurement uncertainty 
               
               
                   
                   
                 model into a special measurement uncertainty model 
               
               
                   
                   
                 clipboard. The clipboard contents will persist until the 
               
               
                   
                   
                 button is pressed again. 
               
               
                 Paste Wizard 
                 Button 
                 Replaces the contents of the measurement uncertainty 
               
               
                   
                   
                 model with the contents of the measurement 
               
               
                   
                   
                 uncertainty model clipboard. 
               
               
                 Type A 
                 Section 
                 This section of the page displays and manages the type 
               
               
                   
                   
                 A uncertainty elements. 
               
               
                 Uncertainty 
                 Grid 
                 Displays the uncertainty elements in the model (all 
               
               
                 Elements 
                   
                 values are displayed in k = 1 terms. 
               
               
                 Add 
                 Button 
                 Adds an uncertainty element and displays the 
               
               
                   
                   
                 Uncertainty Element Editor page. 
               
               
                 Delete 
                 Button 
                 Deletes the currently selected uncertainty element. 
               
               
                 Edit 
                 Button 
                 Displays the Uncertainty Element Editor page. 
               
               
                 Up 
                 Button 
                 Moves the currently selected uncertainty element up 
               
               
                   
                   
                 one row in the Uncertainty Elements grid. 
               
               
                 Down 
                 Button 
                 Moves the currently selected uncertainty element down 
               
               
                   
                   
                 one row in the Uncertainty Elements grid. 
               
               
                 Type B 
                 Section 
                 This section of the page displays and manages the type 
               
               
                   
                   
                 B uncertainty elements. 
               
               
                 Uncertainty 
                 Grid 
                 Displays the uncertainty elements in the model (all 
               
               
                 Elements 
                   
                 values are displayed in k = 1 terms. Items automatically 
               
               
                   
                   
                 managed by Calibration Process Manager are 
               
               
                   
                   
                 displayed in blue. 
               
               
                 Add 
                 Button 
                 Adds an uncertainty element and displays the 
               
               
                   
                   
                 Uncertainty Element Editor page. 
               
               
                 Delete 
                 Button 
                 Deletes the currently selected uncertainty element. 
               
               
                 Edit 
                 Button 
                 Displays the Uncertainty Element Editor page. 
               
               
                 Up 
                 Button 
                 Moves the currently selected uncertainty element up 
               
               
                   
                   
                 one row in the Uncertainty Elements grid. 
               
               
                 Down 
                 Button 
                 Moves the currently selected uncertainty element down 
               
               
                   
                   
                 one row in the Uncertainty Elements grid. 
               
               
                 Auto Include 
                 Check Box 
                 When checked, the Calibration Process Manager will 
               
               
                 Standards 
                   
                 automatically insert an element that represents each 
               
               
                 Uncertainty 
                   
                 Specification Tracking instrument range assigned to 
               
               
                   
                   
                 the test point. As standards are substituted, added, or 
               
               
                   
                   
                 deleted, the entries will be updated automatically. If 
               
               
                   
                   
                 this option is turned off, the user must make sure that 
               
               
                   
                   
                 standards information is kept current. 
               
               
                 Auto Include 
                 Check Box 
                 When Checked, the Calibration Process Manager will 
               
               
                 Resolution 
                   
                 automatically insert an element that represents the 
               
               
                 Uncertainty 
                   
                 resolution of the test point in terms of the unit of 
               
               
                   
                   
                 measure. 
               
               
                 Value 
                 A/N 
                 Indicates the Calibration Process Manager&#39;s perception 
               
               
                   
                   
                 of the test point resolution. 
               
               
                 Override 
                 Check Box 
                 The limitation of the automatic resolution detection 
               
               
                   
                   
                 system is that one count at the resolution indicated by 
               
               
                   
                   
                 the data sheet row&#39;s Nominal Value cell&#39;s primary 
               
               
                   
                   
                 parameter is assumed to be the resolution. If more 
               
               
                   
                   
                 than one count is applicable, the Override check box 
               
               
                   
                   
                 must be checked and the correct resolution must be 
               
               
                   
                   
                 entered into the Override Value text box. 
               
               
                 Override Value 
                 Field 
                 This field is only visible when the Override Check Box 
               
               
                   
                   
                 is checked. Sets the actual resolution that should be 
               
               
                   
                   
                 used in the measurement uncertainty model. The 
               
               
                   
                   
                 override value will be used if the override check box is 
               
               
                   
                   
                 checked. 
               
               
                 k= 
                 Field 
                 Sets the coverage factor for the measurement 
               
               
                   
                   
                 uncertainty calculation. 
               
               
                 EMU= 
                 A/N 
                 Displays the calculated Estimated Measurement 
               
               
                   
                   
                 Uncertainty. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 36 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts inputted values and closes the 
               
               
                   
                   
                 page. 
               
               
                 Cancel 
                 Button 
                 Ignores edits and closes the page. 
               
               
                 Fetch Wizard 
                 Button 
                 Imports all wizards from the selected 
               
               
                   
                   
                 folder. 
               
               
                 Description 
                 Field 
                 Sets the display name of the uncertainty 
               
               
                   
                   
                 element. 
               
               
                 Sense Coeff. 
                 Field 
                 Sets the sensitivity coefficient of the 
               
               
                   
                   
                 uncertainty element. 
               
               
                 Value (k = 1) 
                 Field 
                 The uncertainty contribution of the 
               
               
                   
                   
                 element expressed in k = 1 terms 
               
               
                   
                   
                 relative to the unit of measure. 
               
               
                 Include Element 
                 Check Box 
                 When Checked, the uncertainty element 
               
               
                 in TAR 
                   
                 is included in TAR calculations. 
               
               
                 Calculations 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
             
               
                 TABLE 37 
               
               
                   
               
               
                 Mode 
                 Application 
               
               
                   
               
             
            
               
                 “Myself” 
                 UUT Self definition 
               
               
                 “Def Stds” 
                 Default standards determined during data sheet design 
               
               
                 “Stds Used” 
                 Standards used in an actual calibration 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 38 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Stds Used 
                 Button 
                 Switches the Specification Tracking Instruments 
               
               
                   
                   
                 Interface to the Standards used mode. 
               
               
                 Def Std 
                 Button 
                 Switches the Specification Tracking Instruments 
               
               
                   
                   
                 Interface to the Default Standards mode. 
               
               
                 Myself 
                 Button 
                 Switches the Specification Tracking Instruments 
               
               
                   
                   
                 Interface to the Myself mode. 
               
               
                 This TP 
                 Button 
                 Instructs the Nominal Value decoder to reset the 
               
               
                   
                   
                 synch parameters for the selected test point. 
               
               
                 All TP&#39;s 
                 Button 
                 Instructs the Nominal Value decoder to reset the 
               
               
                   
                   
                 synch parameters for all the test points. 
               
               
                 Instruments 
                 Section 
                 All items in this section are data sheet in scope. 
               
               
                 Instrument List 
                 Grid 
                 Displays all instruments assigned to the data sheet 
               
               
                   
                   
                 and loaded into memory 
               
               
                 Add 
                 Button 
                 This button opens the Instrument Specifications 
               
               
                   
                   
                 Catalog. 
               
               
                 Remove 
                 Button 
                 Removes the selected instrument from the list. 
               
