An electrode addressing protocol for imaging brain function with electrical impedance tomography using a 16-channel semi-parallel system. 
Electrical impedance tomography of brain function poses special problems because applied current is diverted by the resistive skull.  In the past, image resolution was maximized with the use of an electrode addressing protocol with widely spaced drive electrode pairs and use of a multiplexer so that many electrode pairs could be flexibly addressed.  The purpose of this study was to develop and test an electrode protocol for a 16-channel semi-parallel system which uses parallel recording channels with fixed wiring, the Kyung Hee University (KHU) Mk1.  Ten protocols were tested, all addressing pairs of electrodes for recording or current drive, based on recording with a spiral, spiral with suboccipital electrodes (spiral s-o) and zig-zag configurations, and combinations of current injection from electrode pairs at 180 degrees , 120 degrees and 60 degrees .  These were compared by assessing the image reconstruction quality of five simulated perturbations in a homogenous model of the human head and of four epileptic foci in an anatomically realistic model in the presence of realistic noise, in terms of localization error, resolution, image distortion and sensitivity in the region of interest.  The spiral s-o with current injection at 180 degrees + 120 degrees + 60 degrees gave the best image quality and permitted reconstruction with a localization error of less than 10% of the head diameter.  This encourages the view that it might be possible to obtain satisfactory images of focal abnormalities in the human brain with 16 scalp electrodes and improved instrumentation avoiding multiplexers on recording circuits.