CELEX: 32002R2091
Language: en
Date: 2002-11-26 00:00:00
Title: Commission Regulation (EC) No 2091/2002 of 26 November 2002 amending Regulation (EC) No 2870/2000 laying down Community reference methods for the analysis of spirits drinks

Avis juridique important

|

32002R2091

Commission Regulation (EC) No 2091/2002 of 26 November 2002 amending Regulation (EC) No 2870/2000 laying down Community reference methods for the analysis of spirits drinks  

Official Journal L 322 , 27/11/2002 P. 0011 - 0027

Commission Regulation (EC) No 2091/2002of 26 November 2002amending Regulation (EC) No 2870/2000 laying down Community reference methods for the analysis of spirits drinksTHE COMMISSION OF THE EUROPEAN COMMUNITIES,Having regard to the Treaty establishing the European Community,Having regard to Council Regulation (EEC) No 1576/89 of 29 May 1989 laying down general rules on the definition, description and presentation of spirit drinks(1), as amended by the Act of Accession of Austria, Finland and Sweden, and in particular Article 4(8) thereof,Whereas:(1) Commission Regulation (EC) No 2870/2000 of 19 December 2000 laying down Community reference methods for the analysis of spirits drinks(2) describes these methods in an Annex.(2) Four methods of analysis for the determination of trans-anethole in aniseed-flavoured spirit drinks, glycyrrhizic acid and chalcones in pastis and egg yolk in egg liqueur and liqueur with egg have been validated according to internationally recognised procedures as part of a research project supported by the Commission.(3) These four methods may be recognised as Community reference methods and must be added to the Annex to Regulation (EC) No 2870/2000.(4) The measures provided for in this Regulation are in accordance with the opinion of the Implementation Committee for Spirit Drinks,HAS ADOPTED THIS REGULATION:Article 1The Annex to Regulation (EC) No 2870/2000 is hereby amended as follows:1. In the summary of the Annex, the term "(p.m.)" under points V, VI, VII and IX shall be deleted.2. Chapters V, VI, VII and IX in the Annex to this Regulation shall be added after Chapter III.Article 2This Regulation shall enter into force on the seventh day following its publication in the Official Journal of the European Communities.This Regulation shall be binding in its entirety and directly applicable in all Member States.Done at Brussels, 26 November 2002.For the CommissionFranz FischlerMember of the Commission(1) OJ L 160, 12.6.1989, p. 1.(2) OJ L 333, 29.12.2000, p. 20.ANNEXV. ANETHOLE. GAS CHROMATOGRAPHIC DETERMINATION OF TRANS-ANETHOLE IN SPIRIT DRINKS1. ScopeThis method is suitable for the determination of trans-anethole in aniseed-flavoured spirit drinks using capillary gas chromatography.2. Normative referencesISO 3696: 1987 Water for analytical laboratory use - Specifications and test methods.3. PrincipleThe trans-anethole concentration of the spirit is determined by gas chromatography (GC). The same quantity of an internal standard, e.g. 4-allylanisole (estragole) when estragole is not naturally present in the sample, is added to the test sample and to a trans-anethole reference solution of known concentration, both of which are then diluted with a 45 % ethanol solution and injected directly into the GC system. An extraction is necessary before sample preparation and analysis for liqueurs that contain large amounts of sugars.4. Reagents and materialsDuring the analysis, use only reagents of a purity of at least 98 %. Water of at least grade 3 as defined by ISO 3696 should be used.Reference chemicals should be stored cold (at 4 oC), away from light, in aluminium containers or in tinted (amber) glass reagent bottles. The stoppers should preferably be fitted with an aluminium seal. Trans-anethole will need to be "thawed" from its crystalline state before use, but in this case its temperature should never exceed 35 oC.4.1. Ethanol 96 % vol. (CAS 64-17-5)4.2. 1-methoxy-4-(1-propenyl) benzene; (trans-anethole) (CAS 4180-23-8)4.3. 