               
                 Replace with Selected 
                 Button 
                 Replaces the selected instrument with another 
               
               
                   
                   
                 instrument. The desired instrument must already be 
               
               
                   
                   
                 on the list. 
               
               
                 Auto Assign 
                 Button 
                 Auto assigns instrument ranges to the data sheet&#39;s rows. 
               
               
                 Instrument Ranges 
                 Section 
                 All items in this section are test point (data sheet 
               
               
                   
                   
                 row) in scope. The currently selected test point to 
               
               
                   
                   
                 be exact. 
               
               
                 Instrument Range List 
                 Grid 
                 Displays all instrument ranges assigned to a test 
               
               
                   
                   
                 point. 
               
               
                 Auto Assign to TP 
                 Button 
                 Auto assigns instrument ranges to the currently 
               
               
                   
                   
                 selected test point in the data sheet. 
               
               
                 Manual Assign to TP 
                 Button 
                 Allows the user to manually assign instrument 
               
               
                   
                   
                 ranges to the selected test point. 
               
               
                 Remove from TP 
                 Button 
                 Allows the user to remove assigned instrument 
               
               
                   
                   
                 ranges from the selected test point. 
               
               
                 Instrument Synch 
                 Grid 
                 Displays the parametric criteria used to 
               
               
                 Parameters 
                   
                 automatically select standards and determine if the 
               
               
                   
                   
                 Calibration Process Manger can perform 
               
               
                   
                   
                 calculations based on instrument ranges assigned. 
               
               
                   
                   
                 If a parametric match is not achieved with assigned 
               
               
                   
                   
                 instrument ranges, no calculations will be performed 
               
               
                   
                   
                 using the Specification Tracking instrument range 
               
               
                   
                   
                 specifications. 
               
               
                 UUT Acc 
                 Field 
                 Displays the accuracy of the UUT. Available only 
               
               
                   
                   
                 when test point specifications are present. 
               
               
                 Std Acc 
                 Field 
                 Displays the accuracy of the assigned instrument 
               
               
                   
                   
                 range/s. Available only when test point 
               
               
                   
                   
                 specifications are present, at least one instrument 
               
               
                   
                   
                 range is assigned to the test point, and parametric 
               
               
                   
                   
                 synchronization is achieved. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 39 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Stds Used 
                 Button 
                 Switches the Specification Tracking Instruments 
               
               
                   
                   
                 Interface to the Standards used mode. 
               
               
                 Def Std 
                 Button 
                 Switches the Specification Tracking Instruments 
               
               
                   
                   
                 Interface to the Default Standards mode. 
               
               
                 Myself 
                 Button 
                 Switches the Specification Tracking Instruments 
               
               
                   
                   
                 Interface to the Myself mode. 
               
               
                 This TP 
                 Button 
                 Resets the synch parameters for the selected test point. 
               
               
                 All TP&#39;s 
                 Button 
                 Resets the synch parameters for all the test points. 
               
               
                 Instruments 
                 Section 
                 All items in this section are data sheet in scope. 
               
               
                 Instrument List 
                 Grid 
                 Displays all instruments assigned to the data sheet and 
               
               
                   
                   
                 loaded into memory 
               
               
                 Add 
                 Button 
                 This button opens the Instrument Specifications 
               
               
                   
                   
                 Catalog. 
               
               
                 Remove 
                 Button 
                 Removes the selected instrument from the list. 
               
               
                 Replace with 
                 Button 
                 Replaces the selected instrument with another 
               
               
                 Selected 
                   
                 instrument. The desired instrument must already be 
               
               
                   
                   
                 on the list. 
               
               
                 Auto Assign 
                 Button 
                 Auto assigns instrument ranges to the data sheet&#39;s 
               
               
                   
                   
                 rows. 
               
               
                 Set “Used” as 
                 Button 
                 Copies instrument and range assignments from the 
               
               
                 “Default” 
                   
                 Standards Used configuration and duplicates them 
               
               
                   
                   
                 in the Default Standards configuration. 
               
               
                 Instrument Ranges 
                 Section 
                 All items in this section are test point (data sheet 
               
               
                   
                   
                 row) in scope. The currently selected test point to 
               
               
                   
                   
                 be exact. 
               
               
                 Instrument Range 
                 Grid 
                 Displays all instrument ranges assigned to a test 
               
               
                 List 
                   
                 point. 
               
               
                 Auto Assign to TP 
                 Button 
                 Auto assigns instrument ranges to all the currently 
               
               
                   
                   
                 selected test point in the data sheet. 
               
               
                 Manual Assign to TP 
                 Button 
                 Allows the user to manually assign instrument 
               
               
                   
                   
                 ranges to the selected test point. 
               
               
                 Remove from TP 
                 Button 
                 Allows the user to remove assigned instrument 
               
               
                   
                   
                 ranges from selected test point. 
               
               
                 Instrument Synch 
                 Grid 
                 Displays the parametric criteria used to 
               
               
                 Parameters 
                   
                 automatically select standards and determine if the 
               
               
                   
                   
                 Calibration Process Manger can perform 
               
               
                   
                   
                 calculations based on instrument ranges assigned. 
               
               
                   
                   
                 If a parametric match is not achieved with assigned 
               
               
                   
                   
                 instrument ranges, no calculations will be 
               
               
                   
                   
                 performed using the instrument ranges. 
               
               
                 UUT Acc 
                 Field 
                 Displays the accuracy of the UUT. Available only 
               
               
                   
                   
                 when test point specifications are present. 
               
               
                 Std Acc 
                 Field 
                 Displays the accuracy of the assigned instrument 
               
               
                   
                   
                 range/s. Available only when test point 
               
               
                   
                   
                 specifications are present, at least one instrument 
               
               
                   
                   
                 range is assigned to the test point, and parametric 
               
               
                   
                   
                 synchronization is achieved. 
               
               
                 TAR 
                 Field 
                 Displays the Test Accuracy Ratio (TAR). TAR&#39;s 
               
               
                   
                   
                 may be calculated using the simple method of 
               
               
                   
                   
                 comparing the accuracy ratios or utilizing the 
               
               
                   
                   
                 measurement uncertainty in the calculation. This 
               
               
                   
                   
                 option is set in the master setup of the Calibration 
               
               
                   
                   
                 Process manger. Available only when test point 
               
               
                   
                   
                 specifications are present, at least one instrument 
               
               
                   
                   
                 range is assigned to the test point, and parametric 
               
               
                   
                   
                 synchronization is achieved. 
               
               
                 EMU 
                 Field 
                 Displays the Estimated Measurement Uncertainty 
               
               
                   
                   
                 (EMU). Available only when test point 
               
               
                   
                   
                 specifications are present, at least one instrument 
               
               
                   
                   
                 range is assigned to the test point, and parametric 
               
               
                   
                   
                 synchronization is achieved. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 40 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Stds Used 
                 Button 
                 Switches the Specification Tracking Instruments 
               
               
                   
                   
                 Interface to the Standards used mode. 
               
               
                 Def Std 
                 Button 
                 Switches the Specification Tracking Instruments 
               
               
                   
                   
                 Interface to the Default Standards mode. 
               
               
                 Myself 
                 Button 
                 Switches the Specification Tracking Instruments 
               
               
                   
                   
                 Interface to the Myself mode. 
               
               
                 Asset ID&#39;s 
                 Button 
                 This button opens the Set Instrument Asset ID&#39;s page. 
               
