     Optical counting 300px|Diagram of a particle counter 300px|Diagram of a vision-based particle counter  Applications of particle counters are separated into three primary categories: Aerosol particle counters Liquid particle counters Solid particle counters Detection methods There are several methods used for detecting and measuring particle size or size distribution — light blocking (obscuration), light scattering, Coulter principle and direct imaging. The light blocking optical particle counter method is typically useful for detecting and sizing particles greater than 1 micrometre in size and is based upon the amount of light a particle blocks when passing through the detection area of the particle counter. This type of technique allows high resolution and reliable measurement.%DIFDELCMD < %DIFDELCMD < %%% The light scattering method is capable of detecting smaller-sized particles. This technique is based upon the amount of light that is deflected by a particle passing through the detection area of the particle counter. This deflection is called light scattering. Typical detection sensitivity of the light scattering method is 0.05 micrometre or larger. However, employment of the condensation nuclei counter (CNC) technique would allow a higher detection sensitivity in particle sizes down to nanometre range. A typical application is monitoring of ultrapure water in semiconductor fabrication facilities. The light blocking method is specified for particle counters that are used for counting in hydraulic and lubricating fluids. Particle counters are used here to measure contamination of hydraulic oil, and therefore allow the user to maintain their hydraulic system, reduce breakdowns, schedule maintenance during no or slow work periods, monitor filter performance, etc. Particle counters used for this purpose typically use ISO Standard 4406:1999 as their reporting standard, and ISO 11171 as the calibration standard. Others also in use are NAS 1638 and its successor SAE AS4059D.%DIFDELCMD < %DIFDELCMD < %%% Direct imaging is a technique that uses the light emitted by a laser as a source to illuminate a cell where particles are passing through. The technique does not measure the light blocked by the particles, but rather measures the area of the particles functioning like an automated microscope. A pulsed laser diode freezes the particle motion. The light transmitted through the fluid is imaged onto an electronic camera with macro focusing optics. The particles in the sample will block the light, and the resulting silhouettes will be imaged onto the digital camera chip. Other types of particle counters Remote particle counters Small particle counters that are used to monitor a fixed location typically inside a cleanroom or mini-environment to continuously monitor particle levels. These smaller counters typically do not have a local display and are connected to a network of other particle counters and other types of sensors to monitoring the overall cleanroom performance. This network of sensors is typically connected to a facility monitoring system (FMS), data acquisition system or programmable logic controller.
Detection methods There are several methods used for detecting and measuring particle size or size distribution — light blocking (obscuration), light scattering, Coulter principle and direct imaging. The light blocking optical particle counter method is typically useful for detecting and sizing particles greater than 1 micrometre in size and is based upon the amount of light a particle blocks when passing through the detection area of the particle counter. This type of technique allows high resolution and reliable measurement. The light scattering method is capable of detecting smaller-sized particles. This technique is based upon the amount of light that is deflected by a particle passing through the detection area of the particle counter. This deflection is called light scattering. Typical detection sensitivity of the light scattering method is 0.05 micrometre or larger. However, employment of the condensation nuclei counter (CNC) technique would allow a higher detection sensitivity in particle sizes down to nanometre range. A typical application is monitoring of ultrapure water in semiconductor fabrication facilities. The light blocking method is specified for particle counters that are used for counting in hydraulic and lubricating fluids. Particle counters are used here to measure contamination of hydraulic oil, and therefore allow the user to maintain their hydraulic system, reduce breakdowns, schedule maintenance during no or slow work periods, monitor filter performance, etc. Particle counters used for this purpose typically use ISO Standard 4406:1999 as their reporting standard, and ISO 11171 as the calibration standard. Others also in use are NAS 1638 and its successor SAE AS4059D. Direct imaging is a technique that uses the light emitted by a laser as a source to illuminate a cell where particles are passing through. The technique does not measure the light blocked by the particles, but rather measures the area of the particles functioning like an automated microscope. A pulsed laser diode freezes the particle motion. The light transmitted through the fluid is imaged onto an electronic camera with macro focusing optics. The particles in the sample will block the light, and the resulting silhouettes will be imaged onto the digital camera chip. Optical counting 300px|Diagram of a particle counter 300px|Diagram of a vision-based particle counter Matrices Applications of particle counters are separated into three primary categories: Aerosol particle counters Liquid particle counters Solid particle counters %DIFDELCMD < %DIFDELCMD < %%% %DIFDELCMD < %DIFDELCMD < %%% Other types Remote particle counters Small particle counters that are used to monitor a fixed location typically inside a cleanroom or mini-environment to continuously monitor particle levels. These smaller counters typically do not have a local display and are connected to a network of other particle counters and other types of sensors to monitoring the overall cleanroom performance. This network of sensors is typically connected to a facility monitoring system (FMS), data acquisition system or programmable logic controller.