4-allylanisole, (estragole) (CAS 140-67-0), suggested internal standard (IS)4.4. Ethanol 45 % vol.Add 560 g of distilled water to 378 g of ethanol 96 % vol.4.5. Preparation of standard solutionsAll standard solutions should be stored at room temperature (15 to 35 oC) away from light in aluminium containers or in tinted (amber) glass reagent bottles. The stopper should preferably be fitted with an aluminium seal.Trans-anethole and 4-allylanisole are practically insoluble in water, and it is therefore necessary to dissolve the trans-anethole and 4-allylanisole in some 96 % ethanol (4.1) before the addition of 45 % ethanol (4.4).The stock solutions must be freshly prepared each week.4.5.1. Standard solution AStock solution of trans-anethole (concentration: 2 g/l)Weigh 40 mg of trans-anethole (4.2) in a 20 ml volumetric flask (or 400 mg in 200 ml, etc.). Add some 96 % ethanol (4.1) and make up to volume with 45 % vol. ethanol (4.4), mix thoroughly.4.5.2. Internal standard solution BStock solution of internal standard, e.g. estragole (concentration: 2 g/l)Weigh 40 mg of estragole (4.3) in a 20 ml volumetric flask (400 mg in 200 ml etc.). Add some 96 % vol. ethanol (4.1) make up to volume with 45 % vol. ethanol (4.4), mix thoroughly.4.5.3. Solutions used to check the linearity response of the flame ionisation detector (FID)The linearity response of the FID must be checked for the analysis taking into account a range of concentrations of trans-anethole in spirits from 0 g/l up to 2.5 g/l. In the procedure of analysis, the unknown samples of spirits to be analysed are diluted 10 times (8.3). For the conditions of the analysis described in the method, stock solutions corresponding to concentrations of 0, 0,05, 0,1, 0,15, 0,2, and 0,25 g/l of trans-anethole in the sample to be analysed are prepared as follows: take 0,5, 1, 1,5, 2, and 2,5 ml of stock solution A (4.5.1) and pipette in separate 20 ml volumetric flasks; pipette into each flask 2 ml of internal standard solution B (4.5.2) and make up to volume with 45 % vol. ethanol (4.4), mix thoroughly.The blank solutions (8.4) is used as the 0 g/l solution.4.5.4. Standard solution CTake 2 ml of standard solution A (4.5.1) and pipette into a 20 ml volumetric flask, then add 2 ml of internal standard solution B (4.5.2) and make up to volume with 45 % vol. ethanol (4.4), mix thoroughly.5. Apparatus and equipment5.1. A capillary gas chromatograph fitted with a flame ionisation detector (FID) and integrator or other data handling system capable of measuring peak heights or areas, and with an automatic sampler or the necessary equipment for manual sample injection.5.2. Split/splitless injector5.3. Capillary column, for example:Length: 50 mInternal diameter: 0,32 mmFilm thickness: 0,2 μmStationary phase: FFAP - modified TPA polyethylene glycol cross-linked porous polymer.5.4. Common laboratory equipment: A grade volumetric glassware, analytical balance (precision: ± 0,1 mg).6. Chromatography conditionsThe column type and dimensions, and the GC conditions, should be such that anethole and the internal standard are separated from each other and from any interfering substances. Typical conditions for the column given as an example in 5.3 are:6.1. Carrier gas: analytical helium6.2. Flow rate: 2 ml/min6.3. Injector temperature: 250 oC6.4. Detector temperature: 250 oC6.5. Oven temperature conditions: isothermal, 180 oC, run time 10 minutes6.6. Injection volume: 1 μl, split 1:40.7. SamplesSamples should be stored at room temperature, away from light and cold.8. Procedure8.1. Sample screening for estragoleTo ensure that there is no estragole naturally present in the sample, a blank analysis should be carried out without the addition of any internal standard. If estragole is naturally present then another internal standard must be chosen (for instance menthol).Pipette 2 ml sample into a 20 ml volumetric flask and make up to volume with 45 % vol. ethanol (4.4), mix thoroughly.8.2. Preparation of unknown samplesPipette 2 ml sample into a 20 ml volumetric flask then add 2 ml of internal standard solution B (4.5.2) and make up to volume with 45 % vol. ethanol (4.4), mix thoroughly.8.3. BlankPipette 2 ml of internal standard solution B (4.5.2) into