Source: http://slideplayer.es/slide/158075/
Timestamp: 2017-03-29 05:37:49+00:00

Document:
Instrumentación Virtual con LabVIEW - ppt descargar
Publicada porMargaretta Verdugo
Presentación del tema: "Instrumentación Virtual con LabVIEW"— Transcripción de la presentación:
Objetivos del Curso Comprender los componentes de un instrumento virtual Introducir LabVIEW y las funciones comunes de LabVIEW Crear una subrutinas en LabVIEW Trabajar con Arreglos, Clusters y Estructuras Desarrollo en arquitecturas de programación. Publicar VIs en el navegador Esta es una lista de los objetivos del curso Este curso le preparara para lo siguiente: • Utilizar LabVIEW para crear aplicaciones. • Entender paneles frontales, diagramas de bloque, íconos y paneles de conexión. • Incorporación de funciones de LabVIEW. • Crear y guardar programas en LabVIEW para que los pueda usar como subrutinas. • Crear aplicaciones que usen dispositivos DAQ. Este curso no describe lo siguiente: • Teoría de programación • Cada función u objeto encontrada en LabVIEW. • Teoría análoga-a-digital (A/D)
El Enfoque de la Instrumentación Virtual.Virtual Instrumentation For more than 25 years, National Instruments has revolutionized the way engineers and scientists in industry, government, and academia approach measurement and automation. Leveraging PCs and commercial technologies, virtual instrumentation increases productivity and lowers costs for test, control, and design applications through easy-to-integrate software, such as NI LabVIEW, and modular measurement and control hardware for PXI, PCI, USB, and Ethernet. With virtual instrumentation, engineers use graphical programming software to create user-defined solutions that meet their specific needs, which is a great alternative to proprietary, fixed functionality traditional instruments. Additionally, virtual instrumentation capitalizes on the ever-increasing performance of personal computers. For example, in test, measurement, and control, engineers have used virtual instrumentation to downsize automated test equipment (ATE) while experiencing up to a 10 times increase in productivity gains at a fraction of the cost of traditional instrument solutions. Last year 25,000 companies in 90 countries invested in more than 6 million virtual instrumentation channels from National Instruments. © National Instruments Corporation 3	Introduction to LabVIEW Hands-On
Aplicaciones de la Instrumentación VirtualDiseño Procesamiento de Señal e Imagen Programación Integrada de Sistemas (PC, DSP, FPGA, Microcontroladores) Creación de Prototipos y Simulación Control Control Automático y Sistemas Dinámicos Mecatrónica y Robótica Programación de PAC´s Mediciones Circuitería y Electrónica Medidas e Instrumentación Una Sola Plataforma de Desarrollo Gráfico Diseño Prototipo Desarrollo Virtual Instrumentation Applications Virtual instrumentation is applicable in many different types of applications, starting from design to prototyping and deployment. The LabVIEW platform provides specific tools and models to solve specific applications ranging from designing signal processing algorithms to making voltage measurements and can target any number of platforms from the desktop to embedded devices – with an intuitive, powerful graphical paradigm. With version 8, LabVIEW scales from design and development on PCs to several embedded targets from ruggedized toaster size prototypes to embedded systems on chips. LabVIEW streamlines system design with a single graphical development platform. In doing so, LabVIEW encompasses better management of distributed, networked systems because as the targets for LabVIEW grow varied and embedded, you will need to be able to more easily distribute and communicate between various LabVIEW code pieces in your system. © National Instruments Corporation 5	Introduction to LabVIEW Hands-On
Sistema de Desarrollo Gráfico de LabVIEWAmbiente de Programación Gráfico Compilación de código para múltiples SO y dispositivos. Utilizable en una amplia gama de aplicaciones National Instruments LabVIEW is an industry-leading software tool for designing test, measurement, and control systems. Since its introduction in 1986, engineers and scientists worldwide who have relied on NI LabVIEW graphical development for projects throughout the product design cycle have gained improved quality, shorter time to market, and greater engineering and manufacturing efficiency. By using the integrated LabVIEW environment to interface with real-world signals, analyze data for meaningful information, and share results, you can boost productivity throughout your organization. Because LabVIEW has the flexibility of a programming language combined with built-in tools designed specifically for test, measurement, and control, you can create applications that range from simple temperature monitoring to sophisticated simulation and control systems. No matter what your project is, LabVIEW has the tools necessary to make you successful quickly. Introduction to LabVIEW Hands-On 4	ni.com
Plataformas IntegradasHigh-Speed Digitizers High-Resolution Digitizers and DMMs Multifunction Data Acquisition Dynamic Signal Acquisition E/S Digital Control de Instrumento Counter/ Timers Visión de Máquina Motion Control Distributed I/O and Embedded Control PC Portátil PDA PC de Escritorio Instrumentación Modular PXI Acondicionamiento de Señal y Conmutación Unidad Bajo Prueba Integrated Hardware Platforms A virtual instrument consists of an industry-standard computer or workstation equipped with powerful application software, cost-effective hardware such as plug-in boards, and driver software, which together perform the functions of traditional instruments. Virtual instruments represent a fundamental shift from traditional hardware-centered instrumentation systems to software-centered systems that exploit the computing power, productivity, display, and connectivity capabilities of popular desktop computers and workstations. Although the PC and integrated circuit technology have experienced significant advances in the last two decades, software truly offers the flexibility to build on this powerful hardware foundation to create virtual instruments, providing better ways to innovate and significantly reduce cost. With virtual instruments, engineers and scientists build measurement and automation systems that suit their needs exactly (user-defined) instead of being limited by traditional fixed-function instruments (vendor-defined). Introduction to LabVIEW Hands-On 6	ni.com
¿Programas en LabVIEW y otros lenguajes es los mismo?Pareciera que no, pero en realidad los planteamientos generales son los mismo: Definir el tipo de dato de las variables antes de usarlas. Cada variable depende de un solo tipo de datos. No todas las operaciones son aplicables a todos los datos. Tipo de dato: Conjunto de valores que puede asumir una variable. Existen dos tipos de variables: Las variables globales, aquellas a las que se puede acceder o llamar desde cualquier parte del programa y variables locales. Clase de datos: Tipo escalar (números enteros, reales, booleanas, etc.) y Tipo estructurado (arrays y clusters).
Tipo de datos en LabVIEW.
Sección I Términos de LabVIEW Componentes de aplicación de LabVIEWHerramientas de programación de LabVIEW Crear una aplicación en LabVIEW
Programas de LabVIEW se llaman instrumentos virtuales (VIs)Panel frontal Controles = entradas Indicadores = salidas Diagrama de bloque Programa de acompañamiento para el panel frontal Componentes “cableados” entre si Los Programas en LabVIEW son llamados instrumentos virtuales (VIs). Haga énfasis en que controles es igual a entradas, e indicadores es igual a salidas. Cada VI contiene tres partes principales: Panel frontal– Cómo el usuario interacciona con el VI. Diagrama de bloque – El código que controla el programa. Icono/Conector – Medios para conectar un VI con otros VIs. El panel frontal es utilizado para interaccionar con el usuario cuando el programa esta corriendo. Usuarios pueden controlar el programa,cambiar entradas, y ver datos actualizados en tiempo real. Haga énfasis en que los controles son usados como entradas - ajustando controles de deslizamiento para colocar un valor de alarma, encendiendo o apagando un switch, o parando un programa. Los indicadores son usados como salidas. Termómetros, luces, y otros indicadores indican valores del programa. Esto puede incluir datos, estados de programa y otra información. Cada control o indicador del panel frontal tiene una terminal correspondiente en el diagrama de bloques. Cuando un VI se ejecuta, los valores de los controles fluyen a través del diagrama de bloques, en donde estos son usados en las funciones del diagrama, y los resultados son pasados a otras funciones o indicadores.
Panel frontal de VI Icono Control Booleano Leyenda del gráficoBarra de herramientas del panel frontal Icono Control Booleano Leyenda del gráfico Gráfico de Forma de onda El panel frontal es la interfase del usuario con el VI. Usted construye el panel frontal con controles e indicadores,que son las entradas y salidas que interactúan con las terminales del VI, respectivamente. Los controles son botones, botones de empuje, marcadores y otro componentes de entradas. Los indicadores son las graficas, luces y otros dispositivos. Los controles simulan instrumentos de entradas de equipos y suministra datos al diagrama de bloques del VI. Los indicadores simulan salidas de instrumentos y suministra datos que el diagrama de bloques adquiere o genera. En esta imagen, el switch de poder (encendido o apagado) es un control booleano. Un valor booleano contiene ambos un valor verdadero o falso. El valor es falso hasta que el switch se presione.Cuando el switch se presiona, el valor se convierte en verdadero. El indicador del historial de temperatura es una grafica de forma de onda. Esta muestra valores múltiples. En este caso, la grafica señalara grados F versus tiempo (seg.). El panel frontal también contiene una barra de herramientas, y sus funciones serán discutidas después. Leyenda de la escala Leyenda de La grafica
VI Diagrama de bloque Barra de Herramientas Del diagrama de BloqueFuncion de dividir SubVI Terminal grafica Cableado de datos El diagrama de bloque contiene el código fuente grafico. Los objetos del panel frontal aparecen como terminales en el diagrama de bloque. Adicionalmente, el diagrama de bloque contiene funciones y estructuras incorporadas en las bibliotecas de LabVIEW VI. Los cables conectan cada uno de los nodos en el diagrama de bloques, incluyendo controles e indicadores de terminal, funciones y estructuras. En este diagrama de bloque, el subVI Temp llama a la subrutina la cual obtiene una temperatura desde una tarjeta de adquisición de datos (DAQ). Esta temperatura es graficada junto con el valor average de la temperatura en la grafica de forma de onda Temperature History. El switch de poder (Power) es un control booleano en el panel frontal el cual va a detener la ejecución de la estructura mientras( While Loop). La estructura mientras (While Loop) también contiene una función de tiempo para controlar que tan frecuentemente la estructura se repite. Estructura While loop Constante numerico Funcion de tiempo Terminal de control booleano
Paleta de controles y funcionesPaletas de control (Ventana del panel frontal) Paleta de funciones (Ventana de diagrama de bloque) Use la paleta de controles (Controls) para colocar los controles e indicadores en el panel frontal. La paleta de controles esta disponible solamente en el panel frontal. Seleccione Window»Show Controls palette o haga clic derecho en el espacio de trabajo en el panel frontal para desplegar la paleta de controles. Usted también puede desplegar la paleta de controles haciendo un clic derecho en una área abierta del panel frontal. Para desaparecer la paleta de controles presione el botón en la parte superior izquierda de la paleta. Use la paleta de funciones (Functions), para construir un diagrama de bloque. La paleta de funciones esta disponible solamente en el diagrama de bloque. Seleccione Window»Show Functions Palette o haga un click derecho en el espacio de trabajo del diagrama de bloque para desplegar la paleta de funciones. Usted también puede desplegar la paleta de funciones dando un click derecho en una área abierta del diagrama de bloques. Para desaparecer la paleta de funciones presione el botón en la parte superior izquierda de la paleta.
Paleta de HerramientasPaleta flotante Utilizado para operar y modificar objetos en el panel frontal y en el diagrama de bloques. Herramienta de selección automatica Herramienta de operación Herramienta de posicionamiento y redimensión Herramienta de etiquetado Herramienta de cableado Herramienta de menú (atajo) Herramienta de desplazamiento Herramienta de punto de paro Herramienta de prueba Herramienta para copia de color Herramienta para colorear Si la herramienta de selección automática esta habilitada y usted mueve el cursor sobre un objeto en el panel frontal o en el diagrama de bloque, LabVIEW automáticamente selecciona la herramienta correspondiente de la paleta de controles. Cambie a la herramienta de selección automática tecleando el botón de Selección de Herramienta Automática en la paleta de Herramientas. Utilice la herramienta de operación para cambiar valores o para seleccionar texto de un control. Utilice la herramienta de posicionamiento para seleccionar, mover o redimensionar objetos. La herramienta de posicionamiento cambia las formas cuando se mueve sobre una esquina de un objeto reajustable. Utilice la herramienta de etiquetado para editar textos y crear etiquetas libres. La herramienta de etiquetado se convierte en cursor cuando usted crea etiquetas libres. Utilice la herramienta de cableado para cablear objetos juntos en el diagrama de bloque.
