Part Number: MicroLite II: LITE5032P-RH Part Number: MicroLite II: LITE5008P, LITE5032P Two internal digital and humidity temperature sensors and USB interface One internal NTC temperature sensor and USB interface Memory Capacity: 32,000 samples Power supply: Battery operated Memory Capacity: Four models: MicroLite II: 8,000 and 32,000 samples supply: Battery operated Part Number: MicroLite II: LITE5032P-4/20 Part Number: MicroLite II: LITE5032P-V One 4-20 mA current input and USB interface. Logger cap contains universal connector for current sensor. When cap is removed, USB interface is available. One 0-10 V voltage input and USB interface. Logger cap contains universal connector for voltage sensor. When cap is removed, USB interface is available. Memory Capacity: 32,000 samples Power supply: Battery operated Memory Capacity: 32,000 samples supply: Battery operated Part Number: MicroLite II: LITE5032P-EXT One NTC -50 to 150 C temperature and USB interface. Logger cap contains universal connector for temperature sensor. When cap is removed, USB interface is available. Sensor is included. Memory Capacity: 32,000 samples Power supply: Battery operated Hardware Accessories A magnet key is provided to stop and activate the logger as well as to mark time stamps. In addition, a green terminal block is attached to the external logger cap for the
Map View Icons The following icons representing the various statuses of the devices are viewed in Map View. Using these icons the user is aware of the status of each unit e.g. offline/online, alarm, power, etc. Alarm status is also indicated: Green = Normal ; Red = Alarm DataNet Icons Receiver is online and connected to the PC. Receiver is offline, possibly as USB cable has been disconnected or it was removed from external power supply (and backup battery drained). Logger newly acquired by network and awaiting Setup. Running from internal battery supply. Logger newly acquired by network and awaiting Setup. Running from external power supply. Logger which has received its Setup but is in Stop mode. Running from internal battery supply. Logger which has received its Setup but is in Stop mode. Running from external power supply. Logger in Run mode and running on internal battery supply. Green indicates there are no alarms. Logger in Run mode and running on external power supply. Green indicates there are no alarms. Logger in Run mode with an alarm alert. Running from internal battery supply. Logger in Run mode with an alarm alert. Running from external supply. Logger is offline and is no longer detected on the network. Its power supply may have failed or it’s out of range of the Receiver. Logger is processing a command sent from the software. Logger is downloading data. Logger is detected by DataSuite software via direct USB connection. Logger is in Stop mode and running from internal battery supply. Logger is detected by DataSuite software via direct USB connection. Logger is in Stop mode and running from external power supply. Logger is detected by DataSuite software via direct USB connection. Logger is in Run mode and running from internal battery supply. Logger is detected by DataSuite software via direct USB connection. Logger is in Run mode and running from external power supply. Repeater is online and is running from external power supply.
and Run Clicking Setup and Run will send the setup to the logger and immediately start logging data. Only Clicking Only will send the setup to the logger but it will not actually start logging data. You will need to select Run from the Logger context menu or go back into the Setup dialog. Clicking Close will close the Setup dialog without saving any of your configurations. Note: Running the logger clears the logger memory. All previously recorded data will be erased when you a new logging session and DaqLink Setup Tab A feature of the DataNet and DaqLink systems is the ability to configure alarm levels into the loggers so any breach of these levels as recorded on the data logger is monitored by the software, and the user is notified via Email/SMS/audible and visual indicators. The Alarm Setup tab lets you configure each sensor for the following alarms: Low Pre-Low Pre-High High Contact Open / Contact Close (when the Contact sensor is selected) Using the Pre Alarms gives the user ample warning of a possible breach of real alarm levels, so necessary measures can be taken to prevent such a breach. Simply select the check box for the type of alarm you wish to be notified. You can select all alarms too. Enter the alarm value in the text box. Figure 89: DataNet and DaqLink Alarm Setup tab Delay and Pre-alarm Delay The time the alarm (or pre-alarm) is activated. You may not want to have the alarm sound immediately after the alarm level is breached as you are only interested in a condition where the alarm level lasted a certain amount of time. For example, if monitoring change of temperature in a freezer, you would only be interested in a High alarm which lasted more than one minute as anything else could just be caused by the freezer door opening for a few seconds, as opposed to a power failure leading to rise in temperature. Duration The duration of the alarm that is set off by one or more sensors breaching the alarm level.
