Spreadsheet: 867321d1714142508-extract-data-from-inconsistent-number-of-rows-packnonconformancehistoryreportv2
Sheet: Sheet4

[NAME]
Date Between 01-04-2024 and 07-04-2024
 | Log Number: | 6532137 |  | Reason: | DIRTY |  |  | Status: Closed
 |  | Date | Time | Operator | [NAME] | Cycle | Comment \ Action |  |  |  |  | Accountable Operator |  |  | Accountable Operator | Date | Time | Description
 | Raised: | 01/04/2024 | 12:20 | [NAME] | WASHER 1 | 22739 | Scope sticky |  |  |  |  | DarrylBurge1 |  |  | DarrylBurge1 | 2024-04-01 00:00:00 | 12:20:00 | CITY STORZ LAPAROSCOPE 10mm 30 DEGREE TH 5/7
 | Actioned: | 01/04/2024 | 20:53 | [NAME] | - | - | Tray returned to [NAME] to be cleaned correctly and rewashed. |  |  |  |  |  |  |  | DarrylBurge2 | 01/04/2024 | 14:38:00 | NEURO CRANIOTOMY (10kg) (HR STANDARD) | FORCEPS DISSECTING CAIRNS 2x3 TEETH 165mm
 | Code | User  Code \ GIAI |  | Decription |  | Lot | Original Position | New Position |  |  |  | Transfer Point |  |  | Do not mind if this area is blank or if it repeats the line above |  |  |  | SCISSORS DURAL McKISSOCK CURVED 146mm
 | 1002868 | 8004505522501002853 |  | CITY STORZ LAPAROSCOPE 10mm 30 DEGREE TH 5/7 |  | 66 | IAP - PACKING | NON CONFORMED TO WASH AREA |  |  |  | CITY Theatre 3 |  |  |  |  |  |  | [NAME] CURVED 180mm
 |  |  |  |  |  |  |  |  |  |  |  |  |  |  |  |  |  |  | SCISSORS MAYO CURVED 165mm
 | Log Number: | [NAME] |  | Reason: | DIRTY |  |  | Status: Closed |  |  |  |  |  |  |  |  |  |  | SCISSORS MAYO STRAIGHT 165mm
 |  | Date | Time | Operator | [NAME] | Cycle | Comment \ Action |  |  |  |  | Accountable Operator |  |  |  |  |  |  | FORCEPS ARTERY CAIRNS STRAIGHT 146mm
 | Raised: | 01/04/2024 | 14:38 | [NAME] | WASHER 2 | 61793 | Various sticky instruments. |  |  |  |  | DarrylBurge2 |  |  |  |  |  |  | NEEDLE HOLDER MAYO HEGAR 180mm
 | Actioned: | 02/04/2024 | 04:31 | [NAME] | - | - | re processed |  |  |  |  |  |  |  |  |  |  |  | NEEDLE HOLDER HALSEY 125mm
 | Code | User  Code \ GIAI |  | Decription |  | Lot | Original Position | New Position |  |  |  | Transfer Point |  |  |  |  |  |  | RETRACTOR S/R WEITLANDER-BECKMANN 3x4 PRONGS
 | 1020562 | 8004505522501020577 |  | NEURO CRANIOTOMY (10kg) (HR STANDARD) |  | 128 | IAP - CHECKING | NON CONFORMED TO WASH AREA |  |  |  | QMC Sterile Services Theatres / A Floor Theatres |  |  | DarrylBurge1 | 2024-04-02 00:00:00 | 09:23:00 | CITY [NAME]
 |  |  |  | Code | GIAI |  | Description |  |  |  | Quantity
 |  |  |  | 2028344 | 8004505522504011804 |  | FORCEPS DISSECTING CAIRNS 2x3 TEETH 165mm |  |  |  | 1
 |  |  |  | 2004022 | 8004505522504020966 |  | SCISSORS DURAL McKISSOCK CURVED 146mm |  |  |  | 1
 |  |  |  | 2032634 | 8004505522504020708 |  | [NAME] CURVED 180mm |  |  |  | 1
 |  |  |  | 2044955 | 8004505522504018364 |  | SCISSORS MAYO CURVED 165mm |  |  |  | 1
 |  |  |  | 2008499 | 8004505522504020960 |  | SCISSORS MAYO STRAIGHT 165mm |  |  |  | 1
 |  |  |  | 2058481 | 8004505522504018269 |  | FORCEPS ARTERY CAIRNS STRAIGHT 146mm |  |  |  | 1
 |  |  |  | 2017141 | 8004505522504021326 |  | NEEDLE HOLDER MAYO HEGAR 180mm |  |  |  | 1
 |  |  |  | 2052964 | 8004505522504018154 |  | NEEDLE HOLDER HALSEY 125mm |  |  |  | 1