Barra de herramientas de EstadoBotón de ejecución (Run) Botón de ejecución continua (Continuous Run) Cancelación de ejecución (Abort Execution) Botón de pausa/continuación Configuración de textos (Text Settings) Alineamiento de objetos (Align Objects) Distribución de objetos (Distribute Objects) Reordenamiento Redimensionamiento de objetos de panel frontal (Resize Objects) Botones adicionales en el diagrama de la barra de herramientas Botón de ejecución resaltada (Highlight Execution) Botón de entrada al ciclo (Step Into) Botón sobre (Step Over) Botón de salida del ciclo (Step Out) Pulse el botón de ejecución (Run) para ejecutar el VI. Mientras el VI se esta ejecutando, el botón de ejecución aparecerá con una flecha negra si es que el VI es un VI principal, lo que significa que no ha sido llamado por otro VI y por lo tanto este no es un subVI. Pulse el botón de ejecución continua (Continuous Run) para ejecutar el VI hasta que el botón de cancelación de ejecución o de pausa sea presionado. Usted también puede pulsar este botón nuevamente para deshabilitar la ejecución continua. Mientras el VI se esta ejecutando, El botón de cancelación de ejecución (Abort Execution) aparece. Presione este botón para detener el VI inmediatamente. Nota: Evite usar el botón de cancelación de ejecución (Abort Execution) para detener el VI. Ya sea deje que el VI complete su flujo de datos o diseñe un método para detener el VI programáticamente. Al hacer esto el VI se encuentra en un estado conocido. Por ejemplo, coloque un botón en el panel frontal que detenga el VI cuando sea presionado. Pulse el botón de pausa (Pause) para detener momentáneamente la ejecución de un VI. Cuando usted presiona el botón de Pausa, LabVIEW señala la posición donde usted detuvo la ejecución en el diagrama de bloque. Pulse el botón de Pausa nuevamente para que el VI continué ejecutándose. Seleccione el menú Configuración de Textos (Text Settings) para cambiar el tipo de fuente (letra) del VI, incluyendo el tamaño, estilo y color. Seleccione el menú Alineamiento de Objetos (Align Objects) para alinear objetos con respecto a los ejes, incluyendo eje vertical, superior, izquierdo, etc. Seleccione el menú Distribución de Objetos (Distribute Objects) para espaciar objetos uniformemente, incluyendo espacios vacíos, compresiones, etc. Seleccione el Redimensionamiento de Objetos (Resize Objects) para cambiar el ancho y alto de objetos del panel frontal.
VIs Expreso, VIs y FuncionesVIs Expreso: VIs interactivos con pagina de dialogo configurable VIs estándar: VIs modulares y personalizables mediante cableado Funciones: Elementos fundamentales de operación de LabVIEW; no hay panel frontal o diagrama de bloque Funcion LabVIEW 7.0 introduce un nuevo tipo de subVI llamado VIs Expreso (Express VIS). Estos son VIs interactivos que tienen una configuración de caja de dialogo que permite al usuario personalizar la funcionalidad del VI Expreso. LabVIEW entonces genera una subVI basado en estos argumentos. VIs estándar son aquellos VIs (que consisten de un panel frontal y un diagrama de bloque) que son usados adentro de otro VI. Las funciones son los bloques de construcción de todos los VIs. Las funciones no tienen un panel frontal o un diagrama de bloque. VI Estandar VI Expreso
No borre esta DiapositivaSeleccione el menú Reordenamiento (Reorder) cuando tenga objetos superpuestos unos con otros y usted quiera definir cual esta enfrente de quien y cual atrás. Seleccione uno de los objetos con la herramienta de posicionamiento y seleccione entre Move Forward (mover hacia adelante), Move Backward (mover hacia atrás), Move to Front (mover al frente) y Move to Back (mover hacia atrás). <Los siguientes puntos solo aparecen en la barra de herramientas del diagrama de bloques.> Pulse en el botón de ejecución resaltada (Highlight Execution) para ver el flujo de información en el diagrama de bloques. Pulse el botón nuevamente para detener este tipo de ejecución resaltada. Pulse el botón de entrada al ciclo (Step Into) para entrar un paso adentro de un ciclo, un SubVI, etc. Entrando un paso a la vez adentro de un VI le permite introducirse en la VI nodo a nodo. Cada nodo se resalta para señalar cuando esta listo para ejecutarse. Al entrar un paso adentro de un nodo usted esta listo para caminar paso a paso adentro del nodo. Pulse el botón de sobre (Step Over) para posicionarse encima de un ciclo, un SubVI, etc. Al posicionarse encima del nodo, usted ejecuta el nodo sin entrar paso a paso adentro del nodo. Pulse el botón de Salida del ciclo (Step Over) para salirse de un ciclo, SubVI, etc. Al posicionarse fuera del nodo, usted completa la ejecución de un nodo y puede moverse al siguiente nodo. No borre esta Diapositiva
Ejemplo :Abra y Ejecute un Instrumento VirtualBuscador de Ejemplos 1. Seleccione Start » Programs » National Instruments » LabVIEW 7.0 » LabVIEW para lanzar LabVIEW. Aparece la ventana de dialogo de LabVIEW. 2. Seleccione Find Examples del menú de ayuda (Help). La ventana de dialogo que aparece da la lista y enlaces a todos los VIs de ejemplo de LabVIEW. 3. Usted puede buscar ejemplo por categorías, o puede utilizar palabras claves en su búsqueda. Haga clic en la viñeta de búsqueda (Search) para abrir el buscador de palabras claves. 4. En la caja de “Enter Keyword(s)” escriba “Signal” 5. Una lista de temas relacionados aparecerá en la ventana de ejemplos (examples). Haga doble-clic en signals, esto generara una lista de ejemplos en el lado derecho. 6. Haga clic en cualquier programa para ver una descripción detallada de el ejemplo. Haga doble clic sobre Signal Generation and Processing.vi para lanzar el ejemplo. Esto abrirá el panel frontal del VI “Signal Generation and Processing.vi” . Examine el VI y ejecútelo. Cambie las frecuencias y tipos de señales de entradas y note como el despliegue de las graficas cambia. Cambie la ventana de procesamiento de señales (Signal Processing Window) y opciones de filtrado (Filter options). Después de que haya examinado el VI y las diferentes opciones que puede cambiar, pare el VI presionando el botón de paro (Stop button). Nota Usted también puede abrir un VI al hacer clic en el botón de Open VI y navegar a labview\examples\apps\demos.llb\Signal Generation and Processing.vi.
Ventana de Panel Frontal Ventana de Diagrama de BloquesCreando un VI Ventana de Panel Frontal Ventana de Diagrama de Bloques Cuando usted crea un objeto en el panel frontal, una terminal es creada en el diagrama de bloques. Estas terminales le dan acceso a los objetos del panel frontal del código creado mediante el diagrama de bloque. Cada terminal contiene información útil referente al objeto al cual corresponde en el panel frontal. Por ejemplo, el color y los símbolos proporcionan el tipo de dato. Números de punto flotante y de doble-precisión, son representados con terminales anaranjadas y las letras DBL. Las terminales booleanas son verdes y son representadas por las letras TF. En general, las terminales anaranjadas deben unirse (cablearse) con las terminales anaranjadas, verdes con verdes, y así sucesivamente. Esta no es una regla que no se puede romper; por ejemplo LabVIEW permitirá al usuario conectar una terminal azul (valor entero) a una terminal anaranjada (valor fraccional). Pero en la mayoría de casos, busque mejor una igualdad en colores. Los controles tienen una flecha en el lado derecho y tienen un borde grueso. Los indicadores tienen una flecha en el lado izquierdo y un borde fino. Reglas lógicas pueden ser aplicadas al conectar en LabVIEW: Cada cable debe tener una (pero solo una) fuente (o control), y cada cable puede tener varios destinos (o indicadores). El programa en esta diapositiva toma datos de A y B y pasa valores a una función de adición y a una función de resta. Los resultados son mostrados en los indicadores apropiados. Terminales de Control Terminales de Indicador
Creando un VI – Diagrama de BloquesAdemás de los terminales del panel frontal, el diagrama de bloques contiene funciones. Cada función puede tener múltiples terminales de entradas y salidas. La conexión de estas terminales es una parte muy importante de la programación en LabVIEW. Una vez que usted tenga cierta experiencia programado en LabVIEW, la conexión de cables se le hará mas fácil. Primero puede que necesite ayuda. En seguida se le muestra algunas recomendaciones para comenzar: La herramienta para conectar o de cableado es utilizada para conectarse a los nodos de las funciones. Cuando usted “apunte” con la herramienta de cableado, apunte con el extremo del cable que cuelga del carrete. Aquí es donde el cable sera colocado. Mientras usted mueve la herramienta de cableado sobre las funciones, observe la viñeta amarilla que aparece. Esto le dirá el nombre de la terminal al que se esta conectando. Mientras usted mueva la herramienta de cableado encima de una terminal, esta va a mostrar información. Esto le ayudara a identificar donde se va a unir el cable. Para mas ayuda con los terminales, haga clic derecho en la función y seleccione Visible Items o Objetos Visibles>>Terminals o Terminales. Un dibujo de la función será colocada atrás para revelar las terminales de la conexión. Note los colores - estos corresponden a los tipos de datos utilizados por los terminales del panel frontal. Para ayuda adicional, seleccione Help >> Show Context Help, o presione CTRL+H. Esto mostrara la ventana de ayuda en contexto. A medida que uno mueva el raton (mouse) sobre la función, esta ventana le mostrara la función, terminales, y una breve descripción. Utilice esto junto con otras herramientas para ayudarse mientras conecta los cables.
Consejos Para Conectar – Diagrama de Bloques“Punto Caliente” de Cableado Haga Clic para Seleccionar los Cables Utilice la Ruta Automática del Cable Limpiando el Cableado Además de las terminales del panel frontal. El diagrama de bloque contiene funciones. Cada función puede tener múltiples terminales de entradas y salidas. La conexión de estas terminales es una parte muy importante de la programación en LabVIEW. Una vez que usted tenga cierta experiencia programado en LabVIEW, la conexión de cables se le hará mas fácil. Primero puede que necesite ayuda. En seguida se le muestra algunas recomendaciones para comenzar: La herramienta para conectar o de cableado es utilizada para conectarse a los nodos de las funciones. Cuando usted “apunte” con la herramienta de cableado, apunte con el extremo del cable que cuelga del carrete. Aquí es donde el cable sera colocado. Mientras usted mueve la herramienta de cableado sobre las funciones, observe la viñeta amarilla que aparece. Esto le dirá el nombre de la terminal al que se esta conectando. Mientras usted mueva la herramienta de cableado encima de una terminal, esta va a mostrar información. Esto le ayudara a identificar donde se va a unir el cable. Para mas ayuda con los terminales, haga clic derecho en la función y seleccione Visible Items o Objetos Visibles>>Terminals o Terminales. Un dibujo de la función será colocada atrás para revelar las terminales de la conexión. Note los colores - estos corresponden a los tipos de datos utilizados por los terminales del panel frontal. Para ayuda adicional, seleccione Help >> Show Context Help, o presione CTRL+H. Esto mostrara la ventana de ayuda en contexto. A medida que uno mueva el raton (mouse) sobre la función, esta ventana le mostrara la función, terminales, y una breve descripción. Utilice esto junto con otras herramientas para ayudarse mientras conecta los cables. Si el cableado no se mira muy bien, haga clic derecho en el cable que se desea arreglar y escoja la opción de Clean Up Wire o Limpieza del Cable para que automáticamente el cable haga su ruta de nuevo.
Ejemplo: Creando un VI Conversión: °F=(9/5)(°C)+32 Señal del sensorVentana del Panel Frontal Indicador Grafico Control Numérico Conversión: °F=(9/5)(°C)+32 When you create an object on the Front Panel, a terminal will be created on the Block Diagram. These terminals give you access to the Front Panel objects from the Block Diagram code. Each terminal contains useful information about the Front Panel object it corresponds to. For example, the color and symbols provide information about the data type. For example: The dynamic data type is a polymorphic data type represented by dark blue terminals. Boolean terminals are green with TF lettering. In general, blue terminals should wire to blue terminals, green to green, and so on. This is not a hard-and-fast rule; LabVIEW will allow a user to connect a blue terminal (dynamic data) to an orange terminal (fractional value), for example. But in most cases, look for a match in colors. Controls have an arrow on the right side and have a thick border. Indicators have an arrow on the left and a thin border. Logic rules apply to wiring in LabVIEW: Each wire must have one (but only one) source (or control), and each wire may have multiple destinations (or indicators). Señal del sensor 20°C/volt Introduction to LabVIEW Hands-On 24	ni.com
No Borrar Esta DiapositivaEl cableado es muy flexible en LabVIEW. Experimente con combinaciones de clics y/o teclas cuando este cableando. Aquí hay algunas de las características mas a menudo utilizadas. El hacer un simple, doble, y triple clic en el cable selecciona el cable para moverlo o para borrarlo. El hacer un clic mientras se esta cableando hace un doblez en el cable. Haciendo clic derecho o presionando el botón Esc mientras se esta cableando cancela la operación de cableado. No se preocupe por el color de los cables. LabVIEW seleccionará automáticamente el cable correcto y adecuado para cada situación. Cableando Objetos Automáticamente LabVIEW cablea automáticamente los objetos a medida que usted los coloca en el diagrama de bloque. También puede cablear automáticamente objetos que ya están colocados en el diagrama de bloque. LabVIEW conecta la terminal que mejor combina (match) y deja las terminales que no combinan (match) sin conectar. A medida que usted mueve un objeto seleccionado cerca de otros objetos en el diagrama de bloque, LabVIEW dibuja cables temporales para mostrarle conexiones validas. Cuando usted suelta el botón del mouse para poner el objeto en el diagrama de bloque, LabVIEW conecta automáticamente los cables. Cambie al cableado automático presionando el espaciador (spacebar) mientras mueve un objeto utilizando la herramienta de posicionamiento (Positioning tool). Puede ajustarlas opciones del cableado automático seleccionando Tools >> Options y seleccionando Block Diagram de el menú superior principal. No Borrar Esta Diapositiva
Programando el Flujo de DatosEl diagrama de bloque se ejecuta dependiendo del flujo de los datos; el diagrama de bloques NO se ejecuta de izquierda a derecha El nodo se ejecuta cuando los datos están disponibles para TODOS los terminales de entrada. Los nodos suministran datos a todos los terminales de salida cuando termina. LabVIEW sigue un modelo de flujo de datos para correr los VIs. Un nodo del diagrama de bloque se ejecuta cuando todas sus entradas están disponibles. Cuando un nodo completa la ejecución, suministra datos a sus terminales de salida y pasa los datos de salida al siguiente nodo en la trayectoria del flujo de datos. Visual Basic, C++, JAVA y otros lenguajes de programación basados en texto, siguen un modelo de control de flujo de la ejecución de un programa. En flujo de control, el orden secuencial de los elementos del programa determina el orden de ejecución de un programa. Considere el diagrama de bloque que se encuentra arriba. Este suma dos números y luego resta 50 del resultado de la suma. En este caso, el diagrama de bloque se ejecuta de izquierda a derecha, no porque los objetos están puestos en ese orden, sino porque una de las entradas de la función de resta no es valida hasta que la función de suma o adición haya terminado su ejecución y pasado los datos a la función de resta. Recuerde que un nodo se ejecuta solamente cuando tiene datos disponibles en todas sus terminales de entrada, y suministra datos a sus terminales de salidas solamente cuando termina su ejecución. En el código de la derecha, considere cual segmento del código se ejecutara primero---la suma, el numero aleatorio, o la función de división. No se puede saber porque las entradas a las funciones de suma y división están disponibles al mismo tiempo, y la función de numero aleatorio no tiene entradas. En una situación en donde un segmento del código se debe ejecutar antes que otro, y no existe dependencia de datos entre las funciones, utilice una estructura de Secuencia para forzar el orden de la ejecución.