A
Detection methods There are several methods used for detecting and measuring particle size or size distribution — light blocking (obscuration), light scattering, Coulter principle and direct imaging.
A
The light blocking optical particle counter method is typically useful for detecting and sizing particles greater than 1 micrometre in size and is based upon the amount of light a particle blocks when passing through the detection area of the particle counter. This type of technique allows high resolution and reliable measurement.
A
The light scattering method is capable of detecting smaller-sized particles. This technique is based upon the amount of light that is deflected by a particle passing through the detection area of the particle counter. This deflection is called light scattering. Typical detection sensitivity of the light scattering method is 0.05 micrometre or larger. However, employment of the condensation nuclei counter (CNC) technique would allow a higher detection sensitivity in particle sizes down to nanometre range. A typical application is monitoring of ultrapure water in semiconductor fabrication facilities.
A
The light blocking method is specified for particle counters that are used for counting in hydraulic and lubricating fluids. Particle counters are used here to measure contamination of hydraulic oil, and therefore allow the user to maintain their hydraulic system, reduce breakdowns, schedule maintenance during no or slow work periods, monitor filter performance, etc. Particle counters used for this purpose typically use ISO Standard 4406:1999 as their reporting standard, and ISO 11171 as the calibration standard. Others also in use are NAS 1638 and its successor SAE AS4059D.
A
Direct imaging is a technique that uses the light emitted by a laser as a source to illuminate a cell where particles are passing through. The technique does not measure the light blocked by the particles, but rather measures the area of the particles functioning like an automated microscope. A pulsed laser diode freezes the particle motion. The light transmitted through the fluid is imaged onto an electronic camera with macro focusing optics. The particles in the sample will block the light, and the resulting silhouettes will be imaged onto the digital camera chip.
A
Matrices
D
Detection methods There are several methods used for detecting and measuring particle size or size distribution — light blocking (obscuration), light scattering, Coulter principle and direct imaging.
D
The light blocking optical particle counter method is typically useful for detecting and sizing particles greater than 1 micrometre in size and is based upon the amount of light a particle blocks when passing through the detection area of the particle counter. This type of technique allows high resolution and reliable measurement.
D
The light scattering method is capable of detecting smaller-sized particles. This technique is based upon the amount of light that is deflected by a particle passing through the detection area of the particle counter. This deflection is called light scattering. Typical detection sensitivity of the light scattering method is 0.05 micrometre or larger. However, employment of the condensation nuclei counter (CNC) technique would allow a higher detection sensitivity in particle sizes down to nanometre range. A typical application is monitoring of ultrapure water in semiconductor fabrication facilities.
D
The light blocking method is specified for particle counters that are used for counting in hydraulic and lubricating fluids. Particle counters are used here to measure contamination of hydraulic oil, and therefore allow the user to maintain their hydraulic system, reduce breakdowns, schedule maintenance during no or slow work periods, monitor filter performance, etc. Particle counters used for this purpose typically use ISO Standard 4406:1999 as their reporting standard, and ISO 11171 as the calibration standard. Others also in use are NAS 1638 and its successor SAE AS4059D.
D
Direct imaging is a technique that uses the light emitted by a laser as a source to illuminate a cell where particles are passing through. The technique does not measure the light blocked by the particles, but rather measures the area of the particles functioning like an automated microscope. A pulsed laser diode freezes the particle motion. The light transmitted through the fluid is imaged onto an electronic camera with macro focusing optics. The particles in the sample will block the light, and the resulting silhouettes will be imaged onto the digital camera chip.
R
Other types of particle counters
Other types