WARNING LETTER
Rechargeable Power Energy North America LLC
MARCS-CMS 570911 —
May 09, 2019
Delivery Method:
United Parcel Service
Product:
Medical Devices
Recipient:
Recipient Name
Ping Gao
Recipient Title
Owner & President
Rechargeable Power Energy North America LLC
6585 Arville St., Suite A
Las Vegas
,
NV
89118-4327
United States
Issuing Office:
Division of Medical Device and Radiological Health Operations West (MDRHW)
19701 Fairchild
Irvine
,
CA
92612-2506
United States
WARNING LETTER
CMS # 570911
UNITED PARCEL SERVICE
OVERNIGHT DELIVERY
May 9, 2019
Ping Gao
Owner & President
Rechargeable Power Energy North America, LLC
6585 Arville St., Suite A
Las Vegas, NV, 89118-4327
Dear Ms. Gao:
The United States Food and Drug Administration (FDA) conducted an inspection of your firm’s medical device operations, Rechargeable Power Energy North America, LLC located in Las Vegas, Nevada, from November 5 - 9, 2018. During the inspection, an FDA investigator determined that your firm is a medical device manufacturer (i.e. Contract Manufacturer) of Class III Automated External Defibrillator (AED) Battery, intended to replace the original equipment manufacturer (OEM) battery used in
(b)(4)
automated external defibrillators. Under section 201(h) of the Federal Food, Drug, and Cosmetic Act (the Act), 21 U.S.C. § 321(h), these products are devices because they are intended for use in the diagnosis of disease or other conditions or in the cure, mitigation, treatment, or prevention of disease, or to affect the structure or any function of the body.
Observations Pertaining to Quality System Regulation Provisions and Regulations
This inspection revealed your devices are adulterated within the meaning of section 501(h) of the Act, 21 U.S.C. § 351(h), in that the methods used in, or the facilities or controls used for, their manufacture, packing, storage, or installation are not in conformity with the current good manufacturing practice requirements of the Quality System regulation found at Title 21, Code of Federal Regulations (CFR), Part 820. We received your response dated December 14, 2018, concerning our investigator's observations noted on the Form FDA 483, List of Inspectional Observations (FDA 483) that was issued on November 9, 2018. We address the response below, in relation to each of the noted observations. These violations include, but are not limited to, the following:
1. Failure to validate with a high degree of assurance, a process whose results cannot be fully verified by subsequent inspection and test, as required by 21 CFR 820.75(a).
During the inspection we observed your battery testing and calibration equipment used on the Automated External Defibrillator (AED) Battery has not been validated to demonstrate the results are repeatable for every battery tested and calibrated. For example:
a. Your firm was unable to demonstrate that the process used for finished product functional testing of the
(b)(4)
Automated External Defibrillator Battery (part number
(b)(4)
) has been validated.
b. Also, your Operations Manager stated the testing equipment communicates the results to a computer and the computer calibrates each battery during installation on the customer’s defibrillator to read it as a new unused battery, has not been validated. Your firm has developed a Validation Procedure (SOP 11, Rev. 1), however this procedure has not been executed to validate the production equipment.
We reviewed your firm's response and conclude it is not adequate. You indicated an independent review will be conducted, and a
(b)(4)
test will be performed to validate the batteries shelf-life. This proposed correction does not address the observation and the proposed timeline is unacceptable. While we agree the shelf-life of your batteries should be verified through appropriate design testing; this does demonstrate your battery testing process is validated as required by 21 CFR 820.75. In response to this Warning Letter, you should provide your plans to ensure that your production equipment (battery testing and calibration equipment) is undergoing appropriate equipment and software validation. FDA issued a Final Order:
Effective Date of Requirement for Premarket Approval for Automated External Defibrillator Systems; Republication
on February 3, 2015, which represents a tailored approach to help manufacturers assure the quality and reliability of AEDs (AED system consists of an AED device, battery, pad electrode, and if applicable, an adapter).
