WARNING LETTER
Insightra Medical Inc.
MARCS-CMS 706921 —
June 03, 2025
Delivery Method:
VIA Electronic Mail
Product:
Medical Devices
Recipient:
Recipient Name
Kai W. Trompeter
Recipient Title
President and Chief Executive Officer
Insightra Medical Inc.
2543 Madison St. Ste. 105
Clarksville
,
TN
37043
United States
kai@tn-mi.com
sandi@tn-mi.com
Issuing Office:
Center for Devices and Radiological Health
United States
WARNING LETTER
CMS #706921
June 3, 2025
Dear Mr. Trompeter:
During an inspection of your firm located in Clarksville, TN, on January 21, 2025, through January 27, 2025, an investigator from the United States Food and Drug Administration (FDA) determined that your firm manufactures the Double Pump as well as the Ultra Intra Aortic Balloon Pump (IABP) Catheter Kits. Under section 201(h) of the Federal Food, Drug, and Cosmetic Act (the Act), 21 U.S.C. § 321(h), these products are devices because they are intended for use in the diagnosis of disease or other conditions, or in the cure, mitigation, treatment, or prevention of disease, or to affect the structure or any function of the body.
We received a response from Sandie Hetland, Chief Financial Officer, dated February 14, 2025, concerning our investigator’s observations noted on the FDA 483, List of Inspectional Observations, that was issued to your firm. We address this response below, in relation to each of the noted violations. These violations include, but are not limited to, the following:
Unapproved Device Violation(s)
Our inspection revealed that the IABP Catheter Kit is adulterated under section 501(f)(1)(A) of the Act, 21 U.S.C. § 351(f)(1)(A), because your firm does not have an approved application for premarket approval (PMA) in effect pursuant to section 515(b) of the Act, 21 U.S.C. § 360e(b), or an approved application for an investigational device exemption under section 520(g) of the Act, 21 U.S.C. § 360j(g) for the device as described and marketed. This device is also misbranded under section 502(o) the Act, 21 U.S.C. § 352(o), because your firm introduced or delivered for introduction into interstate commerce for commercial distribution this device with major changes or modifications to the intended use without submitting a new premarket notification to FDA as required by section 510(k) of the Act, 21 U.S.C. § 360(k), and 21 CFR 807.81(a)(3)(ii).
Specifically, your device was cleared under K082746. In the supplement to your firm’s premarket notification, you recommended the device for use with the Datascope System 97, System 98, System 98XT, CS100 and CS300 devices and the Arrow ACAT, AutoCat, AutoCat 2, AutoCat 2 Wave, and KAAT II devices. However, your firm’s labeling, including promotional materials and information on your website (www.insightra.com), now state that, in addition to the devices mentioned above, your device is compatible with the Cardiosave IABP Hybrid and the AutoCat 3.
Specifically, your website (www.insightra.com) contains a certificate of compatibility listing these devices. Additionally, a section of your website also states that there is “safe compatibility of the ULTRA 7FR IABP with the most common pumping consoles on the market,” and lists the Cardiosave IABP Hybrid and the AutoCat 3, in addition to the pump consoles that are within the scope of the current clearance. Finally, there is a document found on the Product Resources page of your website entitled, “Operator Presentation,” Although the presentation includes at the bottom of each slide the language, “THIS PRESENTATION IS STRICTLY CONFIDENTIAL – FOR INTERNAL TRAINING PURPOSES ONLY,” the document is publicly available on your website. The second slide of that presentation includes the statement, “Compatible with all common Datascope and Arrow consoles – kit comes with adaptors.” The pump console device is critical to the safe and effective operation of your device because your device cannot function without it. Pump consoles may have different performance characteristics and differences in performance characteristics between pump consoles may impact the safety and effectiveness of your device.
The listing of the Cardiosave IABP Hybrid and the AutoCat 3 in the labeling for your device as compatible pump consoles constitutes a significant change or modification to your device’s intended use that requires a new premarket notification. As explained in Section V.A of FDA’s Guidance Document, Deciding When to Submit a 510(k) for a Change to an Existing Device - Guidance for Industry and Food and Drug Administration Staff, on pages 20 and 21, when making changes concerning the compatibility or interoperability of a device with other devices, components or accessories, manufacturers should consider the criticality of the other device, component or accessory. The more critical the other device, component, or accessory is to overall system function, the more likely a labeling change regarding compatibility or interoperability could significantly affect safety or effectiveness. As we have noted, the pump console is required for the operation of your device and therefore has substantial impact on the safety and effectiveness of your device. A change to a different pump console that may have different performance characteristics could impact the safety and effectiveness of an IABP catheter. The addition of the Cardiosave IABP Hybrid and the Arrow AutoCat 3 required the submission of a new 510(k), per 21 CFR 807.81(a)(3)(ii).
