Source: https://cuthbertsonlaird.co.uk/acdc-clamp-meters/134-fluke-355-true-rms-2000-a-clamp-meter-.html
Timestamp: 2019-04-24 09:47:54+00:00

Document:
Includes: Fluke 355, C43 soft carrying case, 6 AA batteries, TL224 1.5 m silicone rubber test leads, TP2 test probes, AC285 alligator clips, User manual.
Confidently take reliable readings with the True-rms, Fluke 355 digital clamp meter; the tool of choice for high current clamp-on amp meter measurement up to 2000 A.
Reliably handle a wide range of high-current applications with 2000 A AC + DC True RMS, 1400 A AC, and 2000 A DC.
The large 58 mm (2.3 in) jaw capacity is suitable for large or multiple conductors.
CAT IV 600 V, CAT III 1000 V rating for added user protection.
In-rush current measurement captures ‘power-on’ surge current with accuracy and repeatability.
High voltage measurement of 1000 V AC + DC True RMS, 600 V AC, and 1000 V DC allows the user to perform multiple tests with only one tool.
Resistance to 400K ohms coupled with a continuity beeper provide the convenience of a multimeter in a clamp meter.
Accurately measure frequency up to 1 kHz for optimum troubleshooting.
Quickly analyze readings using the MIN, MAX, and AVG functions.
A large backlit display allows for easy visibility in low-lit areas.
Use the display hold feature to capture readings even when the display cannot be viewed.
Use the low-pass filter to smooth out noisy loads and stabilize readings.
Current measurement DC and AC 10 Hz to 100 Hz Range 40 A / 400 A / 2000 A / 1400 AC rms.
Resolution 10 mA / 100 mA / 1 A.
A: 1.5% rdg + 5 digits.
Trigger level for Inrush 0.50 A / 5.0 A / 5 A.
Trigger level for Hz filter OFF 2.50 A / 2.5 A / 8 A.
Trigger level for Hz filter ON 0.50 A / 2.5 A / 8 A.
Crest Factor (50/60 Hz) Range 40 A / 400 A / 2000 A / 1400 AC rms.
2 @ 1000 A, 2.4 @ 833 A.
Current measurement AC 100.1 Hz to 1 kHz Range 40 A / 400 A / 2000 A; 1400 AC rms.
10 A: 3.5% rdg + 5 digits.
(600 V and 1000 V ranges have 10% over range to 660 V and 1100 V respectively.) Range 4 V / 40 V / 400 V / 600 V AC rms / 1000 VDC.
Resolution 1 mV / 10 mV / 100 mV / 1 V.
1% rdg + 5 digits.
Trigger level for Hz filter OFF 0.25 V / 6 V.
Trigger level for Hz filter ON 0.25 V / 6 V.
(600 V and 1000 V ranges have 10% over range to 660 V and 1100 V respectively.) Range 4 V / 40 V / 400 V / 600 V ac rms.
3% rdg + 5 digits.
Trigger level for Hz filter OFF 0.050 V / 0.25 V / 6 V.
Trigger level for Hz filter ON 0.050 V / 0.25 V / 6 V.
Ohms measurement Range 4 kΩ / 40 kΩ / 400 kΩ.
Resolution 0.1 Ω / 1 Ω / 10 Ω / 100 Ω.
Accuracy 1.5% + 5 digits.
Off at ≥ 100 Ω.
Frequency measurement Measurement range 5.0 Hz to 1 kHz.
Resolution 0.1 Hz (15 Hz to 399.9 Hz); 1 Hz (400 Hz to 1 kHz).
Accuracy – 5.0 Hz to 100 Hz 0.2% + 2 counts.
Accuracy – 100.1 Hz to 1 kHz 0.5% + 5 counts.
Trigger level Refer to current and voltage tables.
Batteries Six 1.5 V AA NEDA 15 A or IEC LR6.
Battery life (with typical usage, backlight off) 100 hours.
Test leads Rated to 1000 V.
Weight .814 kg (1.8 lb).
Jaw size 58 mm (2.28 in).
Dimensions (L x W x D) 300 x 98 x 52 mm (12 x 3.75 x 2 in).
Safety rating IEC 61010-2-032, 600 V CAT IV, 1000 V CAT III.
Operating temperature 32°F to +122°F (0°C to +50°C).
Storage temperature -4°F to 140°F (-20°C to +60°C).
Operating humidity 0 to 95% (non-condensing).
IP rating 42 (indoor use only).
Drop test requirements 1 m.
IEC/EN 61326:1997 3 V/m, performance criteria B, EN61325.
Temperature coefficients Current 0.1% of reading per °C outside 22°C to 24°C.
Voltage 0.1% of reading per °C outside 22°C to 24°C.
Fluke 355 warranty 2 years.
PDF Specifications of the Fluke 353 & 355 2000A Clamp Meters.

References: V.

 V.

 V.

 V.

 V.

 V.