               
                 This TP 
                 Button 
                 Resets the synch parameters for the selected test 
               
               
                   
                   
                 point. 
               
               
                 All TP&#39;s 
                 Button 
                 Resets the synch parameters for all the test points. 
               
               
                 Instruments 
                 Section 
                 All items in this section are data sheet in scope. 
               
               
                 Instrument List 
                 Grid 
                 Displays all instruments assigned to the data sheet 
               
               
                   
                   
                 and loaded into memory 
               
               
                 Add 
                 Button 
                 This button opens the Instrument Specifications 
               
               
                   
                   
                 Catalog. 
               
               
                 Remove 
                 Button 
                 Removes the selected instrument from the list. 
               
               
                 Replace with 
                 Button 
                 Replaces the selected instrument with another 
               
               
                 Selected 
                   
                 instrument. The desired instrument must already be 
               
               
                   
                   
                 on the list. 
               
               
                 Auto Assign 
                 Button 
                 Auto assigns instrument ranges to the data sheet&#39;s 
               
               
                   
                   
                 rows. 
               
               
                 Use “Default” Stds 
                 Button 
                 Copies instrument and range assignments from the 
               
               
                   
                   
                 Default Standards configuration and duplicates 
               
               
                   
                   
                 them in the Standards Used configuration. The 
               
               
                   
                   
                 user must still set the asset ID numbers for the 
               
               
                   
                   
                 actual standards used. 
               
               
                 Instrument Ranges 
                 Section 
                 All items in this section are test point (data sheet row) 
               
               
                   
                   
                 in scope. The currently selected test point to be exact. 
               
               
                 Instrument Range 
                 Grid 
                 Displays all instrument ranges assigned to a test 
               
               
                 List 
                   
                 point. 
               
               
                 Auto Assign to TP 
                 Button 
                 Auto assigns instrument ranges to all the currently 
               
               
                   
                   
                 selected test point in the data sheet. 
               
               
                 Manual Assign to TP 
                 Button 
                 Allows the user to manually assign instrument 
               
               
                   
                   
                 ranges to the selected test point. 
               
               
                 Remove from TP 
                 Button 
                 Allows the user to remove assigned instrument 
               
               
                   
                   
                 ranges from selected test point. 
               
               
                 Instrument Synch 
                 Grid 
                 Displays the parametric criteria used to 
               
               
                 Parameters 
                   
                 automatically select standards and determine if the 
               
               
                   
                   
                 Calibration Process Manger can perform 
               
               
                   
                   
                 calculations based on instrument ranges assigned. 
               
               
                   
                   
                 If a parametric match is not achieved with assigned 
               
               
                   
                   
                 instrument ranges, no calculations will be 
               
               
                   
                   
                 performed using the instrument ranges. 
               
               
                 UUT Acc 
                 Field 
                 Displays the accuracy of the UUT. Available only 
               
               
                   
                   
                 when test point specifications are present. 
               
               
                 Std Acc 
                 Field 
                 Displays the accuracy of the assigned instrument 
               
               
                   
                   
                 range/s. Available only when test point 
               
               
                   
                   
                 specifications are present, at least one instrument 
               
               
                   
                   
                 range is assigned to the test point, and parametric 
               
               
                   
                   
                 synchronization is achieved. 
               
               
                 TAR 
                 Field 
                 Displays the Test Accuracy Ratio (TAR). TAR&#39;s 
               
               
                   
                   
                 may be calculated using the simple method of 
               
               
                   
                   
                 comparing the accuracy ratios or utilizing the 
               
               
                   
                   
                 measurement uncertainty in the calculation. This 
               
               
                   
                   
                 option is set in the master setup of the Calibration 
               
               
                   
                   
                 Process manger. Available only when test point 
               
               
                   
                   
                 specifications are present, at least one instrument 
               
               
                   
                   
                 range is assigned to the test point, and parametric 
               
               
                   
                   
                 synchronization is achieved. 
               
               
                 EMU 
                 Field 
                 Displays the Estimated Measurement Uncertainty 
               
               
                   
                   
                 (EMU). Available only when test point 
               
               
                   
                   
                 specifications are present, at least one instrument 
               
               
                   
                   
                 range is assigned to the test point, and parametric 
               
               
                   
                   
                 synchronization is achieved. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 41 
               
               
                   
               
               
                 Mode 
                 Indicator 
                 Information conveyed 
               
               
                   
               
             
            
               
                 “Myself” 
                 None 
                 Calibration Processor does not anticipate an instrument 
               
               
                   
                   
                 range assignment to this test point. This condition exists 
               
               
                   
                   
                 when no test point specifications are available or no 
               
               
                   
                   
                 Standards Synchronization parameters are available. 
               
               
                   
                 (Silver 
                 Calibration Processor anticipates an instrument range 
               
               
                   
                 square) 
                 assignment to this test point, but has not received one. This 
               
               
                   
                   
                 condition exists when test point specifications are available 
               
               
                   
                   
                 and Standards Synchronization parameters are available. 
               
               
                   
                 (Green 
                 An instrument range is assigned and the test point and 
               
               
                   
                 square) 
                 instrument range&#39;s calculated accuracies do agree within 0.1% 
               
               
                   
                   
                 of each other. 
               
               
                   
                 (Red square) 
                 An instrument range is assigned, but the test point and 
               
               
                   
                   
                 instrument range&#39;s calculated accuracies do not agree within 
               
               
                   
                   
                 0.1% of each other or Calibration Process Manger is unable 
               
               
                   
                   
                 to perform calculations using the data. 
               
               
                 “Def Stds” 
                 None 
                 Calibration Processor does not anticipate an instrument 
               
               
                 and “Stds 
                   
                 range assignment to this test point. This condition exists 
               
               
                 Used” 
                   
                 when no test point specifications are available or no 
               
               
                   
                   
                 Standards Synchronization parameters are available. 
               
               
                   
                 (Silver) 
                 Calibration Processor anticipates an instrument range 
               
               
                   
                   
                 assignment to this test point, but has not received one. This 
               
               
                   
                   
                 condition exists when test point specifications are available 
               
               
                   
                   
                 and Standards Synchronization parameters are available. 
               
               
                   
                 (Green) 
                 An instrument range is assigned and the test resulting test 
               
               
                   
                   
                 accuracy ration is &gt;=4:1. 
               
               
                   
                 (Red) 
                 An instrument range is assigned, but the test accuracy ratio is 
               
               
                   
                   
                 &lt;4:1. 
               
               
                   
                 (Fuchsia) 
                 An instrument range was manually assigned to the test point 
               
               
                   
                   
                 that the Calibration Process Manager can not use, but will 
               
               
                   
                   
                 store with the data sheet, as directed by the user. This 
               
               
                   
                   
                 condition exists when test point specifications are not 
               
               
                   
                   
                 available, Standards Synchronization parameters are not 
               
               
                   
                   
                 available, or Standards Synchronization parameters do not 
               
               
                   
                   
                 directly match the instrument range selected. This condition 
               
               
                   
                   
                 will most often occur when an instrument range is manually 
               
               
                   
                   
                 selected and is not a direct parametric match with the test 
               
               
                   
                   
                 point. As of this version, the Calibration Processor is not 
               
               
                   
                   
                 able to divine accuracy information from a combination of 
               
               
                   
                   
                 standards that indirectly comprise a parametric match with 
               
               
                   
                   
                 the test point. i.e.: The test point is measuring Ohms and a 
               
               
                   
                   
                 voltage source and current shunt are selected as standards. 
               