We also note that your device’s current indications listed in your operator’s manual are as follows:
The Insightra Intra Aortic Balloon Catheter Kit has the following indications for use:
A. Refractory Unstable Angina.
B. Impending Infarction.
C. Post infarction angina.
D. Refractory left ventricular failure.
E. Complications of Acute MI (i.e. Acute MR or VSD or papillary muscle rupture).
F. Cardiogenic shock.
G. Support for diagnostic, percutaneous revascularization and interventional procedures.
H. Ischemic related intractable ventricular arrythmias.
I. Septic shock.
J. Intraoperative pulsatile flow generation.
K. Weaning from cardiopulmonary bypass.
L. Cardiac support for non-cardiac surgery.
M. Prophylactic support in preparation for cardiac surgery.
N. Post-surgical myocardial dysfunction/low cardiac output syndrome.
O. Cardiac contusion.
P. Mechanical bridge to other assist devices.
Q. Cardiac support following correction of anatomical defects.
While we acknowledge that these indications were cleared under K082746, on December 30, 2013, FDA issued a final order to reclassify intra-aortic balloons and control systems (FR 2013-Doc 31218) (78 FR 79300). It states that:
“The Food and Drug Administration (FDA) is issuing a final order to reclassify intra-aortic balloon and control system (IABP) devices when indicated for acute coronary syndrome, cardiac and non-cardiac surgery, or complications of heart failure, a preamendments class III device, into class II (special controls), and to require the filing of a premarket approval application (PMA) or a notice of completion of a product development protocol (PDP) for IABPs when indicated for septic shock or pulsatile flow generation.”
It further instructs that “[a] PMA or notice of completion of a PDP is required to be filed with the Food and Drug Administration on or before March 31, 2014, for any intra-aortic balloon and control system indicated for septic shock or pulsatile flow generation that was in commercial distribution before May 28, 1976, or that has, on or before March 31, 2014, been found to be substantially equivalent to any intra-aortic balloon and control system indicated for septic shock or pulsatile flow generation that was in commercial distribution before May 28, 1976.”
In addition, on March 20, 2014, FDA sent your firm a letter informing you of the final order requirements and clarifying the steps you should take to ensure compliance with the final order. We informed you that you must file a PMA or notice of completion of a PDP within 90 days of the date of issuance of the final order for IABP devices indicated for septic shock or pulsatile flow generation. We noted that if you did not file a PMA or notice of completion of a PDP for an IABP device with these indications within 90 days of the date of issuance of the final order, you must cease commercial distribution of your device for these indications because the device is deemed adulterated under section 501(f)(1)(A). We also noted that you may notify FDA that you wish to revise your indications by submitting an add-to-file to your previously cleared 510(k) on or before March 31, 2014.
FDA has not received a notice of completion of a PDP or a PMA from your firm nor did you notify FDA that you wish to revise your indications by submitting an add-to-file to your previously cleared 510(k). Because you continue to market your device with a septic shock and pulsatile flow indication, your device is adulterated under section 501(f)(1)(A) of the Act, 21 U.S.C. § 351(f)(1)(A), because your firm does not have an approved PMA in effect pursuant to section 515(b) of the Act, 21 U.S.C. § 360e(b), or an approved application for an investigational device exemption under section 520(g) of the Act, 21 U.S.C. § 360j(g) for the device as described and marketed.
Medical Device Reporting (MDR) Violation(s)
1. Our inspection revealed that your firm has failed to adequately develop, maintain, and implement written MDR procedures as required by 21 CFR 803.17. For example, during the inspection, your firm provided a draft document titled “Adverse Event & Vigilance Reporting,” SOP‐16, Version 2.1, undated, and identified it as your written MDR procedure. After reviewing the procedure, the following deficiencies were noted:
a. There is no evidence that the draft document identified as your firm’s written MDR procedure has been implemented, as required by 21 CFR 803.17. For example, there is no effective date.
b. The document does not establish internal systems that provide for timely and effective identification, communication, and evaluation of events that may be subject to MDR requirements, as required by 21 CFR 803.17(a)(1). For example, the procedure, as written, does not define the term “serious injury” as defined in 21 CFR 803.3.
