Source: https://www.scribd.com/document/141124364/SMRT1-DS-en-V02
Timestamp: 2019-04-24 06:23:00+00:00

Document:
function buttons are provided to quickly and easily select the desired test function. Tests results can be saved to the PowerDBTM ONBOARD for download to a memory stick to transfer or print test reports. For full automatic testing the SMRT1 may be controlled by Megger Advanced Visual Test Software (AVTS). AVTS is a Microsoft® Windows® XP®/ Vista™/7 compatible software program designed to manage all aspects of protective relay testing using the new Megger SMRT. APPLICATIONS The current channel is rated for 30 Amps @ 200 VA continuous, up to 60 Amps @ 300 VA for short durations. It has a unique flat power curve from 4 to 30 Amps that insures maximum compliance voltage to load at all times. With a high compliance voltage of 50 Volts the SMRT1 has the capability to test high impedance overcurrent relays. The voltage channel can provide a variable output of 0- 30/150/ 300 Volts at 150 VA of output power, and has a unique flat power curve from 30 to 150 Volts insuring maximum output power to the load at all times. With the voltage channel converted to current, it can perform minimum operating point, slope, and timing on current differential relays, including harmonic restraint transformer differential relays (which can be tested one phase at a time). It is also designed to operate in conjunction with other SMRT family units. Using the Ethernet ports, SMRT1 is literally a “plug-and-play” unit, where voltage and current outputs can be seamlessly synchronized with other SMRT units voltage and current outputs for testing more complex relays like three phase directional power, distance, loss of excitation, or 6, 9 up to12 phase current test applications.
6. 37/76 DC Under/Over Voltage/Current 40 46 46N 47 50 51 55 60 67 67N 78 79 81 85 87 91 92 94 Loss of Field Phase Balance Current Negative Sequence Overcurrent Phase Sequence Voltage (Open Delta) Instantaneous Overcurrent Time Overcurrent Power Factor Voltage/Current Balance (Open Delta) Directional Overcurrent Ground Directional Overcurrent Out of Step Reclosing Frequency Carrier or Pilot Wire Differential Voltage Directional (Open Delta) Voltage and Power Directional (Open Delta) Tripping ✓ + ✓ ✓ ✓ ✓ ✓ ✓ ✓ + + ✓ +	Requires additional SMRT1 for each + 1	For operating times less than 1.	Binary Input: Rated 5 to 300 V AC/DC 3. Customers who want to test a panel of relays at one time find it impossible using lower VA rated voltage sources. 4. New PowerV™ Voltage Amplifier High Power Output – The SMRT provides a new higher VA power output on the voltage channel at the lower critical test voltages (from 30 to 150 Volts).	Current Channel: 0 – 30 Amps at 200 VA continuously. or for connecting to the IEC 61850 substation bus.	POWER ON/OFF: Switch illuminates when power is on. 2. The current amplifier delivers maximum compliance voltage to the load constantly during the test. 8.	PC/IN: Ethernet Port is the primary PC connection port.5 seconds. This insures better test results.	Voltage Channel: 0 .	Binary Output: Rated for 300 V at 8 Amps. High Output Current – Provides up 30 Amps at 200 VA per phase continuous for timing tests. 5. FEATURES AND BENEFITS Constant Power Output – New higher powered VoltageCurrent amplifiers. and can provide up to 60 Amperes at 300 VA for testing instantaneous overcurrent relays. convertible to current rated for 5 Amps at 150 VA continuously. and range changing is done automatically under load.300 V at 150 VA. 15 A at 120 VA for short durations. For longer trip times output is rated for 35 Amps continuous with convertible channel in parallel.	BlueTooth: Bluetooth® provides wireless control.	61850/OUT: Ethernet Port may be used to interconnect multiple SMRT units together for synchronous multi-unit operation. 7. and saves time by not having to turn the outputs off to change ranges.SMRT1 Single Phase Relay Test System APPLICATIONS SELECTION GUIDE Protective Relays by IEEE Device # 2 21 21 21 24 25 Time Delay Distance Single Phase Distance Open Delta Distance Three Phase wye Volts/Hz Synchronizing SMRT1 1 2 3 ✓ ✓ + ++ ✓ + ✓ ✓ + ✓ ✓ ✓ ✓ + Up to 75 Amps Up to 75 Amps1 8 5 6 4 7 27/59 Under/Over Voltage 32 32 Directional Power Single Phase Directional Power Three Phase (Open Delta) 1. 2 . Constant power output in many cases eliminates the need to parallel or series current channels together to test high burden relays. This port may also be used for connecting to other SMRT units. up to 60 Amps at 300 VA for short durations.
