LEAGY VICTOR VC60B+ Digital Insulation Resistance Tester Megohm Meter DC250/500/1000V AC750V Orange with Black
Brand: LEAGY
Color: Black
<b> ntroduction: The whole instrument is stylish and superior grade in appearance and highly stable in performance. It can be operated by both hands through using a shoulder strap. It is most suitable for the needs of insulation resistance check, such as electric, cable, electrical and mechanical equipment, telecommunication equipment and electrical facilities. <br> <br>Specification:</b> <br>Model: VC60B+ <br>Measuring Voltage: 50V±10% 500V±10% 1000V±10% <br>Insulation Range: 0.1MΩ~20 MΩ 0.1MΩ~50 MΩ 0.1MΩ~100 MΩ 20MΩ~500 MΩ 50MΩ~1000 MΩ 100MΩ~2000 MΩ <br>Accuracy: ±(4%+2) <br>Current of Short Circuit: 1.8mA 1.8mA 1.6mA <br>Median Resistance: 2MΩ 2MΩ 5MΩ <br>Terminal Position: L & E <br>Voltage Test(AC750V) Accuracy: ±(1.0% of reading + 6 digits) <br>Input Impendance: 1MΩ <br>Frequency Response: (50~200) Hz <br>Terminal Position: ACV & G <br><b> Alarm Function Red indicator light is illuminated</b> <br>Low Battery Indication: Yes <br>Over Load Indication: Yes <br>Unit Symbol Display: Yes <br>Auto Power Off: Yes <br>Max. Display: 1999 <br>Over Range Display: “1” <br>Operating Temperature: 0℃~40℃ <br>Relative Humidity: 30%RH~85%RH <br>Dimension: 105×164×70mm <br>Weight: 900g <br>Power: 9V,6 x AA/1.5V battery (Included) <br> <b><br>Package Include:</b> <br>1 x Digital Insulation Tester <br>1 x 10A Test Leads <br>2 x Alligator Clip <br>1 x Shoulder Strap <br>1 x User Manual <b><br>6 x AA/1.5V battery
Measure Range up to 2000MΩ 1000V with Accuracy of ±(4% of reading +2 digits)
Large LCD display (84.8*59.8mm), with backlight, max display "1999"
Data hold and auto power off
Over Range indication: display "1"
Low power consumption: Less than 300mw at unload measurement