***This is a repeat Observation from Warning Letter# 20-15 issued on May 21, 2015***
Your firm’s response dated February 14, 2025, did not address the noted violation, as it was not listed on the FDA 483.
Our inspection also revealed that your firm’s IABP Catheter Kit is misbranded under section 502(t)(2) of the Act, 21 U.S.C. § 352(t)(2), in that your firm failed or refused to furnish material or information respecting the device that is required by or under Section 519 of the Act, 21 U.S.C. § 360i, and 21 CFR Part 803 ‐ Medical Device Reporting. Significant violations include, but are not limited to, the following:
2. Failure to submit a report to FDA no later than 30 calendar days after the day that your firm received or otherwise became aware of information, from any source, that reasonably suggests that a device that your firm markets may have caused or contributed to a death or serious injury, as required by 21 CFR 803.50(a)(1).
For example, the information included in CMP‐36 reasonably suggests that a patient experiencing worsening heart failure underwent a procedure involving your firm’s IABP Catheter Kit, which malfunctioned (i.e., ruptured balloon) during use, potentially causing or contributing to a serious injury requiring the patient to be upgraded on the heart transplant list. There is no information included in the complaint indicating that the device did not cause or contribute to the patient’s serious injury. Your firm became aware of the event in October 2023. Based on our records, FDA has not yet received an MDR for the adverse event referenced in Complaint# CMP-36.
Your firm’s response dated February 14, 2025, did not address the noted violation, as it was not listed on the FDA 483.
3. Failure to submit a report to FDA no later than 30 calendar days after the day that your firm received or otherwise became aware of information, from any source, that reasonably suggests that a device that your firm markets has malfunctioned and this device or a similar device that you market would be likely to cause or contribute to a death or serious injury, if the malfunction were to recur, as required by 21 CFR 803.50(a)(2).
For example, the information included in CMP‐44 reasonably suggests that your firm’s IABP Catheter Kit malfunctioned (i.e., ruptured ballon) during use. In October 2023, your firm became aware through its investigation results for Complaint# CMP‐36, that the referenced malfunction (i.e., ruptured balloon) for the same or similar device, caused or contributed to the patient sustaining a serious injury. This prior incident established a precedent demonstrating that such a malfunction (i.e., balloon rupture leading to IABP failure) was previously linked to a serious injury. It is reasonable to conclude that a recurrence of this malfunction would be likely to cause or contribute to a death or serious injury. As such, the malfunction referenced in CMP‐44 represents an MDR reportable malfunction for which an MDR must be submitted. Your firm became aware of the event on January 22, 2024. Based on our records, FDA has not yet received an MDR for the malfunction event subject of Complaint #CMP‐44.
***This is a repeat Observation from the 2015 FDA Inspection***
Your firm’s response dated February 14, 2025, did not address the noted violation, as it was not listed on the FDA 483.
Quality System Regulation Violation(s)
This inspection revealed that these devices are adulterated within the meaning of section 501(h) of the Act, 21 U.S.C. § 351(h), in that the methods used in, or the facilities or controls used for, their manufacture, packing, storage, or installation are not in conformity with the current good manufacturing practice requirements of the Quality System regulation found at Title 21, Code of Federal Regulations (CFR), Part 820.
4. Failure to adequately document corrective and preventive action activities and results, as required by 21 CFR 820.100(b). For example:
a. SOP-15, Corrective and Preventive Action, effective 10/26/2022 indicates that CAPAs are generated and documented in the Qualio system, an electronic record system used by your firm to manage quality documents. The following CAPAs were not adequately documented:
i. CAPA-10 (closure date: 3/6/2023) related to batch release not being performed per the firm’s procedure was closed without completion of all the CAPA sections on the form.
ii. CAPA-12 (closure date: 7/19/2023) related to the complaint reporting the Double Pump had an unspecified failure was closed without completion of all the CAPA sections on the form.
iii. CAPA-11 which lists CASE-2023-00163655 on the top of the form did not contain any filled-out information on the form.