25. 33. Steady-State and Dynamic testing capability – The SMRT1 provides.4 V rms Continuous 6. Perform End-to-End tests – Using AVTS software and a GPS satellite receiver. with an extended operating range up to 60 Amperes at 300 VA rms. the SMRT can use virtually any standard source in the world.00 V rms 90 Cycles peak) peak) peak) peak) Current Amplifier .SMRT1 Single Phase Relay Test System Convertible Voltage Channel – Provides second current source for testing single phase current differential relays.0 V rms Continuous 50.67 V rms Continuous 5. 3 . 150. 180.5 Amps 1 Megger reserves the right to change product specifications at any time. Current Amplifier Output Power Curve BlueTooth – Optional Bluetooth provides more flexibility. for more complex power system simulations. and up to 75 Amps short time. Output current and voltage sine waves are generated digitally – Outputs do not vary with sudden changes in input voltage or frequency. using Boolean logic. Digital binary input and output – The programmable binary input. and lockout are easily tested. AC Voltage Output Outputs are rated with the following Ranges: Output Volts Power 30 Volts 150 VA 150 Volts 150 VA	300 Volts 150 VA	DC 150 Watts Duty Cycle: Continuous Max I 5 Amps See “PowerV” 0. timing. Immediate error indication – Audible and visual alarms indicate when amplitude or waveforms of the outputs are in error. All outputs are regulated so changes in load impedance do not affect the output. Eliminates uncertainty with setting values. which increases test accuracy and reduces testing time. with metered values what you see is what you get. 250. either through manual control or computer control. Circuit breaker simulator – Binary output provide programmable normally closed or normally open contacts to simulate circuit breaker operation for testing reclosing relays. 50. It is also used when chaining multiple SMRT units together. Popular test frequencies such as 16. This provides stable outputs not affected by sudden changes in the mains source. 50/60 Hz. 100. Sequence of operation.Extended Power Range The SMRT current amplifier provides a unique flat power curve from 4 to 30 Amperes to permit testing of electromechanical high impedance relays. in conjunction with the SMRT IEC 61850 Ethernet port. This includes programmable waveforms with harmonics. Output Current 1 Ampere 4 Amperes 15 Amperes 30 Amperes 60 Amperes DC 200 Watts Power 15 VA	200 VA (282 200 VA (282 200 VA (282 300 VA (424 Max V / Duty Cycle 15. High resolution and accuracy – Metered outputs provides extremely high accuracy needed for testing a wide variety of devices.0 V rms Continuous 13. Output Current Output power ratings are specified in AC rms values and peak power ratings. Performs transient tests – Perform acceptance or troubleshooting tests by replaying digitally recorded faults or EMTP/ATP simulations in the IEEE. or it may be used to connect to the IEC 61850 substation bus. both steady-state and dynamic testing of protective relays. A wireless interface between the PC and SMRT. 60. and programmable output provide timing and logic operations in real-time with the output voltage and currents. Universal input voltage – Operation from 90 to 264 VAC. 125. 50/60 Hz. The Binary Input can be programmed. Multi-purpose test system saves time and money.C37. SPECIFICATIONS1 Input Power 100 to 240 Volts (± 10%) AC. provides the isolation required for a secure substation access interface between the SMRT and the IEC 61850 substation network. 1Ø. Two Ethernet ports – PC/IN Ethernet Port is the primary PC connection port.111.120. 700 VA Outputs All outputs are independent from sudden changes in line voltage and frequency. the SMRT performs satellitesynchronized end-to-end dynamic multi-state or playback transient COMTRADE files either for commissioning or troubleshooting tests. including harmonic restraint transformer differential relays. COMTRADE Standard format. Parallel with main current channel to increase output current to 35 Amps continuous.66. Wide-ranging output frequency – The output frequency of the current and voltage outputs can be set for any frequency from dc to 1 kHz. 300 and 400 Hz are easily set and controlled. The 61850/OUT Ethernet Port is primarily used to interconnect multiple SMRT units together for synchronous multi-unit operation. and other high burden applications.