a 20 ml volumetric flask and make up to volume with 45 % vol. ethanol (4.4), mix thoroughly.8.4. Linearity testPrior to the commencement of the analysis the linearity of the response of the FID should be checked by successively analysing in triplicate each of the linearity standard solutions (4.5.3).From the integrator peak areas or peak heights for each injection plot a graph of their mother solution concentration in g/l versus the ratio R for each.R= trans-anethole peak height or area divided by the estragole peak height or area.A linear plot should be obtained.8.5. DeterminationInject the blank solution (8.3), followed by standard solution C (4.5.4), followed by one of the linearity standards (4.5.3) which will act as a quality control sample (this may be chosen with reference to the probable concentration of trans-anethole in the unknown), followed by five unknowns (8.2); insert a linearity (quality control) sample after every five unknown samples, to ensure analytical stability.9. Calculation of response factorMeasure either peak areas (using an integrator or other data system) or peak heights (manual integration) for trans-anethole and internal standard peaks.9.1. Response factor (RFi) calculationThe response factor is calculated as followsRFi = (Ci/area or heighti )*(area or heightis/Cis)where:Ci is the concentration of trans-anethole in the standard solution A (4.5.1)Cis is the concentration of internal standard in the standard solution B (4.5.2)areai is the area (or height) of the trans-anethole peakareais the area (or height) of the internal standard peakRFi is calculated from the five samples of solution C (4.5.4).9.2. Analysis of the linearity response test solutionsInject the linearity response test solutions (4.5.3).9.3. Analysis of the sampleInject the unknown sample solution (8.2).10. Calculation of resultsThe formula for the calculation of the concentration of trans-anethole is the following:ci = Cis * (area or heighti/area or heightis)*RFiwhere:ci is the unknown trans-anethole concentrationCis is the concentration of internal standard in the unknown (4.5.2)Area or heighti is the area or height of the trans-anethole peakArea or heightis the area or height of the internal standard peakRFi is the response coefficient (calculated as in 9.1)The trans-anethole concentration is expressed as grams per litre, to one decimal place.11. Quality assurance and controlThe chromatograms should be such that anethole and the internal standard are separated from each other and from any interfering substances. The RFi value is calculated from the results for the five injections of solution C (4.5.4). If the coefficient of variation (CV % = (standard deviation/mean)*100)) is within plus or minus 1 %, the RFi average value is acceptable.The calculation above should be used to calculate the concentration of trans-anethole in the sample selected for the quality control from the linearity control solutions (4.5.3).If the mean calculated results from analysis of the linearity solution selected for internal quality control sample (IQC) are within plus or minus 2,5 % of their theoretical value, then the results for the unknown samples can be accepted.12. Treatment of spirits sample containing large amount of sugar and of liqueur sample prior to GC analysisExtraction of alcohol from spirit drink containing a large amount of sugar, in order to be able to determine the trans-anethole concentration using capillary gas chromatography.12.1. PrincipleAn aliquot of the liqueur sample is taken and to this is added the internal standard, at a concentration similar to that of the analyte (trans-anethole) in the liqueur. To this are added sodium phosphate dodecahydrate and anhydrous ammonium sulphate. The resulting mixture is well shaken and chilled, two layers develop, and the upper alcohol layer is removed. An aliquot of this alcohol layer is taken and diluted with 45 % ethanol solution (4.4) (Note: no internal standard is added at this stage, because it has already been added). The resulting solution is analysed in gas chromatography.12.2. Reagents and materialsDuring the