Source: https://www.law.cornell.edu/cfr/text/10/part-35/subpart-H
Timestamp: 2019-04-20 08:23:52+00:00

Document:
§ 35.600 Use of a sealed source in a remote afterloader unit, teletherapy unit, or gamma stereotactic radiosurgery unit.
§ 35.604 Surveys of patients and human research subjects treated with a remote afterloader unit.
§ 35.605 Installation, maintenance, adjustment, and repair.
§ 35.610 Safety procedures and instructions for remote afterloader units, teletherapy units, and gamma stereotactic radiosurgery units.
§ 35.615 Safety precautions for remote afterloader units, teletherapy units, and gamma stereotactic radiosurgery units.
§ 35.632 Full calibration measurements on teletherapy units.
§ 35.633 Full calibration measurements on remote afterloader units.
§ 35.635 Full calibration measurements on gamma stereotactic radiosurgery units.
§ 35.642 Periodic spot-checks for teletherapy units.
§ 35.643 Periodic spot-checks for remote afterloader units.
§ 35.645 Periodic spot-checks for gamma stereotactic radiosurgery units.
§ 35.647 Additional technical requirements for mobile remote afterloader units.
§ 35.655 Five-year inspection for teletherapy and gamma stereotactic radiosurgery units.
§ 35.657 Therapy-related computer systems.
§ 35.690 Training for use of remote afterloader units, teletherapy units, and gamma stereotactic radiosurgery units.

References: § 35

§ 35

§ 35

§ 35

§ 35

§ 35

§ 35

§ 35

§ 35

§ 35

§ 35

§ 35

§ 35

§ 35