['To present a novel method for analyzing the voltage waveform from high-frequency X-ray generators for radiographic systems.', 'The output signal of the actual voltage across the tube of a high-frequency generator was measured using the built-in voltage sense taps that are used for voltage regulation feedback in X-ray generators. The output signal was stored in an analyzing recorder, and the waveforms were analyzed using FFT analysis. The FFT analysis of high-frequency generators consisted of obtaining the power spectrum, comparing the major frequency components in the tube voltage waveforms, and examining the intensity of each frequency component.', 'FFT analysis enables an objective comparison of the complex tube voltage waveforms in high-frequency X-ray generators. FFT analysis detected the change in the X-ray tube voltage waveform that occurred when there were problems with the high-frequency generator.'] Do fFT analysis of the X-ray tube voltage waveforms of high-frequency generators for radiographic systems?