Spreadsheet: 854311d1704233457-need-to-export-2-csv-files-from-two-different-pdfs-and-merge-into-one-line-in-excel-master-data-from-both-forms
Sheet: 2013

Completed | Waiting for [NAME] or Parts | Percussionaire
Pending approval | In Repair
Customer | PO# | Contact | Phone # | Unit Serial# | Received | Services Approved | Tech | CSO | ISO | Status | Comments | [NAME] | Tracked | [NAME]
Angels of Care | PO# 2013WG11113 | [NAME] | [PHONE] Ext 5647 | IM120010054 | 2013-01-03 00:00:00 | PM and repair | RKR | CSO 130102-1 | N/A | completed | ship to Dallas | 2013-01-16 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F25213 | 2013-01-04 00:00:00 | 10kPM | EP | N/A | ISO 130103-1 | completed |  | 2013-01-23 00:00:00 | yes | N/A
IME | IME | IME | IME | F25203 | 2013-01-04 00:00:00 | 10kPM | EP | N/A | ISO 130103-2 | completed |  | 2013-01-17 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11003 | 2013-01-04 00:00:00 | Annual PM | RKR | N/A | ISO 130103-3 | Completed |  | 2013-01-07 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10818 | 2013-01-04 00:00:00 | OH | RKR | N/A | ISO 130103-4 | completed | OH kit 1/7/2013 | 2013-01-18 00:00:00 | yes | N/A
IME | IME | IME | IME | [NAME] | 2013-01-04 00:00:00 | OH | RKR | N/A | ISO 130103-5 | completed | OH kit 1/7/2013 | 2013-01-19 00:00:00 | yes | N/A
Apple Homecare | PO# 7868 | [NAME] | [PHONE] | IPVHC00363 | 2013-01-11 00:00:00 | PM/spattering/case removed | Percuss | CSO 130110-2 | N/A | completed | R11422 | 2013-03-18 00:00:00 | yes | [EMAIL]
S.TX. Monitoring | PO# 050313KF-PM | [NAME] | [PHONE] | F49194 | 2013-01-08 00:00:00 | 10K PM | EP | CSO 130111-3 | N/A | completed |  | 2013-05-14 00:00:00 | yes | [EMAIL]
Angel Medical/Firmco | compressors | BJ Prendergast | [PHONE] | IPVHC00121 | 2013-01-10 00:00:00 | compressor upgrade | Percuss | CSO 130111-4 | N/A | completed | R11414 | 2013-03-18 00:00:00 | yes | [EMAIL]
Angel Medical/Firmco | compressors | BJ Prendergast | [PHONE] | IPVHT00112 | 2013-01-10 00:00:00 | PM & compressor upgrade | Percuss | CSO 130111-5 | N/A | completed | R11415 | 2013-03-18 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5190 | Pam/Bryan | [PHONE] | C06354 | 2013-01-10 00:00:00 | 10K PM | EP | CSO 130111-6 | N/A | completed |  | 2013-04-23 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | PO# 25787 | [NAME] | [PHONE] | IPVHC00069 | 2013-01-10 00:00:00 | PM | RKR | CSO 130111-7 | N/A | completed |  | 2013-01-16 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F25211 | 2013-01-11 00:00:00 | 10k PM | EP | N/A | ISO 130111-6 | completed |  | 2013-02-22 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10841 | 2013-01-11 00:00:00 | OH | RKR | N/A | ISO 130111-7 | completed | OH KI 1/11/2013 | 1/23./2013 | yes | N/A
IME | IME PO# 1840 | IME | IME | IPV1950065 | 2013-01-14 00:00:00 | upgrade to IPV1C2066 | Percuss | N/A | ISO 130114-8 | completed | R11416 | 2013-02-06 00:00:00 | yes | N/A
IME | IME | IME | IME | F48755 | 2013-01-11 00:00:00 | 10k PM | EP | N/A | ISO 130114-9 | completed |  | 2013-01-17 00:00:00 | yes | N/A
Medical Plus Supplies | compressors | [NAME] | [PHONE] | IPVHT00120 | 2013-01-11 00:00:00 | OH compressor upgrade | Percuss | CSO 130111-8 | N/A | completed | R11419 | 2013-03-18 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5191 | Pam/Bryan | [PHONE] | C16976 | 2013-01-16 00:00:00 | 30K PM | EP | CSO 130116-9 | N/A | completed |  | 2013-05-02 00:00:00 | yes | [EMAIL]
The Care Group | compressors | [NAME] | [PHONE] | IPVHT00226 | 2013-01-16 00:00:00 | OH compressor upgrade | Percuss | CSO 130116-10 | N/A | completed | R11421 | 2013-03-18 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IPV2C0146 | 2013-01-17 00:00:00 | PM | RKR | N/A | ISO 130117-10 | completed |  | 2013-01-17 00:00:00 | yes | N/A
Children's Medical Ctr | PO# 1492078 | [NAME] | [PHONE] | IPV2C0264 | 2013-02-01 00:00:00 | PM | RKR | CSO 130118-11 | N/A | completed |  | 2013-02-13 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | C11657 | 2013-01-18 00:00:00 | 30k PM | EP | N/A | ISO 130118-11 | completed |  | 2013-01-29 00:00:00 | yes | N/A
IME | IME | IME | IME | IPVHC00245 | 2013-01-18 00:00:00 | OH compressor upgrade | Percuss | N/A | ISO 130122-12 | completed | R11462 | 2013-03-18 00:00:00 | yes | N/A
Medical Plus Supplies | PO# 26159 | [NAME] | [PHONE] | IPVHC00368 | 2013-01-18 00:00:00 | PM & power switch | RKR | CSO 130118-12 | N/A | completed |  | 2013-01-25 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO # 5201 | Pam/Bryan | [PHONE] | C09498 | 2013-01-22 00:00:00 | 10K PM | EP | CSO 130122-13 | N/A | completed |  | 2013-04-29 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10838 | 2013-01-23 00:00:00 | OH | RKR | N/A | ISO 130124-13 | completed | [NAME] 1/24/2013 | 2013-02-05 00:00:00 | yes | N/A
Angel Medical/Firmco | compressors | BJ Prendergast | [PHONE] | IPVHT00203 | 2013-01-28 00:00:00 | OH compressor upgrade | Percuss | CSO 130128-14 | N/A | completed | R11446- Change to HC | 2013-03-18 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP11012 | 2013-01-30 00:00:00 | PM | RKR | N/A | ISO 130130-14 | completed |  | 2013-01-31 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10857 | 2013-01-30 00:00:00 | OH | RKR | N/A | ISO 130130-15 | completed | OH Kit 1/29/2013 | 2013-02-14 00:00:00 | yes | N/A
Dallas Life Support | PO#5197 | Pam/Bryan | [PHONE] | [NAME] | 2013-01-30 00:00:00 | 30K PM | RKR | CSO 130130-15 | N/A | completed |  | 2013-06-12 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5183 | Pam/Bryan | [PHONE] | IMP11107 | 2013-01-31 00:00:00 | compressor gasket overhaul | Percuss | CSO 130131-16 | N/A | completed | R11454 | 2013-04-24 00:00:00 | yes | [EMAIL]
The Care Group | PO#22615 | [NAME] | [PHONE] | [NAME] | 2013-02-01 00:00:00 | not due for PM/VPOW | EP | CSO 130201-17 | N/A | completed |  | 2013-02-20 00:00:00 | yes | [EMAIL]
The Care Group | PO#22615 | [NAME] | [PHONE] | B04618 | 2013-02-01 00:00:00 | 30K PM | EP | CSO 130201-18 | N/A | completed |  | 2013-02-27 00:00:00 | yes | [EMAIL]
The Care Group | PO#22615 | [NAME] | [PHONE] | B04881 | 2013-02-01 00:00:00 | 30kPM/alarm | EP | CSO 130201-19 | N/A | completed |  | 2013-02-27 00:00:00 | yes | [EMAIL]
The Care Group | warranty | [NAME] | [PHONE] | IPVHT00213 | 2013-02-01 00:00:00 | PM/compressor | Percuss | CSO 130201-20 | N/A | completed | R11459 | 2013-03-21 00:00:00 | yes | [EMAIL]
The Care Group | PO#22615 | [NAME] | [PHONE] | F40367 | 2013-02-01 00:00:00 | 10K PM error message | EP | CSO 130201-21 | N/A | completed |  | 2013-03-05 00:00:00 | yes | [EMAIL]
The Care Group | PO#22615 | [NAME] | [PHONE] | F30671 | 2013-02-01 00:00:00 | alarm | EP | CSO 130201-22 | N/A | completed |  | 2013-02-21 00:00:00 | yes | [EMAIL]
Seton/Trimedx | warranty | Rebecca/Dell Childrens | [PHONE] | IPV1C2025 | 2013-02-04 00:00:00 | calibration | RKR | CSO 130204-23 | N/A | completed |  | 2013-02-07 00:00:00 | yes
North Central Baptist | PO#1011554 | [NAME] | [PHONE] | IPV1C1205 | 2013-02-04 00:00:00 | PM | RKR | CSO 130204-24 | N/A | completed |  | 2013-02-13 00:00:00 | yes