               
                   
                   
                 The manual assignments should still be made to facilitate 
               
               
                   
                   
                 instrument range utilization reporting and statistics, but the 
               
               
                   
                   
                 user is on his own for TAR and EMU calculations. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 42 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 OK 
                 Button 
                 Accepts all changes and closes the page. 
               
               
                 Cancel 
                 Button 
                 Cancels all changes and closes the page. 
               
               
                 Search For 
                 Field 
                 Allows the user to input specific search criteria. 
               
               
                 Test No 
                 Radial 
                 Searches the catalog by test number. 
               
               
                 (Exact Match) 
                 Button 
               
               
                 Asset No 
                 Radial 
                 Searches the catalog by asset number. 
               
               
                 (Exact Match) 
                 Button 
               
               
                 Description 
                 Radial 
                 Searches the catalog by description. 
               
               
                 (contains) 
                 Button 
               
               
                 Author 
                 Radial 
                 Searches the catalog by author. 
               
               
                 (contains) 
                 Button 
               
               
                 Revision 
                 Field 
                 Displays the revision number of the file in the 
               
               
                   
                   
                 catalog. 
               
               
                 Manufacturer 
                 Field 
                 Displays the manufacturer of the unit for the 
               
               
                   
                   
                 file in the catalog. 
               
               
                 Model 
                 Field 
                 Displays the model number of the file in the 
               
               
                   
                   
                 catalog. 
               
               
                 Description 
                 Field 
                 Displays the description of the unit for the file 
               
               
                   
                   
                 in the catalog. 
               
               
                 Revision 
                 Field 
                 Displays the revision number. 
               
               
                 Manufacturer 
                 Field 
                 Displays the equipment manufacturer. 
               
               
                 Model 
                 Field 
                 This field displays model number. 
               
               
                 Description 
                 Field 
                 This field displays the equipment description. 
               
               
                 Specifications 
                 Field 
                 Displays the specifications referenced in the 
               
               
                 Reference 
                   
                 selected item. 
               
               
                 Author 
                 Field 
                 Displays the author. 
               
               
                 Verified by 
                 Field 
                 Displays the user who verified the 
               
               
                   
                   
                 specifications. 
               
               
                 Approved by 
                 Field 
                 Displays the name of the use who approved the 
               
               
                   
                   
                 instrument. 
               
               
                 Data 
                 Field 
                 Displays the date approved. 
               
               
                 Approved 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 43 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts all changes and closes the page. 
               
               
                 Cancel 
                 Button 
                 Cancels all changes and closes the page. 
               
               
                 Assign all 
                 Radio 
                 This selection will allow the application to assign 
               
               
                 applicable 
                 Button 
                 all standards that are applicable to the UUT to 
               
               
                 standards 
                   
                 the data sheet. 
               
               
                 Assign only 
                 Radio 
                 This selection will allow the application to assign 
               
               
                 the selected 
                 Button 
                 only the selected standards to the UUT to the 
               
               
                 standards 
                   
                 data sheet. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 44 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 + 
                 Button 
                 Adds an instrument to the list 
               
               
                 − 
                 Button 
                 Removes an instrument from the list 
               
               
                 Use Range 
                 Button 
                 Assigns the selected range to the test point. 
               
               
                 Remove 
                 Button 
                 Removes selected range assignment from the 
               
               
                 Range 
                   
                 test point. 
               
               
                 Exit 
                 Button 
                 Closes the page. 
               
               
                 Show Only 
                 Check 
                 Filters the display to only show ranges that fit 
               
               
                 Ranges That 
                 Box 
                 the instrument range search criteria, as 
               
               
                 Match All 
                   
                 determined by the data sheet&#39;s Nominal Value 
               
               
                 Critical 
                   
                 column contents. 
               
               
                 Criteria 
               
               
                 Instruments 
                 Field 
                 Displays the available instruments 
               
               
                 Functions 
                 Field 
                 Displays the functions of the selected 
               
               
                   
                   
                 instrument. 
               
               
                 Ranges 
                 Field 
                 Displays the Ranges of the selected instrument 
               
               
                   
                   
                 function. 
               
               
                 Range Detail 
                 Field 
                 Gives details of the selected range. 
               
               
                 Composite 
                 Field 
                 Displays each of the instrument ranges assigned 
               
               
                 Selection of 
                   
                 to the test point. 
               
               
                 Standards For 
               
               
                 This Test 
               
               
                 Point 
               
               
                 Standards 
                 Field 
                 Displays the instrument range search criteria, as 
               
               
                 Search 
                   
                 determined by the data sheet&#39;s Nominal Value 
               
               
                 Criteria 
                   
                 column contents. 
               
               
                 Function 
                 Field 
                 Displays the function of the chosen instrument. 
               
               
                 Nominal 
                 Field 
                 Displays the test point&#39;s nominal value. 
               
               
                 TP Accuracy 
                 Field 
                 Displays the UUT&#39;s test point accuracy. 
               
               
                 Accuracy 
                 Field 
                 Displays the composite accuracy of the 
               
               
                 (RSS) 
                   
                 assigned instruments relative to the test point&#39;s 
               
               
                   
                   
                 nominal value. This value is calculated using 
               
               
                   
                   
                 the Root Sum Squared (RSS) method. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 45 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Removes the currently selected instrument&#39;s range 
               
               
                   
                   
                 assignments from all data sheet test point rows, 
               
               
                   
                   
                 assigns ranges from the instrument selected in the 
               
               
                   
                   
                 Substitutes grid, and closes the page. 
               
               
                 Cancel 
                 Button 
                 Cancels the operation and closes the page. 
               
               
                 Replace 
                 Field 
                 The instrument to be replaced. 
               
               
                 With 
                 Grid 
                 List of instruments available to act as a substitute 
               
               
                 (Substitutes) 
                   
                 for the instrument to be replaced. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 46 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts changes, and closes the screen. 
               
               
                 Cancel 
                 Button 
                 Cancels all changes, and closes the screen. 
               
               
                 Standards 
                 Grid 
                 List of standards currently assigned to the 
               
               
                   
                   
                 calibration. 
               
               
                 Accept Asset 
                 Button 
                 Accepts the inputted asset number. 
               
               
                 No 
               
               
                 Undo Asset 
                 Button 
                 Undo the changed asset number. This cannot be 
               
               
                 No 
                   
                 undone once the Accept Asset No button is 
               
               
                   
                   
                 clicked. 
               
               
                 Asset No 
                 Field 
                 This field allows the user to assign an asset 
               
               
                   
                   
                 number to each of the standards used in the 
               
               
                   
                   
                 calibration. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 47 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Execute Command 
                 Button 
                 Runs the selected command. 
               
               
                 Append New 
                 Button 
                 Inserts a new command at the end of the 
               
               
                 Command to the End 
                   
                 command list. Opens the Automation 
               
               
                 of the List 
                   
                 Command Editor page. 
               
               
                 Insert a New 
                 Button 
                 Inserts a new command before the 
               
               
                 Command Before the 
                   
                 specified row in the command list. 
               
               
                 Selected Row 
                   
                 Opens Automation Command Editor 
               
               
                   
                   
                 Page. 
               
               
                 Delete Selected 
                 Button 
                 Deletes the selected command/s. 
               