typical. 300V AC Current Amplitude Accuracy: ±0. as a sequence of events recorder. The Timer function displays in Seconds or Cycles.000 Hz Output amplifiers can provide transient signals with a range of DC to 10 kHz for transient playback using COMTRADE files.5 mA whichever is greater Resolution: .15% reading + 0.. optional STVI touch screen.02% range typical.9 (Auto Ranging) Accuracy: ±0.02% range typical. In addition. Ranges DC 0. trip. the binary input controls enable the user to perform logic AND/OR functions. ±0. Imax: 8 amps.05% reading + 0.. with the following range and resolution: Seconds: 0.01 Measurements: AC RMS Ranges: 30. The binary output can be configured to change state based on binary input logic.001 Hz Frequency Accuracy: 2. continuity light is provided for the input gate. and Time may be simultaneously displayed on the large. ±0.05 % reading + 0.001 Measurements: AC RMS Range: 5. DC Volts or DC Amperes.01 Measurements: RMS Ranges: 30A Convertible Source in AC Current Mode Accuracy: ±0. Resolution*: .00 to 359. sine wave.25° max at 50/60 Hz Frequency The output modules provide a variable frequency output with the following ranges and accuracy. galvanically isolated. 60A DC Voltage Amplitude Accuracy: 0.05% range or ±12. ±0. half waves. 2000 VA max. exponential decays. ±0.005% of reading from 0 to 50° C maximum 4 .1% range typical. ±0.111 COMTRADE standard format. 150. The AC and DC outputs display the approximate voltage/current output prior to initiation of the outputs.0 Vrms Continuous 15 Amperes 120 VA 8. or 0.0 Vrms 90 Cycles Phase Angle Ranges: 0. color TFT LCD.05% range maximum Resolution: . Metering Measured output quantities such as AC Amperes. AC Voltage Amplitude Accuracy: ±0. at 50/60 Hz Maximum Total Harmonic Distortion Less than 0.00 degrees Accuracy: ±0. sine wave with percent harmonics at various phase angles.99 degrees.001/. breaking capacity DC Rating: 300 V max.05% reading + 0.5 ppm typical 25 ppm 0° to 50° C.15 % reading + 0.01 Measurements: RMS Ranges: 30.9 (Auto Ranging) Cycles: 0. AC Rating: 400 V max.15% reading + 0. 15A “PowerV” Voltage Amplifier . Voltage Amplifier in Current Mode: The voltage amplifier is convertible to a second current source with the following output capability. or Clock Wise rotation. 0. and conditionally control the binary output relay to simulate circuit breaker.SMRT1 Single Phase Relay Test System Binary Input – Start/Stop/Monitor Gate To monitor operation of relay contacts or trip SCR.001 to 1000.1% typical.05 % range maximum Resolution: . ±2 least significant digit. output relay contact to accurately simulate relay or power system inputs to completely test relays removed from the power system.0001 to 99999.02° typical. The binary output simulates normally open / normally closed contacts for testing breaker failure schemes. 300V DC Current Amplitude Accuracy: ±0.05% range maximum Resolution: . Counter Clock Wise. which conform to the IEEE C37. Input Rating: up to 300 V AC/DC Binary Output Relay SMRT1 has an independent. Output power ratings are specified in rms values and peak power ratings. ±0. Imax: 8 amps.0001/. periodic transient waveforms from digital fault recorders.001% of reading.25% range maximum Resolution: .Extended Power Range The SMRT voltage amplifier provides a flat power curve from 30 to 150 Volts in the 150V range to permit testing of high current applications such as panel testing.01 to 99999.05% reading + 0. reclose and carrier control operation in real-time.001/. Output Current Power Max V Duty Cycle 5 Amperes 150 VA (212 peak) 30. In addition to serving as wet/dry contacts the Binary Input may be programmed to trigger binary output sequence(s). square waves with variable duty cycles. Upon sensing continuity the lamp will glow. 80 W Response Time: < 10ms Waveform Generation Each output channel can generate a variety of output waveforms such as: DC. AC Volts.15% reading + 0. relays with waveform recording capability or EMTP/ATP programs.00 to ±180. 2% maximum at 50/60 Hz Timer The Timer-Monitor Input is designed to monitor and time-tag inputs.02 % range typical.01 Measurements: AC RMS Ranges: 30. 150.02% range typical.