extraction use only reagents of a purity greater than 99 %.12.2.1. Ammonium sulphate, anhydrous, (CAS 7783-20-2).12.2.2. Sodium phosphate, dibasic, dodecahydrate, (CAS 10039-32-4).12.3. Apparatus and equipmentConical flasks, separating flasks, refrigerator.12.4. Procedure12.4.1. Sample screening for estragoleTo ensure that there is no estragole naturally present in the sample, a blank extraction (12.6.2) and analysis should be carried out without the addition of any internal standard. If estragole is naturally present then another internal standard must be chosen.12.4.2. ExtractionPipette 5 ml of 96 % ethanol (4.1) into a conical flask, weigh into this flask 50 mg of internal standard (4.3), and add 50 ml of the sample. Add 12 g of ammonium sulphate, anhydrous (12.2.1), and 8.6 g of dibasic sodium phosphate, dodecahydrate (12.2.2). Stopper the conical flask.Shake the flask for at least 30 minutes. A mechanical shaking device may be used, but not a Teflon coated magnetic stirring bar, as the Teflon will absorb some of the analyte. Note that the added salts will not dissolve completely.Place the stoppered flask in a refrigerator (T &lt;  5 C) for at least two hours.After this time, there should be two distinct liquid layers and a solid residue. The alcohol layer should be clear; if not, replace in the refrigerator until a clear separation is achieved.When the alcohol layer is clear, carefully take an aliquot (e.g. 10 ml), without disturbing the aqueous layer, place in an amber vial and close securely.12.4.3. Preparation of the extracted sample to be analysedAllow extract (12.4.2) to reach room temperature.Take 2 ml of the alcohol layer of the attemperated extracted sample and pipette into a 20 ml volumetric flask, make up to volume with 45 % ethanol (4.4), mix thoroughly.12.5. DeterminationFollow the procedure as outlined in 8.5.12.6. Calculation of resultsUse the following formula to calculate the results:>REFERENCE TO A GRAPHIC>where:mis is the weight of internal standard (4.3) taken (12.4.2) (in milligrams)V is the volume of unknown sample (50 ml)RFi is the response factor (9.1)areai is the area of the trans-anethole peakareais is the area of the internal standard peakThe results are expressed in grams per litre, to one decimal place.12.7. Quality control and assuranceFollow the procedure as outlined in 11 above.13. Method performance characteristics (precision)Statistical results of the interlaboratory test:the following tables give the values for anethole.The following data were obtained from an international method performance study carried out to internationally agreed procedures.>TABLE>>TABLE>Sample types:A pastis, blind duplicatesB pastis, blind duplicatesC pastis, blind duplicatesD pastis, blind duplicatesE pastis, single duplicatesF pastis, single duplicates>TABLE>Sample types:G ouzo, split levels (*)H anis, blind duplicatesI aniseed-flavoured liqueur, duplicatesJ aniseed-flavoured liqueur, duplicates.VI. GLYCYRRHIZIC ACID. DETERMINATION OF GLYCYRRHIZIC ACID USING HIGH PERFORMANCE LIQUID CHROMATOGRAPHY1. ScopeThis method is suitable for the determination of glycyrrhizic acid in aniseed-flavoured spirit drinks using high performance liquid chromatography (HPLC). Regulation (EEC) No 1576/89 specifies that any aniseed-flavoured spirit called "pastis" must contain between 0,05 and 0,5 g of glycyrrhizic acid per litre.2. Normative referencesISO 3696: 1987 Water for analytical laboratory use - Specifications and test methods.3. PrincipleThe glycyrrhizic acid concentration is determined using high-performance liquid chromatography (HPLC) with UV detection. A standard solution and the test sample are filtered and they are separately injected directly into the HPLC system.4. Reagents and materialsDuring the analysis, use only reagents of HPLC grade, absolute ethanol and water of grade 3 as defined by ISO 3696.4.1. Ethanol 96 % vol. (CAS 64-17-5).4.2. Ammonium glycyrrhizinate, C42H62O16.NH3 (Glycyrrhizic acid ammonium salt)(Mol. Wt.: 839,98)(CAS 53956-04-0): purity at least 90 %(Mol. Wt.: glycyrrhizic acid 822,94).4.3. Glacial acetic acid, CH3COOH, (CAS 