               
                 command/s 
               
               
                 Edit the Selected 
                 Button 
                 Opens the Automation Command Editor 
               
               
                 Command 
                   
                 Page. 
               
               
                 Move Command Up 
                 Button 
                 Moves the selected command up in the 
               
               
                   
                   
                 command list. 
               
               
                 Move Command 
                 Button 
                 Moves the selected command down in 
               
               
                 Down 
                   
                 the command list. 
               
               
                 Copy Selected 
                 Button 
                 Copies all selected commands to the 
               
               
                 Commands 
                   
                 clipboard. 
               
               
                 Append Copied 
                 Button 
                 Pastes the copied commands to the end 
               
               
                 Commands to the 
                   
                 of the command list. 
               
               
                 End of the List 
               
               
                 Insert Copied 
                 Button 
                 Pastes the copied commands before the 
               
               
                 Commands Before 
                   
                 selected command. 
               
               
                 the Selected 
               
               
                 Command 
               
               
                 View Command 
                 Button 
                 Loads the ATE Clipboard Viewer. 
               
               
                 Clipboard 
               
               
                 Command List 
                 A/N 
                 listing of commands assigned to the 
               
               
                   
                   
                 currently selected data sheet row. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 48 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Command 
                 Drop 
                 Lists all available commands and is used 
               
               
                   
                 Down Box 
                 to select a command. 
               
               
                 Group 
                 Drop 
                 Lists all of the available groups and is 
               
               
                   
                 Down Box 
                 used to associate a group with the 
               
               
                   
                   
                 selected command. 
               
               
                 Use 
                 Button 
                 Displays the wizard applicable to the 
               
               
                 Command 
                   
                 selected command. The button is only 
               
               
                 Wizard 
                   
                 enabled for commands that have a 
               
               
                   
                   
                 command wizard. 
               
               
                 Command 
                 Button 
                 Displays the Command Group Editor 
               
               
                 Group 
                   
                 page. 
               
               
                 Editor 
               
               
                 Accept 
                 Button 
                 Accepts the current command 
               
               
                   
                   
                 configuration. 
               
               
                 Cancel 
                 Button 
                 Clears and closes the page. 
               
               
                 Copy Var Name 
                 Button 
                 Copies the selected variable&#39;s name onto 
               
               
                   
                   
                 the Windows clipboard for pasting into 
               
               
                   
                   
                 command parameters. 
               
               
                 Move Variable 
                 Button 
                 Moves the selected variable up in the list. 
               
               
                 Up 
               
               
                 Move 
                 Button 
                 Moves the selected variable down in the 
               
               
                 Variable Down 
                   
                 list. 
               
               
                 Add Variable 
                 Button 
                 Adds a new variable. 
               
               
                 Delete Variable 
                 Button 
                 Deletes the selected variable. 
               
               
                 User Defined 
                 Button 
                 Opens the User Defined Variable page. 
               
               
                 Variable Editor 
               
               
                 Name 
                 A/N 
                 Name of the Variable. 
               
               
                 Value 
                 A/N 
                 Value of the Variable. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 49 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts the edits to the variable and closes the 
               
               
                   
                   
                 form. 
               
               
                 Cancel 
                 Button 
                 Cancels the edits to the variable and closes the 
               
               
                   
                   
                 form. 
               
               
                 Variable 
                 Field 
                 The Name given to the user defined variable. This 
               
               
                 Name 
                   
                 name must be unique and is used to identify the 
               
               
                   
                   
                 variable in script commands, equations, etc. There 
               
               
                   
                   
                 is no restriction on the name of a variable. It is 
               
               
                   
                   
                 recommended that the variable name represent it&#39;s 
               
               
                   
                   
                 utilization to make the script easily readable. 
               
               
                 Default 
                 Field 
                 The default value assigned to the variable. The 
               
               
                 Value 
                   
                 value may be text or numeric. It is the 
               
               
                   
                   
                 responsibility of the script author to ensure that the 
               
               
                   
                   
                 content of a variable is applicable to its intended 
               
               
                   
                   
                 use. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 50 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Move Group Up 
                 Button 
                 Moves the selected group up in the group list. 
               
               
                 Move Group Down 
                 Button 
                 Moves the selected group down in the group list. 
               
               
                 Insert a New Group 
                 Button 
                 Inserts a new group into the command list. 
               
               
                 Delete the Selected 
                 Button 
                 Deletes the selected group. 
               
               
                 Group 
               
               
                 Edit Group Name 
                 Button 
                 This button opens the Group Editor Panel. 
               
               
                 Add the Group to 
                 Button 
                 Adds the currently selected command group to the 
               
               
                 My List 
                   
                 data sheet group list. 
               
               
                 Accept 
                 Button 
                 Accepts all changes to the command group. This 
               
               
                   
                   
                 button is only available when the Edit Group button 
               
               
                   
                   
                 is selected. 
               
               
                 Clear 
                 Button 
                 Clears all changes to the command group. This 
               
               
                   
                   
                 button is only available when the Edit Group button 
               
               
                   
                   
                 is selected. 
               
               
                 Close 
                 Button 
                 Closes the Command Group Editor page. 
               
               
                 Menu Key 
                 Numeric 
                 Displays the command menu key that executes the 
               
               
                   
                   
                 command group. 
               
               
                 Group Name 
                 Text 
                 Displays the name of the command group. 
               
               
                 Keypad Exec (0-9) 
                 Field 
                 This field allows the user to change or set the 
               
               
                   
                   
                 command menu key for the selected command 
               
               
                   
                   
                 group. This field is only available when the Edit 
               
               
                   
                   
                 Group button is selected. 
               
               
                 Command Groups 
                 Selections 
                 Lists the available command groups in the System 
               
               
                   
                   
                 Group List: 
               
               
                 Configure 
                 Group 
                 This item is an Interface Group template (Not a 
               
               
                   
                   
                 system Directive.) 
               
               
                 Energize 
                 Group 
                 This item is an Interface Group template (Not a 
               
               
                   
                   
                 system Directive.) 
               
               
                 De-Energize 
                 Group 
                 This item is an Interface Group template (Not a 
               
               
                   
                   
                 system Directive.) 
               
               
                 Execute Step 
                 Group 
                 This item is an Interface Group template (Not a 
               
               
                   
                   
                 system Directive.) 
               
               
                 Measure 
                 Group 
                 This item is an Interface Group template (Not a 
               
               
                   
                   
                 system Directive.) 
               
               
                 Join all Groups 
                 Group 
                 This item is a System Group: Directs the script 
               
               
                   
                   
                 system to execute command members regardless of 
               
               
                   
                   
                 which command group was selected from the 
               
               
                   
                   
                 command menu. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 51 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Cancel 
                 Button 
                 Cancels command selection and closes the page. 
               
               
                 Command 
                 Field 
                 Lists the name of the Command to execute. The 
               
               
                 Names 
                   
                 command name is the name of Interface Group it 
               
               
                 [multiple] 
                   
                 represents. 
               
               
                 Command 
                 Button 
                 Clicking on this button or pressing the key 
               
               
                 Key 
                   
                 displayed on the button will execute the 
               
               
                 [multiple] 
                   
                 associated group of commands. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 52 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Instruction 
                 Drop 
                 The name of the currently selected calibration 
               
               
                   
                 Down List 
                 instruction. Any of the data sheet&#39;s calibration 
               
               
                   
                   
                 instructions can be selected from the list. 
               