Opciones de Ayuda Contexto de la Ayuda Referencias en LíneaAyuda en línea Congelar Ayuda Ayuda del Diagrama Simple/Complejo Ctrl + H Referencias en Línea Todos los menús en línea Clic en las funciones del diagrama para tener acceso directo a la información en línea. Utilice la ventana de Context Help (Ayuda Contextual) y LabVIEW Help (Ayuda de LabViEW) para ayudarle a construir o editar los VIs. Refiérase a la ayuda de LabVIEW y a los manuales para mas información. Ventana de Ayuda Contextual (Context Help Window ) Para desplegar la ventana de Context Help, seleccione Help>>Show Context Help o presione las teclas <Ctrl-H>. Cuando mueva el cursor sobre el panel frontal y los objetos del diagrama de bloque, la ventana de Context Help despliega el icono de los subVIs, funciones, constantes, controles e indicadores, con cables adjuntos a cada una de sus terminales. Cuando mueva el cursor sobre la caja de opciones de dialogo, la ventana de Context Help despliega descripciones de esas opciones. En la ventana, las conexiones requeridas están en negrilla, las conexiones recomendadas en texto normal, y las conexiones opcionales están poco claras o no aparecen. Arriba se presenta un ejemplo de la ventana de Context Help. Haga un clic en el botón de Simple/Detailed Context Help localizado en la esquina inferior izquierda de la ventana de Context Help para distinguir entre la ayuda contextual simple y detallada. El modo simple enfatiza las conexiones importantes. Las terminales opcionales son mostradas por segmentos de cables, informándole de la existencia de otras conexiones. Haga clic en el botón de Lock Context Help para congelar el contenido actual de la ventana de la Context Help. Cuando los contenidos están congelados, el mover el cursor sobre otros objetos no cambia los contenidos de la ventana. Para descongelar la ventana, haga clic nuevamente en el botón. Usted también puede tener acceso a esta opción desde el menú de ayuda (Help).
No Borrar Esta DiapositivaHaga clic en el botón de More Help (mas ayuda) para desplegar el tema correspondiente en LabVIEW Help, el cual describe el objeto en forma detallada. LabVIEW Help (Ayuda de LabVIEW) Puede acceder a LabVIEW Help ya sea haciendo clic en el botón de More Help en la ventana de Context Help, seleccionando Help»VI, Function, & How-To Help, haciendo clic en la oración Click here for more help en la ventana de Context Help o presionando <Ctrl-?>. La LabVIEW Help contiene descripciones detalladas de casi todos los menús, herramientas, VIs y funciones. También incluye instrucciones paso-a-paso para la utilización de muchas características de LabVIEW y enlaces al Tutorial de LabVIEW, versiones en PDF de todos los manuales y Notas de Aplicación de LabVIEW, además de recursos de soporte técnico en el sitio web de National Instruments. No Borrar Esta Diapositiva
Nodos de Formula En la subpaleta de Estructuras (Structures).Implementa ecuaciones complicadas. Variables creadas en los márgenes Nombres de variables son sensibles (mayúsculas/minúsculas) Cada declaración debe de terminar con un punto y coma(;) La ventana de ayuda muestra las funciones disponibles. Note el Punto y coma Algunas veces es preferible programar expresiones matemáticas con funciones basadas en texto, en lugar de hacerlo con iconos (los que pueden tomar mucho espacio en el diagrama). Nodo de Formula: Nos permite implementar ecuaciones complicadas usando instrucciones basadas en texto. Localizada en la subpaleta de Structures (estructuras). La cajas pueden cambiar de tamaño para introducir formulas algebraicas directamente a los diagramas de bloque. Para agregar variables, se presiona el botón derecho del mouse y se escoge la opción de agregar entrada (Add Input) o agregar salida (Add Output). Hay que nombrar las variables de la misma manera en que serán usadas en la formula. (Los nombres son sensibles a capitalización) Las declaraciones deben terminar con un punto y coma (;) Al usar varias formulas en un solo nodo de formula, cada variable asignada (las que aparecen en la parte izquierda de cada formula) deben de tener una terminal de salida en el nodo de formula. Sin embargo, estas terminales de salida no necesitan ser cableadas. Compare los ejemplos en la diapositiva.
Técnicas para Eliminar ErroresEncontrando los Errores Resaltar la Ejecución Herramienta de Prueba Haga clic en el botón de “correr” que esta roto; Aparece una ventana mostrando los errores Haga clic en el botón de ejecución resaltada; el flujo de datos es animado utilizando burbujas. Los valores se despliegan en los cables. Haga clic con el botón derecho sobre el cable para exhibir la ventana de prueba y así mostrar los datos mientras fluyen por el segmento de cable. También puede seleccionar la herramienta de prueba desde la paleta de herramientas y hacer un clic en el cable. Cuando su VI no es ejecutable, se despliega una flecha quebrada en el botón de correr en la paleta de herramientas. Encontrando los Errores: Para hacer una lista de los errores, haga clic en la flecha quebrada. Para localizar el objeto malo, haga clic en el mensaje del error. Resaltando la Ejecucion: Anima el diagrama y traza el flujo de datos, permitiéndole ver los valores intermedios. Haga clic en el bombillo incandescente (light bulb) en la barra de herramientas. Probe: Utilizado para ver los valores en los arrays (arreglos) y clusters . Haga clic en los cables con la herramienta Probe o haga clic derecho en el cable para ajustar los probes. Punto de Paro (Breakpoint): Coloca pausas en diferentes lugares del diagrama. Haga clic en los cables o en los objetos con la herramienta de Punto de Paro para colocar los puntos de paro. Utilice el VI Debug Demonstrate del BASICS.LLB para demostrar las opciones y las herramientas.
Sección II – SubVIs Que es un subVI?Elaboración de un icono y un conector para un subVI Utilizando un VI como un subVI
Nodos del Diagrama de BloquesIcono Nodo Expandible Nodo Expandido VI de Generador de Funciones El mismo VI, visto en tres maneras diferentes. El campo amarillo designa un VI Estándar. El campo azul designa un VI Expreso Así como en las terminales de control o de indicador en el diagrama de bloque pueden ser vistos como un icono o como una simple terminal, los subVIs pueden ser vistos como un icono, un nodo expandible o un nodo expandido. Las diferentes vistas dependen en la preferencia del usuario y no cambia la funcionalidad del subVI.
SubVIs Un SubVI es un VI que puede ser utilizado dentro de otro VISimilar a una subrutina Ventajas Modular Fácil para eliminar errores No tiene que crear códigos Requiere menos memoria Después de haber construido un VI, y creado su propio icono y panel de conexión, este puede ser utilizado en otro VI. Un VI dentro de otro VI se llama un subVI. Un subVI corresponde a una subrutina en lenguajes de programación basados en texto. La utilización de subVIs le ayuda a manejar cambios y a eliminar errores del diagrama de bloque rápidamente.
Las Funciones de LabVIEW y los SubVIs operan como Funciones en otros lenguajesSubVI Diagrama de Bloques Función de Pseudo Código function average (in1, in2, out) { out = (in1 + in2)/2.0; } Llamando Programa con Pseudo main { average (in1, in2, pointavg) } Código Llamando al VI en el Diagrama de Bloques A subVI node corresponds to a subroutine call in text-based programming languages. The node is not the subVI itself, just as a subroutine call statement in a program is not the subroutine itself. A block diagram that contains several identical subVI nodes calls the same subVI several times. The modular approach makes applications easier to debug and maintain. The functionality of the subVI does not matter for this example. The important point is the passing of two numeric inputs and one numeric output. Introduction to LabVIEW Hands-On 42	ni.com
Icono y Conector Un icono representa un VI en otro diagrama de bloqueEl conector muestra terminales disponibles para transferir datos Connector Terminals Cada VI muestra un icono, mostrado arriba, en la esquina superior derecha de las ventanas del panel frontal y del diagrama de bloque. Un icono es una representación grafica de un VI. Puede contener texto, imágenes, o una combinación de ambos. Si usted utiliza un VI como subVI, el icono identifica el subVI en el diagrama de bloque del VI. El conector muestra terminales disponibles para la transferencia de datos hacia y desde el subVI. Hay varios patrones de conectores para elegir. Haga un clic-derecho sobre el conector y seleccione el patrón del menú de patrones. A partir de ahí usted puede asignar controles e indicadores en el panel frontal al conector terminal, como veremos mas adelante.
SubVIs Sub VIs El diagrama de bloque mostrado en la parte superior contiene dos subVIs. Para ver el panel frontal de un subVI, simplemente haga un doble clic sobre el subVI. Usted también puede ver la jerarquía de subVIs dentro de un VI principal haciendo un clic en Browse>> Show VI Hierarchy.
Pasos para crear un SubVICrear el icono Crear el conector Asignar terminales Salvar el VI Insertar el VI dentro del VI principal
Crear el Icono Haga click derecho sobre el icono en el diagrama de bloque o panel frontal Cree iconos personalizados para reemplazar los iconos predeterminados al hacer clic-derecho en la esquina superior derecha del panel frontal o del diagrama de bloques y seleccionando el Edit Icon del menú de atajo o haciendo un doble clic en el icono en la esquina superior derecha del panel frontal. Usted también puede corregir iconos seleccionando File >> VI Properties, seleccionando General del menu Category, y haciendo clic en el botón de editar (Edit Icon). Utilice las herramientas del lado izquierdo de la casilla de dialogo del Icon Editor para crear el diseño del icono en el área de edición. El tamaño normal de la imagen del icono aparece en la casilla apropiada en la parte derecha del área de edición. Usted también puede arrastrar un gráfico desde cualquier archivo y colocarlo en la esquina superior derecha del panel frontal o del diagrama de bloque. LabVIEW convierte el gráfico a un icono de 32 × 32 píxeles.
Haga click derecho sobre el icono (solamente en el panel frontal)Crear el Conector Haga click derecho sobre el icono (solamente en el panel frontal) Para utilizar un VI como un subVI, usted necesita crear un conector. El conector es una serie de terminales que corresponden a los controles y a los indicadores de ese VI, similar a la lista de parámetros de una función basado en un lenguaje de programación de texto. El conector define las entradas y las salidas que usted puede cablear al VI para que pueda utilizarlo como subVI. Defina las conexiones asignando un control del panel frontal o un indicador a cada una de las terminales del conector. Para definir un conector, de un clic-derecho sobre el icono en la esquina superior derecha de la ventana del panel frontal y seleccione Show Connector del menú. El conector substituye el icono. Cada rectángulo en el conector representa una terminal. Utilice los rectángulos para asignar entradas y salidas. El número de terminales que LabVIEW muestra en el conector depende del número de controles e indicadores en el panel frontal. El panel frontal de arriba tiene cuatro controles y un indicador, así que LabVIEW muestra cuatro terminales de entradas y una terminal de salida en el conector.
Asignar Terminales Después que seleccione un patrón para utilizar en su conector, debe definir las conexiones asignando un control del panel frontal o indicador a cada uno de los terminales del conector. Cuando usted hace vínculos entre controles e indicadores y el conector, coloque las entradas en la izquierda y las salidas a la derecha para evitar patrones complicados y poco claros en su VI. Para asignar una terminal a un control de panel frontal o indicador, presione una terminal del conector. Presione el control del panel frontal o indicador que usted desea asignar a la terminal. Haga clic en un área abierta del panel frontal. La terminal cambia al color del tipo de datos del control para indicar que usted conectó la terminal. Usted también puede seleccionar el control o el indicador primeramente y después seleccionar la terminal. Asegurase que salvó el VI después de haber hecho las asignaciones de las terminales.
Salvar el VI Escoja un sitio o un fólder fácil de recordarOrganizar por funcionalidad -salvar los VIs similares en un mismo directorio (Ej. Utilidades matemáticas) Organizar por aplicación Salvar todos los VIs Usados para una Aplicación Especifica dentro de un directorio o un archivo de librería (Ej. Lab 1 – Frecuencia de respuesta) Archivo de librería (.llbs) combina todos los VIs en un solo archivo, ideal para transferir aplicaciones enteras a través de computadoras Hay varias maneras de organizar sus subVIs. La manera más común es organizarlos por aplicación. En este caso, son salvados en el mismo directorio o dentro de un archivo de librería de VI todos los VI’s para una aplicación en particular. Salvarlos en un archivo de librería permite que usted transporte una aplicación entera dentro de un solo archivo. Salvar dentro de una librería es simple. Después de presionar Save As… (salvar como), presione New VI Library (nueva librería de VI). Esto permitirá que usted nombre la librería, y entonces salvar su VI en él. Para agregar VI subsecuentes, simplemente haga un doble-clic al archivo .llb de la ventana estándar de Salvar, y dar un nombre al VI.