64-19-7).4.4. Methanol, CH3OH (CAS 67-56-1).4.5. Ethanol 50 % vol.For 1000 ml at 20 oC:- 96 % vol. ethanol (4.1): 521 ml- Water (2.0): 511 ml.4.6. Preparation of the HPLC elution solutions4.6.1. Elution solvent A (example)80 parts (by volume) of water (2.0)20 parts (by volume) of acetic acid (4.3).Degas the elution solvent for five minutes.Note:If the water being used has not been microfiltered, it is advisable to filter the prepared elution solvent on a filter for organic solvents with a pore size less than or equal to 0,45 μm.4.6.2. Elution solvent BMethanol (4.4).4.7. Preparation of standard solutionsAll standard solutions must be freshly prepared after two months.4.7.1. Reference solution CWeigh to the nearest 0,1 mg, 25 mg of ammonium glycyrrhizinate (4.2) in a 100 ml volumetric flask. Add some 50 % vol. ethanol (4.5) and dissolve the ammonium glycyrrhizinate. When it has dissolved make up to the mark with 50 % vol. ethanol (4.5).Filter through a filter for organic solvents.4.7.2. Standard solutions used to check the linearity of the response of the instrumentationA 1,0 g/l stock solution is prepared by weighing, to the nearest 0,1 mg, 100 mg of ammonium glycyrrhizinate in a 100 ml volumetric flask. Add some 50 % vol. ethanol (4.5) and dissolve the ammonium glycyrrhizinate. When it has dissolved make up to the mark with 50 % vol. ethanol (4.5).At least four other solutions corresponding to 0,05, 0,1, 0,25 and 0,5 g/l of ammonium glycyrrhizinate are prepared by pipetting respectively 5 ml, 10 ml, 25 ml and 50 ml of the 1,0 g/l stock solution in separate 100 ml volumetric flasks. Then make up to the mark with 50 % vol. ethanol (4.5) and mix up thoroughly.Filter all solutions though a filter for organic solvents.5. Apparatus and equipment5.1. Separation system5.1.1. High-performance liquid chromatograph.5.1.2. Pumping system enabling one to achieve and maintain a constant or programmed rate of flow with great precision.5.1.3. UV spectrophotometric detection system: to be set at 254 nm.5.1.4. Solvent degassing system.5.2. Computational integrator or recorder, the performance of which is compatible with the rest of the system.5.3. Column (example):Material: stainless steel or glassInternal diameter: 4 to 5 mmLength: 100 to 250 mmStationary phase: cross-linked silica with a (preferably spherical) octadecyl functional group (C18), maximum particle size: 5 μm.5.4. Laboratory equipment5.4.1. Analytical balance with a precision of 0,1 mg5.4.2. A-grade volumetric glassware5.4.3. Micromembrane filtration arrangement for small volumes.6. Chromatography conditions6.1. Elution characteristics: (example)- flow rate: 1 ml/minute,- solvent A = 30 %,- solvent B = 70 %.6.2. Detection:- UV = 254 nm7. Procedure7.1. Preparation of the spirit sampleFilter, if necessary, through a filter for organic solvents (pore diameter: 0,45 μm).7.2. DeterminationOnce the chromatography conditions have stabilised,- inject 20 μl of the reference solution C (4.7.1),- inject 20 μl of the sample solution,- compare the two chromatograms. Identify the glycyrrhizic acid peaks from their retention times. Measure their areas (or heights) and calculate the concentration in g/l to two decimal figures using the following equation:>REFERENCE TO A GRAPHIC>where:c is the concentration in grams per litre of glycyrrhizic acid in the spirit being analysedC is the concentration in grams per litre of ammonium glycyrrhizinate in the reference solutionh is the area (or height) of the glycyrrhizic acid peak of the spirit being analysedH is the area (or height) of the glycyrrhizic acid peak of the reference solutionP is the purity of the reference ammonium glycyrrhizinate (in %)823 is the mass of one mole of glycyrrhizic acid840 is the mass of one mole of ammonium glycyrrhizinate.8. Method performance characteristics (precision)Statistical results of the interlaboratory test:the following table give the values for glycyrrhizic acid.The following data were obtained from an international method performance study carried out to internationally agreed procedures.