High-intensity focused ultrasound
High-intensity focused ultrasound (HIFU) is a non-invasive therapeutic technique that uses non-ionizing ultrasonic waves to heat or ablate tissue. HIFU can be used to increase the flow of blood or lymph, or to destroy tissue, such as tumors, via thermal and mechanical mechanisms. Given the prevalence and relatively low cost of ultrasound, HIFU has been subject to much research and development. The premise of HIFU is that it is a non-invasive low-cost therapy that can at minimum outperform the current standard of care. The technology is similar to ultrasonic imaging, although  lower frequencies and continuous, rather than pulsed waves are used to achieve the necessary thermal doses. However, pulsed waves may also be used if mechanical rather than thermal damage is desired. Acoustic lenses are often used to achieve the necessary intensity at the target tissue without damaging the surrounding tissue. An analogy is using a magnifying glass to focus sunlight; only the focal point of the magnifying glass has high intensity. Although lenses have traditionally been used, phased arrays are increasingly common as they allow the focal position to be easily changed. HIFU is traditionally combined with other imaging techniques  such as medical ultrasound or MRI to enable guidance of the treatment and monitoring. History. The first investigations of HIFU for non-invasive ablation were reported by Lynn et al. in the early 1940s. Extensive important early work was performed in the 1950s and 1960s by Fry and Sanghvi at the University of Illinois and Carl Townsend, Howard White and George Gardner at the Interscience Research Institute of Champaign, Ill., culminating in clinical treatments of neurological disorders. In particular High Intensity ultrasound and ultrasound visualization was accomplished stereotaxically with a Cincinnati precision milling machine to perform accurate ablation of brain  tumors. Until recently, clinical trials of HIFU for ablation were few (although significant work in hyperthermia was performed with ultrasonic heating), perhaps due to the complexity of the treatments and the difficulty of targeting the beam noninvasively. With recent advances in medical imaging and ultrasound technology, interest in HIFU ablation of tumors has increased. The first commercial HIFU machine, called the Sonablate 200, was developed by the Focal Surgery and launched in Europe in 1994 after receiving CE approval, bringing a first medical validation of the technology for benign prostatic hyperplasia (BPH). Comprehensive studies by practitioners at more than one  site using the device demonstrated clinical efficacy for the destruction of prostatic tissue without surgical loss of blood or long term side effects. Later studies on localized prostate cancer by Murat and colleagues at the Edouard Herriot Hospital in Lyon in 2006 showed that after treatment with the Ablatherm (EDAP TMS, Lyon, France)?, progression-free survival rates are very high for low- and intermediate- risk patients with recurrent prostate cancer (70% and 50% respectively) HIFU treatment of prostate cancer is currently an approved therapy in Europe, Canada, South Korea, Australia. Use of magnetic resonance-guided focused ultrasound was first cited  and patented in 1992. The technology was later transferred to InsighTec in Haifa Israel in 1998. The InsighTec ExAblate 2000 was the first MRgFUS system to obtain FDA market approval in the United States. Medical uses. There is no clear consensus on the boundaries between HIFU and other forms of therapeutic ultrasound. In academic literature, HIFU usually refers to the high levels of energy required to destroy tissue through ablation or cavitation, although it is also sometimes used to describe lower intensity applications such as occupational therapy and physical therapy. Either way, HIFU is used to non-invasively heat tissue  deep in the body without the need for an incision. The main applications are the destruction of tissue, increasing perfusion and physical therapy. The use of ultrasound in the treatment of musculoskeletal conditions is another use in the physiotherapy setting. Neurological disorders. One of the first applications of HIFU was the treatment of Parkinson's disease in the 1940s. Although ineffective at the time, HIFU has the capacity to lesion pathology. A focused ultrasound system is approved in Israel, Canada, Europe, Korea and Russia to treat essential tremor, neuropathic pain, and Parkinsonian tremor. This approach enables treatment of the brain  without an incision or radiation. In 2016, the US Food and Drug Administration ("FDA") approved Insightec's Exablate system to treat essential tremor. Treatment for other thalamocortical dysrhythmias and psychiatric conditions