Air-O Tech | [NAME] | [NAME] | [PHONE] | IMP10866 | 2013-02-04 00:00:00 | PM | RKR | CSO 130204-25 | N/A | completed |  | 2013-02-13 00:00:00 | yes
Dallas Life Support | customer refused PM | Pam/Bryan | [PHONE] | D01326 | 2013-02-05 00:00:00 | get history | EP | CSO 130205-26 | N/A | completed |  | 2013-02-27 00:00:00 | yes | [EMAIL]
Advance Medical | need PO | [NAME] |  | power cord | 2013-02-05 00:00:00 | Never Serviced | EP | CSO 130205-27 | N/A | completed |  | 2014-01-27 00:00:00 | yes
Dallas Life Support | PO# 5198 | Pam/Bryan | [PHONE] | C05318 | 2013-02-05 00:00:00 | 30K PM | EP | CSO 130205-28 | N/A | completed |  | 2013-06-07 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10835 | 2013-02-08 00:00:00 | OH | RKR | N/A | ISO 130208-16 | completed | OH kit 2/7/2013 | 2013-02-19 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10845 | 2013-02-08 00:00:00 | OH | RKR | N/A | ISO 130208-17 | completed | OH kit 2/7/2013 | 2014-08-25 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10862 | 2013-02-08 00:00:00 | OH | RKR | N/A | ISO 130208-18 | completed | OH kit 2/7/2013 | 2013-03-27 00:00:00 | yes | N/A
IME | IME | IME | IME | IPVHC00384 | 2013-02-08 00:00:00 | Cartridge Replacement | RKR | N/A | ISO 130208-19 | completed | PM DUE: 09/06/2013 | 2013-03-11 00:00:00 | yes | N/A
IME | IME | IME | IME | F37874 | 2013-02-08 00:00:00 | Internal grill broken | EP | N/A | ISO 130208-20 | completed | PM DUE: 07/24/2014 | 2013-02-22 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10865 | 2013-02-13 00:00:00 | OH | RKR | N/A | ISO 130214-21 | completed | OH kit 3/13/2013 | 2013-03-28 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10942 | 2013-02-20 00:00:00 | OH | RKR | N/A | ISO 130220-22 | completed |  | 2014-08-25 00:00:00 | yes | N/A
IME | IME | IME | IME | IPVHC00418 | 2013-02-20 00:00:00 | PM | RKR | N/A | ISO 130220-23 | completed |  | 2013-02-20 00:00:00 | yes | N/A
Medical Plus Supplies | warranty | [NAME] | [PHONE] | IPVHC00565 | 2013-02-11 00:00:00 | "burning smell" tested | RKR | CSO 130211-29 | N/A | completed |  | 2013-02-20 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22618 | [NAME] | [PHONE] | B05022 | 2013-02-11 00:00:00 | 30K PM | EP | CSO 130211-30 | N/A | completed |  | 2013-03-07 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22618 | [NAME] | [PHONE] | F40369 | 2013-02-11 00:00:00 | 10K PM | EP | CSO 130211-31 | N/A | completed |  | 2013-03-08 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22618 **DO NOT PM PER CUSTOMER** | [NAME] | [PHONE] | [NAME] | 2013-02-11 00:00:00 | 30K PM & **ROACHES** | EP | CSO 130211-32 | N/A | INFESTED | **DO NOT PM PER CUSTOMER** | 2013-03-27 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22618 | [NAME] | [PHONE] | B05056 | 2013-02-11 00:00:00 | 30K PM | EP | CSO 130211-33 | N/A | completed |  | 2013-03-27 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | warranty- unit replacement | [NAME] | [PHONE] | IPVHT00107 | 2013-02-11 00:00:00 | compressor/**New Unit** | Percuss | CSO 130225-34 | N/A | completed | New Unit IPVHC00720 | 2013-04-17 00:00:00 | yes | [EMAIL]
[NAME] | warranty | [NAME] | [PHONE] | VDR40252 | 2013-02-20 00:00:00 | Pressure issues | Percuss | CSO 130225-35 | N/A | completed | R11487 | 2013-04-26 00:00:00 | yes
[NAME] | warranty | [NAME] | [PHONE] | WF20466 | 2013-02-20 00:00:00 | screen goes blank | Percuss | CSO 130225-36 | N/A | completed | R11487 | 2013-04-26 00:00:00 | yes
The Care Group | PO# 22621 | [NAME] | [PHONE] | B04489 | 2013-02-21 00:00:00 | 30K PM | EP | CSO 130225-37 | N/A | completed |  | 2013-03-05 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22621 | [NAME] | [PHONE] | F46776 | 2013-02-21 00:00:00 | 10K PM | EP | CSO 130225-38 | N/A | completed |  | 2013-03-20 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22621 | [NAME] | [PHONE] | F50691 | 2013-02-21 00:00:00 | 10K PM | EP | CSO 130225-39 | N/A | completed |  | 2013-03-13 00:00:00 | yes | [EMAIL]
[NAME] | Joseph022513 | [NAME] | 210-696-2626x1165 | C14910 | 2013-02-21 00:00:00 | 30K PM | EP | CSO 130225-40 | N/A | completed |  | 2013-03-22 00:00:00 | yes | [EMAIL]
[NAME] | Joseph022513 | [NAME] | 210-696-2626x1165 | C08982 | 2013-02-21 00:00:00 | 30K PM | EP | CSO 130225-41 | N/A | completed | completed 4/30/2013 | 2013-03-21 00:00:00 | yes | [EMAIL]
[NAME] | PO# 1057832 | Caleb | [PHONE] | IPV1C1536 | 2013-03-15 00:00:00 | OH & case | RKR | CSO 130225-42 | N/A | completed | OH kit 3/20/2013 | 2013-04-09 00:00:00 | yes
IME | IME | IME | IME | IPVHC00217 | 2013-02-25 00:00:00 | OH/Compressor Upgrade | Percuss | N/A | ISO 130225-24 | completed | New Unit IPVHC00692 | 2013-03-21 00:00:00 | yes | N/A
[NAME] | PO# [PHONE] | [NAME] | [PHONE] | IPV1C1565 | 2013-02-21 00:00:00 | PM | RKR | CSO 130225-43 | N/A | completed |  | 2013-02-27 00:00:00 | yes | [EMAIL]
[NAME] | PO# DB022713 | [NAME] | [PHONE] | IMP11049 | 2013-02-26 00:00:00 | PM & Cartridge | RKR | CSO 130227-44 | N/A | completed |  | 2013-03-08 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5239 6/11 | Pam/Bryan | [PHONE] | IPVHC00227 | 2013-02-26 00:00:00 | PM & repair | RKR | CSO 130227-45 | N/A | completed |  | 2013-06-12 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP11115 | 2013-02-28 00:00:00 | PM | RKR | N/A | ISO 130228-25 | completed |  | 2013-02-28 00:00:00 | yes | N/A
IME | IME | IME | IME | IPVHT00119 | 2013-03-05 00:00:00 | compressor/**New Unit** | Percuss | N/A | ISO 130305-26 | completed | New Unit IPVHC00726 | 2013-05-02 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP1960395 | 2013-03-01 00:00:00 | OH/Upgrade      Ebay Purchase | Percuss | N/A | ISO 130306-27 | completed | R11529 | 2013-05-09 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10811 | 2013-03-06 00:00:00 | OH | RKR | N/A | ISO 130308-28 | completed | OH kit 3/6/2013 | 2013-03-25 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10840 | 2013-03-06 00:00:00 | OH | RKR | N/A | ISO 130308-29 | completed | OH kit 3/6/2013 | 2013-03-26 00:00:00 | yes | N/A
Angels of Care | PO# 41713 | [NAME] | [PHONE] Ext 5647 | IM120010054 | 2013-03-04 00:00:00 | compressor gasket overhaul | Percuss | CSO 130304-46 | N/A | completed | R11530 | 2013-04-26 00:00:00 | yes | [EMAIL]
Angels of Care | PO# 31313 | [NAME] | [PHONE] Ext 5647 | IMP1200004 | 2013-03-04 00:00:00 | Annual PM | RKR | CSO 130304-47 | N/A | completed |  | 2013-03-15 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO# 051413KF-PM | [NAME] | [PHONE] | F49204 | 2013-03-08 00:00:00 | 10K PM | EP | CSO 130313-48 | N/A | completed |  | 2013-05-24 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO# 062413KFPM | [NAME] | [PHONE] | F46957 | 2013-03-08 00:00:00 | 10K PM | EP | CSO 130313-49 | N/A | completed |  | 2013-07-17 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO# 062513KFPM | [NAME] | [PHONE] | F49244 | 2013-03-08 00:00:00 | 10K PM | EP | CSO 130314-50 | N/A | completed |  | 2013-07-24 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO# 080113KF-PM | [NAME] | [PHONE] | F00684 | 2013-03-08 00:00:00 | 10K PM | EP | CSO 130314-51 | N/A | completed |  | 2013-09-13 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO# 062613KFPM | [NAME] | [PHONE] | F00690 | 2013-03-08 00:00:00 | 30K PM | EP | CSO 130314-52 | N/A | completed |  | 2013-07-30 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | C14278 | 2013-03-18 00:00:00 | 30K PM | EP | N/A | ISO 130318-30 | completed |  | 2013-03-20 00:00:00 | yes | N/A