>TABLE>>TABLE>Sample types:A pastis, blind duplicatesB pastis, split levels (*)C pastis, blind duplicatesD pastis, blind duplicatesE pastis, blind duplicatesVII. CHALCONES. HIGH-PERFORMANCE LIQUID CHROMATOGRAPHY METHOD FOR VERIFYING THE PRESENCE OF CHALCONES IN PASTIS1. ScopeThis method is suitable for determining whether chalcones are present in aniseed-flavoured drinks or not. Chalcones are natural colorants of the flavonoid family that are present in liquorice root (Glycyrrhiza glabra).For an aniseed-flavoured spirit to be called "pastis", it must contain chalcones (Regulation (EEC) No 1576/89).2. Normative referencesISO 3696: 1987, Water for analytical laboratory use - Specifications and test methods.3. PrincipleA reference liquorice extract solution is prepared. The presence or absence of chalcones is determined using high-performance liquid chromatography (HPLC) with UV detection.4. Reagents and materialsDuring the analysis, use only reagents of HPLC grade. The ethanol should be 96 % vol. Only water of grade 3 as defined by ISO 3696 should be used.4.1. Ethanol 96 % vol. (CAS 64-17-5)4.2. Acetonitrile, CH3CN, (CAS 75-05-8)4.3. Reference substance: Glycyrrhiza glabra: liquorice, "sweet root"Coarsely ground liquorice roots (Glycyrrhiza glabra). Average dimensions of the rodlike particles: length: 10 to 15 mm, thickness: 1 to 3 mm.4.4. Sodium acetate, CH3COONa, (CAS 127-09-3)4.5. Glacial acetic acid, CH3COOH, (CAS 64-19-7)4.6. Preparation of solutions4.6.1. Ethanol 50 % volumeFor 1000 ml at 20 °C:- 96 % vol. ethanol (4.1): 521 ml,- Water (2.0): 511 ml.4.6.2. Solvent A: acetonitrileAcetonitrile (4.2) of HPLC analytical purity.Degas4.6.3. Solvent B: 0,1 M sodium acetate buffer solution, pH 4,66.Weigh 8,203 g of sodium acetate (4.4), add 6,005 g of glacial acetic acid (4.5) and make up to 1000 ml with water (2) in a volumetric flask.5. Preparation of the reference extract from Glycyrrhiza glabra (4.3)5.1. Weigh 10 g of ground liquorice root (Glycyrrhiza glabra) (4.3) and place in a round-bottomed distillation flask- add 100 ml of 50 % vol. ethanol (4.6.1),- boil under reflux for one hour,- filter,- set the filtrate aside for later use.5.2. Recover the liquorice extract from the filter- place in a round-bottomed distillation flask,- add 100 ml of 50 % vol. ethanol (4.6.1),- boil under reflux for one hour,- filter. Set aside the filtrate for later use.5.3. The liquorice root extraction must be performed three times in succession.5.4. Combine the three filtrates.5.5. Evaporate the solvent phase (of 5.4) on a rotary evaporator.5.6. Take up the residual extract (of 5.5) with 100 ml 50 % vol. ethanol (4.6.1).6. Apparatus and equipment6.1. Separation system.6.1.1. High-performance liquid chromatograph.6.1.2. Pumping system capable of achieving and maintaining a constant or programmed rate of flow at high pressure.6.1.3. UV/visible spectrophotometric detection system that can be set at 254 and 370 nm.6.1.4. Solvent degassing system:6.1.5. Column oven that can be set at a temperature of 40 ± 0,1 °C.6.2. Computational integrator or recorder, the performance of which is compatible with the rest of the separation system.6.3. ColumnMaterial: stainless steel or glassInternal diameter: 4 to 5 mmStationary phase: cross-linked silica with an octadecyl derived functional group (C18), particle size: 5 μm at most (cross-linked phase).6.4. Common laboratory equipment, including:6.4.1. analytical balance. (precision: ± 0,1 mg);6.4.2. distillation apparatus with a reflux condenser, comprising, for example:- a 250 ml round-bottomed flask with a standardised ground-glass joint,- a 30 cm long reflux condenser, and- a heat source (any pyrogenic reaction involving the extractive matter must be avoided by using an appropriate arrangement).6.4.3. Rotary evaporation apparatus.6.4.4. Filtration set-up (i.e. Buchner funnel).6.5. Chromatography conditions (example).6.5.1. Elution characteristics of solvents A (4.6.2) and B (4.6.3):- shift from 20/80 (v/v) to 50/50 (v/v) gradient in 15 minutes,- shift from 50/50 (v/v) to 75/25 (v/v) gradient in five minutes,- equal strength at 75/25 (v/v) for five minutes,- stabilisation of the column between