['Although various zirconia abutments have been introduced, insufficient data exist regarding the maximum load capacity of internal tri-channel connection zirconia implant abutments with various implant-abutment interfaces.', 'The purpose of this in vitro study was to compare the maximum load capacity of 3 different types of internal tri-channel connection zirconia abutments and to assess their mode of failure.', 'The study investigated 3 groups (n=20) of zirconia implant abutments with different implant-abutment interfaces. Group AllZr consisted entirely of zirconia (Aadva CAD/CAM Zirconia Abutment), group FrZr of a titanium insert friction-fitted to the zirconia abutment component (NobelProcera Abutment Zirconia), and group BondZr of a titanium insert bonded to the zirconia abutment component (Lava Zirconia abutment). All the abutments were thermal cycled for 20\u2008000 cycles between 5°C and 55°C. Sixty test implants made of titanium (Dummy NobelReplace) were embedded in autopolymerizing acrylic resin, and 60 zirconia copings (Lava Zirconia) with a uniform thickness of 2.0 mm were fabricated and bonded to the abutments. A universal testing machine was used to statically load all the specimens at a crosshead speed of 1 mm/min. The maximum load was recorded and used as the failure load. The fractured specimens were collected and representative specimens were studied with a stereomicroscope and scanning electron microscope (SEM). One-way ANOVA and post hoc comparisons with the Tukey HSD tests were used for statistical analysis (α=.05).', 'The mean (SD) maximum load capacity was 484.6 (56.6) N for NobelProcera, 503.9 (46.3) N for Aadva, and 729.2 (35.9) N for Lava abutments. The maximum load capacity of Lava abutments was significantly higher than that of Aadva or NobelProcera (P< 05). No significant difference between Aadva and NobelProcera abutments was noted. The mode of failure among the Aadva, NobelProcera, and Lava abutments was different.'] Do in vitro assessment of three types of zirconia implant abutments under static load?