ASPEN OneLiner™ or Electrocon CAPE™ SS1 File Converter for dynamic testing.5W x 2. FCC Subpart B of Part15 Class A Immunity: EN 61000-4-2/3/4/5/6/8/11 Protection Voltage outputs are protected from short circuits and thermally protected against prolonged overloads. review and printing whenever needed.88D cm) Rack Mount Enclosure 19W x 2.05 g rms) Transit Drop: MIL-RFP-28800F (10 drops. Online Ramp control (for automatic ramping of voltage. See AVTS bulletin for more information. Test results may be exported directly to Microsoft Word. Optional enclosure for 19 inch rack mount is available.5W x 6.85 lb.9 lb. 20 cm. These connectors are used to amplify an external analog signal using the SMRT amplifiers. (47. and easy to use programming Tools for creating and editing test modules.88D cm) Weight Standard Enclosure: 8.75D in. See AVTS bulletin for more information. Rack mount enclosure includes two BNC connectors on the back panel. AVTS Advanced The AVTS Advanced version includes all the features in AVTS Basic plus the powerful Test Editor. Dimensions Standard Enclosure 13. Communication Interfaces Ethernet (2) Bluetooth (optional) SOFTWARE AVTS – STVI Basic Every unit comes with AVTS Basic software and the PC version of the STVI Basic software packages. 5 . See AVTS bulletin for more information. (4. EN 61000-3-2/3. and Waveform Digitizer to digitize scanned waveforms of electromechanical over current time curves. which can also provide the necessary information needed for system reliability audits. See STVI bulletin for more information.75W x 6H x 16. Non-condensing Unit Enclosures The SMRT1 unit comes housed in a rugged.4H x 6. in conjunction with the Megger GOOSE Configurator (MGC) software.5H x 21. AVTS Professional The AVTS Professional version includes all of the features of the Basic and Advanced versions plus some other powerful test tools and features. can be used in the testing or commissioning of IEC 61850 compliant devices. 2. IEC Enclosure Rating IP20. It includes the DFR Waveform Viewer. AVTS Basic version includes Online Vector control (for single and multi-state timing tests).6H x 8. AVTS software includes a database for saving test results. Application of ±10 Volts Peak will provide Full Scale output from the selected output. Modbus communication test capability for fully automatic tests. without transit case) EMC Emissions: EN 61326-2-1. Current outputs are protected against open circuits and thermally protected against prolonged overloads. phase angles or frequency) and Online Click-On-Fault (for dynamic tests of impedance relays).SMRT1 Single Phase Relay Test System Temperature Range Operating: 32 to 122° F (0 to 50° C) Storage: -13 to 158° F (-25 to 70° C) Relative Humidity: 5 . One-Touch™ Test.90% RH. Dynamic Control (includes dynamic end-to-end testing capability. The PC version of the STVI software includes the ability to bring all STVI test data (from other STVI units) into file folders for retrieval. See AVTS bulletin for more information.75D in. (4 kg) Rack Mount Enclosure: 10. current.9 kg) Conformance Standards Safety: EN 61010-1 Shock: MIL-PRF-28800F (30g/11ms half-sine) Vibration: MIL-RFP-28800F (5-500Hz. (33. lightweight enclosure. metal. IEC 61850 GOOSE The SMRT with the GOOSE enabled. and waveform recording capability).
