HiLetgo Battery Capacity Meter Discharge Tester Analyzer 1.5V-12V Battery Capacity Meter Discharge Tester
Brand: HiLetgo
Color: Blue
<b>Product description</b><br> Instructions:<br> 1. The first test should be fully charged battery.<br> 2. Connect the battery to be tested positive to positive input, a negative input to the negative, can not be reversed (with load reversal may damage the circuit)! Load connected to the output of the positive and negative output, to work through the tester micro USB power supply (not available desktop or laptop USB powered), then the battery voltage.<br> 3. Direct start test need only one press of the "OK" button, the tester can automatically develop an appropriate termination voltage of the battery full charge voltage, and will start flashing after the test three times. Need to develop artificial termination voltage only when the battery voltage display state by pressing "+" or "-" buttons to modify the termination voltage display beginning with P, behind the representative voltage resolution 0.1V, set up after pressing the "OK "start testing.<br> 4. After the start of the test the tester will control the load of the electronic switch is turned on, the test data shows that the process will release capacity (Ah), current discharge current (A) and battery voltage (V) between the wheel was. When the battery voltage reaches the set cut-off voltage, load control switches off the tester display data stays in capacity (Ah) and above and the corresponding indicator flashes quickly together, now displays the actual capacity of the battery is discharging capacity, about the press "OK" to terminate flashing allows stable data display, press again "OK" button to return to the power-on state can just replace the battery test the next section ......<br><br> Error codes and their meanings:<br> Err1: battery voltage is above 15V<br> Err2: battery voltage is lower than the termination voltage<br> Err3: Battery can not afford to load or discharge current line too much resistance<br> Err4: overcurrent (current exceeds 3.1A)<br>
Power supply voltage: DC4.5-6V (micro USB interface) ; Working current: less than 70mA. Supported by current: the maximum 3.000A resolution 0.001A.
The maximum measurement error voltage: 1%+0.03V ; The maximum measurement error of current: 2%+-0.010A.
The discharge voltage: 1.00V-15.00V resolution 0.01V ; Termination voltage range: 0.5-11.0V.
Board size: 50mmX34mm ; Product size: 50mmX36mmX17mm (length X width X high maximum position size, containing copper foot height)
Note: this circuit in order to improve the voltage measurement accuracy, specially designed DC bias.when the terminal is what all don't pick will display a small voltage, does not affect the actual measurement.if you are short connected with the input terminal (absolute 0V) will display 0. To understand the principle can consult the superposition theorem in electrical engineering.