['The insulation of St. Jude Medical Riata® leads contains a polytetrafluoroethylene (PTFE) liner, silicone tubing, and ethylenetetrafluoroethylene (ETFE) coating on individual cable conductors. ETFE has sufficient dielectric strength to assure electrical function. This investigation intended to analyze performance of leads with and without externalized conductors and with intact and breached ETFE.', 'Testing was performed on ETFE-coated conductors to determine their ability to deliver high-voltage therapy. Tests were performed on samples under different conditions and current leakage was measured. A high-voltage test and a cyclic pulse test were performed, and the effect of lead modifications on the potential gradient from a high-voltage shock was used to determine functionality. Measurements from modified Riata® leads were compared with a control lead with all insulation and conducting elements intact.', 'Current leakage for all conditions tested, was within the acceptance criteria for the high-voltage test and the cyclic pulse test. In conductors that underwent cyclic testing, the highest value of current leakage was within the limit of acceptability for both phases of the test. Testing of leads with externalized conductors and breached ETFE showed similar potential gradients compared with a control lead.'] Does contribution of ethylenetetrafluoroethylene ( ETFE ) insulation to the electrical performance of Riata® silicone leads have externalized conductors?