S.TX. Medical Supply- SA | Betty | Betty | [PHONE] | sprint pack | 2013-03-18 00:00:00 | Never Serviced | EP | CSO 130318-53 | N/A | completed |  | 2014-01-27 00:00:00 | yes
[NAME] | Tripp050213 | [NAME] | [PHONE] | IMP10989 | 2013-03-15 00:00:00 | PM | RKR | CSO 130320-54 | N/A | completed |  | 2013-05-07 00:00:00 | yes | [EMAIL]
[NAME] | Tripp050213 | [NAME] | [PHONE] | IMP10872 | 2013-03-15 00:00:00 | OH & case | RKR | CSO 130320-55 | N/A | completed | OH kit 5/3/2013 | 2013-05-15 00:00:00 | yes | [EMAIL]
[NAME] | Trip032613 **PM CAN WAIT, SEND BACK** | [NAME] | [PHONE] | F35832 | 2013-03-15 00:00:00 | PM due 8/22/2014 or 13,907 | EP | CSO 130320-56 | N/A | completed |  | 2013-03-26 00:00:00 | yes | [EMAIL]
The Care Group | warranty- unit replacement | [NAME] | [PHONE] | IPVHT00066 | 2013-03-18 00:00:00 | compressor/ **New Unit** | Percuss | CSO 130320-66 | N/A | completed | New Unit IPVHC00715 | 2013-04-09 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22625 | [NAME] | [PHONE] | IPVHT00210 | 2013-03-18 00:00:00 | PM | RKR | CSO 130320-67 | N/A | completed |  | 2013-03-22 00:00:00 | yes | [EMAIL]
[NAME] | PO# [NAME] | [NAME] | [PHONE] | IMP11116 | 2013-03-20 00:00:00 | PM | RKR | CSO 130321-68 | N/A | completed |  | 2013-03-29 00:00:00 | yes | [EMAIL]
[NAME] | Tripp050213 | [NAME] | [PHONE] | IMP11085 | 2013-03-20 00:00:00 | Repair- case, gauge | RKR | CSO 130321-69 | N/A | completed | Case 5/3/2013 | 2013-05-15 00:00:00 | yes | [EMAIL]
Angel Medical/Firmco | warranty- unit replacement | BJ Prendergast | [PHONE] | IPVHT00211 | 2013-03-21 00:00:00 | compressor/ **New Unit** | Percuss | CSO 130321-70 | N/A | completed | New Unit IPVHC00711 | 2013-04-09 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F25210 | 2013-03-21 00:00:00 | 10k PM | CF | N/A | ISO 130321-31 | completed |  | 2013-10-14 00:00:00 | yes | N/A
IME | IME | IME | IME | F48828 | 2013-03-21 00:00:00 | 10k PM | EP | N/A | ISO 130321-32 | completed |  | 2013-06-28 00:00:00 | yes | N/A
Apple Homecare | PO# 8206 | [NAME] | [PHONE] | IPVHC00502 | 2013-03-26 00:00:00 | PM & not holding pressure | RKR | CSO 130325-71 | N/A | completed |  | 2013-04-03 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22626 | [NAME] | [PHONE] | F40377 | 2013-03-18 00:00:00 | 10K PM | EP | CSO 130325-72 | N/A | completed |  | 2013-04-22 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22626 | [NAME] | [PHONE] | F49084 | 2013-03-18 00:00:00 | 10K PM | EP | CSO 130325-73 | N/A | completed |  | 2013-04-25 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | C11632 | 2013-03-27 00:00:00 | 30k PM | EP | N/A | ISO 130327-33 | completed |  | 2013-04-02 00:00:00 | yes | N/A
IME | IME | IME | IME | C11663 | 2013-03-27 00:00:00 | 30k PM | EP | N/A | ISO 130327-34 | completed |  | 2013-04-02 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11007 | 2013-03-27 00:00:00 | OH | RKR | N/A | ISO 130327-35 | completed | [NAME] | 2013-04-10 00:00:00 | yes | N/A
IME | IME | IME | IME | IPV2C0146 | 2013-03-22 00:00:00 | OH kit received 4/24/2013 | RKR | N/A | ISO 130327-36 | completed | [NAME] | 2013-08-09 00:00:00 | yes | N/A
Medical Plus Supplies | warranty- unit replacement | [NAME] | [PHONE] | IPVHT00104 | 2013-03-22 00:00:00 | compressor/ **New Unit** | Percuss | CSO 130328-74 | N/A | completed | New Unit IPVHC00718 | 2013-04-09 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22628 | [NAME] | [PHONE] | B04919 | 2013-03-27 00:00:00 | PM not due- VPOW | EP | CSO 130328-75 | N/A | completed |  | 2013-04-17 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22628 | [NAME] | [PHONE] | B04585 | 2013-03-27 00:00:00 | 30K PM | EP | CSO 130328-76 | N/A | completed |  | 2013-08-28 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22628 | [NAME] | [PHONE] | B04920 | 2013-03-27 00:00:00 | 30K PM | EP | CSO 130328-77 | N/A | completed |  | 2013-08-02 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22628 | [NAME] | [PHONE] | B27368 | 2013-03-27 00:00:00 | 10K PM | EP | CSO 130328-78 | N/A | completed |  | 2013-06-07 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22628 | [NAME] | [PHONE] | B05120 | 2013-03-27 00:00:00 | 30K PM | EP | CSO 130328-79 | N/A | completed |  | 2013-06-07 00:00:00 | yes | [EMAIL]
US Med Equip | PO# RP77729LA ordered kit 4/23 | Tosha | [PHONE] | IPVC20020175 | 2013-03-29 00:00:00 | OH broken case, missing bracket | RKR | CSO 130328-80 | N/A | completed | OH kit 4/23/2013 | 2013-05-08 00:00:00 | yes | [EMAIL]
[NAME] | [NAME] | [NAME] | 210-696-2626x1165 | C08982 | 2013-03-28 00:00:00 | Just PM'd- membrane & overlay | EP | CSO 130402-81 | N/A | completed | completed 4/24 | 2013-05-03 00:00:00 | yes | [EMAIL]
[NAME] | [NAME] | [NAME] | 210-696-2626x1165 | C08793 | 2013-03-28 00:00:00 | 30K PM | EP | CSO 130402-82 | N/A | completed | completed 4/23 | 2013-05-03 00:00:00 | yes | [EMAIL]
[NAME] | [NAME] | [NAME] | 210-696-2626x1165 | C15503 | 2013-03-28 00:00:00 | 30K PM | EP | CSO 130402-83 | N/A | completed | completed 4/25 | 2013-05-03 00:00:00 | yes | [EMAIL]
[NAME] | [NAME] | [NAME] | 210-696-2626x1165 | C14910 | 2013-04-02 00:00:00 | Just PM'd- passed testing | EP | CSO 130403-84 | N/A | completed | completed 4/3 | 2013-04-03 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10832 | 2013-03-29 00:00:00 | OH | RKR | N/A | ISO 130403-37 | completed | OH kit 3/29/2013 | 2013-04-15 00:00:00 | yes | N/A
IME | IME | IME | IME | C14278 | 2013-03-29 00:00:00 | power board defective | EP | N/A | ISO 130403-38 | completed |  | 2013-04-17 00:00:00 | yes | N/A
IME | IME | IME | IME | F37873 | 2013-04-03 00:00:00 | Bad pigtail & flowvalve | EP | N/A | ISO 130403-39 | completed |  | 2015-02-13 00:00:00 | yes | N/A
IME | IME | IME | IME | C16055 | 2013-04-03 00:00:00 | 30k PM | EP | N/A | ISO 130403-40 | completed |  | 2013-04-16 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11101 | 2013-04-03 00:00:00 | PM | RKR | N/A | ISO 130403-41 | completed |  | 2013-04-09 00:00:00 | yes | N/A
Key Med | PO# 130C3171 | [NAME] | [PHONE] | IMP10806 | 2013-04-03 00:00:00 | Annual PM | RKR | CSO 130405-85 | N/A | completed |  | 2013-04-12 00:00:00 | yes
IME | IME | IME | IME | IMP10828 | 2013-04-08 00:00:00 | OH | RKR | N/A | ISO 130408-42 | completed | [NAME] 3/29/2013 | 2013-04-12 00:00:00 | yes | N/A
IME | IME | IME | IME | [NAME] | 2013-04-08 00:00:00 | OH | RKR | N/A | ISO 130408-43 | completed | [NAME] | 2013-04-24 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10893 | 2013-04-15 00:00:00 | OH | RKR | N/A | ISO 130417-44 | completed | OH kit 4/22/2013 | 2013-06-06 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11025 | 2013-04-15 00:00:00 | PM | RKR | N/A | ISO 130417-45 | completed |  | 2013-04-19 00:00:00 | yes | N/A