Insertar el SubVI dentro de un VI PrincipalAcceder subVIs creados por el usuario Funciones >> All Functions >> Seleccione un VI o arrastre el icono hasta el diagrama deseado Después de construir un VI y crear su icono y conector, usted puede utilizarlo como un subVI. Para colocar un subVI en el diagrama de bloque, seleccione Functions>> Select VI. Navegue y haga un doble-clic en el VI que usted desea utilizar como subVI y colocaquelo en el diagrama de bloques. También puede colocar un VI abierto en el diagrama de bloque de otro VI abierto usando la herramienta de colocación (Positioning Tool) para hacer clic en el icono en la esquina superior derecha del panel frontal o diagrama de bloque de el VI que usted desea utilizar como subVI y arrastrar el icono al diagrama de bloque del otro VI.
Ejemplo: Crear un SubVI del convertidor de temperatura.
Tips para trabajar en LabVIEWAtajos desde el teclado <Ctrl-H> – Activa/Desactiva la Ventana de Ayuda <Ctrl-B> – Remueve todos los cables rotos del diagrama de bloques <Ctrl-E> – Cambiar entre el Panel Frontal y el Diagrama de Bloques <Ctrl-Z> – Deshacer cambios – Undo (también disponible en el menu de edición) Herramientas » Opciones… – Establecer Preferencias en LabVIEW Propiedades del VI – Configurar la Apariencia del VI, Documentación, etc. LabVIEW tiene muchas teclas de atajo que hacen el trabajo mas fácil. Las mas comunes se listan en la diapositiva. Mientras la Herramienta de Selección Automática (Automatic Selection Tool) es genial para escoger la herramienta que usted desearía utilizar en LabVIEW, a veces hay casos cuando usted desea controles manuales. Utilice la tecla Tab para cambiar entre las cuatro herramientas mas comunes (Operate Value (valor de operacion), Position/Size/Select, (Posición/Tamaño/Selección), Edit Text (editar texto), Set Color on Front Panel (establecer color en panel frontal), Connect wire on Block Diagram (conectar cable en el diagrama de bloque)). Una vez que se ha terminado con la selección de herramientas, puede presionar <Shift-Tab> para activar la Herramienta de Selección Automática. En el dialogo de Herramientas >> Opciones… , hay muchas opciones configurables para el Panel Frontal, Diagrama de Bloque, Colores, Impresión y mucho mas. Similar a las opciones de LabVIEW, se pueden configurar propiedades especificas del VI al ir a File >> VI Properties… Ahí se puede documentar el VI, cambiar la apariencia de la ventana, y personalizarlos de varias maneras.
Adquisición de Datos en LabVIEWNI-DAQ Tradicional VIs específicos para realizar: Entrada Análoga Salida Análoga I/O (entrada/salida) Digital Operaciones de conteo NI-DAQmx Controlador (Driver) de siguiente generación: VIs para ejecutar una tarea Un serie de VIs para todos los tipos de mediciones La plataforma de Adquisición de Datos en LabVIEW contiene una plataforma para la NI-DAQ tradicional y una para NI-DAQmx. Los VIs tradicionales están divididos por el tipo de medición, los VIs DAQmx están divididos por el tipo de tarea. Antes de utilizar la Adquisición de Datos de VIs se debe completar varios pasos. Los dispositivos deben se configurados para las computadoras en esta clase. 1. El software NI-DAQ debe estar instalado en la computadora. 2. Debe tener instalado una tarjeta E-series DAQ y esta debe ser configurada usando el Explorador de Automatización & Mediación (Measurement & Automation Explorer (MAX)). Para mas información sobre la instalación y configuración de hardware de National Instruments, consultar el DAQ QUick Start Guide: 45
DAQ – Adquisición De DatosAdquisición de temperatura utilizando el Asistente de DAQ (DAQ Assistant) Arriba esta la ventana de ayuda del DAQ (DAQ Assistant) que puede ser configurada rápidamente para leer temperaturas de una tarjeta de Adquisición de Datos (DAQ).
Terminología De la Adquisición De DatosResolución – Determina cuantos diferentes cambios de voltajes pueden ser medidos Resolución mas grande  Una representacion mas exacta de la señal Rango – Voltajes mínimos y máximos Rango mas pequeño  Una representación mas precisa de la señal Gain (ganancia) – Amplifica o atenúa la señal para un mejor ajuste del rango Resolución: Al adquirir datos a una computadora, un convertidor de Análogo-a-Digital (ADC) toma una señal análoga y la convierte a un número binario. Por lo tanto, cada número binario del ADC representa cierto nivel de voltaje. El ADC devuelve el nivel más alto posible sin pasar el nivel de voltaje real de la señal análoga. La resolución se refiere al número de niveles binarios que el ADC puede utilizar para representar una señal. Para encontrar el número de niveles de voltaje disponibles basados en la resolución usted simplemente toma 2Resolucion. Por lo tanto, cuanto más alta es la resolución, más son los niveles que usted tendrá para representar su señal. Por ejemplo, un ADC con 3-bits de resolución puede medir 23 o 8 niveles de voltajes, mientras que un ADC con 12-bit de resolución puede medir 212 o 4096 niveles de voltajes. Rango: A diferencia de la resolución del ADC, el rango del ADC es seleccionable. La mayoría de los dispositivos DAQ ofrecen un rango desde o -10 a +10. Se elige el rango cuando usted configura su dispositivo en NI-DAQ. Tenga presente que la resolución del ADC será extendida por cualquier rango que usted elija. Cuanto más grande sea el rango, mas esparcida será su resolución, y tendrá una peor representación de su señal. Entonces es importante escoger su rango para apropiadamente encajar su señal de entrada.
No Borrar esta DiapositivaGanancia: Escogiendo apropiadamente el rango de su ADC es una manera de asegurarse de que usted esta maximizando la resolución de su ADC. Otra manera de ayudar a su señal a maximizar la resolución del ADS es aplicando una ganancia. Ganancia se refiere a cualquier amplificación o atenuación de una señal. El ajuste de ganancia (gain setting) es un factor de escala. Cada nivel de voltaje en su señal entrante es multiplicado por el ajuste de ganancia para obtener la señal amplificada o atenuada. A diferencia de la resolución que es un ajuste fijo (fixed setting) del ADC, y el rango que es escogido cuando el dispositivo DAQ es configurado, la ganancia es especificada indirectamente a través de un ajuste llamado límites de entrada. Los límites de entrada se refieren a los valores máximos y mínimos de su señal de entrada analógica actual. Basado en los límites de entrada que usted defina, la ganancia posible más grande que mantendrá la señal entre el rango escogido del ADC es aplicada a su señal. Así que en lugar de necesitar calcular la mejor ganancia basada en su señal y en el rango escogido, todo lo que usted necesita saber son los valores máximos y mínimos de su señal. No Borrar esta Diapositiva
Conexiones de HardwareSC-2075 BNC-2120 NI-ELVIS SCB-68 Hay muchos arreglos diferentes posibles de hardware para adquirir datos. Todos los sistemas de la Adquisición de Datos requieren algún tipo de conexión terminal que acepta una señal de su transductor y lo transmite a la tarjeta DAQ. Cuatro de estos bloques de terminales son el BNC-2120, SC-2075, SCB-68, y NI-ELVIS. El BNC-2120 es un bloque de conexiones aislado (shielded) con conectores BNC de señal-etiquetada para la conectividad fácil a su dispositivo DAQ. También proporciona un generador de funciones, encoder (codificador) de cuadratura, referencia de la temperatura, conector de termocupla, y LED para que usted pueda probar la funcionalidad de su hardware. El SC-2075 proporciona áreas para conexión (breadboard areas) para hacer prototipos y conectividad BNC y de terminales de resorte (spring terminal). Los ±15 V de alimentación fija asi como la alimentación ajustable de 0 a 5 V y LEDs hacen del SC-2075 un instrumento ideal para laboratorios académicos. El SCB-68 es un bloque de conexión de E/S (I/O) aislado (shielded) para señales resistentes y de bajo-ruido. Incluye dos áreas de conexión (breadboard areas) de propósito-general así como un sensor de temperatura IC para la compensación de empalme frío en mediciones de temperatura. NI-ELVIS (Sistema de Laboratorio Educacional de Instrumentación Virtual) es un ambiente de prototipo y diseño basado en LabVIEW y consiste de instrumentos virtuales basados en LabVIEW, un dispositivo de adquisición de datos multifuncional, y en una estación de trabajo con un banco y tablero de diseño y prototipo.
Ejemplo– Adquisición de Datos SimplesComplete el VI de conversión de °C a °F (Conversion C a F.vi), y después cree el VI Termómetro (Thermometer.vi). Nota: Para completar este ejercicio, usted necesitará el sensor de la temperatura IC disponible en el BNC-2120, SCB-68 o Accesorio de Señal DAQ. El tiempo estimado para completarse: 30 minutos. Instrucciones: Este ejercicio tiene tres partes. Primero, cree un icono y conector para Conversion C a F.vi (Ejercicio 1). El icono debe recordarle la funcionalidad del VI (e.g. CF o C a F). El conector debe tener una entrada y una salida, permitiendo una terminal para °C de entrada, y °F de salida. Segundo, cree un VI principal que adquiera un punto de datos del canal 0 (el sensor de la temperatura) de su tarjeta DAQ y permita al usuario mostrar la temperatura en Centígrados o Fahrenheit. Para hacer esto usted necesitará adquirir un solo punto de datos de su tarjeta DAQ y escalarlo por un factor de 100. Esto le dará °C. Usted debe tener un interruptor o botón Booleano que le permita al usuario escoger Centígrados o Fahrenheit. Si el usuario escoje Centígrados, el valor escalado se debe mostrar en un indicador de termómetro. Si el usuario escoje Fahrenheit, el valor en Centígrados se debe de pasar por el Conversion C a F.vi (utilizado como un subVI), y el valor de salida Fahrenheit debe ser desplegado. Sugerencia: Utilice la función Select (seleccionar) en la paleta de Comparison (comparación).
No Borrar esta DiapositivaFinalmente, cree un Icono y Conector para Termómetro.vi. Un posible Icono seria una imagen de un termómetro. El conector debería tener dos terminales. Uno para la entrada Booleana (°C or °F), y el segundo para la salida de la temperatura escalada. Guarde el VI como termómetro.vi No Borrar esta Diapositiva
Sección IV – Ciclos y GráficasFor Loop (Ciclo Para) While Loop (Ciclo Mientras) Gráficas Multiplots Dele clic para agregar notas
Ciclos Ciclos For Ciclo While Tiene una Terminal de IteraciónSiempre corre al menos una vez Corre de acuerdo a la Terminal Condicionante Ciclos For Tiene una Terminal de Iteración Corre de acuerdo a la entrada N de la Terminal de Conteo Ambos los ciclos Mientras (While) y Para (For) están localizados en la paleta Functions»Structures. El ciclo Para difiere del ciclo Mientras en que el ciclo Para ejecuta una serie (set) de veces predeterminada. El ciclo Mientras deja de ejecutar el sub diagrama solo si el valor en la condición terminal existe. Ciclo Mientras Similar al ciclo Haga (Do) o al ciclo Repita-hasta (Repeat-Until) en lenguajes de programación basados en texto, un ciclo Mientras, mostrado arriba a la derecha, ejecuta un sub diagrama hasta que la condición sea cumplida. El ciclo Mientras ejecuta el sub diagrama hasta que la terminal dependiente, recibe un valor Booleano específico. El comportamiento incumplido y la apariencia de la terminal dependiente (valor por defecto) es Continue if True (continue si es Verdadero), mostrado a la izquierda. Cuando una terminal dependiente es Continue if True, el ciclo Mientras ejecuta su sub diagrama hasta que la terminal dependiente recibe un valor FALSO. La terminal de iteración (una terminal de salida), mostrada a la izquierda, contiene el numero de iteraciones completas. El conteo de iteraciones siempre empieza en cero. Durante la primera iteración, la terminal de iteración regresa a cero. Ciclo Para Un ciclo Para, mostrado a la izquierda, ejecuta una serie varias veces. El valor en la terminal de conteo (una terminal de entrada) representada por la N, indica cuantas veces repetir el sub diagrama. La terminal de iteración (una entrada de salida), mostrada a la izquierda, contiene el numero de iteraciones completas. El conteo de iteraciones siempre empieza en cero. Durante la primera iteración, la terminal de iteración regresa a cero.
Ciclos (cont.) 1. Seleccionar el ciclo2. Encerrar código que va a ser repetido 3. Arrastrar nodos adicionales y luego cablear Coloque ciclos en su diagrama seleccionándolos desde la paleta de Estructuras de la paleta de Funciones (demostrar) Cuando este seleccionado, el cursor del ratón se convierte en un puntero especial que usted usa para encerrar la sección del código que desea repetir. Déle clic al botón del ratón para definir la esquina superior-izquierda, déle clic otra vez al botón del ratón en la esquina inferior izquierda, y la frontera del ciclo Mientras es creada alrededor del código seleccionado. Arrastre nodos adicionales adentro del ciclo Mientras si es necesario.