               
                 FIGS. No. [1-5] 
                 Field 
                 The figure number to be displayed on the Graphical 
               
               
                   
                   
                 Instruction Viewer&#39;s tabs. 
               
               
                 Image [1-5] 
                 Drop 
                 The name of the image selected or blank if not in use. 
               
               
                   
                 Down List 
                 Any of the data sheet&#39;s binary resource images can be 
               
               
                   
                   
                 selected from the list. 
               
               
                 Show Image 
                 Button 
                 Clicking on one of these buttons will cause the image to 
               
               
                 [1-5] 
                   
                 be displayed in the Image Viewer box. 
               
               
                 Image Viewer 
                 Image Box 
                 Clicking on one of the Show Image buttons will cause 
               
               
                   
                   
                 the image to be displayed. 
               
               
                 Edit Instructions 
                 Button 
                 Opens the Calibration Instruction Editor page. 
               
               
                 Edit Binary List 
                 Button 
                 Opens the Binary Resources List page. 
               
               
                 Accept 
                 Button 
                 Accepts edits, and closes the page. 
               
               
                 Cancel 
                 Button 
                 Clears edits, and closes the page. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 53 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Resource 
                 Grid 
                 A list of the data sheet&#39;s binary resource files. The size 
               
               
                 Name 
                   
                 of the resource&#39;s memory utilization is also shown. 
               
               
                 Image Viewer 
                 Image Box 
                 Clicking on an image in the Resource Name grid will 
               
               
                   
                   
                 cause the image to be displayed. 
               
               
                 Add New 
                 Button 
                 Opens the Binary Resource Editor page. 
               
               
                 Resource 
               
               
                 Remove 
                 Button 
                 Removes the selected resource file from the list. 
               
               
                 Resource 
               
               
                 Edit Resource 
                 Button 
                 Opens the Binary Resource Editor page. 
               
               
                 Combined 
                 Field 
                 Shows the combined memory utilization of all binary 
               
               
                 Resource Size 
                   
                 resources 
               
               
                 Close 
                 Button 
                 Closes the page. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 54 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Name 
                 Field 
                 Edit the name of the binary resource. This name is 
               
               
                   
                   
                 used to reference this specific resource on the list. 
               
               
                   
                   
                 There is no restriction on the name, but it should be 
               
               
                   
                   
                 unique within the binary resource list. 
               
               
                 Item 
                 Field 
                 Indicates the Calibration Process Manager&#39;s perception 
               
               
                 Type 
                   
                 of the resource. As of this version, only items of type 
               
               
                   
                   
                 “Image” can be added to the resource list. 
               
               
                 Source 
                 Field 
                 The name of the file that the resource was 
               
               
                 File 
                   
                 created from. 
               
               
                   
                   
                 This name will be included in the file name when the 
               
               
                   
                   
                 resource is exported to a file, via data sheet export: 
               
               
                   
                   
                 [data sheet file name] + “_(“ + [source file 
               
               
                   
                   
                 name] + “).” + [source file extension] 
               
               
                 File 
                 Field 
                 The file date of the file that the resource was created 
               
               
                 Date 
                   
                 from. 
               
               
                 Size 
                 Field 
                 Displays the memory utilization of the resource item. 
               
               
                 Load 
                 Button 
                 Displays the windows image dialog to select an image 
               
               
                 Image 
                   
                 file that will be the source of the binary resource. 
               
               
                 Accept 
                 Button 
                 Accepts edits, and closes the page. 
               
               
                 Cancel 
                 Button 
                 Clears edits, and closes the page. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 55 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts inputted values and closes the page. 
               
               
                 Cancel 
                 Button 
                 Ignores edits and closes the page. 
               
               
                 Caption [1-10] 
                 A/N 
                 Required field (when the row is in use). This 
               
               
                   
                   
                 is where the user sets the caption title 
               
               
                   
                   
                 for the variable. 
               
               
                 Default Value 
                 A/N 
                 Displays the value of the variable. 
               
               
                 [1-10] 
               
               
                 Tail [1-10] 
                 A/N 
                 A short caption displayed to the right of the 
               
               
                   
                   
                 default value i.e. (%, ppm, mv, etc.). 
               
               
                 Target Variable 
                 Drop 
                 Required Field (when the row is in use). 
               
               
                 [1-10] 
                 Down Box 
                 Lists the options of all user defined 
               
               
                   
                   
                 variables. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 56 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept these 
                 Button 
                 Accepts edits to the command parameters, Closes the 
               
               
                 command 
                   
                 form, and returns to the Command Editor form. 
               
               
                 parameters 
               
               
                 Cancel 
                 Button 
                 Ignores edits to the command parameters, Closes the 
               
               
                   
                   
                 form, and returns to the Command Editor form. 
               
               
                 GPIB Reset 
                 Button 
                 Clears the GPIB interface. 
               
               
                 Send 
                 Button 
                 Sends a test command to the GPIB interface. 
               
               
                 Command 
               
               
                 Fetch 
                 Button 
                 Sends a test command from the GPIB interface and 
               
               
                 Command 
                   
                 reads the response. 
               
               
                 &gt;Direct 
                 A/N 
                 Indicates the response received from the fetch 
               
               
                 Response&lt; 
                   
                 operation. 
               
               
                 Cut Left 
                 Numeric 
                 The number of characters to remove from the 
               
               
                   
                   
                 response string. An entry of −1 indicates no 
               
               
                   
                   
                 characters are removed. 
               
               
                 Keep Left 
                 Numeric 
                 The number of characters to retain after the “cut 
               
               
                   
                   
                 left”. An entry of −1 indicates all characters are 
               
               
                   
                   
                 retained. 
               
               
                 Decimals 
                 Numeric 
                 This option forces the numeric format to use the 
               
               
                   
                   
                 listed number of decimal places. 
               
               
                 GPIB Address 
                 A/N 
                 1-31 
               
               
                 Rcv Buffer 
                 Numeric 
                 Max number of characters to Receive. When in 
               
               
                 Size 
                   
                 doubt set this to 100. 
               
               
                 Exponent 
                 Drop 
                 Selects the desired exponential format of the result. 
               
               
                   
                 Down 
               
               
                   
                 Box 
               
               
                 Formatted Results 
                 Numeric 
                 The results from the entered data. 
               
               
                 String 
               
               
                 Resulting 
                 Numeric 
                 The results from the entered data. 
               
               
                 Numeric Value 
               
               
                 Continuous Run 
                 Check 
                 When this check box is selected the machine will 
               
               
                   
                 Box 
                 continually run the fetch command option. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 57 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept these 
                 Button 
                 Accepts edits to the command parameters, Closes the 
               
               
                 command 
                   
                 form, and returns to the Command Editor form. 
               
               
                 parameters 
               
               
                 Cancel 
                 Button 
                 Ignores edits to the command parameters, Closes the 
               
               
                   
                   
                 form, and returns to the Command Editor form. 
               
               
                 Send 
                 Button 
                 Sends a command. 
               
               
                 Command 
               
               
                 Fetch 
                 Button 
                 Sends a command and displays the returned value. 
               
               
                 Command 
               
               
                 Configure Port 
                 Button 
                 Opens the Com Port Setup page. 
               
               
                 &gt;Direct 
                 A/N 
                 Indicates the response received from the fetch 
               
               
                 Response&lt; 
                   
                 operation. 
               