This is a repeat observation from the previous FDA inspection, dated January 20 – 22, 2016.
2. Failure to establish procedures for receiving, reviewing, and evaluating complaints by a formally designated unit, as required by 21 CFR 820.198(a). For example:
a. Your firm has not established complaint handling procedures and records of complaints received do not include an evaluation for MDR reporting to FDA.
b. During the inspection your Operations Manager stated your customer (
(b)(4)
) handles all complaints unless an investigation was required, which would be evaluated by your firm to determine the root cause. Also, complaints received by the firm are mostly resolved over the phone with instructions by your customer on how to properly install the battery into the device. In addition, the FDA investigator reviewed your Returned Goods Authorization (RGA) forms returned by customers for the use of
(b)(4)
batteries for issues such as disconnected battery wires, defective cells, and battery reporting not as calibrated, were not documented as complaints and fails to include an evaluation for reporting adverse events to FDA.
We reviewed your firm's response and conclude that it is not adequate. You indicated complaints are handled by your firm and your customer (
(b)(4)
) while operating with a general customer satisfaction procedure. Your procedure is unclear if the requirements for maintaining complaint files as required by 21 CFR 820.198 have been met. You indicated, if corrective action is required or the customer raises concerns about product and/or service related issues, you will email or phone the customer to resolve the problem. In response to this Warning Letter, you should provide assurance that a complaint handling procedure has been established for receiving, reviewing, and evaluating complaints by a formally designated unit and that complaints are evaluated for MDR reporting to FDA. You should provide evidence, if possible, based on retrospective review that corrections have been implemented to ensure compliance with the requirements of this part.
This is a repeat observation from the previous FDA inspection, dated January 20 – 22, 2016.
3. Failure to establish procedures for device history records, as required by 21 CFR 820.184. For example:
a. Your firm has not established procedures to ensure device history records (DHRs) for each lot or unit are maintained to demonstrate the device is manufactured in accordance with device master record (DMR).
b. Your production records do not document each step of production to demonstrate the device was manufactured in accordance with the established specifications in the device master record (DMR).
c. During the inspection, your Operations Manager indicated approximately
(b)(4)
AED batteries are produced
(b)(4)
. As such, there are no records of components used and results of finished product testing to demonstrate the devices met established requirements prior to product release.
We reviewed your firm's response and conclude it is not adequate. You indicated you will implement a procedure for inspection and testing requirements that include the use of device history records. You referenced the use of an Inspection/Testing checklist to identify the DHR requirements. There is no assurance that procedures for DHRs were established to demonstrate the device is manufactured is accordance with the DMR and the requirements of 21 CFR 820.184. In response to this Warning Letter, we request you conduct retrospective review of your manufactured products to determine if additional lots of DHRs are deficient. Please provide examples of your device history records (DHRs) for the products manufactured.
4. Failure to establish procedures to ensure that all purchased or otherwise received product and services conform to specified requirements, as required by 21 CFR 820.50. For example:
a. Your firm has no procedures for supplier evaluation and approval, including requirements to be met by suppliers, and that suppliers will notify the firm of any changes to materials and services.
b. Evaluation of your printed circuit boards (PCBs) and PCB component suppliers used for production of the AED Battery were not documented.
We reviewed your firm's response and conclude it is not adequate. Your firm provided information regarding purchase order requirements, recordkeeping, and purchasing review. You indicated you will periodically review the purchase order log for variances, trends, etc. and that the QA Manager will audit the purchasing process. However, it is not clear how your firm determines a supplier to be critical and if the evaluation of your supplier is based on their ability to meet quality requirements. In addition, your response does not address how your firm defines the type and extent of control to be exercised over your suppliers and how they will notify you with any chances to product or service.
This is a repeat observation from the previous FDA inspection, dated January 20 – 22, 2016.
