Network Analysis (Scattering Parameters, Group Delay, Phase) Modulation Digital Analog Video Data Acquisition Systems National Instruments various various Cal EXECUTIVE & METCAL (ALL FP, SCXI, PXI, PXIe, NI) Electrical Data Acquisition Systems Dewetron Call for details Call for details with METCAL (ALL Scanivalves, Precision Filters Call for details Call for details with METCAL Fluid Properties and Quantities Gas Flow Flow Rate 1 SCCM - 60 SLPM 0.2-0.3% of reading Molbox system using nitrogen Flow Rate 1 SCCM - 40 SLPM 0.8% of reading Brooks Stand using air, carbon dioxide, carbon monoxide, helium, or argon Flow Rate 1 SCCM - 40 SLPM 0.8% of reading Brooks Stand using gaseous fuels (Propane, hydrogen, etc.) Flow Rate 14-50,000 SLPM 0.5% of reading Nitrogen Stand using nitrogen at ambient to 1800 psig at ambient temperature Flow Rate 7-75,000 SLPM 0.5% of reading Air Stand using air at ambient to 225 psig at ambient temperature Liquid Flow Flow Rate 0.01-285 GPM 0.3% of reading Cox Stand using water from 50 to 90 0F Flow Rate 1-500 lbm/hr 0.3% of reading Stoddard Stand-Stoddard solvent Fluid Properties and Quantities Liquid Flow Flow Rate 0.01 -75 LPM 0.1% of reading Stoddard Calibrator – Jet Fuel simulated flow Mechanical Pneumatic Pressure Pressure 0.5-30 inch water 0.0004-0.0029 inch water Direct comparison with the low pressure standard RUSKA 7250 LP:xi MUA includes drift of all weights and piston
Pressure 1 PSIA to 1000 PSIA/1000 PSIG 15- 30 ppm Direct comparison with the deadweight pressure standard. Pressure 300 to 13,343 psi 40-44 ppm Direct comparison with the High Pressure Gas Piston Gage Pressure From 15 capable to calibrate 10 simultaneously pressure transducer, mV output, 0-5 VDC and 4-20mA 2 Automated systems with 20 unit capacity with temperature capability of -73 to +200 C Force Load Cells (compression, Tension) 100, 200, 500, 1000 lbf, 0.036% range Comparison Method to standard load ( Interface System, hydraulic) "Range" is of standard load cell Mechanical Force Load (compression, Tension) 100, 200, 500, 1000 lbf, 0.036% range Comparison Method to standard load cells ( Interface System, hydraulic) "Range" is of standard load cell Load Cells (compression, Tension) 2000 lbf 0.012% range Comparison Method to standard load cells ( Interface System, hydraulic) "Range" is of standard load cell Load Cells (compression, Tension) 5.5K lbf 0.032% range Comparison Method to standard load cells ( Interface System, hydraulic) is of standard load cell Load Cells (compression, Tension) 10K lbf 0.043% range Comparison Method to standard load cells ( Interface System, hydraulic) is of standard load cell Load Cells (compression, Tension) 25K lbf 0.071% range Comparison Method to standard load cells ( Interface System, hydraulic) is of standard load cell Load Cells (compression, Tension) 50K lbf 0.018% range Comparison Method to standard load ( Interface System, hydraulic) "Range" is of standard load cell Mechanical Load Cells (compression, Tension) 55K lbf 0.044% Comparison Method to standard load ( Interface System, hydraulic) "Range" is of standard load cell Mass Mass (scale measurement) 1 mg to 12 Kg Weight sets 0.000015-0.0066g Direct comparison Mass (scale measurement) 0.5 oz – 50 lb 0.000003-0.03g Direct comparison
Humidity 10% to 95 % 0.5 % RH Thunder scientific 2500: Humidity Generator Resistance Thermometry Temperature: Probes & thermistors 0 to 500 kΩ Resistance range Typical accuracy, Internal Reference: 25 Ω SPRT (0 °C, 100 Ω refr, 1 mA) 0.002°C, 10 kΩ thermistor (25°C, 10 kΩ refr, 0.01 mA) 0.0005 °C Super -Thermometer: Hart Scientific 1575 Temp measurement -200-420 °C 0.013°C (max) quartz Fluke 5683 MUA includes only drift and cal uncertainty of probe Temp measurement 420-990 °C 0.009°C (max) quartz Flue 5684 MUA includes only and cal uncer of probe Temp measurement 990-1200 °C 1.5°C (for max temp) S type TC MUA includes only drift and cal uncer of probe Thermodynamic Thermocouples Temperature/ Precision Thermocouple Module _10 mV to 100 mV 0.05°C Fluke 2565 with Hart 1560 Temperature/ Precision Thermocouple Module Reference Temperature 0°C ±0.02°C Kaye Ice point K150 Thermodynamic Thermometers Temperature/ Precision Thermometer/all kinds of TC & PRT100: for instance, J type Thermocouple _210 0C to 1200 0C 0.03% of reading + 0.005% scale Isotech F150 Precision Thermometer Thermodynamic Thermodynamic Sources (temperature baths and drywells) Temperature/ source _80 0C to 110 0C 0.007°C (max) Hart 7080SD: stirred bath (must be monitored with external std) Does not include uncertainty of measuring standard