2 x brackets in steel for installation against the wall. Or Equivalent 12 Suction Machine Heavy Duty with 4 Liters Jars less suction machine with a noise level of not more than 47db for smooth and calm operation. Should have 4 extra strong wheels (1of them antistatic and 2 of them with brakes) Should have suction capacity of 40l/min Vacuum pressure 650 -700 mmhg ON/OFF switch Should have Knob for Suction Control Should be piston type, oil free , maintenance free Should have 2 Polycarbonate collection jar of 4 liters with safety valve to prevent overfilling Should be Complete With Suction tube of 3-4 m 2 No Collection jar of 4 liters each and 1 no. extra jar. With foot switch cord Anti-Bacterial filter Or Equivalent USA/UK/EEC/Japan 32 13 Duty Nebulizer Compressor: Piston. Operating Pressure: 0.90 bar or better. Air flow: 12~ 14 1/min. Level: 50db or better Accessories: Adult face mask 01. Pediatric Face Mask . Faster jet Bulb. Mouthpiece. Nasal prong. Connection tubing. Or Equivalent 05 14 Therapeutic Ultrasound for Rehabilitation 1& 3 MHZ Frequency Therapy for treatment to increase blood flow in the tissues 01
XY Fine orientation controls located on the left side of the specimen head – user can adjust settings with one hand and does not have to reach over the blade. 64 mm vertical cutting stroke – Accommodates a broad range of specimen sizes including super mega cassettes. Parallel guide rails – preserve stability of the knife holder for accurate and reliable sectioning Cross roller bearings – Provide exceptionally smooth operation and helps maintain calibration. Removable levers on the blade holder – Avoids accidental adjustment and frees work are from less frequently used levers. Large wraparound waste tray completely surrounds the blade holder – User can cleanup both faster and safer without having to reach behind the blade holder. The tray can be removed very quickly and easily to tip waste directly into appropriate waste receptacle. Selectable retraction from the left side of the instrument – Makes switching retraction on and off very simple. Enabling rapid trimming. Or equivalent USA/UK/EEC/Japan 22 Cystopin USA/UK/EEC/Japan 1 23 Wax Dispenser USA/UK/EEC/Japan 1 24 Tissue Floating USA/UK/EEC/Japan 2 25 Slide Warmer Touch-screen interface Stain protocols with recall and loading of reagent layouts User protection from fume extraction through a carbon filter Internal LED lighting battery backup password protected Protocols. 1
USA/UK/EEC/Japan 49 Bone Marrow Aspiration Needles USA/UK/EEC/Japan 10 50 Bone Marrow Trephine Biopsy Needles USA/UK/EEC/Japan 10 51 DLC Manual Counter USA/UK/EEC/Japan 5 52 Storage Cabinets USA/UK/EEC/Japan 1 53 Micro Pipits/Jesters/Fixed/Adjustable Adjustable pipettes. Robust body construction. smooth fatigue free operation. precisely calibrated. Corrosion free ceramic construction piston. Tip holder & ejector fully autoclave able. Measuring capacity of the following sizes. Adjustable range of 0.2 to 2 Ul/Adjustable range of 10 uL , Adjustable range of 0-50 ml Adjustable range of 50-100mL Adjustable range of 100-1000mL Or Equivalent USA/UK/EEC/Japan 10 54 Lab Incubator Size Microprocessor Control. PID temperature control. LED digital display to show actual PV and preset chamber. Over temperature safety device. LED 3 digit display . 0.1°C stability Stainless steel interior and powder coated exterior. Capacity 36 Liters . chamber size : 330x330x330. 2 x shelves Or Equivalent USA/UK/EEC/Japan 2 55 Laminar Flow (Cabinet) Nominal size 0.6 (2”) Internal Work Area, Space: 0.39 m2 ( 4.2 sq.ft) Average Airflow Velocity: 0.45 m/s (90 fpm) at initial setpoint Air Volume: 603 m3 /h (355 cfm) ULPA Filter Typical Efficiency: >99.999% at particle size between 0.1 to 0.3µm 1
Vertical movement of the Arm to lower or raise the tube collimator assembly Rotation of the Collimator with reference to the module (+/- 90deg) or better . Or Equivalent USA/UK/EEC/Japan 72 Color Doppler Ultrasound Machine with 3 Probes 21 inch medical grade LED monitor with up/down and left/right rotation or better. 12 inch medical grade touch screen or better with adjustable angle of 10° or more. Articular arm, flexible adjusting control panel including height and left/right rotation. Backlit alphanumeric keyboard. 5 transducer ports with at least 4 active ports and one additional Port for Pencil probe with automatic recognition. Built-in Gel Warmer Complete DICOM 3.0 Connectivity 500 GB Hard Disk or better. Wireless Wi-Fi Connection. Inbuilt ECG Module as standard. 4 USB Ports or more DVD-RW Modes as standard: B Mode. 3D Mode. Dual Mode / Quad Mode. Stress Echo Protocol. User Defined Pre-set settings. User Define Keys for Calculation. Speckle reduction Technology. 5-band adjustable frequency in B mode. Tissue characteristic index. Tissue Harmonic Imaging mode. Pulse Inversion Harmonic Imaging mode. Image rotation function. Spatial Compound imaging. Frequency Compound imaging. Trapezoidal imaging (with multiple selection for the trapezoid shape). mode. PDI mode. 1
Anesthesia Instrument Trolley (Qty: 1) Entirely made of AISI304 stainless steel Anesthesia instruments connection system 8 wheels Recognition through magnet Should be from Washer Disinfector manufacturer Air-Water Gun (Qty: 1) Spray gun made of stainless steel AISI304, suitable for water and compressed air, supplied with 8 nozzles Tube with respective connection to the power supply of water or air under the tub or at floor Handle thermally insulated Accessories contained in a structure suitable for wall mounting The length of the connecting hose: 2m Heat Sealer (Qty: 1) Standards: CE / CEM / EN 868-5 / DIN 58953-7 Sealing: 15 mm / flat and full weld for better checking, unlimited sealing length. Display: 1 line LCD Temperature range: 20-230C° with a 2% regulation tolerance Adjustable pouches guide (0-30 mm from edge). Welds counter with reset. Should seals every kind of pouches (steam, Tyvek, OE, alu, Ultra, etc). Starts automatically when the set temperature is reached. Environmentally responsible: automatic start and stop by photocell. Safety: critical parameters control complying to ISO 11607-2 standard (temperature & pressure) Automatic stop if critical parameters exceed limits