Indoor Air Quality 9999ppm Digital Carbon Dioxide Temperature Humidity NDIR Sensor IAQ CO2 Monitor WB DP Tester
Brand: Gain Express
This meter notify <b>CO2 level, Air Temperature, Dew Point </b>(DP), <b><br>Wet Bulb Temperature (WB) </b>and <b>Humidity </b>(RH) and is an ideal instrument for Indoor Air Quality (IAQ) diagnosis. <br> <br> <b>Features:</b> <br> <br> Backlight, Hold data, Min/Max <br> Dual power source, battery (inlcuded) or 9V adaptor <b> (**note: adaptor is not included on set).</b> Power from battery is automatically cut off when 9V adaptor is in use <br> Easy manual calibration on CO2 and humidity <br> PC connection for on-line loggin and data analysis (not included on set) <br> Manual calibration of CO2 and Humidity is available <br> Auto / Manual power off <br> <br> <b>Specifications:</b> <br> <br> Measuring Range <br> * CO2: 0~9999ppm (2001~9999 out of scale) <br> * Temperature : -10~60°C (14~140°F) <br> * Humidity : 0.1% ~ 99.9%RH <br> * Dew Point : -20.0~59.9°C <br> * Wet Bulb : -5.0~59.9°C <br> * Temp : ±0.6°C / ±0.9°F <br> Accuracy : <br> * C02 : ±50ppm ±5% of reading (0~2000ppm) other ranges are not specified <br> * Temp : ±0.6°C / ±0.9°F <br> * Humidity : ±3%RH (at 25°C, 10~90% RH); ±5%RH (at 25°C, others) <br> Response time <br> * CO2 : <30 seconds (90% step change) <br> *Temp : lt;2 mins (90% step change) <br> *RH : lt;10 mins (90% step change) LCD size : 26*44mm Operating condition : 0~50°C, 0~95%RH (avoid condensation) <br> Storage Condition : -20~50°C, 0~95%RH (avoid condensation) <br> Item Weight : 200g <br> Package Weight: 1.16kg<br> Package Size: 37cm x 23cm x 7cm <br> <br> <br> <b>Set Includes:</b> <br> 1 x Main Unit <br> 4 x AA batteries <br> English Operation Manual <br>Hard Carrying Case
Triple displays of CO2 level, Temperature (°F/ °C) and Humidity (Air, Wet Bulb, or Dew Point Info are all available)
Stable NDIR Sensor for CO2 Detection. Ideal indoor air quality meter for home, laboratories, hotels, etc
Sound Warning Alarm notifies operator when CO2 levels reach or exceed a preset threshold
Power Supplied by Battery (Included) or 9V Adaptor . Optional Software CD for PC Connection Online Logging is also possible.
Measuring Range:CO2: 0~9999ppm,(2001~9999 out of scale), Resolution : 1ppm,0.1°C/°F, 0.1%RH