[NAME] | PO#[PHONE] | [NAME] | [PHONE] | VDR420030006 | 2013-04-11 00:00:00 | Repair | RKR | CSO 130422-86 | N/A | completed |  | 2013-04-18 00:00:00 | yes | [EMAIL]
[NAME] | PO#[PHONE] | [NAME] | [PHONE] | VDR4990006 | 2013-04-11 00:00:00 | Repair | RKR | CSO 130422-87 | N/A | completed |  | 2013-04-18 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | F46799 | 2013-04-11 00:00:00 | 10K PM | EP | CSO 130422-88 | N/A | completed |  | 2013-05-02 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | F49081 | 2013-04-11 00:00:00 | 10K PM | EP | CSO 130422-89 | N/A | completed |  | 2013-05-14 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | F64147 | 2013-04-11 00:00:00 | VPOW | EP | CSO 130422-90 | N/A | completed |  | 2013-04-23 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | F46982 | 2013-04-11 00:00:00 | 10K PM/repair | EP | CSO 130422-91 | N/A | completed |  | 2013-05-08 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22643 | [NAME] | [PHONE] | B27373 | 2013-04-11 00:00:00 | address charge issue only | EP | CSO 130422-92 | N/A | completed |  | 2013-05-29 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | B27361 | 2013-04-11 00:00:00 | 10K PM | EP | CSO 130422-93 | N/A | completed |  | 2013-05-22 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | B27359 | 2013-04-11 00:00:00 | 10K PM | EP | CSO 130422-94 | N/A | completed |  | 2013-05-28 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | B04887 | 2013-04-11 00:00:00 | 30K PM | EP | CSO 130422-95 | N/A | completed |  | 2013-08-23 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | B04892 | 2013-04-11 00:00:00 | 30K PM | EP | CSO 130422-96 | N/A | completed |  | 2013-08-19 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | B04631 | 2013-04-11 00:00:00 | 10K PM | EP | CSO 130422-97 | N/A | completed |  | 2013-06-19 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | F34418 | 2013-04-11 00:00:00 | 10K PM | EP | CSO 130422-98 | N/A | completed |  | 2013-04-29 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | B04905 | 2013-04-11 00:00:00 | 30K PM | EP | CSO 130422-99 | N/A | completed |  | 2013-08-21 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22630 | [NAME] | [PHONE] | IPVHC00221 | 2013-04-11 00:00:00 | PM & repair | RKR | CSO 130422-100 | N/A | completed |  | 2013-05-02 00:00:00 | yes | [EMAIL]
GE | PO# 401157204 | [NAME] | [PHONE] | IPVC0999 | 2013-04-11 00:00:00 | Annual PM | RR | CSO 130422-101 | N/A | completed |  | 2013-04-25 00:00:00 | yes | [EMAIL]
[NAME] | [NAME] | [NAME] | 210-696-2626x1165 | C08670 | 2013-04-12 00:00:00 | requested history | EP | CSO 130423-102 | N/A | completed |  | 2013-05-08 00:00:00 | yes | [EMAIL]
Children's Medical Ctr | PO# 1501083 | [NAME] | [PHONE] | IPV2C0292 | 2013-04-19 00:00:00 | Annual PM | RR | CSO 130423-103 | N/A | completed |  | 2013-05-09 00:00:00 | yes | [EMAIL]
Children's Medical Ctr | PO# 1501083 | [NAME] | [PHONE] | IPV1C0505 | 2013-04-19 00:00:00 | Annual PM & repair | RR | CSO 130423-104 | N/A | completed |  | 2013-05-09 00:00:00 | yes | [EMAIL]
TX Children's Hospital | PO# 13-599479 | Curley Green | [PHONE] | IPV1C0206 | 2013-04-19 00:00:00 | [NAME] | RR | CSO 130423-105 | N/A | completed | [NAME] 5/9/2013 | 2013-05-24 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F48833 | 2013-04-23 00:00:00 | 10k PM | CF | N/A | ISO 130423-46 | completed | RMA218555 | 2013-11-07 00:00:00 | yes | N/A
[NAME] | PO# [PHONE] | [NAME] | [PHONE] | VDR20030007 | 2013-04-12 00:00:00 | Calibration | RR | CSO 130423-106 | N/A | completed |  | 2013-04-25 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | PO# 30349 | [NAME] | [PHONE] | IPVHC00367 | 2013-04-12 00:00:00 | PM & power switch | RR | CSO 130423-107 | N/A | completed |  | 2013-04-29 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | compressor | [NAME] | [PHONE] | IPVHT00102 | 2013-04-12 00:00:00 | compressor/ **New Unit** | Percuss | CSO 130423-108 | N/A | completed | New Unit IPVHC00727 | 2013-05-06 00:00:00 | yes | [EMAIL]
Peak Medical/Med One | warranty repair | [NAME] | [PHONE] | IPV1C2017 | 2013-04-22 00:00:00 | Warranty | Percuss | CSO 130423-109 | N/A | completed | R11562 | 2013-05-17 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F48835 | 2013-04-23 00:00:00 | 10K PM | CF | N/A | ISO 130423-47 | completed | RMA218557 | 2013-11-07 00:00:00 | yes | N/A
Alliance Medical | RM04242013 | [NAME] | [PHONE] | F46539 | 2013-04-10 00:00:00 | 10K PM | EP | CSO 130424-110 | N/A | completed |  | 2013-05-10 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IPVHC00464 | 2013-04-24 00:00:00 | Annual PM | RKR | N/A | [NAME] | completed |  | 2013-05-06 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11042 | 2013-04-24 00:00:00 | Power Switch Plug Repalcement | RKR | N/A | [NAME] | completed |  | 2013-05-03 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11068 | 2013-04-30 00:00:00 | Annual PM | RKR | N/A | ISO 130430-50 | completed |  | 2013-05-03 00:00:00 | yes | N/A
IME | IME | IME | IME | F48826 | 2013-05-01 00:00:00 | 10k PM | CF | N/A | ISO 130501-51 | completed |  | 2013-10-14 00:00:00 | yes | N/A
GE | PO# 401157205 | [NAME] | [PHONE] | IPVC0897 | 2013-04-30 00:00:00 | Annual PM | RR | CSO 130430-111 | N/A | completed |  | 2013-05-01 00:00:00 | yes | [EMAIL]
GE | PO# 401157203 | [NAME] | [PHONE] | IPVC20020146 | 2013-04-30 00:00:00 | Annual PM | RR | CSO 130430-112 | N/A | completed |  | 2013-05-02 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | compressor, emailed 5/2 | [NAME] | [PHONE] | IPVHT00086 | 2013-04-25 00:00:00 | compressor/new unit | Percuss | CSO 130502-113 | N/A | completed | New Unit IPVHC00733 | 2013-05-14 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5223 | Pam/Bryan | [PHONE] | C17070 | 2013-04-26 00:00:00 | 30K | EP | CSO 130502-114 | N/A | completed |  | 2013-06-26 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22634 | [NAME] | [PHONE] | B04487 | 2013-04-29 00:00:00 | 30K PM | EP | CSO 130502-115 | N/A | completed |  | 2013-08-26 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22634 | [NAME] | [PHONE] | F50037 | 2013-04-29 00:00:00 | 10K PM | EP | CSO 130502-116 | N/A | completed |  | 2013-05-16 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22634 | [NAME] | [PHONE] | F39923 | 2013-04-29 00:00:00 | 10K PM | EP | CSO 130502-117 | N/A | completed |  | 2013-05-16 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22634 | [NAME] | [PHONE] | B27363 | 2013-04-29 00:00:00 | 30K | EP | CSO 130502-118 | N/A | completed |  | 2013-05-28 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22634 | [NAME] | [PHONE] | B05041 | 2013-04-29 00:00:00 | 30K PM | EP | CSO 130502-119 | N/A | completed |  | 2013-05-22 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | PO#31020 | [NAME] | [PHONE] | IPVHC00328 | 2013-04-30 00:00:00 | PM | RR | CSO 130502-120 | N/A | completed |  | 2013-05-07 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | PO#31020 | [NAME] | [PHONE] | IPVHC00354 | 2013-04-30 00:00:00 | PM | RR | CSO 130502-121 | N/A | completed |  | 2013-05-07 00:00:00 | yes | [EMAIL]