•	E Option . Enter 0 for the unit without Test Leads. The rack unit will come in a metal enclosure with 19 inch rack mount hardware installed. enter the number 1 for the unit to come with the Bluetooth option installed. Enter the number 1 for the unit to come with the IEC 61850 option enabled. Enter 0 for without. the IEC 61850 feature needs to be enabled. North American Line cord. Enclosure Option: The options are S for Standard. the unit will come with one of the following.International color coded wires (light blue. Continental Europe with CEE 7/7 Schuko Plug Line cord. Bluetooth Option: For customers who wish to have a wireless control of the SMRT unit. •	A Option – NEMA 5-15 to IEC 60320 C13 connectors. Enter 0 for the unit without IEC 61850 enabled. brown and green with yellow stripe) insulation jacket stripped ready for male connector with IEC 60320 C13 connector. International color coded wire Ethernet crossover cable for interconnection to PC. Unit comes with hardware for rack mount in 19 in rack IEC61850 Option 1 = With IEC 61850 GOOSE 0 = Without Power Cord Option A = North American Power Cord I = International Power Cord E = Continental Europe Power Cord Smart Touch View Interface Option 1 = With STVI 0 = Without Common Returns Option F = Floating Ungrounded Common Return Bluetooth Option 1 = With Bluetooth 0 = Without DESCRIPTION OF SOFTWARE OPTIONS Included Software AVTS Basic with STVI Application CD AVTS Basic with IEC 61850 Megger GOOSE Configurator. Reserved for future use. Power Cord Option: Customers can choose which type of power cord they want the unit to come with. and STVI Application CD AVTS Professional with STVI Application CD AVTS Professional Test with IEC 61850 Megger GOOSE Configurator. •	I Option . or enter the number 0 for without. Line cord. 210cm (7 ft. It is F for floating returns. Smart Touch View Interface Option: Enter the number 1 for the unit to come with the STVI. and STVI Application CD Software Options AVTS Advanced with STVI Application CD AVTS Advanced Test with IEC 61850 Megger GOOSE Configurator. for Standard. UL & CSA approved for countries with NEMA outlets. Enter S. Base Unit Options: SMRT1 does not have any Base Unit options. Common Returns Option: The SMRT1 unit has only one option. for the return terminals associated with each output channel. CE marked. Future: 0. rugged metal field type enclosure. and STVI Application CD 81570 1001-106 81571 1002-102 Part Number 81302 1002-103 DESCRIPTIONS OF HARDWARE OPTIONS Voltage/Current Module: The SMRT1 by default the first digit will always be 1.) long (Qty. at this time. and R for rack mount.Depending on the style number. Test Leads Option: Enter the number 1 for the unit to come with Test Leads. DESCRIPTION Included Standard Accessories Power Cord . 1 ea) Instruction manual CD 620000 50425 15065 620094 80989 Part Number 6 . IEC 61850 Option: The SMRT1 in conjunction with the Megger GOOSE Configurator (MGC) software can be used in the testing or commissioning of IEC 61850 compliant devices. In order for the SMRT1 to be able to subscribe as well as publish GOOSE messages. By default this digit is N for None.SMRT1 Single Phase Relay Test System ORDERING INFORMATION STYLE NUMBER IDENTIFICATION Model SMRT1 Voltage/Current Modules Default = 1 for a SMRT1 unit Reserved for Future Use 1 0 N F Test Leads Option 1 = With Leads 0 = Without Leads Enclosure Options S = Standard field enclosure R = Rack-mount Option.CEE 7/7 “Schuko” plug to IEC 60320 C13 connector is CE marked.
