Intraocular lens
Intraocular lens (IOL) is a lens implanted in the eye as part of a treatment for cataracts or myopia. If the natural lens is left in the eye, the IOL is known as phakic, otherwise it is a pseudophakic, or false lens. Such a lens is typically implanted during cataract surgery, after the eye's cloudy natural lens (cataract) has been removed. The pseudophakic IOL provides the same light-focusing function as the natural crystalline lens. The phakic type of IOL is placed over the existing natural lens and is used in refractive surgery to change the eye 's optical power as a treatment for myopia (nearsightedness). This is an alternative to LASIK. IOLs usually consist of a small plastic lens with plastic side struts, called haptics, to hold the lens in place in the capsular bag inside the eye. IOLs were conventionally made of an inflexible material (PMMA), although this has largely been superseded by the use of flexible materials, such as silicone. Most IOLs fitted today are fixed monofocal lenses matched to distance vision. However, other types are available, such as a multifocal intraocular lens that provide multiple-focused vision at far and reading distance, and  adaptive IOLs that provide limited visual accommodation. Multifocal IOLs can further be, Trifocal IOLs or extended depth of focus (EDOF) lenses. More than six million lenses are implanted annually. The procedure can be done under local or topical anesthesia with the patient awake throughout the operation. The use of a flexible IOL enables the lens to be rolled for insertion into the capsular bag through a very small incision, thus avoiding the need for stitches. This procedure usually takes less than 30 minutes in the hands of an experienced ophthalmologist, and the recovery period is about 2–3 weeks. After surgery , patients should avoid strenuous exercise or anything else that significantly increases blood pressure. They should visit their ophthalmologists regularly for 3 weeks to monitor the implants. IOL implantation carries several risks associated with eye surgeries, such as infection, loosening of the lens, lens rotation, inflammation and nighttime halos, but a systematic review of studies has determined that the procedure is safer than conventional laser eye treatment. Though IOLs enable many patients to have reduced dependence on glasses, most patients still rely on glasses for certain activities, such as reading. These reading glasses may be avoided if Multifocal IOLs, Trifocal IOLs  or EDOF lenses are used. Medical uses. Intraocular lenses have been used since 1999 for correcting larger errors in near-sighted, far-sighted, and astigmatic eyes. This type of IOL is also called phakic intraocular lens (PIOL), as it is implanted without removing the patient's natural crystalline lens. Phakic IOL appear to be less dangerous than excimer laser surgery (LASIK) in those with significant nearsightedness. More commonly IOLs are implanted via "Clear Lens Extraction And Replacement" (CLEAR) surgery. During CLEAR, the crystalline lens is extracted and an IOL replaces it in a process that is very similar to cataract  surgery: both involve lens replacement, local anesthesia, last approximately 30 minutes, and require making a small incision in the eye for lens insertion. People recover from CLEAR surgery 1–7 days after the operation. During this time, they should avoid strenuous exercise or anything else that significantly raises blood pressure. They should visit their ophthalmologists regularly for several weeks to monitor the IOL implants. CLEAR has a 90% success rate (risks include wound leakage, infection, inflammation, and astigmatism). CLEAR can be performed only on patients ages 40 and older. This is to ensure that eye growth, which disrupts IOL lenses, will  not occur post-surgery. Once implanted, IOL lenses have three major benefits. First, they are an alternative to the excimer laser procedure (LASIK), a form of eye surgery that does not work for people with serious vision problems. Effective IOL implants also eliminate the need for glasses or contact lenses post-surgery for most patients. The cataract will not return, as the lens has been removed. The disadvantage is that the eye's ability to change focus (accommodate) has generally been reduced or eliminated, depending on the kind of lens implanted. Some of the risks that FDA have been found  so far during a three-year study of the Artisan are: Other risks include: One of the causes of the risks above is that the lens can rotate inside the eye if the PIOL is too short, if the eye was incorrectly measured, or because the sulcus has a slightly oval shape (the height is slightly smaller than the width). Toric IOLs must be powered and aligned inside the eye on a meridian that corrects the patient's preexisting astigmatism. Again, these lenses can rotate inside the eye postoperatively or be placed incorrectly by the operating surgeon. Either way, the  patient's preexisting astigmatism may not be corrected completely or may even increase. When standard IOLs are implanted with a CLEAR procedure, in substitution of the patient's