Key Med | PO# 13OC3246 | [NAME] | [PHONE] | IMP10806 | 2013-05-01 00:00:00 | repair | RR | CSO 130507-122 | N/A | completed |  | 2013-05-07 00:00:00 | yes
US Med Equip | PO# R79017MW | Melody | [PHONE] | IPV1C0132 | 2013-05-08 00:00:00 | OH ordered kit 5/9 | RKR | CSO 130507-123 | N/A | completed | OH kit 5/9/2013 | 2013-05-20 00:00:00 | yes | [EMAIL]
Univ of Texas Med Branch | PO# 10908963 s/w [NAME] | [NAME] | [PHONE] | IPVC0982 | 2013-05-15 00:00:00 | PM | RKR | CSO 130508-124 | N/A | completed |  | 2013-06-03 00:00:00 | yes | [EMAIL]
Univ of Texas Med Branch | PO# 10908968 s/w [NAME] | [NAME] | [PHONE] | IPVC0973 | 2013-05-15 00:00:00 | PM | RKR | CSO 130508-125 | N/A | completed |  | 2013-06-03 00:00:00 | yes | [EMAIL]
Covenant Medical Center | PO# 401159807 | [NAME] | [PHONE] | IPV1C1491 | 2013-05-10 00:00:00 | OH ordered kit 5/15 | RKR | CSO 130510-126 | N/A | completed | OH kit 5/15/2013 | 2013-05-30 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10998 | 2013-05-10 00:00:00 | PM | RKR | N/A | ISO 130510-52 | completed |  | 2013-06-03 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10906 | 2013-05-10 00:00:00 | OH | RKR | N/A | ISO 130510-53 | completed | OH kit 5/10/2013 | 2013-06-07 00:00:00 | yes | N/A
IME | IME | IME | IME | F48729 | 2013-05-10 00:00:00 | 10k PM | CF | N/A | ISO 130510-54 | completed | RMA218554 | 2013-11-07 00:00:00 | yes | N/A
IME | IME | IME | IME | F48718 | 2013-05-10 00:00:00 | 10k PM | CF | N/A | ISO 130510-55 | completed |  | 2013-10-25 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11103 | 2013-05-10 00:00:00 | PM | RKR | N/A | ISO 130510-56 | completed |  | 2013-05-20 00:00:00 | yes | N/A
IME | IME | IME | IME | IPVHC00564 | 2013-05-14 00:00:00 | PM | RKR | N/A | ISO 130514-57 | completed |  | 2013-05-17 00:00:00 | yes | N/A
[NAME] | [NAME] | [NAME] | 210-696-2626x1165 | [NAME] | 2013-05-10 00:00:00 | 30k PM | EP | CSO 130514-127 | N/A | completed |  | 2013-06-26 00:00:00 | yes | [EMAIL]
[NAME] | [NAME] | [NAME] | 210-696-2626x1165 | B01384 | 2013-05-10 00:00:00 | 10k PM | EP | CSO 130514-128 | N/A | completed |  | 2013-06-26 00:00:00 | yes | [EMAIL]
[NAME] | PO# [PHONE] | [NAME] | [PHONE] | WF20101 | 2013-05-13 00:00:00 | Recalibration/False Readings | Percuss | CSO 130514-129 | N/A | completed | R11592 | 2013-06-19 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F48756 | 2013-05-16 00:00:00 | 10k PM | EP | N/A | ISO 130516-58 | completed |  | 2014-08-26 00:00:00 | yes | N/A
US Med Equip | PO# R79295MW | Melody | [PHONE] | IPV1C0751 | 2013-05-21 00:00:00 | OH/not in range | RKR | CSO 130520-130 | N/A | completed | OH kit 05/29/2013 | 2013-06-12 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F50223 | 2013-05-21 00:00:00 | 10k PM | EP | N/A | ISO 130521-59 | completed |  | 2014-11-10 00:00:00 | yes | N/A
The Care Group | PO# 22643 | [NAME] | [PHONE] | F39926 | 2013-05-17 00:00:00 | 10k PM | EP | CSO 130521-131 | N/A | completed |  | 2013-05-30 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP11005 | 2013-05-20 00:00:00 | PM | RKR | N/A | ISO 130521-60 | completed |  | 2013-06-04 00:00:00 | yes | N/A
Dallas Life Support | PO# 5259 | Pam/Bryan | [PHONE] | C17239 | 2013-05-22 00:00:00 | 30k PM | EP | CSO 130522-132 | N/A | completed |  | 2013-07-05 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO# 090313KF-PM | [NAME] | [PHONE] | F00699 | 2013-05-22 00:00:00 | 30K PM | EP | CSO 130522-133 | N/A | completed |  | 2013-12-11 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO# 090213KF-PM | [NAME] | [PHONE] | F49184 | 2013-05-22 00:00:00 | 10k PM | EP | CSO 130522-134 | N/A | completed |  | 2013-09-30 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10904 | 2013-05-23 00:00:00 | OH | RKR | N/A | ISO 130523-61 | completed | OH kit 06/06/2013 | 2013-06-20 00:00:00 | yes | N/A
[NAME] | PO# [NAME] | [NAME] | 210-696-2626x1165 | C15991 | 2013-05-24 00:00:00 | VPOW | EP | CSO 130524-135 | N/A | completed |  | 2013-06-07 00:00:00 | yes | [EMAIL]
Peak Medical/Med One | emailed [NAME] 5/29 | [NAME] | [PHONE] | IPV1C2026 | 2013-05-28 00:00:00 | Warranty  Purchase 11/21/2012 | Percuss | CSO 130528-136 | N/A | completed | R11607 | 2013-07-05 00:00:00 | yes | [EMAIL]
[NAME] | warranty repair | [NAME] | [PHONE] | IMP11206 | 2013-05-28 00:00:00 | Warranty  Purchase 9/7/2012 | RKR | CSO 130528-137 | N/A | completed |  | 2013-05-29 00:00:00 | yes | [EMAIL]
[NAME] | PO#Tripp053013 | [NAME] | [PHONE] | IMP10873 | 2013-05-28 00:00:00 | OH | RKR | CSO 130528-138 | N/A | completed | OH kit 05/30/2013 | 2013-06-19 00:00:00 | yes | [EMAIL]
US Med Equip | PO# RP80084AH | [NAME] | [PHONE] | IPVC20020034 | 2013-06-07 00:00:00 | OH | RKR | CSO 130607-139 | N/A | completed | OH kit 06/13/2013 | 2013-06-28 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F48832 | 2013-06-07 00:00:00 | 10k PM | CF | N/A | ISO 130607-62 | completed | RMA218556 | 2013-11-07 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10814 | 2013-06-07 00:00:00 | OH | RKR | N/A | ISO 130607-63 | completed |  | 2013-06-11 00:00:00 | yes | N/A
Univ of Texas Med Branch | PO# I0916184 | [NAME] | [PHONE] | IPVC0993 | 2013-06-07 00:00:00 | OH | RKR | CSO 130610-140 | N/A | completed | OH kit 06/12/2013 | 2013-06-24 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO# 090413KF-PM | [NAME] | [PHONE] | F00689 | 2013-06-10 00:00:00 | 30K PM | EP | CSO 130610-141 | N/A | completed |  | 2013-12-06 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO# 090513KF-PM | [NAME] | [PHONE] | F49241 | 2013-06-10 00:00:00 | 10k PM | EP | CSO 130610-142 | N/A | completed |  | 2013-10-09 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP11133 | 2013-06-10 00:00:00 | PM | RKR | N/A | ISO 130612-64 | completed |  | 2013-06-12 00:00:00 | yes | N/A
TX Children's Hospital | PO# 13-609195/ emailed about case 6/24 | Curley Green | [PHONE] | IPV1C0595 | 2013-06-13 00:00:00 | OH & case replacement | RKR | CSO 130613-143 | N/A | completed | OH kit 06/13/2013 | 2013-07-01 00:00:00 | yes | [EMAIL]
US Med Equip | PO# 80515AH Given by [NAME] | Melody | [PHONE] | IPV1C0299 | 2013-06-19 00:00:00 | OH | RKR | CSO 130617-144 | N/A | completed | OH kit 06/20/2013 | 2013-07-15 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F55642 | 2013-06-20 00:00:00 | 10k PM Due 03/2014 or 10,000 | EP | N/A | ISO 130620-65 | completed | Issue w/ unit | 2013-06-28 00:00:00 | yes | N/A
IME | IME | IME | IME | C11652 | 2013-06-20 00:00:00 | 10k PM Due 09/21/2013 or 35,021 | EP | N/A | ISO 130620-66 | completed |  | 2013-09-23 00:00:00 | yes | N/A
Children's Medical Ctr | PO# 1509512 7/23 | [NAME] | [PHONE] | IPV2C0258 | 2013-06-20 00:00:00 | PM | RKR | CSO 130620-145 | N/A | completed |  | 2013-07-24 00:00:00 | yes | [EMAIL]
Angels of Care | Return w/o service-OH by Bemes 2/12/13- Warranty | [NAME] | [PHONE] | IMP20000142 | 2013-06-21 00:00:00 | PM 02-2014 Recently OH'd FEB 2013 | RKR | CSO 130621-146 | N/A | completed | NOT SERVICED | 2013-07-23 00:00:00 | yes | N/A