Intuitive menu screens and buttons are provided to quickly and easily select the desired test function. Additional Software Options AVTS Advanced with STVI Application	Part No.SMRT1 Single Phase Relay Test System Table of Accessories Accessories are supplied with the selection of the Test Leads Option and/or the STVI Option. These playback test files can also be used in end-to-end tests to recreate the transient event and evaluate the protection scheme. Also included is the One-Touch Test Editor Control Tool for fully automatic testing of microprocessor based relays using VB script files or Modbus communications for automatic download of settings. Online Ramp and Online Click-On-Fault controls.: 81302 AVTS Basic includes Online Vector. pulse ramp. Frequency and Distance relays. Optional Accessories Descriptions Accessory Carry Case: Use to carry power cord.7”) long. 4 pr. Optional STVI and test leads. Additional run keys can be purchased seperately. save and execute relay specific test modules. use with test leads up to 1000 V/ 20 Amps CAT II Jumper Lead: Used to common returns together on units with floating ground returns. 4 ea. Part No.: 81571 Qty. The PC version of the STVI software includes the ability to bring all STVI test data (from other STVI units) into file folders for retrieval and review whenever needed. or Ramp. The user can select from a variety of test options including manual control using the cursor up down arrows or use the mouse control wheel to vary outputs. Test Leads and Accessories can be ordered individually.5 cm (5”) long. Ethernet cable. Each copy of the PC version of the STVI software is licensed for running on one PC. Lug adapter.1 mm. or perform relay specific timing tests using the Timing Test Screen. Test files created in Professional can be used with Advanced Test and Basic.111 COMTRADE Standard files from digital fault recorders and microprocessor based relays. Overcurrent. 2001-394 The powerful STVI screens can be run directly from a PC providing both manual and automatic test capabilities. 1 ea. one black. black. or have setting values displayed. 200 cm (78. to automatically download relay settings. red. If desired. use with test leads up to 1000 V/ 20 Amps CAT II Lug adapter.1 mm. An automatic ramp. Part No. Part No. or dropout tests as well as timing and multi-state dynamic tests. and automatically test all the measuring elements within the relay based upon those settings. with the ability to import. it also includes SS1 File Converter for ASPEN and CAPE dynamic test files. End-to-End DFR Playback test macros and basic programming Tools for creating and editing test modules. A vector graph indicates the relative phase angles of all of the outputs. Pulse Ramp. 4 ea. The Waveform Digitizer feature is also included in the Professional Test version of AVTS. 12. black. 7 . Voltage. The user may select to have all output amplitudes metered to provide real time verification of all of the selected outputs. The Online Click-On-Fault tool is used to automatically determine the reach characteristics of single or multi-zone Distance relays using shot for single point tests. It includes the capability to extend the prefault data as well as start the timer associated with the event to time relay operation. 1 ea. 600 V. one red. In addition a dynamic sequence test includes trip and reclose up to 9 operations. It provides tools to create digital time curves for virtually any electromechanical relay time curve (that do not fit a time curve algorithm).: 81570 Qty. 4. The DFR Waveform Viewer and Playback tools are used for viewing and analyzing IEEE C37. Part Number 2001-573 Descriptions of Software Included Software – Every unit comes with AVTS Basic and the PC version of the STVI Basic Test software packages	AVTS Basic with STVI Application Software (PC Version)	Part No. The DFR Waveform Viewer includes tools to recreate the analog and digital channels for playback into protective relays for troubleshooting or evaluation. 