Gráficas Gráfica de forma de onda – indicador númerico especial que puede mostrar una historia de valores Controles >> Indicadores de Gráficas >> Gráfica de forma de onda La grafica de forma de onda es un indicador numérico especial que muestra una o más graficas (plots). La grafica de forma de onda esta localizada en la paleta Controls»Graph Indicators. Las graficas de forma de onda pueden mostrar graficas simples o múltiples. El siguiente panel frontal muestra un ejemplo de una grafica de forma de onda múltiple. Usted puede cambiar los valores min y max del eje x y y dándole doble clic al valor con la herramientas de etiquetado y escribiendo el nuevo valor. Similarmente, usted puede cambiar la etiqueta del eje. También puede darle clic derecho a la leyenda de la grafica y cambiar el estilo, forma y color del dibujo que se muestra en la gráfica.
Cableando Datos a las GráficasGrafica de diagrama simple Grafica de diagrama múltiple Usted puede cablear una salida escalar directamente a una grafica de forma de onda para mostrar un diagrama. Para mostrar múltiples diagramas en una grafica, use la función Merge Signals encontrada en la paleta Functions >> Signal Manipulation. La función Merge Signal junta múltiples salidas para diagramar en la gráfica de forma de onda. Para agregar mas diagramas, utilice la herramienta de Posicionamiento para cambiar el tamaño de la función Merge Signal. La ayuda de contexto contiene muy buena información en como cablear de diferentes maneras los datos dentro de las graficas.
Ejemplo – Usando ciclosEstudiantes construyen un VI con un Ciclo.vi. Este ejercicio debe tomar de minutos. Instrucciones Cree un VI que genere un número aleatorio a una velocidad especifica y mustre las lecturas en una grafica de forma de onda hasta que el usuario lo pare. Conecte la terminal de terminación a un botón de paro en el panel frontal, y agregue un control de deslizamiento (slider control) al panel de control. Este control de deslizamiento debe recorrer de 0 a 2000 en valor, y será conectado a la función Express VI Time Delay (demora de tiempo) dentro de su ciclo Mientras. Guarde el VI como Use a Loop.vi
Sección V – Arreglos & Manejo de Archivo (File I/O)Construir arreglos manualmente Dejar que LabVIEW construya arreglos automáticamente Escribir a un archivo de hoja de cálculo (spreadsheet) Leer desde un archivo de hoja de cálculo (spreadsheet) Los arreglos agrupan elementos de datos del mismo tipo. Un arreglo consiste de elementos y dimensiones. Los elementos son los datos que componen el arreglo. Una dimensión es la longitud, altura, o profundidad de un arreglo. Un arreglo puede tener una o mas dimensiones y hasta 2^31 – 1 elementos por dimensión, si lo permite la memoria. Usted puede construir arreglos de tipos de datos numéricos, Booleanos, rutas, cadenas, formas de onda y cluster data. Considere usar arreglos cuando trabaje con una colección de datos similares y cuando realice computaciones repetitivas. Los arreglos son ideales para almacenar información que se obtiene de formas de ondas o información generada en los ciclos, donde cada iteración de un ciclo produce un elemento del arreglo. Los elementos del arreglo son ordenados. Un arreglo utiliza un índice para que usted pueda tener acceso rápido a cualquier elemento en particular. El índice es basado en cero, lo que significa que tiene un rango de 0 a n – 1, donde n es el numero de elementos en el arreglo. Por ejemplo, n = 9 para los nueve planetas, así que el índice recorre de 0 a 8. La Tierra es el tercer planeta, así que tiene un índice de 2. Las operaciones de Entrada/Salida de archivo (I/O) pasan los datos desde y a los archivos. Utilice los VI de File I/O y las funciones localizadas en la paleta Functions » File I/O para manejar todos aspectos de I/O de archivo. En esta clase nosotros cubriremos la lectura y escritura de archivos de hoja de cálculo (spreadsheet) utilizando el Express VIs para File I/O.
Agregando un Arreglo al Panel FrontalDesde la sub-paleta de Controls >> All Controls >> Array and Cluster, seleccione el Array Shell Colóquelo en la pantalla. Para crear un control o indicador de un arreglo como el mostrado, escoja un arreglo (array) en la paleta Controls » All Controls » Arrays & Cluster, colóquelo en el panel frontal, y arrastre un control o un indicador adentro de la estructura (shell) del arreglo. Si usted procura arrastrar un control o indicador inválidos tales como una gráfica XY adentro de la estructura del arreglo, usted sera incapaz de dejar caer el control o el indicador en la estructura del arreglo. Usted debe insertar un objeto en la estructura del arreglo antes de que usted utilice el arreglo en el diagrama de bloques. De otro modo, la terminal del arreglo aparece negra con un paréntesis vacío.
Agregando un Arreglo (cont.)Situé el objeto de datos dentro del shell (Ej. Control Numérico) Para agregar las dimensiones a un arreglo de uno en uno, hace clic derecho en el despliegue del índice y seleccione Add Dimension del menú de atajo. Usted puede utilizar también la herramienta que Posicionamiento para cambiar el tamaño del despliegue del índice hasta que usted tenga las dimensiones que usted quiera.
Creando un Arreglo con un CicloLos ciclos acumulan arreglos en sus limites. Si se cablea un arreglo a un tunel de entrada de un ciclo Para (For Loop) o un ciclo Mientras (While Loop), usted puede leer y procesar cada elemento en ese arreglo al activar el auto-indexing (auto-indicación). Cuando usted utiliza el auto-índice en un arreglo de túnel de salida el arreglo de salida recibe un nuevo elemento por cada iteración del ciclo. El cable que va del túnel de salida al arreglo indicador se vuelve mas grueso al cambiar a un arreglo en el borde del ciclo, y el túnel de salida contiene corchetes cuadrados representando un arreglo, como se muestra en la siguiente ilustración. Desactive el auto-indicador al hacer un clic derecho en el tunel y seleccionar Disable Indexing (desactivar Indicacion) desde el menu de atajo. Por ejemplo, desactive el auto-indicador si necesita únicamente el ultimo valor pasado al tunel en ejemplo anterior, sin crear un arreglo. Nota: Debido a que usted puede utilizar ciclos Para (For Loops) para procesar arreglos y elemento a la misma vez, LabVIEW, activa el auto-indicador como opción por defecto para cada arreglo que se cablee a un Ciclo Para. El Auto-Indicador para Ciclos Mientras (While Loops) esta desactivados por defecto. Para activar el auto-indicador, haga clic derecho en un tunel y seleccione Enable Indexing del menu de atajo. Si usted activa el auto-indicador en un arreglo cableado a una terminal de entrada de un Ciclo Para, LabVIEW arregla el contador de la terminal al tamaño del arreglo para que así usted no tenga que cablear la terminal del contador. Si usted activa el auto-indicador para mas de un túnel, o si cablea la terminal del contador, el contador se vuelve la más pequeña de las opciones. Por ejemplo, si usted cablea un arreglo con 10 elementos al túnel de entrada de un Ciclo Para y pone la terminal del contador a 15, el ciclo se ejecuta 10 veces.
Creando Arreglos Bidimensionales (2D)Se pueden utilizar dos Ciclos Para, uno dentro del otro, para crear arreglos bidimensionales. El Ciclo Para externo crea las filas de elementos y el Ciclo Para interno crea las columnas de elementos.
Construyendo Arreglos con Ciclos (Auto-Indexando)Los ciclos pueden acumular arreglos auto indexados en sus bordes Los ciclos For se auto indexan por defecto Los ciclos While sacan solo el valor final por defecto Un clic derecho en el túnel habilita / deshabilita el auto indexado Auto-Indexado Habilitado El cable engrosa su tamaño Arreglo de 1D Auto-Indexado Deshabilitado El cable conserva su tamaño For Loops and While Loops can index and accumulate arrays at their boundaries. This is known as auto-indexing. The indexing point on the boundary is called a tunnel. The For Loop default is auto-indexing enabled. The While Loop default is auto-indexing disabled. Examples: Enable auto-indexing to collect values within the loop and build the array. All values are placed in array upon exiting loop. Disable auto-indexing if you are interested only in the final value. Solo un valor (última iteración) pasa fuera del ciclo 5 © National Instruments Corporation 55	Introduction to LabVIEW Hands-On
Funciones de los Arreglos – Lo BasicoFunctions >> All Functions >> Array Utilice las funciones de los arreglos situadas en la paleta Functions» All Functions» Array para crear y para manipular arreglos. Las funciones de los arreglo incluyen las siguientes: Array Size (Tamaño del Arreglo) – devuelve el número de elementos en cada dimensión de un arreglo. Si el arreglo es de n-dimensiones, el tamaño (size) de la salida es un arreglo de n elementos. Initialize Array (Inicializar el Arreglo) – crea un arreglo de n dimensiones en donde cada elemento ha sido inicializado al valor del elemento (element). Cambie el tamaño de la funcion para incrementar el numero de dimensiones del arreglo de salida. Build Array (Construcción del Arreglo) -- concatena arreglos múltiples o añade elementos a un arreglo de n dimensiones. Cambie el tamaño de la funcion para incrementar el numero de dimensiones del arreglo de salida. Array Subset (Subconjunto de Arreglo) – devuelve parte de un arreglo que comienza en el índice y que contiene elementos de la longitud. Index Array (Indice del Arreglo) – devuelve un elemento de un arreglo en el índice. Usted también puede utilizar la función Indice del Arreglo para extraer una fila o una columna de un arreglo bidimensional para asi crear un subarreglo del original. Para hacer esto cablee un arreglo bidimensional a la entrada de la función. Dos terminales de índice están disponibles. La terminal del indice superior indica la fila, y la segunda terminal indica la columna. Usted puede cablear entradas a ambas terminales de índice para indicar un solo elemento, o usted puede cablear solamente una terminal para extraer una fila o una columna de datos. Por ejemplo, cablee el arreglo siguiente a la entrada de la función.
Funciones de los Arreglos – Construcción de un ArregloBuild Array (Construcción de Arreglos) puede realizar dos funciones distintas. Concatena múltiples arreglos o añade elementos a un arreglo de n dimensiones. Cambie el tamaño de la función para incrementar el numero de dimensiones del arreglo de salida. Para concatenar las entradas en un arreglo más largo de la misma dimensión según lo demostrado en el siguiente arreglo, haga click derecho en el nodo de la función y seleccione Concatenate Inputs (Concatenar Entradas) desde el menú de atajo.
¿Cómo tomo decisiones en LabVIEW?Estructuras Caso (case) Selección (select) (a) (b) Case Structure The Case Structure has one or more subdiagrams, or cases, exactly one of which executes when the structure executes. The value wired to the selector terminal determines which case to execute and can be boolean, string, integer, or enumerated type. Right-click the structure border to add or delete cases. Use the Labeling tool to enter value(s) in the case selector label and configure the value(s) handled by each case. It is found at Functions»Programming»Structures»Case Structure. Select Returns the value wired to the t input or f input, depending on the value of s. If s is TRUE, this function returns the value wired to t. If s is FALSE, this function returns the value wired to f. The connector pane displays the default data types for this polymorphic function. It is found at Functions»Programming» Comparison»Select. Example a: Boolean input: Simple if-then case. If the Boolean input is TRUE, the true case will execute; otherwise the FALSE case will execute. Example b: Numeric input. The input value determines which box to execute. If out of range of the cases, LabVIEW will choose the default case. Example c: When the Boolean passes a TRUE value to the Select VI, the value 5 is passed to the indicator. When the Boolean passes a FALSE value to the Select VI, 0 is passed to the indicator. (c) © National Instruments Corporation 47	Introduction to LabVIEW Hands-On
Repaso de los tipos de Datos hallados en LabVIEWLabVIEW utilizes many common datatypes. These Datatypes include: Boolean, Numeric, Arrays, Strings, Clusters, and more. The color and symbol of each terminal indicate the data type of the control or indicator. Control terminals have a thicker border than indicator terminals. Also, arrows appear on front panel terminals to indicate whether the terminal is a control or an indicator. An arrow appears on the right if the terminal is a control, and an arrow appears on the left if the terminal is an indicator. Definitions Array: Arrays group data elements of the same type. An array consists of elements and dimensions. Elements are the data that make up the array. A dimension is the length, height, or depth of an array. An array can have one or more dimensions and as many as (231) – 1 elements per dimension, memory permitting. Cluster: Clusters group data elements of mixed types, such as a bundle of wires in a telephone cable, where each wire in the cable represents a different element of the cluster. See Help»Search the LabVIEW Help… for more information. The LabVIEW User Manual on ni.com provides additional reference for data types found in LabVIEW. © National Instruments Corporation 67	Introduction to LabVIEW Hands-On
¿Cómo controlo el Tiempo de un Ciclo?1. Ciclo de Retardo de Tiempo (Time Delay) Configura el VI Expres Time Delay para segundos de espera en cada iteración del ciclo (funciona en ciclos For y While). 2. Ciclos Temporizados Configura el ciclo especial temporizado While para un retraso en tiempo deseado. Time Delay The Time Delay Express VI delays execution by a specified number of seconds. Following the rules of Data Flow Programming, the while loop will not iterate until all tasks inside of it are complete, thus delaying each iteration of the loop. Timed Loops Executes each iteration of the loop at the period you specify. Use the Timed Loop when you want to develop VIs with multi-rate timing capabilities, precise timing, feedback on loop execution, timing characteristics that change dynamically, or several levels of execution priority. Double-click the Input Node or right-click the Input Node and select Configure Timed Loop from the shortcut menu to display the Loop Configuration dialog box, where you can configure the Timed Loop. The values you enter in the Loop Configuration dialog box appear as options in the Input Node. Retardo de Tiempo Ciclo Temporizado Wait Until Next ms Multiple Waits until the value of the millisecond timer becomes a multiple of the specified millisecond multiple. Use this function to synchronize activities. You can call this function in a loop to control the loop execution rate. However, it is possible that the first loop period might be short. This function makes asynchronous system calls, but the nodes themselves function synchronously. Therefore, it does not complete execution until the specified time has elapsed. Functions»Programming»Timing»Wait Until Next ms Multiple Introduction to LabVIEW Hands-On 58	ni.com
Sección VI – Funciones de Arreglos y GraficosFunciones Básicas de los Arreglos Uso de gráficos Crear Diagramas Múltiples con Gráficos Haga click para adicionar notas.
