['The observation of intact non-covalent complexes by electrospray ionization mass spectrometry (ESI-MS) hinges on the ability to minimize in-source activation processes that take place during analyte desolvation. We explored the merits of replacing the sampling cone of a standard Z-spray source with a heated capillary that makes the desolvation process slower and more gradual. We employed well-characterized protein-RNA, RNA-RNA, and DNA-DNA assemblies to compare the alternative configurations.', 'Mass analysis evaluated the integrity of the complexes, whereas traveling wave ion mobility experiments assessed the stability of biomolecular structure. Analyses were performed back-to-back on the same samples on a Synapt G2 HDMS equipped with either the standard sampling cone or the heated-capillary apparatus. In each configuration, the source/capillary temperature was varied in controlled fashion, while keeping all other desolvation parameters constant to monitor the in-source dissociation of selected DNA duplexes. Ion mobility data were obtained from the same precursor by using the alternative configurations under the same settings.', 'Monitoring the percentage of associated complex demonstrated that the heated capillary provided softer desolvation that was more conducive to the detection of intact non-covalent interactions. This configuration failed to produce complete dissociation of 14\u2009bp and 24\u2009bp duplexes, even when the source/capillary temperature was increased well above their solution melting points. Analyzed by IMS-MS, a selected construct displayed just one conformation with the heated capillary, but two with the standard sampling cone.'] Does a simple heated-capillary modification improve the analysis of non-covalent complexes by Z-spray electrospray ionization?