Angels of Care | PO# Jessica080213 | [NAME] | [PHONE] | IM120000147 | 2013-06-21 00:00:00 | OH Pressure Gauge & Retain Ring | RKR | CSO 130621-147 | N/A | completed | OH kit 08/02/2013 | 2013-08-14 00:00:00 | yes | N/A
Angels of Care | PO# Jessica080213 | [NAME] | [PHONE] | IPV2C0164 | 2013-06-21 00:00:00 | OH | RKR | CSO 130621-148 | N/A | completed | OH kit 08/02/2013 | 2013-08-14 00:00:00 | yes | N/A
Alliance Medical | PO# Ray0624 | Ray | [PHONE] | C42813 | 2013-06-24 00:00:00 | 10k PM | EP | CSO 130624-149 | N/A | completed |  | 2013-06-27 00:00:00 | yes | [EMAIL]
Apple Homecare | PO# 8862 | [NAME] | [PHONE] | IPVHC00364 | 2013-06-24 00:00:00 | PM & replace harness | RKR | CSO 130624-150 | N/A | completed |  | 2013-07-01 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5260 | Pam/Bryan | [PHONE] | C17012 | 2013-06-24 00:00:00 | 10k PM | EP | CSO 130624-151 | N/A | completed |  | 2013-07-23 00:00:00 | yes | [EMAIL]
US Med Equip | Repair cancelled per [NAME] device placed on patient | [NAME] | [PHONE] | IPVC20020175 | 2013-07-30 00:00:00 | Warranty manometer guage | RKR | CSO 130625-152 | N/A | completed |  | 2013-07-30 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F55169 | 2013-06-25 00:00:00 | 10k PM Due 03/2014 or 10,000 | EP | N/A | ISO 130625-67 | completed |  | 2014-11-03 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11213 | 2013-06-25 00:00:00 | Annual PM Due 09/20/2013 | RKR | N/A | ISO 130625-68 | completed |  | 2013-07-05 00:00:00 | yes | N/A
IME | IME | IME | IME | C06629 | 2013-06-26 00:00:00 | 10k PM | EP | N/A | ISO 130626-69 | completed |  | 2013-08-08 00:00:00 | yes | N/A
IME | IME | IME | IME | F37891 | 2013-06-26 00:00:00 | 10k PM | CF | N/A | ISO 130626-70 | completed |  | 2013-10-28 00:00:00 | yes | N/A
Medical Plus Supplies | warranty- shipped 7/5 to [NAME] | [NAME] | [PHONE] | IPVHC00120 | 2013-06-27 00:00:00 | Warranty | Percuss | CSO 130627-153 | N/A | completed | R11635 | 2013-08-02 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10982 | 2013-06-27 00:00:00 | Annual PM Due 08/07/2013 | RKR | N/A | ISO 130327-71 | completed |  | 2013-07-03 00:00:00 | yes | N/A
[NAME] | [NAME] | [NAME] | 210-696-2626x1213 | C15543 | 2013-06-27 00:00:00 | 30k PM | EP | CSO 130627-154 | N/A | completed |  | 2013-07-30 00:00:00 | yes | [EMAIL]
[NAME] | [NAME] | [NAME] | 210-696-2626x1213 | C15966 | 2013-06-27 00:00:00 | 30k PM | EP | CSO 130627-155 | N/A | completed |  | 2013-11-07 00:00:00 | yes | [EMAIL]
[NAME] | [NAME] | [NAME] | 210-696-2626x1213 | C15513 | 2013-06-27 00:00:00 | 30k PM | EP | CSO 130627-156 | N/A | completed |  | 2013-08-22 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP11073 | 2013-06-27 00:00:00 | PM | RKR | N/A | ISO 130627-72 | completed |  | 2013-07-03 00:00:00 | yes | N/A
[NAME] Hospital | PO# WC26242 7/23 | [NAME] | (281) 293-7774 ext. 6626 | IPVC20020164 | 2013-07-02 00:00:00 | OH | RKR | CSO 130628-157 | N/A | completed | OH kit 07/03/2013 | 2013-07-23 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22651 | [NAME] | [PHONE] | F53913 | 2013-07-01 00:00:00 | 10k PM | EP | CSO 130702-158 | N/A | completed |  | 2013-07-05 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22651 | [NAME] | [PHONE] | F53911 | 2013-07-01 00:00:00 | 10k PM | EP | CSO 130702-159 | N/A | completed |  | 2013-07-17 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | /received PO#102213 KFPM | [NAME] | [PHONE] | F00522 | 2013-06-28 00:00:00 | 30k PM | EP | CSO 130702-160 | N/A | completed |  | 2013-11-14 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | /Received PO#102313 KFPM | [NAME] | [PHONE] | F48042 | 2013-06-28 00:00:00 | 10k PM | EP | CSO 130702-161 | N/A | completed |  | 2013-11-13 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | /received PO#102413 KFPM | [NAME] | [PHONE] | F49193 | 2013-06-28 00:00:00 | 10k PM | EP | CSO 130702-162 | N/A | completed |  | 2013-11-13 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#9105 | [NAME] | [PHONE] | F49202 | 2013-06-28 00:00:00 | 10k PM | EP | CSO 130702-163 | N/A | completed |  | 2014-04-25 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | NOT SERVICED | [NAME] | [PHONE] | F49205 | 2013-06-28 00:00:00 | NOT Serviced 10kpm/flow valve Cancelled by Customer | EP | CSO 130702-164 | N/A | completed |  | 2015-01-09 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5263 | Pam/Bryan | [PHONE] | C17014 | 2013-07-02 00:00:00 | 30k PM | EP | CSO 130702-165 | N/A | completed |  | 2013-07-29 00:00:00 | yes | [EMAIL]
Dallas Life Support | recently PM'd, could not duplicate problem | Pam/Bryan | [PHONE] | C17070 | 2013-07-02 00:00:00 | Pressure Rise issues | EP | CSO 130702-166 | N/A | completed | 30kPM performed 06/26/2013 | 2013-07-03 00:00:00 | yes | [EMAIL]
Alliance Medical | PO# Ray070313 | Ray | [PHONE] | C42814 | 2013-07-03 00:00:00 | 30k PM | EP | CSO 130703-167 | N/A | completed |  | 2013-07-19 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IPVHC00563 | 2013-07-03 00:00:00 | PM | RKR | N/A | ISO 130703-73 | completed |  | 2013-07-08 00:00:00 | yes | N/A
IME | IME | IME | IME | VDR40187 | 2013-07-08 00:00:00 | PM | RKR | N/A | ISO 130708-74 | completed |  | 2013-07-08 00:00:00 | yes | N/A
IME | IME | IME | IME | IPV2C0186 | 2013-07-08 00:00:00 | PM | RKR | N/A | ISO 130708-75 | completed |  | 2013-07-09 00:00:00 | yes | N/A
IME | IME | IME | IME | IPV1C1758 | 2013-07-08 00:00:00 | PM | RKR | N/A | ISO 130708-76 | completed |  | 2013-07-10 00:00:00 | yes | N/A
IME | IME | IME | IME | WF20148 | 2013-07-10 00:00:00 | Calibration | Percuss | N/A | ISO 130710-77 | completed | R11637 | 2013-08-20 00:00:00 | yes | N/A
Dallas Life Support | UNIT WAS NOT SERVICED AT CUSTOMER REQUEST | Pam/Bryan | [PHONE] | C17057 | 2013-07-09 00:00:00 | 30k PM | EP | CSO 130710-168 | N/A | completed |  | 2014-07-28 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5277 | Pam/Bryan | [PHONE] | C17079 | 2013-07-09 00:00:00 | 30k PM | EP | CSO 130710-169 | N/A | completed |  | 2013-08-07 00:00:00 | yes | [EMAIL]
US Med Equip | PO#81508AH | [NAME] | [PHONE] | IPV1C0302 | 2013-07-11 00:00:00 | [NAME] case | RKR | CSO 130710-170 | N/A | completed | OH kit 07/17/2013 | 2013-08-02 00:00:00 | yes | [EMAIL]
US Med Equip | PO#81508AH | [NAME] | [PHONE] | IPVC20020168 | 2013-07-11 00:00:00 | PM | RKR | CSO 130710-171 | N/A | completed |  | 2013-07-22 00:00:00 | yes | [EMAIL]
Children's Medical Ctr | PO# 1516105 | [NAME] | [PHONE] | [NAME] | 2013-09-06 00:00:00 | OH Due 07/25/2014 | RKR | CSO 130716-172 | N/A | completed | OH kit 09/19/2013 | 2013-10-16 00:00:00 | yes | [EMAIL]
In Depth Respiratory | PO# [NAME] 7/31 | [NAME] | [PHONE] | C08126 | 2013-07-16 00:00:00 | 10k PM | EP | CSO 130718-173 | N/A | completed | 10k Due 7-2013 or 36,720 hours | 2013-08-07 00:00:00 | yes | N/A
The Care Group | PO#22659 | [NAME] | [PHONE] | F46799 | 2013-07-16 00:00:00 | could not duplicate problem | EP | CSO 130718-174 | N/A | completed | Last 10k 05/02/2013 | 2013-07-26 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | VDR4970020 | 2013-07-17 00:00:00 | PM | RKR | N/A | ISO 130718-78 | completed |  | 2013-07-22 00:00:00 | yes | N/A