Ejemplo: Realizar un programa generar la secuencia de movimiento de un motor de pasos. Colocar un dial (control numérico) para poder variar el tiempo entre cada dato. Colocar un control booleano para el cambio del sentido de giro del motor. Valor decimal Valor Hexadecimal 3 0011 6 0110 12 1100 9 1001
Gráficos Seleccionados desde la Paleta de Controles del MenuControls >> All Controls >> Graphs Grafico de Forma de Onda (Waveform Graph) – Grafica un arreglo de numeros en contra de sus indices Grafico XY Expreso (Express XY Graph) – Grafica un array en contra de otro Grafico de Forma de Onda Digital (Digital Waveform Graph) – Grafica bits de datos binarios VIs con gráficos usualmente recolectan los datos en un arreglo y después dibujan estos datos en una gráfica. Los gráficos situados en la paleta Controls» All Controls» Graph incluyen el gráfico de forma de onda y el gráfico XY. El diagrama del gráfico de forma de onda traza solamente funciones de una sola variable, como en y = f(x), con puntos distribuidos uniformemente a lo largo del eje x, tal como formas de onda adquiridas con tiempo variable. Los gráficos expresos XY exhiben cualquier set de puntos, ya sea que estos sean muestreados uniformemente o no. Cambie el tamaño de la leyenda de la grafica para exhibir múltiples diagramas. Utilice diagramas múltiples para ahorrar espacio en el panel frontal y para hacer hacer comparaciones entre diagramas. Los gráficos XY y de forma de onda se adaptan automáticamente a los diagramas múltiples. Grafica de Forma de Onda de Trazo-Simple (Single-Plot Waveform Graph) Los gráficos de forma de onda aceptan un solo arreglo de valores e interpreta los datos como puntos en el gráfico e incrementa el indice x por uno comenzando en x = 0. El gráfico también acepta un cluster de un valor inicial x, un .x, y de un arreglo de datos y. Refierase al VI del gráfico de forma de onda (Waveform Graph VI) en examples\general\graphs\gengraph.llb para encontrar ejemplos de los tipos de datos que los gráficos de forma de onda de trazo-simple aceptan.
Gráficos Haga Clic Derecho en la grafica y seleccione PropertiesLos Gráficos son indicadores muy poderosos en LabVIEW. Pueden ser altamente adaptables y pueden ser usados para mostrar de forma concisa una gran cantidad de información. La pagina de propiedades de las Graficas le permite mostrar las opciones de configuración, escala, cursores y muchas otras características de las gráficas. Haga Clic Derecho en la grafica y seleccione Properties (propiedades) para poder personalizar interactivamente
Ejercicio 5 – Usando Graficos de Forma de Onda (Waveform )Tiempo De Completación Estimado: 20 minutos. Cree un VI como el que esta representado arriba. El VI debe utilizar un Ciclo Mientras (While Loop) con 100 milisegundos de retraso para generar continuamente formas de onda seno y cuadradas y mostrarlas en un gráfico de forma de onda. Utilice el VI expreso Simulate Signal Express VI de la paleta Functions» Input para generar las señales. La entrada de la frecuencia para cada función es elegida por el usuario. Cambie los colores, los artículos visibles, y los estilos de diagramado del gráfico. Experimente con algunas de las opciones de cursor y de zoom disponibles.
No borrar esta DiapositivaGrafico de Forma de Onda de Trazo- Múltiple Los gráficos de forma de onda de trazo múltiple aceptan un arreglo bidimensional de valores, en donde cada fila del arreglo es una grafica sencilla. La gráfica interpreta los datos como puntos en la gráfica e incrementa el indice x por uno, comenzando en x = 0. Cablee un arreglo de datos bidimensionales a la gráfica, haga clic derecho en la gráfica, y seleccione Transpose Array (arreglo transpuesto) del menú de atajo para manejar cada columna del arreglo como un diagrama aparte. Refiérase al (Y) Multi Plot 1 graph en el VI de Waveform Graph en examples\general\graphs\gengraph.llb para encontrar un ejemplo de un gráfico que acepte este tipo de datos. Un gráfico de forma de onda de diagramas múltiples también acepta un cluster de un valor x, un valor x , y de un arreglo bidimensional de datos y. El gráfico interpreta los datos de y como los puntos en el gráfico e incrementa el indice x por x, comenzando en x = 0. Refierase a (Xo, x, Y) Multi Plot 3 graph en el VI de Waveform Graph que se encuentra en examples\general\graphs\gengraph.llb para encontrar un ejemplo de un gráfico que acepte este tipo de datos. Un gráfico de forma de onda de diagramas múltiples acepta un cluster de un valor inicial x, de un valor .x, y de un arreglo que contenga clusters. Cada cluster contiene un arreglo de puntos que contiene los datos de y. Usted utiliza la función de Bundle (empaquetado) para empaquetar los arreglos en clusters, y usted utiliza la función de Build Array (construir arreglo) para construir los clusters resultantes en un arreglo. Usted también puede utilizar el Build Cluster Array (construir arreglo de clusters), el cual crea arreglos de clusters que contienen entradas que usted especifica. Refiérase a (Xo, x, Y) Multi Plot 2 en el VI de Waveform Graph que se encuentra en examples\general\graphs\gengraph.llb para encontrar un ejemplo de un gráfico que acepte este tipo de datos. Single-Plot XY Graph (Grafica XY de Trazo Simple) Los gráficos XY de trazo simple aceptan un cluster que contenga un arreglo x y un arreglo y. El gráfico de XY también acepta un arreglo de puntos, en donde un punto es un cluster que contiene un valor de x y un valor de y. Refiérase al VI de XY Graph en examples\general\graph\gengraph.llb para encontrar un ejemplo del tipo de datos de los gráficos de XY de trazo simple. MultiPlot XY Graph (Grafica XY de diagramas múltiples) La Grafica XY de diagramas múltiples acepta un arreglo de diagramas, en donde un diagrama es un cluster que contiene un arreglo x y un arreglo y. La Grafica XY de diagramas múltiples también acepta un arreglo de clusters de diagramas, en donde una grafica es un arreglo de puntos. Un punto es un cluster que contiene un valor x y un valor y. Refiérase al VI del XY Graph encontrado en examples\general\graph\gengraph.llb para encontrar un ejemplo de los tipos de datos del Grafico XY de diagramas múltiples. No borrar esta Diapositiva
Shift Register • Disponible en los bordes derecho e izquierdo de los ciclos y estructuras • Con un clic derecho en el borde y selecciona Add Shift Register • El terminal derecho guarda el dato al terminar la iteración • La terminal izquierda provee el dato al comenzar la nueva iteración Valor 3 Valor Inicial Shift registers transfer data from one iteration to the next: Right-click on the left or right side of a For Loop or a While Loop and select Add Shift Register. The right terminal stores data at the end of an iteration. Data appears at the left terminal at the start of the next iteration. A shift register adapts to any data type wired into it. An input of 0 would result in an output of 5 the first iteration, 10 the second iteration and 15 the third iteration. Said another way, shift registers are used to retain values from one iteration to the next. They are valuable for many applications that have memory or feedback between states. The feedback node is another representation of the same concept. (pictured below) Both programs pictured behave the same. See Help»Search the LabVIEW Help… for more information. Antes de que Inicie el Ciclo Primera Iteración Segunda Iteración Ultima Iteración Introduction to LabVIEW Hands-On 74	ni.com
Variables Locales Las Variables Locales permiten que los datos pasen a través de los ciclos. Un solo control o indicador puede ser escrito o leído por más de una ubicación en el programa Las Variables Locales rompen con el paradigma de que el flujo de datos debe ser usado en un orden concreto. Sometimes you may need to access a front panel object from more than one place on the block diagram or to pass data between structures that cannot be connected by a wire. To accomplish these tasks, you would use a local variable. Local variables are located in the Structures subpalette of the Functions palette. When you place a local variable on the diagram, it contains by default the name (owned label) of the first object you placed on the front panel. You use a local variable by first selecting the object you want to access. You can either click on the local variable with the Operating tool and select the object (by owned label) you want to access, or pop up on the local variable and choose the object from the Select Item menu. Next, you must decide to either read or write to the object. Right click on the local variable and choose Change To Read or Change to Write. © National Instruments Corporation 75	Introduction to LabVIEW Hands-On
Estructuras de SecuenciaEn la subpaleta de control de ejecución (Execution Control) de la paleta de funciones. Ejecuta diagramas secuencialmente. Presionar botón derecho del mouse para agregar un nuevo cuadro (frame). En un lenguaje basado en texto ,las declaraciones del programa se ejecutan en el orden en que aparecen. En un flujo de datos, un nodo se ejecuta cuando los datos están disponibles en todas sus terminales de entrada. Algunas veces es difícil determinar el orden exacto de ejecución. A menudo, ciertos eventos deben suceder antes que otros. Cuando usted necesite controlar el orden de la ejecución del código en su diagrama de bloques, usted puede utilizar una Sequence Structure (estructura de secuencia). Estructura de Secuencia: Usada para controlar el orden en que los nodos en un diagrama se ejecutaran. In la paleta de Execution Control (control de ejecución). Luce como un pedazo de rollo de película. Utilizada para ejecutar diagramas secuencialmente. Hacer clic derecho con el mouse en los bordes para crear un nuevo cuadro.
Ejemplo: Realizar un programa para indicar las secuencia de encendido de las luces de un semáforo. La luz verde debe durar 5 segundos y la luz ámbar 3 segundos.
Almacenamiento de datos en un archivo.Para guardar datos en un archivo nuevo se han de seguir tres pasos: Crear un archivo nuevo Escribir los datos en el archivo Cerrar el archivo También podemos abrir uno ya existente y añadir información al final de ese fichero o sobrescribir su contenido.
Abrir Abrir/crear Crear/remplazar Crear Abrir (solo lectura)Para crear un archivo nuevo, hay que proporcionar el nombre del archivo y de la localización del archivo (pathname) dentro del VI. Indicar que función se quiere hacer con el archivo: Abrir Abrir/crear Crear/remplazar Crear Abrir (solo lectura) Dicha función devuelve un número de referencia (refnum), este parámetro es un identificador de archivo generado cuando se crea un archivo o se abre uno ya existente, identifica el archivo en la operaciones subsiguientes. Este número de referencia hay que pasárselo de una función File I/O a otra. un directorio y escribir el nombre del archivo.
Modelos para Programación de Archivos E/SAbre / Crea / Remplaza Archivos Lee y / o Escribe Archivos Cierra Archivos Busca Errores Programming Model for the Intermediate File VIs This same programming model applies to data acqusion, instrument control, file I/O, and most other communication schemes. In most instances you will open the file or communication channel, read and write multiple times, and then the communication will be closed or ended. It is also good programming practice to check for errors at the end. Remember this programming model when you move on to more advanced programming or look inside DAQ, communication, or file I/O Express VIs. File I/O VIs and Functions Use the File I/O VIs and functions to open and close files, read from and write to files, create directories and files you specify in the path control, retrieve directory information, and write strings, numbers, arrays, and clusters to files. Use the high-level File I/O VIs located on the top row of the palette to perform common I/O operations, such as writing to or reading from various types of data. Acceptable types can include characters or lines in text files, 1D or 2D arrays of single-precision numeric values in spreadsheet text files, 1D or 2D arrays of single-precision numeric values in binary files, or 16-bit signed integers in binary files. Use low-level File I/O VIs and functions located on the middle row of the palette, and the Advanced File Functions to control each file I/O operation individually. Use the principal low-level functions to create or open, write data to, read data from, and close a file. You also can use low-level functions to create directories; move, copy, or delete files; list directory contents; change file characteristics; or manipulate paths. Refer to the NI Developer Zone for more information about choosing a file format. Introduction to LabVIEW Hands-On 50	ni.com
Recuperación de datos desde un archivo.Cuando se lee desde un archivo, normalmente lo abrimos, leemos el contenido en forma de string o cualquier otro tipo de datos y cerramos el fichero. Los datos se deben leer con el mismo formato con el que se almacenaron. Indicar el número de datos a leer del archivo.