crystalline, astigmatism is typically not corrected, as astigmatism is mainly attributable to a deformation of the cornea. Toric IOLs may be used during the CLEAR procedure to correct astigmatism. The surgeon can ascertain the astigmatic, or steepest, meridian in a number of ways, including manifest refraction or corneal topography. Manifest refraction is the familiar test where the eye doctor rotates lenses in front of the eye, asking the patient, "Which is better  (or clearer), this one or this one?" Corneal topography is considered a more quantitative test, and for purposes of aligning a toric IOL, most surgeons use a measurement called simulated keratometry (SimK), which is calculated by the internal programming of the corneal topography machine, to determine the astigmatic meridian on the surface of the cornea. The astigmatic meridian can also be identified using corneal wavefront technology or paraxial curvature matching. Type of surgery. Implants with or without removal of natural crystalline. Many aphakic and pseudophakic IOLs such as anterior chamber IOLs or 3 piece posterior chamber IOLs can be used  interchangeably. The exception are one piece IOLs, which must be placed within the capsular bag at the time of cataract surgery and hence cannot be used as secondary implants. Pseudophakic IOLs. Pseudophakic IOLs are lenses implanted during cataract surgery, immediately after removal of the patient's crystalline lens. Monofocal. Monofocal IOLs are standard lenses used in cataract surgery. One of the major disadvantages of these conventional IOLs is that they can only be focused for one particular distance - either optical infinity (rendering the eye emmetropic) or a fixed finite distance (rendering the eye myopic). Patients who undergo a standard  IOL implantation no longer experience clouding from cataracts, but they are unable to accommodate (change focus from near to far, far to near, and to distances in between). This is not a concern for most cataract surgeries, as they are primarily performed on elderly people that are already completely presbyopic. However, it can be a problem for patients that are not yet presbyopic (or are in the early stages of presbyopia) undergoing refractive lens exchange for the sake of correcting refractive errors. Monovision, in which one eye is made emmetropic and the other myopic, can partially compensate for the loss  of accommodation and enable clear vision at multiple distances. More versatile types of lenses (multifocal and accommodating IOLs) were introduced in 2003 in the United States, with the approval by the Food and Drug Administration. These come at an additional cost to the recipient beyond what Medicare will pay and each has advantages and disadvantages. Multifocal. Multifocal IOLs attempt to provide simultaneous viewing of distance vision and near vision. Trifocal IOLs can provide intermediate vision. Many multifocal IOL designs attempt to achieve this simultaneous viewing focus using a concentric ring design, which alternates distance and near focal points. However, many  concentric ring multifocal lenses used today are prone to glare and mildly compromised focus at all ranges of vision. People who have a multifocal IOL after their cataract is removed may be less likely to need additional glasses compared with people who have standard monofocal lenses. However, people receiving multifocal lenses may experience more visual problems than with monofocal lenses. The most common adverse visual effects from multifocal IOLs include glare, halos (rings around lights), and a loss of contrast sensitivity in low-light conditions. Accommodating. Some newer lens designs attempt to allow the eye to regain some partial focusing  ability in order to change focus from distance to near (accommodation). However, many accommodating IOLs used today only achieve a very limited improvements in near vision which reduced over time. Accommodative intraocular lenses may also have a slightly higher risk of developing posterior capsule opacification (PCO), though there is some uncertainty around this finding. PCO is a common side-effect of many cataract surgeries and is easily treatable with a one-time laser capsulotomy procedure (see below). Accommodating IOLs interact with ciliary muscles and zonules, using hinges at both ends to "latch on" and move forward and backward inside the  eye using the same mechanism as normal accommodation. These IOLs have a 4.5-mm square-edged optic and a long hinged plate design with polyimide loops at the end of the haptics. The hinges are made of an advanced silicone called BioSil that was thoroughly tested to make sure it was capable of unlimited flexing in the eye. Toric. A toric IOL is a type of toric lens used to correct preexisting corneal astigmatism at the time of cataract surgery. Astigmatism can also be treated with limbal relaxing incisions or an excimer laser procedure. About 40% of Americans have significant  astigmatism and thus may be candidates for a toric IOL. Cataract surgery with implantation of a toric IOL is essentially the same as cataract surgery with a conventional