5. Failure to establish procedures for acceptance activities, as required by 21 CFR 820.80(b). For example:
a. Your firm has not established procedures for receiving acceptance activities to verify incoming products are inspected, tested, or otherwise verified as conforming to specified requirements.
b. During the inspection, your Operations Manager stated that all materials received are tested in production. The FDA investigator explained that the firm must have procedures for incoming acceptance to describe the firm’s practices and that the firm’s documentation must reflect that materials are being tested.
This observation was verbally discussed during the closeout meeting on 11/9/2018 and not included on the Form FDA 483. However, after further review by the Agency and the information provided during the inspection, it was determined that this observation is significant and noted on the Warning Letter to ensure compliance with the requirements of 21 CFR 820.80(b).
6. Failure to establish procedures for corrective and preventative action, as required by 21 CFR 820.100(a). For example:
? Your firm has not established procedures for implementing corrective and preventive actions. CAPA procedures must be established and followed in the event a nonconforming product and other quality problems are found.
We reviewed your firm's response and conclude it is not adequate. Your firm provided information regarding CAPA requirements to detect recurring quality problems. However, once nonconformities have been investigated and action(s) have been identified, your procedure fails to identify the requirements for verifying and/or validating CAPA's to ensure that such action is effective and is not adversely affecting the finished device. In response to the Warning Letter, we request that you revise your procedure to include the CAPA requirements as required by 21 CFR 820.100. If possible, you should provide evidence based on retrospective review that correction has been implemented.
This is a repeat observation from the previous FDA inspection, dated January 20 – 22, 2016.
7. Failure to establish procedures for quality audits, as required by 21 CFR 820.22. For example:
? Your firm has not established procedures for conducting and documenting internal audits of the firm’s quality system. You have not conducted any internal quality audits.
We reviewed your firm's response and conclude it is not adequate. Your firm indicated your Internal Audit Program procedure has been established for conducting and documenting internal quality audits. In response to the Warning Letter, you should provide evidence that a quality audit checklist or audit plan has not been established to determine when the firm plans to conduct their scheduled audits, and what processes or areas will be audited to assure that the quality system in compliance with the QSR.
During the previous FDA inspection (1/20-22/2016), it was noted that the firm failed to conduct quality audit related to the firm's operations.
8. Failure to establish procedures for management review, as required by 21 CFR 820.20(c). For example:
? Your firm has not established procedures for conducting and documenting management review meetings. Your 2018 management review meeting does not document the agenda items covered in the meeting.
We reviewed your firm's response and conclude it is not adequate. Your firm provided a Management's QMS Commitment procedure for conducting and documenting management review meetings. However, your response does not include any record that a management meeting was conducted or is scheduled for 2019. In response to the Warning Letter, we request you provide a timeline for when you plan to have the management meeting and to ensure that the agenda items covered satisfies the quality system requirements.
During the previous FDA inspection (1/20-22/2016), it was noted that the firm failed to conduct management reviews related to the firm's operations.
Observation Pertaining to Medical Device Reporting Provisions and Regulations
Our inspection also revealed that your firm’s Class III Automated External Defibrillator (AED) Batteries are misbranded under section 502(t)(2) of the Act, 21 U.S.C. § 352(t)(2), in that your firm failed or refused to furnish material or information respecting the device that is required by or under section 519 of the Act, 21 U.S.C. §360i, and 21 CFR Part 803 – Medical Device Reporting. Significant violations include, but are not limited to, the following:
1. Failure to develop and implement written MDR procedures, as required by 21 CFR 803.17. For example:
? Your firm has not established procedures for handling reports of serious injury or death and
reporting such incident as Medical Device Reports (MDRs) to the FDA. There were no
documented complaints reviewed during the inspection which would require filing an MDR.