are under investigation. Uterine adenomyosis and fibroids. Treatment for symptomatic uterine fibroids became the first approved application of HIFU by the US Food and Drug Administration (FDA) in October 2004. Studies have shown that HIFU is safe and effective, and that patients have sustained symptomatic relief is sustained for at least two years without the risk of complications involved in surgery or other more invasive approaches. Up to 16 -20% of patients will require additional treatment. Cancers. HIFU is an attractive option for tumors in locations that are hard to reach or non-resectable. Of particular interest are intestinal cancers and brain cancer. When recommending treatment, a clinician must consider other modalities to determine if HIFU is a viable option. Prostate cancer. HIFU is being studied in men with prostate cancer. HIFU was requested 501k multiple safety data in the United States for prostate tissue ablation in 2015. HIFU is being used for ablation of prostate cancer. Liver cancer. HIFU is well studied in liver cancer and in many  studies report a high response rate and positive patient outcome. Abscopal effect. During the treatment of metastasized liver cancer with HIFU, immune responses have been observed in locations that are distant from the focal region. Although the mechanism of this systemic response is unknown, it is thought to be elicited by the release of tumor antigens with retained immunogenicity through histotripsy. Other cancers. HIFU has been successfully applied in treatment of cancer to destroy solid tumors of the bone, brain, breast, pancreas, rectum, kidney, testes, prostate. Palliative care. HIFU has been found to have palliative effects. CE approval has been  given for palliative treatment of bone metastasis. Experimentally, a palliative effect was found in cases of advanced pancreatic cancer. Prostate enlargement. Treatment of prostate enlargement (benign prostatic hyperplasia) by HIFU from inside the intestine (transrectal) has turned out to be unsuccessful. In some countries, not in USA, HIFU has also been offered from the inside of the prostate, that is, via a catheter in the prostatic urethra. Evidence as of 2019 is lacking. In England the National Institute for Health and Care Excellence (NICE) in 2018 classified the method as "not recommended". In the US (as of 2019) not even  the technical device needed for the treatment has been approved. Mechanism. HIFU beams are precisely focused on a small region of diseased tissue to locally deposit high levels of energy. The focusing effect of the transducer allows high sound pressures to be delivered to a focal point without causing unwanted damage to other tissue. This increase in pressure can cause a number of effects including heating and cavitation. In 2015 the FDA had requested the 501K Safety tests for SonaCare HIFU device for the ablation of prostate tissue. Temperature. The temperature of tissue at the focus will rise to  between 65 and 85 °C, destroying the diseased tissue by coagulative necrosis. If tissue is elevated above the threshold of 60 °C for longer than 1 second this process is irreversible. Higher temperatures are usually avoided to prevent boiling of liquids inside the tissue. Each sonication (individual ultrasound energy deposition) treats a precisely defined portion of the targeted tissue. The entire therapeutic target is treated by using multiple sonications to create a volume of treated tissue, according to a protocol developed by the physician. Anesthesia is not required, but sedation is generally recommended. The amount of damage caused  in the tissue can be modeled using Cumulative Equivalent Minutes (CEM). The treatment will be done by means of the shape of the cylinder in the tissue, therefore, the ideal formula will be adopted by the primitive Bessel's (Cylinder) function. The ideal formulations of the CEM equation have been supposed as Dewey and Sapareto: formula_1 with the integral being over the treatment time, R=0.5 for temperatures over 43 °C and 0.25 for temperatures between 43 °C and 37 °C, a reference temperature of 43 °C, and time T is in minutes. The equations and methods  described in this report are not intended to represent the only approach for thermal dose estimation; . As an acoustic wave propagates through the tissue, part of it is absorbed and converted to heat. With focused beams, a very small region of heating can be achieved deep in tissues (usually on the order of millimeters). Tissue damage occurs as a function of both the temperature to which the tissue is heated and how long the tissue is exposed to this heat level in a metric referred to as "thermal dose". By focusing at more than one place or by scanning the  focus, a volume can be thermally ablated. Thermal doses