               
                 Cut Left 
                 Numeric 
                 The number of characters to remove from the 
               
               
                   
                   
                 response string. An entry of −1 indicates no 
               
               
                   
                   
                 characters are removed. 
               
               
                 Keep Left 
                 Numeric 
                 The number of characters to retain after the “cut left”. 
               
               
                   
                   
                 An entry of −1 indicates all characters are retained. 
               
               
                 Decimals 
                 Numeric 
                 This option forces the numeric format to use the listed 
               
               
                   
                   
                 number of decimal places. 
               
               
                 Exponent 
                 Drop 
                 Selects the desired exponential format of the result. 
               
               
                   
                 Down 
               
               
                   
                 Box 
               
               
                 Formatted Results 
                 Numeric 
                 The results from the entered data. 
               
               
                 String 
               
               
                 Resulting 
                 Numeric 
                 The results from the entered data. 
               
               
                 Numeric Value 
               
               
                 Continuous Run 
                 Check 
                 When this check box is selected the machine will 
               
               
                   
                 Box 
                 continually run the fetch command option until 
               
               
                   
                   
                 halted by the user. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 58 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 View the 
                 Button 
                 Displays the text of a calibration instruction. 
               
               
                 Instructions 
               
               
                 Exit 
                 Button 
                 Closes the ATE Clipboard Viewer. 
               
               
                 Calibration 
                 List 
                 Displays the title of the calibration instructions 
               
               
                 Instructions 
                   
                 used by the copied commands. 
               
               
                 User Defined 
                 Grid 
                 This group contains the following fields: 
               
               
                 Variable List 
               
               
                 Name 
                 Field 
                 Lists the name of the individual user defined 
               
               
                   
                   
                 variable. 
               
               
                 Value 
                 Field 
                 Displays the value currently stored in the user 
               
               
                   
                   
                 defined variable. 
               
               
                 Command 
                 Grid 
                 This group has the following fields: 
               
               
                 Groups 
               
               
                 Key 
                 Field 
                 Displays the key pad selector key for the group. 
               
               
                 Group Name 
                 Field 
                 Displays the group name. 
               
               
                 Commands 
                 List 
                 Lists the commands currently copied to the 
               
               
                   
                   
                 clipboard. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 59 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Open Equation 
                 Button 
                 This button allows the user to open a saved equation 
               
               
                   
                   
                 file. 
               
               
                 Save Equation 
                 Button 
                 This button allows the user to save an equation. 
               
               
                 Copy Function to 
                 Button 
                 Copies the selected function to the clipboard. 
               
               
                 Clipboard 
               
               
                 Copy Variable to 
                 Button 
                 Copies the selected variable to the clipboard. 
               
               
                 Clipboard 
               
               
                 Variables 
                 Grid 
                 A list of available variables. 
               
               
                 Variable Up 
                 Button 
                 Moves the variable up in the list. 
               
               
                 Variable Down 
                 Button 
                 Moves the variable down in the list. 
               
               
                 Add Variable 
                 Button 
                 Adds a new variable to the list. 
               
               
                 Remove Variable 
                 Button 
                 Deletes the variable from the list. 
               
               
                 Edit Variable 
                 Button 
                 Opens the equation variable editor page. 
               
               
                 Execute Equation 
                 Button 
                 Runs the current variable. 
               
               
                 OK 
                 Button 
                 Accepts changes and closes the page. 
               
               
                 Equation File 
                 Field 
                 Displays the path to the file. 
               
               
                 Prototype 
                 Field 
                 Displays the parametric requirements of the selected 
               
               
                   
                   
                 function. 
               
               
                 Description 
                 Field 
                 Describes the selected function. 
               
               
                 Set Value 
                 Field 
                 This field allows the user to set the numeric values 
               
               
                   
                   
                 for the variables. 
               
               
                 Equation 
                 Field 
                 Displays the equation. 
               
               
                 Result 
                 Field 
                 Displays the result of the executed equation. 
               
               
                 Auto insert 
                 Check 
                 This check box allows the user to insert selected 
               
               
                 functions and 
                 Box 
                 variables and functions simply by double clicking the 
               
               
                 variables into 
                   
                 mouse on them. 
               
               
                 equation on “dbl 
               
               
                 click” 
               
               
                 Include prototype 
                 Check 
                 This check box allows the application to include all 
               
               
                 on auto insert. 
                 Box 
                 prototype data on auto inserts. 
               
               
                 Functions 
                 Selection 
                 This field has a plurality of mathematical functions. 
               
               
                   
                 Field 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 60 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Load Calculator 
                 Button 
                 Loads a previously saved Custom Calculator. 
               
               
                 Save Calculator 
                 Button 
                 Saves a Custom Calculator. 
               
               
                 “user prompt” 
                 Button 
                 Moves the selected “user prompt” up in the command 
               
               
                 Up 
                   
                 list. 
               
               
                 “user prompt” 
                 Button 
                 Moves the selected “user prompt” down in the 
               
               
                 Down 
                   
                 command list. 
               
               
                 Edit the user 
                 Button 
                 Opens the Enter the User Query Setup page. 
               
               
                 query list 
               
               
                 Variable Up 
                 Button 
                 Moves the selected variable up in the variable list. 
               
               
                 Variable Down 
                 Button 
                 Moves the selected variable down in the variable list. 
               
               
                 Add Variable 
                 Button 
                 Opens the Equation Variable Editor page. 
               
               
                 Delete Variable 
                 Button 
                 Deletes the selected variable 
               
               
                 Edit Variable 
                 Button 
                 Opens the Equation Variable Editor page . . . 
               
               
                 Use as Equation 
                 Button 
                 Sets equation variable as the target variable. 
               
               
                 Target 
               
               
                 Equation Up 
                 Button 
                 Moves the selected equation up in the equation list. 
               
               
                 Equation Down 
                 Button 
                 Moves the selected equation down in the equation list. 
               
               
                 Add Equation 
                 Button 
                 Opens the Equation Editor page. 
               
               
                 Delete Equation 
                 Button 
                 Deletes the selected equation. 
               
               
                 Edit Equation 
                 Button 
                 Opens the Equation Editor page. 
               
               
                 Exec All 
                 Button 
                 Executes a test of the custom calculators function. 
               
               
                 Use as Equation 
                 Button 
                 Sets the currently selected Variable to be the target 
               
               
                 Target 
                   
                 variable for the custom equation. Note: the 
               
               
                   
                   
                 “Sys_Result” system variable is always the target of 
               
               
                   
                   
                 the last equation executed. This is the return value of 
               
               
                   
                   
                 the custom calculator. 
               
               
                 Exit 
                 Button 
                 Clears and closes the Custom Calculator Workshop 
               
               
                   
                   
                 page. 
               
               
                   
                   
                 Only available when running from the “tools” menu. 
               
               
                 Calculator 
                 A/N 
                 The name given to the custom calculator by the 
               
               
                 Name 
                   
                 creating user. 
               
               
                 Author 
                 A/N 
                 The name of the user who creates the custom 
               
               
                   
                   
                 calculator. 
               
               
                 Version 
                 A/N 
                 The version of the custom calculator as determined by 
               
               
                   
                   
                 the user. 
               
               
                 Calculator User 
                 A/N 
                 Displays user query prompts that will take place during 
               
               
                 Queries 
                   
                 calculation execution. 
               