injections,- equal strength at 20/80 (v/v) for five minutes.6.5.2. Flow rate: 1 ml/minute.6.5.3. UV detector settings:the detector must be set at 370 nm to detect the presence of chalcones and then at 254 nm to detect glycyrrhizic acid.Note:the change of wavelength (from 370 nm to 254 nm) must be carried out 30 seconds before the beginning of the peak of elution of glycyrrhizic acid.7. Procedure7.1. Preparation of the spirit sampleFilter through a filter for organic solvents (pore diameter: 0,45 μm).7.2. Preparation of the residual liquorice extract (5.6)Make a one in ten dilution with 50 % vol. ethanol (4.6.1) before analysis.7.3. Determination7.3.1. Inject 20 μl of the prepared liquorice extract (7.2). Perform the analysis using the chromatography conditions described above (6.5).7.3.2. Inject 20 μl of the sample (7.1) (aniseed-flavoured spirit sample). Perform the analysis using the chromatography conditions described above (6.5).7.3.3. Compare the two chromatograms. There must be a great similarity between the two chromatograms in the chalcone exit zone (during the detection at 370 nm under the analysis conditions described above) (see Figure 1).8. Characteristic chromatogram for a pastisFigure 1Chromatogram obtained by the method described above, showing the presence of chalcones in a "pastis". Peaks 1 to 8 are chalcones and peak 9 is glycyrrhizic acid.>PIC FILE= "L_2002322EN.002301.TIF">9. Method performance characteristics (precision)Results of the interlaboratory test:the following table gives the performance for recognition of presence or absence of chalcones in pastis and aniseed-flavoured spirits.The following data were obtained from an international method performance study carried out to internationally agreed procedures.>TABLE>>TABLE>>TABLE>Sample types:A pastis, blind duplicatesB pastis, single sampleC pastis, single sampleD "pastis" (not containing chalcones), blind duplicatesE "pastis" (not containing chalcones), blind duplicatesF aniseed-flavoured liqueur (not containing chalcones), blind duplicatesG aniseed-flavoured liqueur (not containing chalcones), blind duplicatesH ouzo (not containing chalcones), single sampleI ouzo (not containing chalcones), single sampleJ anis (not containing chalcones), blind duplicatesK "pastis" (not containing chalcones), blind duplicates.IX. EGG YOLK. DETERMINATION OF EGG YOLK CONCENTRATION IN SPIRIT DRINKS - PHOTOMETRIC METHOD1. ScopeThis method is suitable for the determination of egg yolk concentration in the range of 40 to 250 g/l in egg liqueur and liqueur with egg.2. Normative referencesISO 3696:1897 Water for analytical laboratory use - Specifications and test methods.3. PrincipleThe ethanol-soluble phosphorus compounds found in egg yolk are extracted and assayed photometrically as a phosphorus molybdate complex.4. Reagents and materials4.1. Double-distilled water4.2. Diatomaceous earth4.3. Ethanol 96 % vol. (CAS 64-17-5)4.4. 15 % magnesium acetate (CAS 16674-78-5) solution4.5. 10 % sulphuric acid (CAS 7664-93-9)4.6. 1 N sulphuric acid.4.7. 0,16 g/l potassium dihydrogen phosphate (CAS 778-77-0), KH2PO4 solution4.8. Reagent for phosphate determination:dissolve 20 g of ammonium molybdate (CAS 12054-85-2), (NH4)6Mo7O24.4H2O in 400 ml water at 50 oC;dissolve, in another vessel, 1 g of ammonium vanadate (CAS 7803-55-6), NH4VO3, in 300 ml hot water, allow to cool, then add 140 ml of concentrated nitric acid (CAS 7697-37-2). Combine the cooled solutions in a 1000 ml volumetric flask and make up to the 1000 ml mark.5. Apparatus and equipment5.1. 100 ml conical flask5.2. Ultrasonic bath (or magnetic stirrer)5.3. 100 ml volumetric flask5.4. 20 oC water bath5.5. Filter (Whatman No 4 or equivalent)5.6. Porcelain (or platinum) crucible5.7. Boiling water bath5.8. Hot plate5.9. Muffle furnace5.10. 50 ml volumetric flask5.11. 20 ml volumetric flask5.12. Spectrophotometer set at 420 nm5.13. 