iv. CAPA-22 was related to internal audit findings noting supplier evaluations and reevaluations were not being performed as per schedule. The verification of effectiveness activities performed on 11/11/2024 for this CAPA which was to “Verify and Check supplier evaluation for all critical suppliers,” failed to address the overdue evaluations for at least 3 suppliers.
v. CAPA-24 was opened to address relevant information of identified quality problems not being captured in records. The verification of effectiveness, “Training has been imparted to new staff” was listed as being completed on 07/13/2023, despite the CAPA being opened on 11/11/2024.
b. Your firm failed to initiate a CAPA to document the investigations, the actions to correct and prevent the recurrence of quality problems, and the verifications that actions were effective for known quality issues. For example:
i. Between July 2023 and May 2024, at least 35 complaints were received from your US customers related to a device malfunction that causes leaking of the Double Pumps used for diagnostic and operative arthroscopic procedures. No CAPA was opened to investigate this known and ongoing issue to identify actions needed to correct and prevent the recurrence of nonconforming product or other quality issues.
ii. CMP-27, dated 8/8/2023, was related to a complainant reporting the inner lumen of the IABP Catheter device being broken during patient use. Your firm failed to open a CAPA to document your investigation activities and corrective actions. Furthermore, a corrective action implemented for this quality issue was to update the distributor training to recommend the insertion technique of pushing the catheter at a 45-degree angle to prevent kinking which may lead the central lumen of the catheter to break. There was no assessment of the verification and/or validation activities needed for this corrective action or an identification of the required changes in methods and procedures needed to correct and prevent the identified quality problem.
***This is a repeat Observation from the 2018 FDA Inspection***
We reviewed your firm’s response and conclude that it is not adequate. The response failed to comprehensively address the systemic deficiencies identified within your CAPA process. Your firm only addressed the examples listed as opposed to addressing how your firm will correct the CAPA issues systemically to prevent the recurrence of the issues identified. For example, your firm fails to include as a corrective action the need to update the CAPA procedure to ensure that CAPAs that are voided have a documented rationale and to ensure that the CAPA void process does not impact product quality. Furthermore, in regard to CAPA-22, your firm failed to address why the supplier you no longer are doing business with was still listed on your firm’s approved supplier list and how your firm was planning on making corrections to prevent this issue from recurring.
Regarding the ongoing leaking Double Pump issue discussed in part b of this observation, your firm’s response does not address whether your firm will open a CAPA to address the ongoing known quality issue. Your contract manufacturer proposed a potential correction to the known leaking issue in a February 2024 complaint report. According to your contract manufacturer, they had already proposed the potential correction to your firm before February 2024, as well, including in August 2021; however, your firm failed to make corrections or to assess this proposed solution. It is a requirement of 21 CFR 820.100 that you as a manufacturer investigate the cause of nonconformities relating to product, processes, and the quality system. Furthermore, you are required to identify actions needed to correct and prevent the recurrence of nonconforming product and other quality problems.
Your firm is planning to have personnel involved in the CAPA documentation be retrained with an emphasis on proper record-keeping and compliance with SOP-15; however, it is important to make the proper corrections to your CAPA procedure to ensure it includes all requirements of 21 CFR 820.100 such as verifying or validating the corrective and preventive action to ensure that such action is effective and does not adversely affect the finished device, and submitting relevant information on identified quality problems, as well as corrective and preventive actions, for management review.
5. Failure to adequately maintain complaint files, as required by 21 CFR 820.198(a). For example:
a. Section 5.1.2.2 of your procedure SOP-14 Customer Complaints, versions 1.0 and 2.0 state, “Complaints are evaluated using the MDV, MDR or MPR Decision Trees (found in SOP-16 Adverse Event & Vigilance Reporting) to determine if the complaint is a Reportable Complaint; the Decision Tree is attached to the Complaint Notification.” The following customer complaint notifications involving patients did not have an MDR decision tree attached:
i. CMP-30
ii. CMP-44
iii. CMP-70
iv. CMP-27
v. CMP-36
b. Section 5.1.2.3 of your procedure SOP-14 Customer Complaints, versions 1.0 and 2.0, state, “Quality Assurance promptly initiates investigation and reporting to meet the required report dates.” Review of complaints revealed complaint records were not opened in a timely manner as evidenced by:
i. Compliant CMP-44, received on 01/22/2024 and entered into the Qualio database on 2/29/2024. CMP-44 was opened due to a complainant reporting a rupture of the intra-aortic balloon.