of 120-240 min at 43 °C coagulate cellular protein and leads to irreversible tissue destruction. There is some evidence that HIFU can be applied to cancers to disrupt the tumor microenvironment and trigger an immune response, as well as possibly enhance the efficacy of immunotherapy. Mechanical. Inertial cavitation. At high enough acoustic intensities, cavitation (microbubbles forming and interacting with the ultrasound field) can occur. Microbubbles produced in the field oscillate and grow (due to factors including rectified diffusion), and can eventually implode (inertial or transient cavitation). During inertial cavitation , very high temperatures occur inside the bubbles, and the collapse during the rarefaction phase is associated with a shock wave and jets that can mechanically damage tissue. Stable cavitation. Stable cavitation creates microstreaming which induces high shear forces on cells and leads to apoptosis. Elaborating, bubbles produced by the vaporization of water due to acoustic forces oscillate under a low-pressure acoustic field. Strong streaming may cause cell damage but also reduces tissue temperature via convective heat loss. Theory. There are several ways to focus ultrasound—via a lens (for example, a polystyrene lens), a curved transducer, a phased array , or any combination of the three. This concentrates it into a small focal zone; it is similar in concept to focusing light through a magnifying glass. This can be determined using an exponential model of ultrasound attenuation. The ultrasound intensity profile is bounded by an exponentially decreasing function where the decrease in ultrasound is a function of distance traveled through tissue: formula_2 formula_3 is the initial intensity of the beam, formula_4 is the attenuation coefficient (in units of inverse length), and z is distance traveled through the attenuating medium (e.g. tissue). In this model, formula_5 is a measure of the  power density of the heat absorbed from the ultrasound field. Sometimes, SAR is also used to express the amount of heat absorbed by a specific medium, and is obtained by dividing Q by the tissue density. This demonstrates that tissue heating is proportional to intensity, and that intensity is inversely proportional to the area over which an ultrasound beam is spread—therefore, focusing the beam into a sharp point (i.e. increasing the beam intensity) creates a rapid temperature rise at the focus. The ultrasound beam can be focused in these ways: Beam delivery. Beam delivery consists of beam steering and  image guidance. The beam has the ability to pass through overlying tissues without harm and focus on a localized area with size limit of 3–4 cm. The area at the focal point of the beam undergoes coagulative necrosis. Following ablation a distinct boundary forms between healthy and necrotic tissue (width less than 50 microns). Beam steering. The most common transducer used is a concave focusing transducer with a fixed aperture and a fixed focal length. Phased array transducers can also be used with different arrangements (flat/bowl). Image guidance. HIFU therapy requires careful monitoring and so is usually performed in  conjunction with other imaging techniques. Pre-operative imaging, for instance CT and MRI, are usually used to identify general parameters of the target anatomy. Real-time imaging, on the other hand, is necessary for safe and accurate noninvasive targeting and therapy monitoring. Both MRI and Medical ultrasound imaging have been used for guidance in FUS treatment. These techniques are known as Magnetic Resonance guided Focused Ultrasound Surgery (MRgFUS) and Ultrasound guided Focused Ultrasound Surgery (USgFUS) respectively. MRgFUS is a 3D imaging technique which features high soft tissue contrast and provides information about temperature, thus allowing to monitor ablation. However, low  frame rate makes this technique perform poorly in real-time imaging and high costs represent a significant limitation to its use. USgFUS, differently, is a 2D imaging technique in which, although no system to provide quantitative information on temperature has been commercially developed so far, several benefits are exploited, such as high frame rate (up to 1000 images per second), low cost and minimal adverse health effects. Another reason why ultrasound is ideal for image guidance is it verifies the acoustic window in real time since it is the same modality as the therapy. The implication of this is that  if the target region is not visualized by ultrasound imaging before and during HIFU therapy, then it is unlikely that HIFU therapy will be effective in that specific region. In addition, treatment outcomes can be estimated in real time through visual inspection of hyperechoic changes in standard B-mode images.