1 cm cuvette.6. SamplesSamples are stored at room temperature prior to analysis.7. Procedure7.1. Sample preparation7.1.1. Weigh 10 g of the sample into a 100 ml conical flask (5.1).7.1.2. Add gradually 70 ml of ethanol (4.3) in small portions, swirling with each addition, and place in an ultrasonic bath (5.2) for 15 minutes (or stir the mixture with a magnetic stirrer (5.2) for 10 minutes at room temperature).7.1.3. Transfer the contents of the flask to a 100 ml volumetric flask (5.3) with washings of ethanol (4.3). Adjust to the calibration mark with ethanol (4.3) and place the flasks in a 20 oC water bath (5.4). Adjust to the calibration mark at 20 oC.7.1.4. Add a small amount of diatomaceous earth (4.2) and filter (5.5), discarding the first 20 ml.7.1.5. Transfer 25 ml of the filtrate to a porcelain (or platinum) crucible (5.6). The filtrate must then be concentrated by gentle evaporation in a boiling water bath (5.7), with the addition of 5 ml of 15 % magnesium acetate solution (4.4).7.1.6. Place the crucibles on a hot plate (5.8) and heat until just dry.7.1.7. Ash the residue by heating to incandescence at 600 oC in a muffle furnace (5.9) until the ash is white, minimum of one and a half hours but can be left overnight.7.1.8. Take up the ash with 10 ml of 10 % sulphuric acid (4.5) and transfer it with washings of distilled water (4.1) to a 50 ml volumetric flask (5.10), and fill to the mark at room temperature with distilled water (4.1). A 5 ml aliquot of this ash solution is to be used to prepare the sample solution of the photometric phosphate assay.7.2. Photometric phosphate assay7.2.1. Comparative solution7.2.1.1. Place 10 ml of 10 % sulphuric acid (4.5) in a 50 ml volumetric flask (5.10) and fill to the mark with distilled water (4.1).7.2.1.2. Add to a 5 ml aliquot of this solution (7.2.1.1), contained in a 20 ml volumetric flask (5.11), 1 ml of 1 N sulphuric acid (4.6) and 2 ml of the phosphate reagent (4.8) and make up to 20 ml with distilled water (4.1).7.2.1.3. Stopper with a loosely inserted stopper, shake, and heat in a boiling water bath (5.7) for 10 minutes, then cool in a 20 oC water bath (5.4) for 20 minutes.7.2.1.4. Fill a 1 cm cuvette (5.13) with this comparative solution.7.2.2. Sample solution7.2.2.1. Add to a 5 ml aliquot of the ash solution (7.1.8), contained in a 20 ml volumetric flask (5.11), 1 ml of 1 N sulphuric acid (4.6) and 2 ml of the phosphate reagent (4.8) and make up to 20 ml with distilled water (4.1).7.2.2.2. Stopper with a loosely inserted stopper, shake, and heat in a boiling water bath (5.7) for 10 minutes, then cool in a 20 oC water bath (5.4) for 20 minutes.7.2.2.3. The yellow solution that develops is immediately analysed spectrophotometrically (5.12) in a 1 cm cuvette (5.13) at 420 nm against the comparative solution (7.2.1.4).7.2.3. Calibration curve7.2.3.1. To construct the calibration curve, add 2 ml aliquots of the phosphate reagent (4.8) to 20 ml volumetric flasks (5.11) each containing 1 ml of 1 N sulphuric acid (4.6) and 0, 2, 4, 6, 8, and 10 ml of the potassium dihydrogen phosphate solution (4.7) respectively, and make up to the 20 ml mark with distilled water (4.1).7.2.3.2. Stopper with a loosely inserted stopper, shake, and heat in a boiling water bath (5.7) for 10 minutes, then cool in a 20 oC water bath (5.4) for 20 minutes and analyse spectrophotometrically (5.12) in a 1 cm cuvette (5.13) at 420 nm against the comparative solution (7.2.1.4).7.2.3.3. Construction of the calibration curve:>TABLE>8. Expression of resultsThe egg yolk content in g/l is calculated from the following formula:>REFERENCE TO A GRAPHIC>where:110 conversion factor for total P2O5 in g in 100 g of egg yolkmg P2O5 value established from the calibration curvedensity mass per unit volume (g/ml) of the egg-based liqueur at 20 oCE weight of the egg-based liqueur in g40 dilution factor for a 5 ml aliquot of ash solution.9. Method performance characteristics (precision)Statistical results of the interlaboratory test:the following table gives the values for egg yolk.The following data were obtained from an international method performance study carried out to internationally agreed procedures.>TABLE>>TABLE>Sample typesA Advocaat, blind duplicatesB Advocaat, blind duplicatesC Advocaat, blind duplicatesD Advocaat (diluted), split levels (*)E Advocaat, blind duplicates