（1）PTZ control 1.1 Coaxial PTZ control Click the icon“PTZ”on the right toolbar to do the corresponding setting. “Port”: select coaxial control Enter PTZ address to perform parameters setting Operation instructions, please refer to “3.3.1 PTZ (1) Video monitor test” The PTZ address in the tester must be consistent with the dome camera or decoder, then the IPC tester can test .After setting the parameter, the tester can control the PTZ and lens.
Hik test tool Hik test tool app is design for activating and debugging Hikvision camera, can auto-identify unactivated hikvision camera, also can display from the Hikvision camera. Tap icon to enter Hikvision activation：When connect the unactivated Hikvison camera to the tester, it will auto identify and display “Unactivate” at safety mode. Select needed activate the camera, click bottom right corner button “Enable”, pop-up “the camera is not activate, activate now”? 2.Input password：Input new password, tap “ok” to activate 3. Confirm activation After activating the camera, the program default modifies the camera IP. Activated multiple cameras in the local area Network, and pop-up menu to modify IP, improve project efficiency.
PoE power output 48V PoE output, Max power 24W Audio test 1 channel signal input and 1 channel audio signal output to connect headphones PTZ control Support RS485 control, Baud 600-115200bps, Compatible with more than 30 protocols such as PELCO-D/P, Samsung, Panasonic, Lilin, Yaan, etc bar generator Output one channel PAL/NTSC color bar video signal for testing monitor or video cable.(red, green ,blue, white and black color ) UTP Cable tester Test UTP cable connection status and display on the screen. Read the number on the screen Data monitor Captures and analyzes the command data from controlling device, also can send hexadecimal Network test IP address scan, link scan, and Ping test. Quickly search the for IP camera’s IP address on your network Cable tracer Find a connected cable from a bundle of cables using audio tones PoE /PSE voltage test Measures PoE switch voltage and displays pin configuration Digital *(Optional) AC/DC Voltage, AC/DC current, Resistance, Capacitance, Data hold, Relative measurement, Continuity testing. Testing speed: 3 times/ seconds, Data range -6600～+6600. Optical meter *(Optional) Calibrated Wavelength(nm) :850/1300/1310/1490/1550/1625nm Power range(dBm) :-70～+10dBm Visual fault locator *(Optional) Test fiber’s bending and breakage ( SM and MM fiber) TDR test *(Optional) Cable’s open circuit (Breakpoint) and short circuit measurement(BNC cable, telephone cable) POWER External power supply DC 12V 2A Battery 7.4V Lithium polymer battery ,5800mAh Rechargeable After charging 6~7 hours, normal working time 10 hours Parameter Operation setting Capacitive screen, OSD menu, select your desired language: English, Chinese, Korean, Russian, Italian or Polish, etc Auto off 1-30 (mins) General Working Temperature -10℃---+50℃ Working Humidity 30%-90% Dimension/Weight 240mm x 154mm x 46mm / 620g 4.2 Multi-meter specifications Counts:-6600～+6600 Conversion rate: 3times/s Current modes for clamp meter with ZERO function
6.600KΩ ±（0.8%+2） 1Ω 66.00KΩ 10Ω 660.0KΩ 100Ω 6.600MΩ 1KΩ 66MΩ ±（1.2%+5） 10KΩ Continuity Range Resolution Function 660.0Ω 0.1Ω The measurement value less 30Ω±3Ω,the tester will sound Range Resolution Function 2.0V 1mV Schottky diode:0.15~0.25V rectifier diode:0.6~1.0V triode PN junction:0.5~0.8V Capacitance Range Accuracy Resolution 6.600nF ±（0.5%+20） 1pF 66.00nF ±（3.5%+8） 10pF 660.0nF 100pF 6.600μF 1nF 66.00μF 10nF 660.0μF ±（5%+8） 6.600mF 1μF 66.00mF 10μF 4.3 Optical power meter specifications Measure Range(dBm) -70～+10dBm