Searched product:
Brand: Danfoss
Part number: 111.20.A52.00
Name: Sauer Danfoss 111.20.A52.00 Zahnradpumpe GR2 FIXED DISPL
Description: Danfoss: Sauer Danfoss 111.20.A52.00 Zahnradpumpe GR2 FIXED DISPL (111.20.A52.00) in 'Hydraulikpumpe' > 'Hydraulikpumpe (nicht klassifiziert)': Order best price offe<supplier> for Sauer Danfoss 111.20.A52.00 Zahnradpumpe GR2 FIXED DISPL direct at <supplier>, the procurement platform for business customers.
Category: Hydraulic pumps, Electro-hydraulic pump: Hydraulic gear pump
Price: 365.4002736932703 EUR
Supplier country: AUT

Found product:
Brand: S+S Regeltechnik GmbH
Part number: 1301-7111-0010-200
Name: Pressure ^Differential Pressure Transducer PremasGard 7111-U, range/DIP: 0..100Pa⁄±100Pa ^0..300Pa⁄±300Pa ^0..500Pa⁄±500Pa ^0..1000Pa⁄±1000Pa, air media -20..50°C, output 0..10V, inc. ASD-06, cv 24VAC/DC, IP65, S+S RegeltechnikKood: 1301-7111-0010-200
Description: Overview Specification Downloadable files Pressure˄differential pressure transducer Premasgard 7111 TYR2, 24VAC/DC; range/DIP: 0..100Pa⁄-100..+100Pa, 0..300Pa⁄-300..+300Pa, 0..500Pa⁄-500..+500Pa, 0..1000Pa⁄-1000..+1000Pa, 0..10V, inc. ASD-06, IP65, S+S Regeltechnik The calibratable, compact pressure sensors PREMASGARD® 7110 (series) are equipped with eight switchable measuring ranges and with or without optional display (eight devices in one) and are used for measuring above-atmospheric, below-atmospheric, or differential pressures in air. The piezoresistive measuring element is temperature-compensated and guarantees a high degree of reliability and accuracy. The pressure transmitters feature a pushbutton for manual zero point calibration or an automatic zero point calibration function as well as an adjustable offset. Applications of these pressure sensors are in clean rooms; medical and filter technology; ventilation and air conditioning ducts; in spray booths; large-scale catering facilities; for monitoring filters; level measurement; or for triggering frequency converters. Media measured with these pressure transducers are air (non-precipitating), or other gaseous non-aggressive, non-combustible media. The pressure sensor has eight selectable measuring ranges and therefore, minimizes the diversity of types and inventory levels while covering a greater range of applications. The differential pressure sensor is supplied including the ASD-06 connection set (2 m connection hose, two pressure connection nipples, screws). Technical data: Power supply: 24 V AC ±20% 15...36 V DC ±10% Power consumption: < 3.5 VA / 24 V AC < 2 VA / 24 V DC Measuring range, pressure: 0 ... +100 Pa / -100 ... +100 Pa 0 ... +300 Pa / -300 ... +300 Pa 0 ... +500 Pa / -500 ... +500 Pa 0 ... +1000 Pa / -1000 ... +1000 Pa Deviation pressure: ±1.5% of the end value (≙ ±15 Pa), at 25°C Operation range, temperature: -35 ... 70 °C Operation range, humidity: 0 ... 95 %r.H. non-precipitating air Storage temperature: -35 ... 70 °C Enclosure: Polyamide, 30% glass-globe reinforced, with quick-locking screws (slotted ⁄ Phillips head combination), Traffic white (similar to RAL 9016) Enclosure dimensions: 126 x 90 x 50 mm (Tyr 2) Enclosure: Polyamide, 30% glass-globe reinforced, with quick-locking screws (slotted ⁄ Phillips head combination), Traffic white (similar to RAL 9016) Sensor protection: 0.14 - 1.5 mm², 3-wire connection, via removable plug-in screw terminal File description File type Download link Operating instructions PDF Download link Operating instructions PDF Download link Catalogue page EN PDF Download link CE - Declaration of Conformity PDF Download link
Price: 141.8 EUR
Supplier country: EST

Same real-world product (1 = yes, 0 = no): 0