ETCR ETCR2100+ Digital Clamp Ground Earth Resistance Meter Tester 0.01-1200Ω
Brand: ETCR
<p>The lead time is about 2-3 business days; FedEx/DHL delivery time is about 5-7 days.</p> <b>Specificaiton:</b> <p></p> Function:Ground resistance test, Loop resistance test <br> Power Supply:6VDC 4 PCS of AA Alkaline batteries <br> Resistance Range:0.01-1200Ω <br> Resistance Resolution:0.001Ω <br> Highest Resistance Accuracy:±1%±0.01Ω <br> Data memory:99 Sets <br> Alarm Threshold Setting Range:Resistance: 1-199Ω <br> Alarm Sound and Light:“Beep-beep-beep” alarm tone, Alarm Symbol Flashing, Press AL key to turn on, turn off Alarm or set Alarm Threshold. <br> Interference Signal Recognition:“Beep-beep-beep” tone, NOISE symbol blinking <br> Working Temperature and Humidity:-20℃-55℃；20％RH-90％RH <br> LCD Screen:4 Digits LCD display, Screen L×W: 47mm×28.5mm <br> Clamp Size:Φ32mm <br> Clamp Opening Extent:32mm <br> Tester Weight:1160g (With batteries) <br> Clamp Dimension:L×W×T: 285mm×85mm×56mm <br> Protection Level:Dual Insulation <br> Structural Feature:Clamp CT <br> Shifting:Automatic shifting <br> External magnetic field:＜40A/m <br> External electrical field:＜1V/m <br> Single measurement time:0.5 second <br> Resistance measurement frequency:＞1KHz <br> Measured current frequency:50/60Hz Automatic measurement <br> Low Battery Indication:Low battery symbol displays when battery is lower then 5.3V <br> Memory Full Indication:MEM symbol blinks when all 99 sets memory is full <br> Over Range Indication:Out of range indication “OLΩ”, “L0.01” or “OL A” <br> Auto Power-off:Auto Power-off <br> Power Consumption:Max 50mA either in booting up or normally working, Continuously working for 30 hours <br>
Boot Time: Self-test in 0.5 second, boot instantly
Measuring Range:0.01-1200Ω, Wider range
Beep Alarming: Yes
Clamp Size:Φ32mm
Stored Data:99 sets