Carbon Dioxide CO2 Detector Monitor Meter Tester Sensor Temperature Humidity Indoor 9999ppm NDIR Air Quality IAQ Handheld Portable Digital Wet Bulb
Brand: TEKCOPLUS
Color: Hand Held CO2 Meter
<p>Shows CO2 level, temp. (°C/°F), Humidity, Dew Point & Wet Bulb. Auto/Manual power off, PC connection loggin and data analysis. Backlight, Hold data, Min/Max<br /><br /></p> <p><b>Features:</b></p> <p>1.Triple displays of CO2 level, temperature (°C/°F) and Humidity<br />2.Stable NDIR sensor for CO2 detection</p> <ul> <li>Statistics of weighted averages</li> <ul> <li>TWA (8 hours weighted average)</li> <li>STEL (15 minutes-weighted average)</li> </ul> </ul> <p>3.Updates reading every second on the meter, output to PC every 2 seconds</p> <p><b>Specifications:</b></p> <ul> <li>Measuring Range</li> <ul> <li>CO2: 0~9999ppm <ul> <li>(2001~9999 out of scale)</li> </ul> </li> <li>Temperature : -10~60°C (14~140°F)</li> <li>Humidity : 0.1% ~ 99.9%RH</li> <li>Dew Point : -20.0~59.9°C</li> <li>Wet Bulb : -5.0~59.9°C</li> </ul> <li>Resolution : 1ppm,0.1°C/°F, 0.1%RH</li> <li>Accuracy : <ul> <li>C02 : ±50ppm ±5% of reading (0~2000ppm) other ranges are not specified</li> <li>Temp : ±0.6°C / ±0.9°F</li> <li>Humidity : ±3%RH (at 25°C, 10~90% RH); ±5%RH (at 25°C, others)</li> </ul> </li> <li>Warm-up time : 30 seconds</li> <li>Response time <ul> <li>CO2 : <30 seconds (90% step change)</li> <li>Temp : <2 mins (90% step change)</li> <li>RH : <10 mins (90% step change)</li> </ul> </li> <li><b>NO RELAY</b></li> <li>Operating condition : 0~50°C, 0~95%RH (avoid condensation)</li> <li>Storage Condition : -20~50°C, 0~95%RH (avoid condensation)<br /><br /><br /></li> </ul> <p><b>Set Includes:</b></p> <p>1 x Portable CO2 meter<br />4 x AA batteries<br />1 x Power Adapter<br />English Operation Manual<br />Hard Carrying Case</p>
MONITORING & ANALYSIS Allows continuous monitoring of CO2, RH & Temperature Level. Data output to PC every 2 seconds.
MULTIPLE MEASUREMENT DISPLAYS AT ONCE- CO2, Temperature, and Relative Humidity readings are all shown on the LCD display at the same time
NORMAL/ MAX/ MIN/ AVERAGE (15mins/ 8hrs) MODE- It can show not only the normal, maximum and minimum CO2 readings but also the weighted average CO2 readings in the past 15mins (STEL mode) and 8 hours (TWA mode).
AUDIBLE CO2 SOUND- The alarm beeps when the CO2 Level exceeds the preset value. The alarm threshold is adjustable.
NDIR SENSOR for CO2 DETECTION- Non-dispersive infrared technology is used to ensure the reliability and long-term stability. Measures 0~9999ppm while 0~2000ppm can maintain a high accuracy at +/- 50ppm