["Image quality and radiation dose to the patient are important factors in computed tomography (CT). To provide constant image quality, tube current modulation (TCM) performed by automatic exposure control (AEC) adjusts the tube current to the patient's size and shape.", 'To evaluate the effects of patient centering on tube current-time product (mAs) and image noise.', 'An oval-shaped acrylic phantom was scanned in various off-center positions, at 30-mm intervals within a 500-mm field of view, using three different CT scanners. Acquisition parameters were similar to routine abdomen examinations at each site. The mAs was recorded and noise measured in the images. The correlation of mAs and noise with position was calculated using Pearson correlation.', 'In all three scanners, the mAs delivered by the AEC changed with y-position of the phantom (P<0.001), with correlation values of 0.98 for scanners A and B and -0.98 for scanner C. With x-position, mAs changes were 4.9% or less. As the phantom moved into the y-positions, compared with the iso-center, the mAs varied by up to +70%, -34%, and +56% in scanners A, B, and C, respectively. For scanners A and B, noise in two regions of interest in the lower part of the phantom decreased with elevation, with correlation factors from -0.95 to -0.86 (P<0.02). In the x-direction, significant noise relationships (P<0.005) were only seen in scanner A.'] Does efficient use of automatic exposure control systems in computed tomography require correct patient positioning?