ii. Complaint CMP-36, received in October 2023 and entered into the Qualio database on 12/04/2023. CMP-36 is associated with a serious injury reported to the FDA as a Medwatch Report by a complainant and brought to your firm’s attention by FDA.
c. Section 5.1.1. of your procedure SOP-14 Customer Complaints, versions 1.0 and 2.0 state, “ Quality Assurance follows-up on any incomplete information to proceed with the complaint.” Review of complaints revealed complete and accurate information was not always included within your complaints as evidenced by:
i. The complaint description within CMP-36 does not align with what was included within the documented investigation or the MedWatch Report submitted for the event. For example, the investigation documented that the Ultra IABP device was used for a matter of minutes; however, the MedWatch report indicates an implant date of 8/23/23 and explant date on 8/27/23. The complaint record does not explain the contradiction.
ii. CMP-52, dated 03/26/2024, was opened due to a complainant reporting four leaking cassettes at the level of the insertion of the tubing into the rigid cassette. This complaint record failed to include that this is a known product issue reported by complainants previously and that there was currently an ongoing investigation by the firm. Furthermore, under the “Investigation Summary” section of the complaint record, it mentions, “There is no complaint product for return to further evaluation.” Your firm failed to document any follow up attempts to obtain information pertinent to conducting a meaningful investigation.
iii. CMP-44, dated 02/29/2024, failed to include any follow up attempts for the firm to carry out an adequate investigation. The “Closure Activities” section of the complaint record states, “No complaint product for return to further evaluation.
The detailed complaint description was not provided. hence [sic], it is difficult to carry out the investigation.”
***This is a repeat Observation from the 2015 FDA Inspection and Warning Letter# 20-15 issued on May 21, 2015***
We reviewed your firm’s response and conclude that it is not adequate. The response was not comprehensive to address the systemic deficiencies identified within your complaint handling process. For example, your response mentions that you will retrieve the MDR decision trees for the complaints referenced in the observation; however, your firm did not clearly address how you will ensure that MDR determinations are made within required timeframes. Section 5.1.2.2. of the procedure SOP-14, version 2.0 states, “These trees do not need to be completed for each complaint file IF the complaint does not impact any patient or surgical procedure.” Please note, complaints are to be evaluated to determine whether the complaint represents an event which is required to be reported to FDA under 21 CFR 803, Medical Device Reporting, as required by 21 CFR 820.198(a).
Your response also mentions that the Qualio software will be modified to include a provision for closure time justification in cases where a complaint closure exceeds 30 days. It does not mention what corrective actions will be implemented to ensure complaints are processed in a uniform and timely manner once your firm becomes aware of the complaints, as required by 21 CFR 820.198(a). Please note, the delay between awareness of a complaint and the date the complaint is entered into the database has the potential for not allowing for an adequate follow up and prolongs the initiation of the investigation. Lastly, your response mentions that you will reopen CMP-52 and add all necessary details to document a clear understanding of events; however, you do not address making the proper corrections to your complaint procedure to ensure complaints undergo adequate investigation and include all relevant known information.
6. Failure to adequately establish and maintain procedures for the identification, documentation, validation or where appropriate verification, review, and approval of design changes before their implementation, as required by 21 CFR 820.30(i). For example, your firm failed to ensure the following design changes were adequately identified, documented, and reviewed prior to implementation:
a. Design change related to changing the shelf life of the IABP Catheter Kit from 1 year to 5 years. This design change was not properly identified and reviewed. Although real time aging studies to support a 5-year shelf life were conducted on October 20, 2015, the studies did not include any evaluation that the sterility for the device can be maintained over the labeled shelf-life. Furthermore, in 2022, the contract sterilizer for the Ultra IABP Catheter Kit was changed from
(b)(4)
to
(b)(4)
, and there is no evidence to demonstrate that you evaluated the impact of this change on the shelf-life for your product.
b. Design change related to implementing the use of a
(b)(4)
with the IABP Catheter Kit. This design change was not properly identified and reviewed. For example, you failed to ensure that the change included a proper update of the instructions for use and that the syringe maintains sterility when performing
(b)(4)
filling of the intra-aortic balloon. Furthermore, the design history file was not updated to reflect the addition of using the
(b)(4)
or the removal of this instruction to use a
(b)(4)
on or about 11/07/2024.