DaqLink Specifications Data Logger Input Types DBSA710 and DBSA720 External Inputs Four external inputs with selectable sensor type for each input: 4 to 20 mA 0 to 1 V 0 to 50 mV PT-100 (2-wire) Contact (Open/Closed) Thermocouple J, K or T counter (Input 4 only) Frequency (Input 4 only) User-defined sensors Internal Inputs DBSA710 PT-100 Temperature DBSA720 Digital Temperature Digital Relative Humidity DBSA710 and DBSA720 Outputs Alarm Output (Output 1) External Power Excitation (transducers usage) Open collector Close position resistance: 50 Ω Max. Load: 50 mA, 3 V DC 12 VDC @ 300 mA Reset fuse overload protection Logger Input Specifications DBSA710 and DBSA720 Internal PT-100 Temperature (DBSA710) Internal Digital Temperature (DBSA720) Internal Digital Relative Humidity (DBSA720) Type: PT-100 Range: -20 to 50 ˚C Resolution: 0.1 ˚C Accuracy: ±0.3 ˚C Type: Digital sensor Range: -20 to 50 ˚C Resolution: 0.1 ˚C Accuracy: ±0.5 ˚C Range: 5 to 95 % RH Resolution: 0.5% RH Accuracy: 3% RH (DataSuite software) 4% RH (Logger LCD) Temperature Thermocouple J Temperature Thermocouple K Temperature Thermocouple T
Range: -200 to 1,000 ˚C Resolution: 0.1 ˚C Accuracy: -200 to – 60 ˚C ±0.5 % -60 to 60 ˚C ±0.5 ˚C 60 to 1,000 ˚C ±0.5 % Cold junction compensation: ±0.3 ˚C Range: -200 to 1,000 ˚C Resolution: 0.1 ˚C Accuracy: -200 to – 60 ˚C ±0.5 % -60 to 60 ˚C ±0.5 ˚C 60 to 1,000 ˚C ±0.5 % Cold junction compensation: ±0.3 ˚C -200 to 400 ˚C Resolution: 0.1 ˚C Accuracy: -200 to – 60 ˚C ±0.5 % -60 to 60 ˚C ±0.5 ˚C 60 to 400 ˚C ±0.5 % Cold junction compensation: ±0.3 ˚C Temperature PT-100 (2-wire) Counter (Input 4 only) Frequency 4 only) Range: -200 to 400 ˚C Resolution: 0.1 ˚C) Accuracy: -200 to - 60 ˚C ±0.5 % -60 to 60 ˚C ±0.3 ˚C 60 to 400 ˚C ±0.5 % 1 to 65,536 counts Resolution: 1 count Frequency range: 0 to 4 KHz Input signal: 0 to 5 V impedance: 470 Ω Range: 20 Hz to 4 KHz Input signal: 0 to 5 V Input impedance: 470 Ω 4 to 20 mA 0 to 1 V 0 to 50 mV 4 to 20 mA Resolution: 47.6 μA Accuracy: ±0.5 % Loop impedance: 21 Ω 0 to 1 V Resolution: 200 μV Accuracy: ±0.5 % impedance: 25 MΩ Maximum voltage: 5.2 V 0 to 50 mV Resolution: 37 μV Accuracy: ±0.5 % Input impedance: 25 MΩ Maximum voltage: 5.2 V Contact Open/Close General Specifications DaqLink Logger Sampling Features Memory capacity: ~38,000 samples in Online Mode ~59,000 samples in Optimized Memory Mode (depending on logger configuration) Sampling rate: Once per second to once every 18 hours Channel separation: 80 dB DBSA710 and DBSA720 Hardware Specifications Display 2-row LCD 16 character display Power Supply : Internal rechargeable 4.8 V, 800 mAh NiMH battery (2 batteries in series) Battery Life: Up to six months depending on logger configuration AC Power:
MicroLite Specifications General Specifications LITE5008 LITE5016 LITE5008P LITE5032P LITE5032P-RH Internal Temp Internal Temp Internal Temp/RH 4-20 mA 0-10 V External NTC Range -40 to 80 °C -40 to 80 °C Temp: -40 to 80 °C RH: 5-95 % 4-20 mA 0-10 V -50 to 150 °C Accuracy ± 0.3 ˚C ±0.3˚C @ 0 to 60 ˚C ±0.3 ˚C @ -40 to 0 ˚C and 60 to 80 ˚C after calibration only Temp: ±0.3 ˚C RH: ± 2 %RH ±20 µA ±5 mV ±0.3 ˚C (after calibration with Fourtec NTC sensor) Resolution 0.1 °C (display) 0.01 °C (software) 0.03 °C or better Temp: 0.01 °C RH: 0.05 %RH 1 µA (display) 5 µA (software) 1 mV (display) 4 mV (software) 0.02 °C or better Maximum Load N/A N/A N/A 25 mA, 3 V 12 V N/A Dew X X √ N/A N/A N/A LCD √ √ √ √ √ √ LED Alarm Indicator X √ √ √ √ √ Sample Memory 8,000 and 16,000 8,000 and 32,000 32,000 (16K per channel) 32,000 Sampling Rate Once per second / once every 2 hours Once per second / once every 18 hours Battery Type / Battery Life CR2032 3V 1.5 years, depending on sampling rate and operating conditions CR2032 3V 2 years, depending on sampling rate and operating conditions Operating Environment -40 to 80 °C Dimensions (with cap) 11 x 3.9 x 2.6 cm 11 x 3.9 x 3 cm 14 x 3.9 x 2.6 cm Weight 45.5 g
Standards Compliance CE, EN12830, IP67 CE, EN12830, IP54 Wall Mount X Logger cradle accessory provided System Requirements Software Hardware OS Platform: Windows XP SP3, Windows and Windows 7 (32 and 64-bit) Screen Resolution: Minimum 1024 x 768 Processor: Pentium 4, 2.0 GHz or better Memory: 512 MB RAM Storage Memory: 250 MB available disk for the DataSuite application