The Chemresistive Properties of SiC Nanocrystalline Films with Different Conductivity Type We have demonstrated the ability of thin nanocrystalline SiC films with various types of conductivity to detect oxidative (O2), reducing gases (CO, CH4) with the maximum allowable concentrations for human safety. It was shown that nncSiC films with electronic conductivity had a higher gas sensitivity Sn than pncSiC films with hole conductivity sensivity Sp to the action of gases in a wide concentration range. So, for the maximum permissible concentrations of O2 (3%), CO (0.1%), CH4 (10%) the sensitivity ratio of the films SnSp was 2.9; 4.8 and 10, respectively. For research, we used a simple resistor geometry optimizing which it is possible to significantly increase the sensitivity of films to gases in order to detect extremely low gas concentrations. Thus, based on ncSiC films, it is possible to develop hightemperature gas sensors to detect reactive gases in a wide range of concentrations, including threshold allowable values.

CO2 and O2 Detection by Electric Field Sensors In this work an array of chemical sensors for gas detection has been developed, starting with a commercial sensor platform developed by Microchip (GestIC), which isxc2xa0normally used to detect, trace, and classify hand movements in space. The system is based on electric field changes, and in this work, it has been used as mechanism revealing the adsorption of chemical species CO2 and O2. The system is composed of five electrodes, and their responses were obtained by interfacing the sensors with an acquisition board based on an ATMEGA 328 microprocessor (Atmel MEGA AVR microcontroller). A dedicated measurement chamber was designed and prototyped in acrylonitrile butadiene styrene (ABS) using an Ultimaker3 3D printer. The measurement cell size is 120 xc3x97 85 mm. Anthocyanins (red rose) were used as a sensing material in order to functionalize the sensor surface. The sensor was calibrated using different concentrations of oxygen and carbon dioxide, ranging from 5% to 25%, mixed with water vapor in the range from 50% to 90%. The sensor exhibits good repeatability for CO2 concentrations. To better understand the sensor response characteristics, sensitivity and resolution were calculated from the response curves at different working points. The sensitivity is in the order of magnitude of tens to hundreds of xc2xb5V% for CO2, and of xc2xb5V% in the case of O2. The resolution is in the range of 10xe2x88x921%xe2x80x9310xe2x88x923% for CO2, and it is around 10xe2x88x921% for O2. The system could be specialized for different fields, for environmental, medical, and food applications.