Source: https://www.broadcaststore.com/store/prod_detail.cfm?eq_id=448582
Timestamp: 2019-04-26 11:59:48+00:00

Document:
All in One Digital Video Analyser from SMPTE to CCIR.
Amplitude - INPUT SIGNAL MAXIMUM LIMITS: ±10 V. MEASUREMENT RANGE: ±1.5 V. RIPPLE MEASUREMENT RANGE (Vrp): <1.5 V. Vrp FREQUENCY RANGE: 0 to 10 kHz. MEASUREMENT RESOLUTION: 12 mV. TEST OUTPUT FOR MEASUREMENTS: Vpp and Vrp identical value to voltage measured at 75 Ohms.
Jitter - PARALLEL SIGNAL JITTER RANGE: ±7.5 ns. SERIAL SIGNAL JITTER RANGE: ±1.5 ns. SERIAL SIGNAL JITTER RESOLUTION: 40 pS +10 pS with 10 m of cable. PARALLEL SIGNAL JITTER BANDWIDTH: 1.4 kHz. PARALLEL TEST OUTPUT JITTER: 100 mV/ns at 75 Ohms. SERIAL TEST OUTPUT JITTER: 300 mV/ns at 75 Ohms. MISCELLANEOUS - MON OUTPUT: 720 mV at 75 Ohms, DC component not transmitted. CLOCK OUTPUT: Parallel signal clock 400 mVpp at 75 Ohms, DC component not transmitted.
Measurements Common To Serial And Parallel Signals - PREAMBLE ERROR DETECTION: The analyser has a "hard" preamble detector, which allows detection of a preamble error on the 4 least significant bits. TEST OUTPUT LEVELS FOR DISPLAY OF RESERVED CODES AND TRS ERRORS: 700 mV at 75 Ohms.
SERIAL SIGNAL: Signal absence or TRS word error frequency of more than 10 errors in 500 ms. PARALLEL SIGNAL: When clock/data phase measurement shows clock phase to be outside the range of ±11 nS. SECONDARY ALARM THRESHOLD: When one parameter falls outside specifications. SERIAL LINK: Adjustable from 2,400 to 19,200 Baud; delivered set at 9,600 Baud. OTHER PARAMETERS: 8 bits, 1 start bit, no parity.
Amplitude - INPUT SIGNAL MAXIMUM LIMITS: 0 V to -5.2 V. CLOCK SIGNAL MEASUREMENT RANGE: -1.29 V ± 1.29 V. VAV DATA MEASUREMENT RANGE: -1.29 V ±1.29 V. MEASUREMENT RESOLUTION: 11mV. MEASUREMENT PRECISION: ±25 mV. TEST OUTPUT FOR VPP AND VRP MEASUREMENTS: Vpp value identical to the measured signal. VAV: Displayed value is measured value +1.79V.
Clock/Data Phase - PRECISION: ±1.5 ns. TEST OUTPUT LEVEL: 700 mV at 75 Ohms.
Outputs - 1 parallel digital output to CCIR Rec. 656, 2 270 Mb/s serial digital outputs at 75 Ohms.
Calibrated errors on parallel signal - Level errors can be introduced in the clock signal and on the two most significant bits. PEAK-TO-PEAK LEVEL ERROR: Vpp minimum = 0.25 V, ±0.1 V, Vpp nominal = 1.1 V, ±0.1 V. AVERAGE LEVEL ERROR: Vav min = 1.60 ±0.05 V, Vav nom = 1.29 ±0.05 V, Vav max = -0.95 ±0.05 V. RIPPLE VOLTAGE ERROR: 1V ±10% at 122 Hz or 7812 Hz. CLOCK/DATA PHASE ERROR: -12 ns to +16 ns in steps of 4 ns, ±2 ns precision for each step.
Errors common to parallel and serial signals - Ability to introduce reserved codes as well as errors in TRS word and preambles. JITTER ERROR: 4 ±1 ns at 122 Hz or 976 Hz.
Patterns - PATTERNS AVAILABLE: 34. OF WHICH: CCIR - 14 patterns, user configurable - 2 patterns, others - 18 patterns. ACCESSIBLE VIA FRONT-PANEL CONTROLS: 16 patterns. ACCESSIBLE VIA SOFTWARE/SERVICE MODULE: 34 patterns. PATTERN 1: 75% colour bars. PATTERN 2: 100% colour bars. PATTERN 3: multi-burst. PATTERN 4: green (270 Mb/s serial test). PATTERN 5: magenta (270 Mb/s serial test). PATTERN 6: grey. PATTERN 7: black and white alternating at 0.1 Hz. PATTERN 8: end-of-lines pulses. PATTERN 9: black/white ramp. PATTERN 10: yellow/grey ramp. PATTERN 11: grey/ blue ramp. PATTERN 12: cyan/grey ramp. PATTERN 13: grey/red amp. PATTERN 14: Cb, Y, Cr, Y ramps. PATTERN 15: programmable pattern. PATTERN 16: programmable pattern. PATTERN 17: white end-of-line porches. PATTERN 18: blue end-of-line porches. PATTERN 19: red end-of-line porches. PATTERN 20: yellow end-of-line porches. PATTERN 21: cyan line end-of-line porches. PATTERN 22-34: grey.
CONSTRUCTION:2U 19" rack mounting unit. DEPTH: 328 mm (13"). WEIGHT: Basic unit - 5 kg (11 lbs), with all options - 6.2 kg (13.64 lbs). AMBIENT TEMPERATURE RANGE: +5°C to +45°C.
POWER SUPPLY: 105-130 V AC, 50-60 Hz (220 V Opt). MAXIMUM CONSUMPTION: 50 watts (all optional modules included). FUSE: 1.6 A (time-delay fuse).

References: V. 
 V. 
 V. 
 V. 
 V. 
 V. 
 V. 
 V.