Nanovna 10KHz-1.5Ghz Vector Network Analyzer Kit,SEESII MF HF VHF UHF Antenna Analyzer Measuring S Parameters, Voltage Standing Wave Ratio, Phase, Delay, Smith Chart with 2.8" Touch Screen
Brand: Seesii
Color: Black
10KHz-1500MHz Vector Network Analyzer Kit MF HF VHF UHF Antenna Analyzer<br /> <br /> <b>Specification:</b><br /> ● PCB: 54mm x 85.5mm x 11mm (without connectors, switches)<br /> ● Measurement Frequency: 50KHz ~ 300MHz (50KHz -900MHz, enable extended firmware)<br /> ● RF output: -13dbm (maximum -9dbm)<br /> ● Measurement range: 70dB (50kHz-300MHz), 60dB (300M-600MHz), 50dB (600M-900MHz) enable extended firmware);<br /> ● Port SWR: < 1.1<br /> ● Display: 2.8 inch TFT (320 x240)<br /> ● USB interface: USB type-C communication mode: CDC (serial)<br /> ● Power: USB 5V 120mA, built-in 400mAh battery , maximum charging current 0.8A<br /> ● Number of scanning points: 101 (fixed)<br /> ● Display Tracking: 4, Marking: 4, Setting Save: 5<br /> Measuring S parameters, voltage standing wave ratio, phase, delay, Smith chart...<br />  <br /> <b>Package Included:</b><br /> 1X NanoVNA Host<br /> 1X USB Type-C DataCable<br /> 2X 30mm SMA male to Male RG174 RF Cable<br /> 1X SMA Simple Calibration Kit<br /> 1X SMA Female to Female Connector<br />
Unit is designed to fit for Original Equipment, form and function Simple and practical PC control software, you can export Touchstone (snp) files for various radio design and simulation software through PC software.
The improved frequency algorithm can use the odd harmonic extension of si5351 to support the measurement frequency up to 1.5GHz. The 50KHz-300MHz frequency range of the si5351 direct output provides better than 70dB dynamic, The extended 300M-900MHz band provides better than 60dB of dynamics, and the 900M-1.5GHz band is better than 40dB of dynamics
The extended 300M-600MHz band provides better than 60dB of dynamics, and the 600M-900M band is better than 50dB of dynamics.
Requirements accomplished by leveraging Spectra Premium's OE engine management systems design expertise.Platinum connector threads reduce the number of connection points for faster reaction to change.Ceramic-built printed circuit boards are protected by silica gel, providing superior resistance to temperature changes and vehicle vibrations.
This is a DIY product that provides perfect vector network measurement capabilities in the size of a credit card. It includes a 2.8-inch touch screen and 500mAh battery, which can be used offline for 2 hours. For longer measurement, please connect external power supply or contact me to make a larger battery version. The TX/RX method can measure the complete S11 and S21 parameters. If you need to obtain S12 and S22, you need to manually replace the transceiver port wiring.