Manejo de Archivos (File I/O)Manejo de Archivos – transferir datos a y desde archivos - Los Archivos pueden ser binarios, texto u hojas de calculo - Escritura/Lectura de Archivos de Medición de LabVIEW (LabVIEW Measurements – *.lvm) Escritura en Archivo LVM Lectura de un Archivo LVM Operaciones de Entradas y Salidas de archivo (File I/O) pasan datos desde y hacia archivos. En LabVIEW, usted puede utilizar funciones de archivo I/O para: Abrir y cerrar archivos de datos. Lectura de datos desde y escribir datos a archivos Lectura y escritura de datos a documentos en formato de hoja de calculo (spreadsheet) Mover y retitular los archivos y directorios Cambiar las características del archivo Crear, modificar, y leer archivos de configuración Escribir a o leer en archivos de Medición LabVIEW (LabVIEW Measurement Files). En este curso examinaremos cómo escribir a o leer desde archivos de Medición LabVIEW (archivos *.lvm).
Escritura de Archivos de Medición LabVIEW (LabVIEW Measurement)Incluye las funciones de apertura, escritura, cierre y manejo de errores Maneja el formateo de cadenas de caracteres (strings) ya sea con delimitación por medio de tab o de coma La función de concatenar señales es usada para combinar datos en datos de tipo dinámico El archivo de escritura LVM puede escribir a archivos de hoja de calculo (spreadsheet). Sin embargo, su propósito principal es el de introducir datos, que luego seran utilizados en LabVIEW. Este VI crea un archivo .lvm que se puede abrir en una aplicación de hoja de calculo. Para los archivos de hoja de calculo simple, utilice el VIs expreso: Write LVM y Read LVM.
Ejemplo– Analizando e Introduciendo DatosLos Estudiantes construyen el VI Logger de Temperatura.vi Tiempo de completación estimado: minutos. Instrucciones: Cree un VI que adquiera y exhiba datos de temperatura a una velocidad fija hasta que el usuario lo detenga. Si usted ha terminado el ejercicio 2 y tiene una tarjeta DAQ, utilice Thermometer.vi para obtener sus datos. Si usted no ha terminado el ejercicio ni tiene una tarjeta de DAQ, usted puede utilizar el Digital Thermometer.vi de la subpaleta Tutorial de la paleta de funciones. Una vez que esté parado, el VI debe realizar análisis en los datos que recogió mientras estaba corriendo. Acumule un arreglo de puntos y de valores en la frontera del túnel del Ciclo Mientras (While Loop). Encuentre el máximo, el mínimo, y el valor medio de los datos de la temperatura y exhíbalos en indicadores numéricos (la función Mean (media) puede ser encontrada en Functions» Analyze» Mathematics» Probability and Statistics, y la función Array Max & Min (máximo y mínimo de un arreglo) puede ser encontrada en Functions» Array). Utilice el Write LabVIEW Measurements File Express VI (VI de escritura expreso de documentos de medición LabVIEW) que puede ser encontrado en Functions» Output. Una vez que se corra, verifique que el archivo fue creado correctamente abriéndolo en Notepad o creando un VI que lo lea de nuevo utilizando el Read LabVIEW Measurements File. Una vez que haya terminado el ejercicio, guarde su VI como Logger de Temperatura.vi.
Sección VII – Cadenas de Caracteres, Clusters, & Manejo de ErroresCreando Clusters Funciones de los Clusters I/O Error Haga click para adicionar notas.
Cadenas de caracteres Una cadena es una secuencia de caracteres desplegables o no desplegables (ASCII) Tienen muchos usos – despliegue de mensajes, Control de instrumentos, Archivos de entrada y Salida (I/O) El controlador/indicador de cadenas se encuentra en Controls »Text Control o Text Indicator Una cadena es una secuencia de caracteres desplegables o no desplegables (ASCII). Las Cadenas se usan frecuentemente para enviar comandos a instrumentos, para proveer información acerca de pruebas (tales como el nombre del operador y la fecha), o para desplegar resultados a el usuario. Los Controles e indicadores de Cadena se encuentran en la subpaleta Text Control o Text Indicator de la paleta de Controles. Introduzca o cambie texto utilizando la herramienta de texto (Text Tool) y presionado con el mouse el control de cadena (String Control). Las Cadenas pueden variar en su tamaño. Las Cadenas de Indicadores y Controles pueden tener barras de movimiento (Scroll Bars): Haga clic con el botón derecho del mouse y seleccione Visible Items» Scrollbar. La barra de movimiento no se activara si el Control o el Indicador no son lo suficientemente grandes.
Clusters Estructura de Datos que agrupa datos juntos.Los Datos pueden ser de diferentes tipos. Semejante a una estructura (struct) en C++ Los elementos deben de ser ya sea, todos Controles o todos Indicadores Se puede considerar como un conjunto de alambres agrupados en un cable. Los Clusters, agrupan componentes iguales o desiguales juntos. Es equivalente a un Record en Pascal o una Struct en C. Los Componentes de los Clusters pueden ser de diferentes tipos de datos. Ejemplos: Información de Error—Agrupando una señal de error Booleano, un codigo de error numerico, y un cadena de origen de un error para especificar el error exacto. Información de Usuario—Agrupando una cadena indicando el nombre del usuario y su numero de identifiacion (ID) especificando el codigo de seguridad. Todos los elementos de un Cluster deben ser ya sea Indicadores o Controles. No se puede tener una Control de Cadena y un Indicador Booleano. Los Clusters se pueden considerar como un grupo de alambres individuales (datos) juntos en un cable (cluster).
Creando un Cluster Seleccionar la pre-forma (shell) del ClusterControls >> All Controls >> Array & Cluster 2. Colocar Objetos dentro de la pre-forma (shell) Demostrar como crear un cluster en el panel frontal escogiendo Cluster desde la paleta Controls » All Controls » Array & Cluster. Esta opción le brinda una pre-forma (shell) parecida a la pre-forma obtenida al crear arreglos. Uno puede cambiarle el tamaño a la pre-forma una vez esta se ha seleccionado. Haga clic derecho adentro de la pre-forma para añadir objetos de cualquier tipo. Nota: Usted inclusive puede tener un cluster adentro de un cluster. El cluster se convierte en un Indicador o Control basado en el primer objeto que fue ingresado dentro del Grupo. Usted también puede crear una constante de cluster en el diagrama de bloque escogiendo Cluster Constant de la paleta Cluster. Esto le brinda una pre-forma (shell) de cluster vacía. Usted puede darle cualquier tamaño al cluster una vez ingresado. Puede Introducir otras constantes dentro de la pre-forma (Shell). Nota: Usted no puede colocar terminales para los objetos del panel frontal en una constante de cluster en el diagrama de bloques, ni tampoco puede colocar constantes especiales como el Tab o una Cadena de caracteres vacías dentro de una pre-forma de cluster vacia.
Funciones del Cluster Están ubicadas en la subpaleta de Cluster de la paleta Functions>>All functions. También puede ser accesada haciendo clic con el botón derecho del mouse en la terminal del Grupo. (Las etiquetas de la terminal reflejan los tipos de datos) Bundle (Agrupar) Función de Bundle (agrupación)—Forma un cluster conteniendo los objetos dados (explique el ejemplo). Función Bundle by Name (agrupe por nombre) —Actualiza valores de objetos localizados en grupos especificos (los objetos deben tener su propia clasificacion). Nota: Usted debe de tener un cluster existente cableado a la terminal de en medio de la función para poder utilizar Bundle by Name. Bundle by name (Agrupar por nombre)
Funciones del Cluster Unbundled (separar) Unbundled by name(separar por nombre) Función Unbundle (desagrupar)—Se utiliza para accesar a todos los objetos del cluster. Función Unbundle by Name (desagrupar por nombre)—Se utiliza para accesar objetos específicos (uno o mas) en el cluster. Nota: Solamente los objetos en el cluster que tengan su propia marca (Label) pueden ser accesados. Cuando se desagrupe por nombre, haga clic en la terminal con la herramienta de operación para escoger el elemento que quiere accesar. La función Unbundle (desagrupar) debe tener exactamente el mismo numero de terminales como se tienen elementos en el cluster. Añadiendo o Removiendo elementos en el cluster rompe cables en el diagrama. Usted también puede obtener las funciones de Bundle (agrupar), Unbundle (desagrupar), Bundle by Name (agrupar por nombre), y Unbundle by Name (Desagrupar por Nombre) al hacer clic derecho con el mouse en la terminal del cluster en el diagrama de bloques y eligiendo Cluster Tools (herramientas de cluster) del Menú. Cuando usted escoge Cluster Tools las funciones Bundle (agrupar) y Unbundle (desagrupar) automáticamente contienen el numero correcto de terminales. Las funciones Bundle by Name (agrupar por nombre) y Unbundle by Name (desagrupar por nombre) aparecen con el primer elemento del Cluster. Ejemplo de un Cluster separado (Unbundled Cluster)
Errores de Cluster Los Errores de Cluster contienen la siguiente información: Booleano (Boolean) Para reportar si surgió algún error Enteros (Integer) Para reportar errores específicos de códigos Cadena (String) Para brindar información acerca del error Los clusters para errores son un medio poderoso para manejar errores. Los VI de adquisición de datos (DAQ VIs) las funciones de manejo de entradas y salidas de archivo (File I/O), los VI de networking, y muchos otros VI utilizan este método para pasar información acerca de errores entre nodos. El cluster para errores contiene los siguientes elementos: status, Un Booleano al que se le asigna un valor verdadero si ocurre algún error. code, Un valor numerico al que se le asigna un numero de código correspondiente al error que ha ocurrido. source, Una cadena de caracteres que identifica el VI en donde ocurrió el error.
Técnicas de Manejo de ErroresLa información de un Error es transmitida de un subVI al siguiente Si un Error ocurre en un subVI, todos los subVI subsecuentes no serán ejecutados de la manera usual Los errores de Cluster contienen todas las condiciones del Error Manejo automático de Errores Los clusters para errores son útiles para determinar la ejecución de un subVI cuando se encuentra un error. Note también que los clusters para errores pueden ser útiles para determinar el flujo del programa debido al paradigma de programación del flujo de datos. Esto puede ser útil especialmente al configurar un muestreo en mas de una tarjeta de adquisición de datos (DAQ board) simultáneamente. La función de Unbundle by Name (desagrupar por nombre) muestra los componentes de un cluster de error. Errores de Cluster
Sección VIII – Estructuras de Caso y Secuenciales, Nodos de Formula
Estructuras Caso En la subpaleta de estructura de las paleta de funciones. Encerrar los nodos o arrastrarlos hacia adentro de la estructura. Colocados como un juego de cartas (naipes), solo un caso es visible a la vez. Functions >> Execution control Las estructuras de caso nos permiten elegir un curso de acción dependiendo del valor de entrada. En la subpaleta de Execution Control (control de ejecución) de la paleta de Functions (funciones). Es semejante a los enunciados if-then-else (si-entonces-de otro modo) en otros lenguajes. Como una baraja de naipes (cartas). Solo se puede ver un caso a la vez. Ejemplo 1: Entrada Booleana: Caso simple if-then (si-entonces). Si el valor booleano de la entrada es verdadero (TRUE), el caso verdadero se ejecutara; de otra manera el caso falso (FALSE) se ejecutara. Ejemplo 2: Valor numérico de entrada. El valor de entrada determina cual caso (caja) se ejecutara. Si esta fuera de rango de los casos LabVIEW escogerá el caso por omisión (DEFAULT). Ejemplo 3: Entrada de cadena de caracteres. Como el caso de los valores numéricos de entrada, el valor de la cadena de caracteres determina cual caso (caja) se ejecutara. Haga hincapié en que el valor debe de ajustarse exactamente al de los casos o sino la estructura ejecutara el caso por omisión.
Sección X – Arquitectura Básica de ProgramaciónArquitectura VI de un simple Arquitectura de un VI General Arquitectura de maquina de estado Usted puede estructurar los VIs dependiendo de la funcionalidad que desea que tengan. En esta sección discutiremos algunas de las arquitecturas de VI mas comunes, a si como sus ventajas/desventajas: simple, general, maquina de estado (state machine).
Arquitectura de un VI simpleVI funcional que produce resultados al ser ejectado No tiene opciones para “comienzo” o “fin” Adecuado: para pruebas de laboratorio, cálculos Ejemplo: Convertir C a F.vi Arquitectura de VI Simple Cuando se hacen pruebas de laboratorio rápidas, no se necesita una arquitectura complicada. Su programa puede consistir de un solo VI que toma una medición, hace cálculos, y ya sea despliega los resultados o los guarda en el disco. La medición puede ser iniciada cuando el usuario le da clic a la flecha de inicio (run arrow). Aparte se ser usada para aplicaciones simples, esta arquitectura es usada para componentes “funcionales” dentro de aplicaciónes mas grandes. Usted puede convertir estos VIs simples en subVIs que son utilizados como componentes de una apicacion mas grande.
Arquitectura de un VI GeneralTres pasos principales Inicio Aplicación principal Cierre (shutdown) Arquitectura de un VI General Generalmente al diseñar una aplicación se siguen tres pasos: Inicio (startup)—Esta area se usa para inicializar el hardware, leer información de configuración de archivos o preguntar al usuario la localización de datos de archivos. Aplicación Principal (main application)—Generalmente consiste de por lo menos un ciclo que se repite hasta que el usuario decida salir del programa, o el programa termina debido a otras razones como la completacion de una entrada/salida (I/O) Apagado (shutdown)—Esta seccion usualmente se encarga de cerrar los archivos, escribir datos de configuracion al disco o reestablecer los datos de entrada/salida (I/O) a su estado predeterminado. Para aplicaciones simples, el ciclo de la aplicación principal puede ser bastante directo. Cuado se tiene interfaces de usuario complicadas o eventos múltiples (acción de usuario, accionamientos de entradas/salidas, etc.), esta sección puede volverse mas complicada.