IME | IME | IME | IME | C11650 | 2013-07-18 00:00:00 | 10k PM | EP | N/A | ISO 130718-79 | completed |  | 2013-08-08 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11041 | 2013-07-18 00:00:00 | PM | RKR | N/A | ISO 130718-80 | completed |  | 2013-07-30 00:00:00 | yes | N/A
Dallas Life Support | PO# 5280 7/23 | Pam/Bryan | [PHONE] | C15331 | 2013-07-18 00:00:00 | 30k PM | EP | CSO 130718-175 | N/A | completed |  | 2014-07-31 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5279 7/23 | Pam/Bryan | [PHONE] | C16994 | 2013-07-18 00:00:00 | 30k PM | EP | CSO 130718-176 | N/A | completed |  | 2013-09-13 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | C09610 | 2013-07-18 00:00:00 | 10k PM | EP | N/A | ISO 130718-81 | completed |  | 2013-08-14 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11031 | 2013-07-18 00:00:00 | PM | RKR | N/A | ISO 130718-82 | completed |  | 2013-07-29 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11028 | 2013-07-18 00:00:00 | PM | RKR | N/A | ISO 130718-83 | completed |  | 2013-07-26 00:00:00 | yes | N/A
IME | IME | IME | IME | C16059 | 2013-07-19 00:00:00 | 30k PM | EP | N/A | ISO 130719-84 | completed |  | 2013-08-14 00:00:00 | yes | N/A
IME | IME | IME | IME | F37892 | 2013-07-19 00:00:00 | 10k PM | EP | N/A | ISO 130719-85 | completed |  | 2014-12-30 00:00:00 | yes | N/A
The Care Group | PO# 22659 | [NAME] | [PHONE] | F53910 | 2013-07-19 00:00:00 | 10k PM | EP | CSO 130722-179 | N/A | completed |  | 2013-08-02 00:00:00 | yes | [EMAIL]
Peak Medical/Med One | Warranty Purchase: 11/14/2012 R11679 | [NAME] | [PHONE] | IPV1C2015 | 2013-07-24 00:00:00 | not nebulizing- could not duplicate | Percuss | CSO 130722-180 | N/A | completed | shipped 8/5- Percuss found nothing wrong 8/16 | 2013-08-23 00:00:00 | yes | [EMAIL]
[NAME] | PO# [PHONE] | [NAME] | [PHONE] | IPV1C1565 | 2013-07-25 00:00:00 | Gauge Not Working OH | RKR | CSO 130724-181 | N/A | completed | OH kit 07/30/2013 | 2013-08-14 00:00:00 | yes | [EMAIL]
[NAME] | PO# Trip080713 | [NAME] | [PHONE] | F49628- L | 2013-08-02 00:00:00 | 10k PM | EP | CSO 130725-182 | N/A | completed |  | 2013-08-30 00:00:00 | yes | [EMAIL]
[NAME] | PO# Trip080713 | [NAME] | [PHONE] | C39187- L | 2013-08-02 00:00:00 | 10K PM LIMV Alarm & Disc/Sense Alarm | EP | CSO 130725-183 | N/A | completed | 10k Due 10/12/2013 or 20,136 hrs | 2013-08-30 00:00:00 | yes | [EMAIL]
[NAME] | PO# Trip080713- emailed on 30K 9/4 | [NAME] | [PHONE] | C40998- L | 2013-08-02 00:00:00 | 10k PM- roaches 30K INFESTATION! | EP | CSO 130725-184 | N/A | completed | CUSTOMER STATES DO NOT SERVICE SHIPPED BACK | 2014-02-24 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#121013KFPM | [NAME] | [PHONE] | F49239 | 2013-07-23 00:00:00 | 10k PM | EP | CSO 130725-185 | N/A | completed |  | 2013-12-17 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#121813 KFPM | [NAME] | [PHONE] | F49198 | 2013-07-23 00:00:00 | 10k PM & low alarm | EP | CSO 130725-186 | N/A | completed |  | 2014-01-20 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#042814KFPM9144 | [NAME] | [PHONE] | F00541 | 2013-07-23 00:00:00 | 30K due 4/27/14 | EP | CSO 130725-187 | N/A | completed |  | 2014-05-08 00:00:00 | yes | [EMAIL]
Angel Medical/Firmco | old compressor/SO#20131100 return to [NAME] | BJ Prendergast | [PHONE] | IPVHT00115 | 2013-07-24 00:00:00 | NEW COMPRESSOR IPVHC00809 | Percuss | CSO 130725-188 | N/A | completed | New Unit IPVHC00809 | 2013-08-09 00:00:00 | yes | [EMAIL]
Angel Medical/Firmco | old compressor/SO# 20131102 return to [NAME] | BJ Prendergast | [PHONE] | IPVHC00135 | 2013-07-24 00:00:00 | NEW COMPRESSOR IPVHC00808 | Percuss | CSO 130725-189 | N/A | completed | New Unit IPVHC00808 | 2013-08-09 00:00:00 | yes | [EMAIL]
Angel Medical/Firmco | PO# BJ Prendergast 7/31 | BJ Prendergast | [PHONE] | IPVHC00416 | 2013-07-24 00:00:00 | PM | RKR | CSO 130725-190 | N/A | completed |  | 2013-08-02 00:00:00 | yes | [EMAIL]
Angel Medical/Firmco | PO# BJ Prendergast 7/31 | BJ Prendergast | [PHONE] | [NAME] | 2013-07-24 00:00:00 | PM | RKR | CSO 130725-191 | N/A | completed |  | 2013-08-02 00:00:00 | yes | [EMAIL]
US Med Equip | PO# 82228AH | [NAME] | [PHONE] | IPV1C0805 | 2013-07-30 00:00:00 | OH | RKR | CSO 130729-192 | N/A | completed | OH kit 08/02/2013 | 2013-08-21 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | [NAME] | [NAME] | [PHONE] | F00530 | 2013-07-29 00:00:00 | 30k PM | EP | CSO 130730-193 | N/A | completed |  | 2014-06-26 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#20913KFPM | [NAME] | [PHONE] | F49237 | 2013-07-29 00:00:00 | 10k PM | EP | CSO 130730-194 | N/A | completed |  | 2014-01-07 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | PO# 35570 | [NAME] | [PHONE] | IPVHC00093 | 2013-07-30 00:00:00 | [NAME] selection switch loose | RKR | CSO 130731-195 | N/A | completed |  | 2013-08-09 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | old compressor/SO# 20131155 return to [NAME] | [NAME] | [PHONE] | IPVHT00100 | 2013-07-30 00:00:00 | NEW COMPRESSOR IPVHC00815 | Percuss | CSO 130731-196 | N/A | completed | New Unit IPVHC00815 | 2013-08-14 00:00:00 | yes | [EMAIL]
[NAME] | PO# [NAME] | [NAME] | [PHONE] | C39186-L | 2013-08-07 00:00:00 | 10k PM | EP | CSO 130731-197 | N/A | completed |  | 2013-09-06 00:00:00 | yes | [EMAIL]
[NAME] | PO# [NAME] | [NAME] | [PHONE] | C47051-L | 2013-08-07 00:00:00 | 10k PM | EP | CSO 130731-198 | N/A | completed |  | 2013-09-06 00:00:00 | yes | [EMAIL]
Dallas Life Support | UNIT WAS NOT SERVICED AT CUSTOMER REQUEST | Pam/Bryan | [PHONE] | C17078-L | 2013-08-02 00:00:00 | 30k PM | EP | CSO 130802-199 | N/A | completed |  | 2014-07-28 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5284/PO#5285 7/25/2014 | Pam/Bryan | [PHONE] | C17084-L | 2013-08-02 00:00:00 | 30k PM | EP | CSO 130802-200 | N/A | IN REPAIR |  | 2015-08-13 00:00:00 | yes | [EMAIL]
[NAME] | Warranty MFG: 2013-02- Returned to customer to send to Care Fusion | [NAME] | [PHONE] | F65903 | 2013-08-07 00:00:00 | Turbine going out- under warranty | EP | CSO 130805-201 | N/A | completed | WARRANTY MFG: 2013-02 | 2013-08-15 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP11034 | 2013-08-06 00:00:00 | PM | RKR | N/A | ISO 130806-86 | completed |  | 2013-08-22 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11004 | 2013-08-06 00:00:00 | OH | RKR | N/A | ISO 130806-87 | completed | OH kit 08/15/2013 | 2013-08-30 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11230 | 2013-08-07 00:00:00 | PM | RKR | N/A | ISO 130807-88 | completed |  | 2013-08-22 00:00:00 | yes | N/A
IME | IME | IME | IME | C11600 | 2013-08-07 00:00:00 | 30kPM-INFESTATION! | EP | N/A | ISO 130807-89 | STAND BY |  |  |  | N/A
American Medical Equipment | Pam Fish | Pam Fish | [PHONE] | C12778 | 2013-08-26 00:00:00 | 30k PM | EP | CSO 130812-202 | N/A | completed |  | 2013-09-11 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5289/PO#5287 7/25/2014 | Pam/Bryan | [PHONE] | C17246-L | 2013-08-07 00:00:00 | 30k PM | EP | CSO 130812-203 | N/A | completed |  | 2014-08-01 00:00:00 | yes | [EMAIL]
