Searched product:
Brand: Phoenix Contact
Part number: 2906242
Name: 2906242 Phoenix Contact - Voltage measuring transducers - MACX MCR-VDC
Description: MACX MCR voltage transducers for DC voltages of 0... (+/-) 20 V DC to 0.. .(+/-) 660 V DC, output signal (+/-) 10 V/ (+/-)20 mA
Category: Drives – Motors – Circuits Protection
Price: 420 EUR
Supplier country: DEU

Found product:
Brand: Phoenix
Part number: 2906242
Name: Voltage Measuring Transducer MACX MCR-VDC, 1pcs/pck, PhoenixKood: 2906242
Description: Overview Specification Downloadable files Voltage Measuring Transducer MACX MCR-VDC, 1pcs/pck, PhoenixNote Utilization restriction EMC: class A product, see manufacturer's declaration in the download area Dimensions Width 22.5 mm Height 113 mm Depth 114 mm Ambient conditions Ambient temperature (operation) -25 °C ... 60 °C Ambient temperature (storage/transport) -40 °C ... 85 °C (non-condensing) Maximum altitude ≤ 2000 m Permissible humidity (operation) 10 % ... 95 % (non-condensing) Degree of protection IP20 Noise immunity EN 61000-6-2 When being exposed to interference, there may be minimal deviations. Input data Input voltage range -550 V DC ... 550 V DC Input resistance 5500 kΩ Input voltage range -370 V DC ... 370 V DC Input resistance 3700 kΩ Input voltage range -250 V DC ... 250 V DC Input resistance 2500 kΩ Input voltage range -170 V DC ... 170 V DC Input resistance 1700 kΩ Input voltage range -120 V DC ... 120 V DC Input resistance 1200 kΩ Input voltage range -80 V DC ... 80 V DC Input resistance 800 kΩ Input voltage range -54 V DC ... 54 V DC Input resistance 800 kΩ Input voltage range -36 V DC ... 36 V DC Input resistance 800 kΩ Input voltage range -24 V DC ... 24 V DC Input resistance 240 kΩ Nom. voltage ± 660 V DC Output data Output name Voltage output Voltage output signal -10 V ... 10 V Max. output voltage ≤ 11 V Load/output load voltage output > 10 kΩ Ripple 50 mV Output name Current output Current output signal -20 mA ... 20 mA Max. output current ≤ 22 mA Load/output load current output < 500 Ω Power supply Nominal supply voltage 24 V DC (-20 % ... +25 %) Supply voltage range 19.2 V DC ... 30 V DC Max. current consumption < 60 mA Connection data Connection method Pluggable screw connection Connection technology Screw connection Stripping length 8 mm Screw thread M3 Conductor cross section solid 0.2 mm² ... 2.5 mm² Conductor cross section flexible 0.2 mm² ... 2.5 mm² Conductor cross section AWG 24 ... 14 General Maximum transmission error < 1 % (of measuring range end value) Maximum temperature coefficient < 0.015 %/K Alignment zero ± 20 % Alignment span ± 20 % Step response (10-90%) < 16 ms Test voltage 5.3 kV AC (50 Hz, 1 min.) Electromagnetic compatibility Conformance with EMC directive Noise emission EN 61000-6-4 Noise immunity EN 61000-6-2 When being exposed to interference, there may be minimal deviations. Color gray Mounting position any Assembly instructions Can be aligned with spacing = 10 mm Standards and Regulations Electromagnetic compatibility Conformance with EMC directive Noise emission EN 61000-6-4 Standards/regulations IEC 61010-1 IEC 61010-2-030 Pollution degree 2 Low Voltage Directive Conformance with Low Voltage Directive Conformance/approvals Designation CE Certificate CE-compliant Designation UL, USA/Canada Identification UL 61010 Listed Environmental Product Compliance REACh SVHC Lead 7439-92-1 China RoHS Environmentally Friendly Use Period = 50 years For details about hazardous substances go to tab “Downloads”, Category “Manufacturer's declaration” ECLASS ECLASS 10.0.1 27210125 ECLASS 11.0 27210125 ECLASS 4.0 27271100 ECLASS 5.0 27200300 ECLASS 5.1 27200300 ECLASS 6.0 27210100 ECLASS 7.0 27210125 ECLASS 9.0 27210125 ETIM ETIM 4.0 EC002477 ETIM 6.0 EC002477 ETIM 7.0 EC002477 UNSPSC UNSPSC 18.0 39121008 UNSPSC 19.0 39121008 UNSPSC 20.0 39121008 UNSPSC 21.0 39121008 File description File type Download link
Price: 292.91 EUR
Supplier country: EST

Same real-world product (1 = yes, 0 = no): 1