IOL. Like toric contact lenses, toric IOLs have different powers in different meridians of the lens, and they must be positioned on the correct meridian to reverse the preexisting astigmatism. If the toric IOL is not on the correct meridian, it may need to be repositioned in a second procedure. Multifocal toric. Standard toric IOLs are monofocal, permanently focused on distant objects. Multifocal toric IOLs are also available. These lenses provide the  patient not only with correction of preexisting astigmatism, but also with multiple-focused vision at far and reading distance Phakic IOLs. Phakic IOLs (PIOLs) are intraocular lenses which are placed in an eye that still contains a natural human crystalline lens. PIOLs are sometimes referred to as an 'implantable contact lenses' (ICLs). As with other IOLs, PIOLs can be either spheric or toric. Toric PIOLs have to be aligned with the meridian of astigmatism; toric IOL misalignment or rotation can lead to residual or even greater astigmatism postoperatively. Depending on their attachment site to the eye, PIOLs can be divided  into three categories: Blue-light filtering IOLs. Blue light filtering IOLs filter the UV and high-energy blue light present in natural and artificial light, both of which can cause vision problems; however too much filtering of blue light can increase depression, especially in the winter months (SAD). The trademarked "Natural Yellow" material is available in three hydrophilic IOLs. Dr. Patrick H. Benz of Benz Research and Development created the first IOL material to incorporate the same UV-A blocking and violet light filtering chromophore that's present in the human crystalline lens in order to attempt to protect the  retina after cataract extraction of the natural crystalline lens. A Cochrane Review found little evidence of important differences between blue‐light filtering and non‐blue‐light filtering lenses for protecting the macula (back of the eye) after cataract surgery. This was due to studies being too small and too short‐term to provide reliable results. Posterior capsule opacification. Posterior capsule opacification (PCO), often referred to as "after cataract", is the most common complication of cataract surgery. In a small percentage of patients, posterior chamber intraocular lenses may form PCOs a few months after implantation. They are easily treatable, and typically only require a one -time capsulotomy procedure (using a ) to clarify. Materials. The materials that have been used to manufacture intraocular lens implants include polymethylmethacrylate (PMMA), silicone, hydrophobic acrylate, hydrophilic acrylate and collamer. Polymethylmethacrylate (PMMA) was the first material to be used successfully in intraocular lenses. British ophthalmologist Sir Harold Ridley observed that Royal Air Force pilots who sustained eye injuries during World War II involving PMMA windshield material did not show any rejection or foreign body reaction. Deducing that the transparent material was inert and useful for implantation in the eye, Ridley designed and implanted the first intraocular lens in a human eye . Advances in technology have brought about the use of silicone and acrylic glass, both of which are soft foldable inert materials. This allows the lens to be folded and inserted into the eye through a smaller incision. Specifically, acrylic lenses are a better choice in people who have a history of uveitis or are likely to have to undergo retinal surgery requiring vitrectomy with replacement by silicone oil, such as persons with proliferative diabetic retinopathy or who are at high risk of retinal detachment, such as persons with high myopia. A study found that in participants with a history of  uveitis, eyes treated with hydrophobic acrylic IOLs were over twice as likely to have a best corrected visual acuity of 20/40 or more, compared to eyes treated with silicone IOLs. History. Sir Harold Ridley was the first to successfully implant an intraocular lens on 29 November 1949, at St Thomas' Hospital at London. That lens was manufactured by the Rayner company of Brighton, East Sussex, England from Perspex CQ polymethylmethacrylate (PMMA) made by ICI (Imperial Chemical Industries). It is said the idea of implanting an intraocular lens came to him after an intern asked him why he was not replacing  the lens he had removed during cataract surgery. The acrylic plastic material was chosen because Ridley noticed it was inert after seeing RAF (Royal Air Force) pilots of World War II with pieces of shattered canopies in their eyes (this acrylic resin is known by several trade names including Lucite and Plexiglas). The intraocular lens did not find widespread acceptance in cataract surgery until the 1970s, when further developments in lens design and surgical techniques had come about. As of the early 2000s, more than a million IOLs were implanted annually in the United States. That number was estimated by  the World Health Organization to have increased to 20 million annually worldwide by 2010 (for cataract surgery), and has projected increased IOL surgeries to reach 32 million worldwide by 2020.