Power supply control thro’ RS485/Ethernet/LXI/ any other. Vendor may give available options and effect on the performance in system Heater harness with connectors Total heater extension wire shall be about 20m length. Due to test set-up constraints the extension wire is divided into two parts (15m & 5m). Vendor shall provide these with 50-pins D-sub connectors and extension wire length. Harness bunch shall be suitably tied. Extension of 15m: This shall be extension of temperature sensor from power supply to intermediate 50-pins female D-sub connectors. Connection of this harness with the Power Supply unit shall be the responsibility of the vendor. Typically, 15 nos. of 2-wire heater wire can be accommodated in each 50-pins D-sub connector. Wire gauge: 20 AWG Extension of 5m: This shall be extension of heater wire from heater element to intermediate 50-pins male D-sub connectors. Typically, 15 nos. of 2-wire heater wire can be accommodated in each 50-pins D-sub connector. Wire gauge: 20 AWG Heater element: SAC shall supply heater element during installation at SAC. Vendor should provide 50-pins female D-sub connector extension with harness length of 15m. 50-pins male D-sub connector extension with harness length of 5m. 50-pins Male and Female Vacuum Compatible/Space grade D-sub Crimpable connectors Make: Standard Make like Amphenol, Positronics, etc. The wires, connectors, sensors and crimping should be compatible in vacuum chambers (<1x10-5torr). All extension cable shall be flexible extension wire, silver plated, multi-stranded copper conductors, individually PTFE insulated, and Twisted. Typical Heater Harness Diagram: Server Server shall be used for controlling and monitoring of real time test data, storage etc. Processor: Intel Xeon Processor E5-2637 V3 Hard Disk and Controller: 1.2TB SAS HDD 10K Qty: 3 Nos. The internal storage should configure in RAID 5.
Storage controller should be capable of providing RAID 0/1/5 configuration RAM: 16GB DDR4 SDRAM - 2133 MHz Network: Ethernet 1Gb 6-port. 6 ports on 2 separate controller. Optical Drive: 9.5mm SATA DVD-RW Power supply: Redundant Hot plug power supply Form factor: Not more than 2U. OS: Window 10 or higher Suggest brand: DELL, HP, Lenovo Equivalent with latest configuration at the time of supply Software: Licensed. As required to operate the OS and application software. Safety Features: Server must have latest cyber security features. Application Software The application software must be suitable right from test configuration, running the test and generating reports. The total solution shall be designed to provide all functionalities and features of the applications software as per detailed specifications laid in this document. Temperature controller is proposed to control temperature at number of locations in Vacuum chamber. The sensors are of PT 1000/100 Class A, Thermistor and Thermocouple T-type. The heat sources (IR lamps or foil heater) will be connected to power supply. Necessary accommodation of web client should be made to view the test parameter though existing SAC intranet. Functionality/Features: History Manager History Access Server Configuration Manager Diagnostics and Fault Management Live Data Viewer (User select-able, Multi Channel) Historical Data Viewer Report generation. History Access Client Naming/ID: Unique ID/naming of each sensor and power supply. Description of these shall be possible. Page Display Utility: The parameters and their sequence should available in page configuration ASCII files There should be a provision to display all parameters including heater enable/disable, heater On/off status etc.