 |  |  |  | 2011529 | 8004505522504011835 |  | RETRACTOR S/R WEITLANDER-BECKMANN 3x4 PRONGS |  |  |  | 1
 |  |  |  | Total Contents: |  |  |  |  |  |  | 9
 | Log Number: | 6532190 |  | Reason: | BYPASSED WASH CYCLE |  |  | Status: Closed
 |  | Date | Time | Operator | [NAME] | Cycle | Comment \ Action |  |  |  |  | Accountable Operator
 | Raised: | 02/04/2024 | 09:23 | [NAME] | N\a | - |  |  |  |  |  | DarrylBurge1
 | Actioned: | 02/04/2024 | 12:34 | [NAME] | - | - | WASHING
 | Code | User  Code \ GIAI |  | Decription |  | Lot | Original Position | New Position |  |  |  | Transfer Point
 | 1017354 | 8004505522501017364 |  | CITY [NAME] |  | 51 | RETURNED FROM OFFSITE | RETURNED FROM OFFSITE |  |  |  | City Sterile Services Theatres / City Sterile Store
 | Log Number: | 6532194 |  | Reason: | DIRTY |  |  | Status: Closed
 |  | Date | Time | Operator | [NAME] | Cycle | Comment \ Action |  |  |  |  | Accountable Operator
 | Raised: | 02/04/2024 | 10:28 | [NAME] | WASHER 2 | 61771 | had not been flushed through, very bloody |  |  |  |  | DarrylBurge2
 | Actioned: | 02/04/2024 | 12:36 | [NAME] | - | - | WASHING
 | Code | User  Code \ GIAI |  | Decription |  | Lot | Original Position | New Position |  |  |  | Transfer Point
 | 1072909 | 8004505522500007349 |  | E.N.T. SUCTION CATHETER 1.5mm 4.5FR (10468D) |  | 2 | WASHER - UNLOAD | NON CONFORMED TO WASH AREA |  |  |  | E.N.T. Theatre Centre
 | Log Number: | 6532220 |  | Reason: | DIRTY |  |  | Status: Closed
 |  | Date | Time | Operator | [NAME] | Cycle | Comment \ Action |  |  |  |  | Accountable Operator
 | Raised: | 02/04/2024 | 20:09 | [NAME] | WASHER 1 | 22754 | Tray pre wash inspected by [NAME] |  |  |  |  | DarrylBurge3
 | Actioned: | 03/04/2024 | 05:01 | [NAME] | - | - | re processed
 | Code | User  Code \ GIAI |  | Decription |  | Lot | Original Position | New Position |  |  |  | Transfer Point
 | 1072777 | 8004505522500007217 |  | CITY [NAME] (2 OF 4) AMIS STEM (SET 1) (2023) |  | 12 | IAP - PACKING | NON CONFORMED TO WASH AREA |  |  |  | City Sterile Services Theatres / City Sterile Store
 |  |  |  | Code | GIAI |  | Description |  |  |  | Quantity
 |  |  |  | 2069450 |  |  | ACETABULAR REAMER 48mm (OR 01.30.10.3448) |  |  |  | 1
 |  |  |  | Total Contents: |  |  |  |  |  |  | 1
 | Log Number: | 6532222 |  | Reason: | DIRTY |  |  | Status: Closed
 |  | Date | Time | Operator | [NAME] | Cycle | Comment \ Action |  |  |  |  | Accountable Operator
 | Raised: | 02/04/2024 | 21:09 | [NAME] | WASHER 6 | 14108 | Tray pre wash inspected by [NAME] |  |  |  |  | DarrylBurge2
 | Actioned: | 03/04/2024 | 05:01 | [NAME] | - | - | re processed
 | Code | User  Code \ GIAI |  | Decription |  | Lot | Original Position | New Position |  |  |  | Transfer Point
 | 1028872 | 8004505522501028921 |  | CITY UROLOGY STORZ 26FR RESECTOSCOPE SET DOUBLE IRRIGATION |  | 81 | IAP - PACKING | NON CONFORMED TO WASH AREA |  |  |  | CITY - Urology Theatres
 |  |  |  | Code | GIAI |  | Description |  |  |  | Quantity
 |  |  |  | 2019947 |  |  | SHEATH INNER 27050 XA |  |  |  | 1
 |  |  |  | 2030492 |  |  | OBTURATOR VISUAL 27050 BE |  |  |  | 1
 |  |  |  | Total Contents: |  |  |  |  |  |  | 2
 | Total Records: |  |  |  |  |  |  |  |  |  | 70