We reviewed your firm's response and conclude it is not adequate. You indicated you have established an MDR procedure and provided information for production and process control requirements. However, you did not explain or reference the requirements for developing, maintaining, and implementing MDRs. In addition, it is not clear if the firm evaluates all complaints to determine whether it represents an event which is required to be reported by FDA under part 803. In response to this Warning Letter, you should provide supporting documents to determine that written MDR procedures have been implemented and that complaints have been evaluated for MDRs.
This is a repeat observation from the previous FDA inspection, dated January 20 - 22, 2016.
Medical Device Registration and Listing
A review of your firm’s 2018 and 2019 device registration and listing revealed that you have not registered or listed any of your device products held for sale for which you are a manufacturer (i.e. contract manufacturer). Your firm’s device is a Class III Automatic External Defibrillator (AED) Battery cleared under
(b)(4)
.
Under section 510 of the Act (21 U.S.C. § 360), manufacturers of medical devices are required to annually register with the FDA. In September 2007, section 510 of the Act was amended by the Food and Drug Administration Amendments Act of 2007 (Pub. L. 110-85) to require domestic and foreign device establishments to submit their annual establishment registration and device listing information to FDA by electronic means [section 510(p) of the Act (21 U.S.C. § 360(p))] during the period beginning October 1st and ending December 31st of each year. Our records indicate that your firm has not fulfilled annual registration and listing requirements for fiscal year 2019.
Therefore, all of your firm’s devices are misbranded within the meaning of section 502(o) of the Act (21 U.S.C. § 352(o)), in that the devices were manufactured, prepared, propagated, compounded, or processes in an establishment not duly registered under section 510 of the Act (21 U.S.C. § 360) and were not included in a list required by section 510(j) of the Act (21 U.S.C. § 360(j)).
Your firm should take prompt action to correct the violations addressed in this letter. Failure to promptly correct these violations may result in regulatory action being initiated by the FDA without further notice. These actions include, but are not limited to, seizure, injunction, and civil money penalties. Also, federal agencies may be advised of the issuance of Warning Letters about devices so that they may take this information into account when considering the award of contracts. Additionally, premarket approval applications for Class III devices to which the Quality System regulation violations are reasonably related will not be approved until the violations have been corrected. Requests for Certificates to Foreign Governments will not be granted until the violations related to the subject devices have been corrected.
Please notify this office in writing within fifteen business days from the date you receive this letter of the specific steps your firm has taken to correct the noted violations, as well as an explanation of how your firm plans to prevent these violations, or similar violations, from occurring again. Include documentation of the corrections and/or corrective actions (which must address systemic problems) that your firm has taken. If your firm’s planned corrections and/or corrective actions will occur over time, please include a timetable for implementation of those activities. If corrections and/or corrective actions cannot be completed within fifteen business days, state the reason for the delay and the time within which these activities will be completed. Your firm’s response should be comprehensive and address all violations included in this Warning Letter.
Your firm's response should be sent via e-mail to: US Food and Drug Administration, Division 3/West, Office of Medical Device and Radiological Health Operations at ORADevices3FirmResponse@fda.hhs.gov. Please identify your response with CMS case #570911 when replying. If you have any questions about the contents of this letter, please contact Compliance Officer Charles J. Chacko at 214-253-4939, or via email at charles.chacko@fda.hhs.gov.
Finally, you should know that this letter is not intended to be an all-inclusive list of the violations at your firm’s facility. It is your firm’s responsibility to ensure compliance with applicable laws and regulation administered by FDA. The specific violations noted in this letter and in the Inspectional Observations, FDA 483, issued at the close of the inspection may be symptomatic of serious problems in your firm’s manufacturing and quality management systems. Your firm should investigate and determine the causes of the violations and take prompt actions to correct the violations and bring the products into compliance.
Sincerely,
/S/
Shari J. Shambaugh
Program Division Director
Office of Medical Device and Radiological Health
Division 3 West
Cc: Sean K. Turner, Management Representative
Content current as of:
05/21/2019
Regulated Product(s)
Medical Devices