['A study was carried out to evaluate the robustness and mutual interplay of two variables concurring to generate modulation patterns of the RapidArc (RapidArc) implementation of volumetric modulated are therapy. Dose rate (DR) and gantry speed (GS) are free parameters optimized alongside field aperture shape by the RapidArc engine; however, they are limited by machine constraints and mutually compensate in order to deliver the proper MU/deg during the gantry rotation.', 'Four test cases (one geometrical and three clinical) were selected and RapidArc plans were optimized using maximum allowed dose rates from 100 to 600 MU/min. The maximum gantry speed was fixed at 4.8 deg/s. Qualitative analysis of DR and GS patterns from these cases was summarized together with quantitative assessment of delivery parameters. Pretreatment quality assurance measurements and scoring of plan quality aimed to determine whether preferable initial conditions might be identified or the optimization engine might be invariant to those variables and capable of providing adequate plans independently from the limits applied.', 'The results of the study were: (i) High dynamic range in MU/deg is achievable across all dose rates by means of gantry speed modulation; (ii) there is a robust compensation mechanism between the two variables; (iii) from a machine delivery point-of-view, slightly improved accuracy is achieved when lower DRs are applied; however, this does not have practical consequences since measurements and plan evaluation showed a lack of clinically relevant deviation; and (iv) reduced total treatment time is a major advantage of high DR.'] Are on the impact of dose rate variation upon RapidArc implementation of volumetric modulated therapy?