***This is a repeat Observation from the 2018 FDA Inspection***
We reviewed your firm’s response and conclude that it is not adequate. It fails to address the need to update your design control procedures to ensure they include the requirements of 21 CFR 820.30(i).
Corrections and Removals Violation(s)
Our inspection also revealed that your firm’s IABP Catheter Kit devices are misbranded under section 502(t)(2) of the Act, 21 U.S.C. § 352(t)(2), in that your firm failed or refused to furnish material or information respecting the device that is required by or under section 519 of the Act, 21 U.S.C. § 360i, and 21 CFR Part 806 Medical Devices; Reports of Corrections and Removals. Violations include, but are not limited to, the following:
7. Failure to submit any report required by 21 CFR 806.10(a) within 10 working days of initiating a correction or removal, as required by 21 CFR 806.10(b). For example, you updated your IABP distributor training materials to add instructions to use an angle of insertion of 45 degrees or less to reduce the incidence of kinking. Kinking may lead to a break of the central lumen of the catheter, cause poor gas filling, or contribute to balloon leak or rupture. Your Failure Modes and Effects Analysis document states that balloon rupture or leak has gas emboli as a potential effect to the failure mode. Gas embolism occurs when gas bubbles enter the bloodstream and obstruct blood flow, potentially leading to serious complications or death. This update to your training materials meets the definition of a medical device correction or removal initiated to reduce a risk to health posed by the device or remedy a violation of the Act caused by the device which may present a risk to health, for which you are required by 21 CFR 806.10 to report the correction or removal to FDA. As of 4/15/2025, you did not submit a Medical Device Report of Correction or Removal to FDA for this action.
Your firm’s response, dated February 14, 2025, did not address the noted violation, as it was not listed on the FDA 483.
Your firm should take prompt action to address any violations identified in this letter. Failure to adequately address this matter may result in regulatory action being initiated by the FDA without further notice. These actions include, but are not limited to, seizure, injunction, and civil money penalties.
Other federal agencies may take your compliance with the FD&C Act and its implementing regulations into account when considering the award of federal contracts. Additionally, should FDA determine that you have Quality System regulation violations that are reasonably related to premarket approval applications for Class III devices, such devices will not be approved until the violations have been addressed. Should FDA determine that your devices or facilities do not meet the requirements of the Act, requests for Certificates to Foreign Governments (CFG) may not be granted.
Please notify this office in writing within fifteen business days from the date you receive this letter of the specific steps your firm has taken to address the noted violations, as well as an explanation of how your firm plans to prevent these violations, or similar violations, from occurring again. Include documentation of the corrections and/or corrective actions (which must address systemic problems) that your firm has taken. If your firm’s planned corrections and/or corrective actions will occur over time, please include a timetable for implementation of those activities. If corrections and/or corrective actions cannot be completed within fifteen business days, state the reason for the delay and the time within which these activities will be completed. Your firm’s response should be comprehensive and address any violations included in this Warning Letter. If you believe that your products are not in violation of the FD&C Act, include your reasoning and any supporting information for our consideration as part of your response.
Your firm’s response should be sent via email to Melissa Michurski, Establishment Assessment Team 2, Assistant Director, at CDRHEnforcement@fda.hhs.gov. Please include in the subject line, CMS Case #706921 when replying. If you have any questions about the contents of this letter, please contact: Ashley Mutawakkil at Ashley.Mutawakkil@fda.hhs.gov.
Finally, you should know that this letter is not intended to be an all-inclusive list of the violations at your firm’s facility. It is your firm’s responsibility to ensure compliance with applicable laws and regulations administered by FDA. The specific violations noted in this letter and in the Inspectional Observations, FDA 483, issued at the close of the inspection may be symptomatic of serious problems in your firm’s manufacturing and quality management systems. Your firm should investigate and determine the causes of any violations and take prompt actions to address any violations and bring the products into compliance.
Sincerely,
/S/
Matthew G. Hillebrenner
Deputy Director
Office of Product Evaluation and Quality
Center for Devices and Radiological Health
Content current as of:
07/08/2025
Regulated Product(s)
Medical Devices