HiLetgo USB Logic Analyzer Device With EMI Ferrite Ring USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug
Brand: HiLetgo
Color: Gray
<b>HiLetgo USB Logic Analyzer Device With USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug </b><br> Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz;<br> The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions;<br> 8-channel;<br><br> <b>Voltage range:</b><br> Input voltage range: -0.5V to 5.25V<br> Input Low Voltage: -0.5V to 0.8V<br> Input High Voltage: 2.0V to 5.25V<br> Works with 5V, 3.3V, 2.5V, 2.0V systems. May work with 1.8V but not recommended.<br> Input Impedance: 1Mohm || 10pF (typical, approximate)<br> Crystal: +/-20ppm, 24MHz<br> Error/Accuracy: pulse-width measurement: +/- 42ns (at 24MHz).<br> <br><b>Packing Included:</b><br> 1 * Logic Analyser<br> 1 * USB Cable<br> 1 * Dupont Cable<br>
The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions; 8-channel
Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz;
The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions;
Input voltage range: -0.5V to 5.25V; Input Low Voltage: -0.5V to 0.8V; Input High Voltage: 2.0V to 5.25V
Input Impedance: 1Mohm || 10pF (typical, approximate); Crystal: +/-20ppm, 24MHz