               
                 System 
                 A/N 
                 Displays the fixed (System) variables. These variables 
               
               
                 Variables 
                   
                 can be used in equations, but are read only. They 
               
               
                   
                   
                 include: 
               
               
                   
                   
                 Sys_Result: Return value of the Custom Calculator. 
               
               
                   
                   
                 Sys_Nominal: The Absolute value of the Nominal 
               
               
                   
                   
                 Value 
               
               
                   
                   
                 Sys_NominalSign: 1 if the nominal value was positive 
               
               
                   
                   
                 and −1 if the nominal value was negative. 
               
               
                   
                   
                 Note: The Calibration Process Manager&#39;s tolerance 
               
               
                   
                   
                 calculator uses all absolute values to calculate 
               
               
                   
                   
                 tolerances. The original sign of the nominal value is 
               
               
                   
                   
                 restored when calculations are completed. The 
               
               
                   
                   
                 Sys_Nominal variable value is also made absolute in 
               
               
                   
                   
                 custom calculators to make them easier to design for 
               
               
                   
                   
                 use with the tolerance calculator. If sign is relevant to 
               
               
                   
                   
                 your custom calculation, use: (Sys_Nominal * 
               
               
                   
                   
                 Sys_NominalSign) to restore sign to the nominal value. 
               
               
                 Variables 
                 A/N 
                 Displays the variables as set in the Equation Editor. 
               
               
                 Equations 
                 A/N 
                 Displays the equations as set in the Equation Editor. 
               
               
                 Nominal Value 
                 Numeric 
                 This number is used to test the custom calculators 
               
               
                   
                   
                 function. 
               
               
                 Results 
                 Numeric 
                 Displays the result of the custom calculator test. 
               
               
                   
                   
                 Note: The result of a custom calculator must be 
               
               
                   
                   
                 relative to the nominal value&#39;s unit of measure when 
               
               
                   
                   
                 used with the Calibration Process Manger&#39;s integrated 
               
               
                   
                   
                 tolerance calculator. The value must also be absolute. 
               
               
                 Accept 
                 Button 
                 Accepts edits to the calculator and closes the 
               
               
                   
                   
                 workshop. Only available when editing one of the 
               
               
                   
                   
                 custom calculators assigned to the data sheet. 
               
               
                 Cancel 
                 Button 
                 Cancels edits to the calculator and closes the 
               
               
                   
                   
                 workshop. Only available when editing one of the 
               
               
                   
                   
                 custom calculators assigned to the data sheet. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 61 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Accept 
                 Button 
                 Accepts inputted values. 
               
               
                 Cancel 
                 Button 
                 Clears and closes Enter the Following Values page. 
               
               
                 Caption 
                 A/N 
                 Required field. This is where the user sets the 
               
               
                   
                   
                 caption title for the variable. 
               
               
                 Default 
                 A/N 
                 Displays the value of the variable. 
               
               
                 Value 
               
               
                 Tail 
                 A/N 
                 A short caption displayed to the right of the default 
               
               
                   
                   
                 value i.e. (%, ppm, mv, etc.). 
               
               
                 Target 
                 Drop 
                 Lists the options of all system and user defined 
               
               
                 Variable 
                 Down 
                 variables. This is a required field. 
               
               
                   
                 Box 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 62 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 GPIB Reset 
                 Button 
                 Clears the GPIB interface. 
               
               
                 Send 
                 Button 
                 Sends a test command to the GPIB interface. 
               
               
                 Command 
               
               
                 Fetch 
                 Button 
                 Sends a test command from the GPIB interface and reads 
               
               
                 Command 
                   
                 the response. 
               
               
                 &gt;Direct 
                 A/N 
                 Indicates the response received from the fetch 
               
               
                 Response&lt; 
                   
                 operation. 
               
               
                 Cut Left 
                 Numeric 
                 The number of characters to remove from the 
               
               
                   
                   
                 response string. An entry of −1 indicates no 
               
               
                   
                   
                 characters are removed. 
               
               
                 Keep Left 
                 Numeric 
                 The number of characters to retain after the “cut 
               
               
                   
                   
                 left”. An entry of −1 indicates all characters are retained. 
               
               
                 Decimals 
                 Numeric 
                 This option forces the numeric format to use the 
               
               
                   
                   
                 listed number of decimal places. 
               
               
                 GPIB Address 
                 A/N 
                 1-31 
               
               
                 Rcv Buffer 
                 Numeric 
                 Max number of characters to Receive. When in 
               
               
                 Size 
                   
                 doubt use 100. 
               
               
                 Exponents 
                 Drop 
                 Selects the desired exponential format of the result. 
               
               
                   
                 Down 
               
               
                   
                 Box 
               
               
                 Formatted 
                 Numeric 
                 The results from the entered data. 
               
               
                 Results String 
               
               
                 Resulting 
                 Numeric 
                 The results from the entered data. 
               
               
                 Numeric Value 
               
               
                 Continuous Run 
                 Check 
                 When this check box is selected the machine will continually 
               
               
                   
                 Box 
                 run the fetch command option. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
             
               
                 TABLE 63 
               
               
                   
               
               
                 Name 
                 Type 
                 Description 
               
               
                   
               
             
            
               
                 Send Command 
                 Button 
                 Sends a command. 
               
               
                 Fetch 
                 Button 
                 Sends a command and displays the returned value. 
               
               
                 Command 
               
               
                 Configure Port 
                 Button 
                 Opens the Com Port Setup page. 
               
               
                 &gt;Direct 
                 A/N 
                 Indicates the response received from the fetch 
               
               
                 Response&lt; 
                   
                 operation. 
               
               
                 Cut Left 
                 Numeric 
                 The number of characters to remove from the response 
               
               
                   
                   
                 string. An entry of −1 indicates no characters are 
               
               
                   
                   
                 removed. 
               
               
                 Keep Left 
                 Numeric 
                 The number of characters to retain after the “cut left”. 
               
               
                   
                   
                 An entry of −1 indicates all characters are retained. 
               
               
                 Decimals 
                 Numeric 
                 This option forces the numeric format to use the listed 
               
               
                   
                   
                 number of decimal places. 
               
               
                 Exponents 
                 Drop 
                 Selects the desired exponential format of the result. for 
               
               
                   
                 Down 
                 a list of valid values. 
               
               
                   
                 Box 
               
               
                 Formatted Results 
                 Numeric 
                 The results from the entered data. 
               
               
                 String 
               
               
                 Resulting Numeric 
                 Numeric 
                 The results from the entered data. 
               
               
                 Value 
               
               
                 Continuous Run 
                 Check 
                 When this check box is selected the machine will 
               
               
                   
                 Box 
                 continually run the fetch command option until 
               
               
                   
                   
                 halted by the user. 
               
               
                   
               
            
           
         
       
     
     
       
         
           
               
               
               
               
             
               
                   
                 TABLE 64 
               
               
                   
                   
               
             
            
               
                   
                 μVDC Range 
                 VAC Range 
                 mA DC Range 
               
               
                   
                 mVDC Range 
                 μOhm Range 
                 A DC Range 
               
               
                   
                 VDC Range 
                 mOhm Range 
                 μA AC Range 
               
               
                   
                 VDC Linearity 
                 kOhm Range 
                 mA AC Range 
               
               
                   
                 mVAC Range 
                 MOhm Range 
               
               
                   
                 mVAC Range 
                 μA DC Range