Máquinas de estado Muestras completas Entrada AdquisiciónEstado presente Muestras completas Entrada Filtrado Estado siguiente
Tiempo Case Stop While Inicialización de variables. Abrir archivos.Inicio Estado 1 Inicialización de variables. Abrir archivos. Abrir y configurar puertos de comunicación. Cerrar archivos. Cerrar puertos de comunicación. Estado 3 Case Stop While Fin
Máquinas de estado Entrada 1 Entrada 2 Entrada 4 Entrada 3 STOP InicioFin STOP
Arquitectura de Maquina de EstadoVentajas Puede ir de cualquier estado a otro Fácil de modificar y depurar Desventajas Se pierden eventos si dos ocurren al mismo tiempo Pasos: 0: Inicio (Startup) 1: Ocio (Idle) 2: Evento 1 3: Evento 2 4: Cierre (Shutdown) Arquitectura de un VI de Maquina de Estado Usted puede hacer sus diagramas mas compactos con el uso de una estructura de caso simple para manejar todos sus eventos. En este modelo usted escanea (scan) la lista de posibles eventos, o estados, y luego correlaciona (map) estos a un caso. Para el VI mostrado arriba, los estados posibles son inicio (startup), ocio (idle), evento 1, evento2 y apagado (shutdown). Estos estados se encuentran guardados en una constante enumerada. Cada estado tiene su propio caso en el lugar que usted coloque los nodos apropiados. Mientras se ejecuta un caso, el siguiente caso es determinado por el resultado actual. El siguiente caso que se ejecutara es guardado en el registro de desplazamiento (shift register). Si ocurre un error en cualquier estado, el caso de apagado (shutdown) el llamado. La ventaja de este modelo es que su diagrama se puede reducirse significativamente (izquierda a derecha), haciendo que sea mas fácil de lee y depurar. Una desventaja de la estructura de secuencia es que no puede saltarse o salirse de un caso. Este método resuelve ese problema ya que cada caso determina cual será el próximo caso que se ejecutara. Una desventaja de esta técnica es que con la arquitectura simple se pueden perder eventos. Si dos eventos ocurren al mismo tiempo, este modelo maneja solo el primero, y el segundo se pierde. Esto puede conducir a errores que son difíciles de depurar porque solo ocurren ocasionalmente. Existen versiones mas complejas de la Arquitectura del VI de la Maquina de Estado que contiene código extra para construir un queue de eventos (estados) de manera que usted no pierda ningún evento.
Ejemplo – Máquina de EstadoTiempo estimado de duración: 25 minutos. En este ejercicio crearemos un VI usando la arquitectura de Maquina de Estado. El VI tendrá un estado de ocio (idle), en donde espera por entradas. Cuando el usuario presiona un botón, el VI se ira al estado 1 (State 1). El estado 1 genera una caja de dialogo que le permite al usuario proceder a ir al estado 2 (State 2) o volver a empezar. Desde el estado 2 el usuario puede decidir cerrar el programa o iniciar nuevamente. En lugar de empezar desde cero, utilizaremos una Plantilla de VI para crear nuestra maquina de estado. Desde el File menu, en lugar de seleccionar New VI, seleccione New…, y haga clic en Start from template. Busque …\Program Files\National Instruments\LabVIEW 6.1\templates\State Machine.vit, Y abralo. Esto abrira una plantilla simple de maquina de estado utilizando cadenas de caracteres para controlar el estado. Examine la plantilla, y guardela en otro directorio antes de empezar a trabajar en ella. El primer estado que haremos es el de inicialización. Note que este estado ya ha sido creado para nosotros, y es el estado predeterminado (default). En el caso de inicialización, coloque un One Button Dialog (Functions » Time & Dialog), y cablee “Initialized. Clic OK para proceder.” en la terminal de mensajes. Reemplace el texto “Next State” con “Idle”.
Haga clic derecho en la estructura de casos, escoja Add Case After, y déle el nombre de “Idle”. Dentro del caso Idle, coloque una segunda estructura de caso. Cablee un Booleano llamado “Begin Test” a la terminal de selección de la estructura de caso anidada, y cablee una constante de cadena de caracteres con el valor “Test” de el caso verdadero a el registro de desplazamiento (Shift register). Esta configuracion mantendra la maquina de estados en el estado Idle hasta que el usuario proceda a presionar el boton para iniciar la prueba. Haga clic derecho en el tunel de salida del Booleano y escoja Create » Constant. Cambie el valor constante a True (verdadero). Haga clic derecho en la estructura principal y adicione otro caso. Llame a este caso “Test”. En el caso Test, coloque un Two Button Dialog, un boton doble de dialogo, (Functions » Time & Dialog). Conecte el Texto “The test has run. Click OK to quit, Cancel to test again.” con la terminal de mensaje. Coloque otra estructura de caso adentro del caso Test, y conecte “T button?” la salida de el Two Button Dialog con la terminal seleccionada. Haga una conexión del valor “Idle” desde el caso falso hasta el registro de desplazamiento (Shift register), y una conexioncon costante Booleana Verdadera con la terminal del ciclo mientras (while loop). El caso Verdadero debera de contener una cadena de caracteres constante vacia, y un valor Boleano Falso conectado con la condicion de terminacion. Esto terminara el programa si el usuario presiona OK, y regresara al estado idle (ocio) si el usuario presiona Cancel. No Borrar Esta Filmina
Ejercicio 5 – Aplica lo aprendidoExercise 5 – Apply What You Have Learned (Track A, B, & C) In this exercise, you will create a VI that uses what you have learned. Design a VI that does the following: Acquire data from your device and graph it (either your DAQ device, your simulated device, or your sound card). Filter that data using the Filter Express VI (Functions»Express»Signal Analysis»Filter). There should be a front panel control for a user configurable cut-off frequency. Take a Fast Fourier Transform to get the frequency information from the filtered data and graph the result. Use the Spectral Measurements Express VI (Functions»Express»Signal Analysis»Spectral). 4. Find the dominant frequency of the filtered data using the Tone Measurements Express VI. Compare that frequency to a user inputted limit. If the frequency is over that limit, light up an LED. If you have a USB-6009, light up the LED on your hardware using the DAQ Assistant. You will need to invert the digital line for the LED to light up when over the limit. You can specify this in the configuration window of the DAQ Assistant or with a “not” boolean function. If you get stuck, open up the solution or view it at the end of this manual. Tomar una muestra cada segundo Almacenar en archivo .txt: Temp Tmin Tmax Tipo de Alarma (End of Exercise) Introduction to LabVIEW Hands-On 68	ni.com
Sección XI – Panel Frontal RemotoObservar y Controlar Paneles Frontales de LabVIEW desde un Navegador en la Web No Requiere Programación Usuarios en lugares Remotos pueden ver en Tiempo Real las Actualizaciones del Panel Frontal Múltiples clientes pueden Observar el Panel Frontal Simultáneamente Solo un Cliente Puede Controlar el Panel Frontal a la vez En LabVIEW, una opción llamada panel frontal remoto le permite al usuario observar y controlar el panel frontal de un VI remotamente desde LabVIEW o un navegador sin necesidad de programación. El usuario puede correr una aplicación o revisar el progreso desde cualquier lugar. Además diferentes usuarios en diferentes lugares pueden tener acceso simultaneo al VI. Los usuarios remotos pueden observar el panel frontal en “vivo” (en tiempo real), en lugar de imágenes estáticas. Pueden existir múltiples clientes observando el mismo panel frontal. Sin embargo, solo un cliente puede controlar el panel frontal a la vez.
Herramienta para la Publicación de un Panel RemotoTools » Web Publishing Tool… Clic Salvar a Disco y el VI es embebido a un archivo HTML Después que el archivo a sido grabado, puede ser reabierto y personalizado en cualquier editor HTML Para incrustar su VI en una pagina web, simplemente abra la herramienta de publicación web (Web Publishing Tool) en herramientas del menú. Web Publishing Tool crea un simple archivo html con un panel frontal incrustado en el. Con esta herramienta usted puede crear un titulo y colocar texto antes y después de colocar el panel frontal. Una vez creado el archivo html, se puede observar en un navegador web y ser grabado al disco. Salvar el archivo coloca el html en su directorio www de LabVIEW, que es el predeterminado para el servidor web de LabVIEW. Usted puede iniciar su servidor web desde Web Publishing Tool o desde Tools » Options… » Web Server : Configuration. (Nota: en Tools » Options… menu usted tambien puede configurar el derecho de acceso y VIs expuestos para su servidor web. Si usted desea personalizar su pagina web mas allá de lo que provee la herramienta de publicación web (Web Publishing Too)l, simplemente abra el archivo html en otro editor de html, y personalice alrededor del código del panel embebido para LabVIEW.
El estándar RS-232 El RS232 ó Recommended Standard 232, está definido en las especificaciones ANSI (American National Standard Institution) como la interface entre un equipo terminal de datos y un equipo de comunicación de datos empleando un intercambio en modo serie de datos binarios. Normalmente, los dispositivos que intervienen en una comunicación serie son el Equipo Terminal de Datos (ETD), que suele ser una PC y el Equipo de Comunicación de Datos (ECD) que suele ser un modem ó un periférico.
Utilización del puerto serie mediante LabVIEW.LabVIEW proporciona herramientas de gran utilidad para el manejo del puerto serie. Todas las funciones que son necesarias a la hora de realizar una comunicación serie entre la PC y un periférico se encuentran ya programadas en forma de Instrumentos Virtuales (VIs). De esta forma, la utilización del puerto serial es transparente al programador de LabVIEW.
Es importante tener en cuenta que los pasos a seguir al utilizar el puerto serie los siguiente:Realizar la configuración del puerto serie, inicializándolo según las características que se deseen para la comunicación. Acceder al puerto serie para recibir o enviar datos tantas veces como se desee. Cerrar el puerto serie para dejarlo libre para otra aplicación.
Configuración del puerto serie.Aquí hay que indicar a la PC cómo ha de actuar en las comunicaciones, es decir, qué puerto se ha de utilizar, con qué velocidad de transmisión ha de emitir y recibir bytes, que tipo de paridad, etc.
Lectura del puerto serieUna vez configurado el puerto serie y establecida la comunicación es posible que el periférico haya transferido datos a la PC. Para el programador, este proceso de recepción de datos es transparente, es decir, es el propio puerto serie quien se encargará de gestionar la comunicación con el periférico si éste desea enviar información. Una vez terminada la transferencia, los datos recibidos quedan almacenados en el buffer de recepción, en espera de que sean leídos Ahora bien en el momento en que se quiera acceder a la información recibida para poder tratarla, es necesario programar un acceso de lectura al puerto.
Esta lectura se realiza con el VI VISA Read
Escritura del puerto serie.El método que se sigue para transferir información a un periférico es utilizar el icono VISA write.
Ejemplo Realizar un programa para leer información por el puerto serial de un lector de RF (Microreader de TIRIS), el cual cada que recibe un TAG valido, envía por su puerto serial (9600,8,N,1) una trama de 12 bytes. De la trama obtenida, sacar el TAG ID y compararlo con uno conocido, si son iguales, almacenar en un archivo el ID, la hora y la fecha e indicar con una variable boolena que el ID coincide.
Byte Contenido Comentario Descripción01 Start Mark 1 09 Length 9 bytes seguidos, excluyendo BCC 2 0C Status Tag valido 3 ** Data Field (1) Tag ID (LSB) 4 Data Field (2) 5 Data Field (3) 6 Data Field (4) 7 Data Field (5) 8 Data Field (6) 9 Data Field (7) 10 Data Field (8) Tag ID (MSB) 11 BCC Detección de error
Qué hago Ahora? Ejemplos de Programas (Help» Find Examples…)LabVIEW Edición de Estudiantes (www.ni.com/labviewse) Recursos Web (ni.com) Zona de Desarrollo NI (zone.ni.com) Notas de Aplicaciones grupo de noticias labview (www.info-labview.org/) Instrument Driver Library (www.ni.com/idnet) Que hago ahora? National Instruments ofrece una amplio rango de cursos de enseñanza para expandir tu conocimiento. Por favor visite ni.com/academic para obtener información de programas y recursos disponibles. LabVIEW Edición de Estudiantes esta disponible en nuestra pagina Web. Incluye aprendiendo con LabVIEW, un libro de texto escrito por el Dr. Bob Bishop de la universidad de Austin Texas. La web es el mejor lugar para buscar información sobre LabVIEW. NI.com ha sido diseñado para ser el recurso principal para encontrar información. La zona de desarrollo NI (“NIDZ”) es un lugar para que los desarrolladores se reunen, discutan temas de diseño y comenten sobre diversos temas. Notas de aplicación pueden ser descargadas de ni.com/support en una gran cantidad de temas. Grupo de Noticias LabVIEW (Info-labview) es un grupo manejado por terceras personas. Existe una extensa librería de drivers de instrumentos de LabVIEW disponibles para ser descargados de NIDZ
Descargar ppt "Instrumentación Virtual con LabVIEW"
Presentaciones similares Lección 1 Introducción. Ventanas de Inicio Clic.
Lección 2 Creación de un SubVI. SubVIs SubVIs Un SubVI es un VI que puede ser utilizado dentro de otro VI Similar a una subrutina Ventajas ◦ Modular.
INTRODUCCIÓN A LA INSTRUMENTACIÓN VIRTUAL LA INSTRUMENTACIÓN VIRTUAL.

References: Resolución 
 resolución 
 resolución 
 resolución 
 resolución 
 resolución 
 resolución 
 resolución 
 resolución 
 resolución