32 Amps CAT II STVI and/or Test Leads Option Qty. AVTS Professional with STVI Application	Part No. Test files created in Advanced Test can be used with AVTS Basic. Professional Test includes all of the features of AVTS Advanced Test version plus the following additional specialized test tools. In addition. It can even be used for digitizing scanned waveforms from a light-beam chart recorder. which includes the Dynamic Control (with dynamic end-to-end test capability. and Recorder features) for developing sequential tests for virtually any function or measuring element within digital relays. The online tools of Vector and Ramp provide automatic pickup. use with voltage / current outputs. Sleeved Test Leads. 684004 Qty. Part Number 684005 AVTS Advanced includes all of the features of AVTS Basic in addition to the powerful Test Editor and test editor tools. Includes enhanced Relay Test Wizards for. binary search or pulse ramp binary search is built in to determine pickup or drop out of relay contacts. 2001-487 Test Leads Option Qty. or parallel of current channels. Sleeved Pair of Test Leads: Keeps the test leads in pairs and from getting entangled. or Binary Search tools along user defined search lines. 600 V. Differential. 32 Amperes CAT II Cable/Spade Lug Adapter (Small): Small lug fit most new relay small terminal blocks. see description and part numbers below. 4. The Manual Test Screen power-up preset default values maybe automatically set from the user defined configuration screen. Jumper lead.
50 cm (20”) long. Flexible test lead adapter. or STATES® Company FTP10 or FTP14 Test paddles. or use to perform interoperability high-speed shared I/O tests between multiple IED’s. Order the appropriate number required. black. black. 32 Amperes CAT II. Float and UTC datasets. 200 cm long (78. 200 cm long (78. use with test leads up to 1000 V/ 32 Amps CAT III. red. Lug adapter. Description Part No.7”) 600 V/32 Amps CAT II. 32 Amperes CAT II.15 A fuse protection. use with voltage/current output . black.7”). or binary I/O.SMRT1 Single Phase Relay Test System IEC 61850 Megger GOOSE Configurator Software The Megger GOOSE Configurator (MGC) provides easy to use tools for testing relays and substations using the IEC 61850 protocol. and/or capture GOOSE messages directly from the substation bus. use with test leads up to 600 V. 200 cm long (78. red. Additional Accessories (Not Included in the SMRT1 Test Leads Option) Additional Optional Test Leads and Accessories can be ordered individually. use with test leads up to 1000 V/20 Amps CAT II. use with test leads up to 1000 V/20 Amps CAT II. Retractable Sleeve Test Lead. Use the MGC Merge feature to compare imported SCL and captured GOOSE messages to verify all GOOSE messages needed to perform tests. Alligator/Crocodile Clip: Excellent for test connections to terminal screws and pins where spade lugs cannot be used. black.7”). 500395 8 . 50 cm (20”) long. This provides both manual and automatic testing of the relay using either the STVI or AVTS software. in-line 500 mA fuse protection. After the appropriate assignments of binary inputs and outputs have been made. black. Individual (Non-Sleeved) Test Leads: Excellent for widely separated individual terminal test connections. The configurator provides relay test engineers and technicians the capability to import parameters from configuration files in the Substation Configuration Language (SCL) format. use with test leads up to 1000 V/32 Amps CAT III. 90001-843 90001-844 In-Line Fused Test Lead: Use with high speed binary outputs 5 or 6 (“P” Option) to protect for accidental switching of currents higher than 1 Amp. 6. see Descriptions of Software Options above. red. 568025 Cable/Spade Lug Adapter (Large): Large spade lug fits older relay terminal blocks. Use them to configure the SMRT to subscribe to preselected GOOSE messages by assigning the data attributes to the appropriate binary inputs of the SMRT.2 mm. 684006 684007 Flexible Test Lead Adapter: Use with rail-mounted terminals or screw clamp connections where spade lugs and crocodile/alligator clips cannot be used. 568026 Test Lead. Test Lead. Alligator clip.7”) 600 V/32 Amps CAT II. Alligator clip. use with test leads up to 1000 V/32 Amps CAT III. see description and part numbers below. In-Line Fused Test Lead: Use with (“P” Option) Battery Simulator output to protect for accidental connection to substation battery. Advanced or Professional versions of AVTS Software. use with test leads up to 600 V. red. Use the configurator to assign the appropriate binary outputs of the SMRT to publish GOOSE messages simulating circuit breaker status. use with test leads up to 1000 V/32 Amps CAT III. red. All imported GOOSE messages will be unconfirmed messages. in-line 3. in-line 100 k Ohm resistor. use with voltage/current output. black. The MGC provides mappings of Boolean and Bit Strings and/or simulation of STRuct. In-Line Resistor Test Lead: Use with old solid state relays with “leaky” SCR trip gates. ABB or General Electric test plugs with screw down terminals. 1. Lug adapter. Use standard test modules in AVTS to perform automatic tests.2 mm. Use the Dynamic Control in AVTS Advanced or Professional to perform high speed trip and reclose tests. 200 cm long (78. Integer/Unsigned. 620143 620144 Test lead. blue. 90001-845 Flexible Test Lead Adapter with Retractable Insulated Sleeve: Use for connection to old style non-safety sockets with retractable protective sleeve on one end. Description Part No. Only captured messages are confirmed messages due to the Capture feature of the MGC. The following accessories and part numbers are in quantities of 1 each. Test lead.8 mm male pin. or binary I/O. the test file can be saved for reuse. It is an optional software tool available with Basic. 6. 684002 684003 Test lead. Retractable Sleeve Test Lead.
ISO STATEMENT Registered to ISO 9001:1994 Reg no. Johannesburg SOUTH AFRICA and Conjure THAILAND. as well as the standard top connections for test leads. Adapter can provide convenient parallel test connections of test currents to two terminals at one time. Use with test leads up to 600 V. while the custom designed foam padded inside provides protection while in transit for the SMRT1. Description Part No. 32 Amperes CATII. Toronto CANADA.com Megger is a registered trademark . Test paddle features knobs which also serve as insulated Ø 4 mm rigid socket accepting spring loaded Ø 4 mm plugs with rigged insulating sleeve. Tested and certified to MIL-SPEC Standards for impact.SMRT1 Single Phase Relay Test System Description Part No.8 pounds (2. Mumbai INDIA. EMS 61597 SMRT1_DS_en_V02 www. Trappes FRANCE. Rugged Carry Case (1ea) 2002-468 Test paddle attachment provides an additional 10 insulated connection points for front connection. Sydney AUSTRALIA. 32 Amperes CAT II. With a SMRT1 and STVI it is small and light enough to hand carry on virtually all major commercial airlines. or retractable sleeve.6 kg). Rugged Carry Case: The fabric covered hard sides protects the unit from light rain and dust.megger. Use with test leads up to 600 V. The case is small. Q 09250 Registered to ISO 14001 Reg no. STVI-2. Kingdom of BAHRAIN. TPA10 UK Archcliffe Road Dover CT17 9EN England T +44 (0) 1304 502101 F +44 (0) 1304 207342 UNITED STATES 4271 Bronze Way Dallas TX 75237-1088 USA T 800 723 2861 (USA only) T +1 214 333 3201 F +1 214 331 7399 OTHER TECHNICAL SALES OFFICES Norristown USA. and weighs only 5. and test leads. FTP10 STATES® 10 Pole Test Paddle Attachment: Use with STATES FTP10 Test Paddle.

References: V. 
 V. 
 V. 
 V. 
 V. 
 V.