Dallas Life Support | PO# 5290 | Pam/Bryan | [PHONE] | C06108-L | 2013-08-07 00:00:00 | 10k PM | EP | CSO 130812-204 | N/A | completed |  | 2014-07-28 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#061014KFPM9252 | [NAME] | [PHONE] | F00535 | 2013-08-13 00:00:00 | [NAME] 30kPM Due: [NAME] | EP | CSO 130815-205 | N/A | completed |  | 2014-06-18 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#020414KFPM | [NAME] | [PHONE] | F49200 | 2013-08-13 00:00:00 | keeps beeping malfunctioning    10k PM | EP | CSO 130815-206 | N/A | completed |  | 2014-02-12 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10851 | 2013-08-13 00:00:00 | OH | RKR | N/A | ISO 130815-90 | completed | OH kit 08/15/2013 | 2013-08-29 00:00:00 | yes | N/A
Texas State University | R11695- shipped to [NAME] 8/20/2013 | [NAME] | [PHONE] | IPV1940014 | 2013-08-13 00:00:00 | OH/Ugrade to IPV1C Model F00001-C | Percuss | CSO 130815-207 | N/A | completed | New SN# IPV1C2249 | 2013-11-01 00:00:00 | yes | N/A
Covenant Medical Center | PO# CM-108943 | [NAME] | [PHONE] | IPV1C1494 | 2013-08-23 00:00:00 | OH Due: 06/04/2013 | RKR | CSO 130820-208 | N/A | completed | OH kit 09/04/2013 | 2013-10-11 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10846 | 2013-08-23 00:00:00 | OH Case Replacement | RKR | N/A | ISO 130823-91 | completed | OH kit w/ case 08/23/2013 | 2013-09-23 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10855 | 2013-08-26 00:00:00 | OH | RKR | N/A | ISO 130826-92 | completed | OH kit 08/26/2013 | 2013-10-02 00:00:00 | yes | N/A
IME | IME | SOLD | SOLD | C16063 | 2013-08-22 00:00:00 | NOTED SOLD: 12/23/2019 | EP | N/A | ISO 130826-93 | completed |  | 2019-12-23 00:00:00 | RKR | N/A
Children's Medical Center | PO# 1514060 | [NAME] | [PHONE] | VDR40189 | 2013-08-30 00:00:00 | Annual PM | RKR | CSO 130827-209 | N/A | completed |  | 2013-12-04 00:00:00 | yes | [EMAIL]
Angel Medical/Firmco | compressor exchange for new unit RMA1041 | BJ Prendergast | [PHONE] | IPVHC00127 | 2013-08-23 00:00:00 | Needs Compressor Upgrade | Percuss | CSO 130827-210 | N/A | completed | New Unit IPVHC00906 | 2014-01-08 00:00:00 | yes | [EMAIL]
IME | IME | SOLD | SOLD | C16060 | 2013-08-27 00:00:00 | NOTED SOLD: 12/23/2019 | EP | N/A | ISO 130827-94 | completed |  | 2019-12-23 00:00:00 | RKR | N/A
[NAME] Hospital | approved via email, need copy of PO | [NAME] | (281) 293-7774 ext. 6626 | IPV1C1759 | 2013-09-03 00:00:00 | PM | RKR | CSO 130828-211 | N/A | completed | Last OH/UGRADE 07/09/2011 PM 07/09/2012 | 2013-09-11 00:00:00 | yes | [EMAIL]
[NAME] | PO# Tripp092013 | [NAME] | [PHONE] | F35832 | 2013-09-11 00:00:00 | VPOW will not stop blowing | EP | CSO 130828-212 | N/A | completed | 10k Due: 08/22/2014 or 13,907 usage hours | 2013-09-20 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | compressor- exchange for new unit RMA1040 | [NAME] | [PHONE] | IPVHT00109 | 2013-08-28 00:00:00 | Needs Compressor Upgrade | Percuss | CSO 130828-213 | N/A | completed | New Unit IPVHC00907 | 2014-01-08 00:00:00 | yes | [EMAIL]
Partsource/Baylor University Medical Center | PO# [PHONE] | [NAME] | [PHONE] | IMP10680 | 2013-09-05 00:00:00 | PM | RKR | CSO 130904-214 | N/A | completed |  | 2013-10-01 00:00:00 | yes | [EMAIL]
Children's Medical Center | PO# 1517775 | [NAME] | [PHONE] | IPV1C0552 | 2013-09-06 00:00:00 | OH Due: 06/22/2013 | RKR | CSO 130904-215 | N/A | completed | OH kit 9/10/2013 | 2013-10-14 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#061214KFPM9254 | [NAME] | [PHONE] | B01469 | 2013-09-03 00:00:00 | 10k PM | EP | CSO 130904-216 | N/A | completed |  | 2014-06-16 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | [NAME] | [NAME] | [PHONE] | C15178 | 2013-09-03 00:00:00 | 30k PM | EP | CSO 130904-217 | N/A | completed |  | 2014-06-13 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#061614KFPM9259 | [NAME] | [PHONE] | F00533 | 2013-09-03 00:00:00 | 30k PM | EP | CSO 130904-218 | N/A | completed |  | 2014-08-27 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#020214KFPM | [NAME] | [PHONE] | F49196 | 2013-09-03 00:00:00 | 10k PM | EP | CSO 130904-219 | N/A | completed |  | 2014-02-10 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#061114KFPM9253 | [NAME] | [PHONE] | F00697 | 2013-08-30 00:00:00 | 30k PM | EP | CSO 130904-220 | N/A | completed |  | 2014-08-21 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | [NAME] | [NAME] | [PHONE] | F49242 | 2013-08-30 00:00:00 | 10k PM | EP | CSO 130904-221 | N/A | completed |  | 2014-02-21 00:00:00 | yes | [EMAIL]
Children's Medical Center | PO# 1514060 | [NAME] | [PHONE] | WF20329 | 2013-08-30 00:00:00 | Calibration @ Percussionaire | Percuss | CSO 130905-222 | N/A | completed | RMA1046 | 2013-12-04 00:00:00 | yes | [EMAIL]
Wyle Integrated Science | PO# T33726003 | [NAME] | [PHONE] | TXP950012 | 2013-08-30 00:00:00 | Annual PM | RKR | CSO 130905-223 | N/A | completed |  | 2013-09-06 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F25481 | 2013-09-05 00:00:00 | 30k PM Due: 01/10/2014 or 27,455 hours | EP | N/A | ISO 130905-95 | completed |  | 2013-10-01 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10864 | 2013-09-05 00:00:00 | OH- ordered kit 9/9 | RKR | N/A | ISO 130905-96 | completed | OH kit 09/06/2013 | 2013-10-03 00:00:00 | yes | N/A
The Care Group | PO# 22673 | [NAME] | [PHONE] | F45102 | 2013-09-05 00:00:00 | 10k PM | EP | CSO 130909-224 | N/A | completed |  | 2013-10-04 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22673 | [NAME] | [PHONE] | F49086 | 2013-09-05 00:00:00 | 10k PM | EP | CSO 130909-225 | N/A | completed |  | 2013-10-09 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22673 | [NAME] | [PHONE] | F54141 | 2013-09-05 00:00:00 | 10k PM | EP | CSO 130909-226 | N/A | completed |  | 2013-10-25 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22673 | [NAME] | [PHONE] | F54158 | 2013-09-05 00:00:00 | 10k PM | EP | CSO 130909-227 | N/A | completed |  | 2013-10-30 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22673 | [NAME] | [PHONE] | F49084 | 2013-09-05 00:00:00 | VPOW PM not due until 04/25/2015 | EP | CSO 130909-228 | N/A | completed |  | 2013-09-20 00:00:00 | yes | [EMAIL]
Children's Medical Center | PO# 1517775 | [NAME] | IME | IPVC0954 | 2013-09-11 00:00:00 | Annual PM | RKR | CSO 130910-229 | N/A | completed |  | 2013-10-04 00:00:00 | yes | [EMAIL]
[NAME] | RMA 092713-2 shipped 9/30 | [NAME] | [PHONE] | VDR40253/WF20468 | 2013-09-13 00:00:00 | Warranty Purchase Date: 10/01/2012 | Percuss | CSO 130911-230 | N/A | completed | RMA092713-2 | 2013-12-13 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10980 | 2013-09-11 00:00:00 | OH Due: 11/11/2013 | RKR | N/A | ISO 130911-97 | completed | OH kit 9/20/2013 | 2013-10-24 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11216 | 2013-09-11 00:00:00 | PM Due: 09/26/2013 | RKR | N/A | ISO 130911-98 | completed |  | 2013-09-19 00:00:00 | yes | N/A
Dallas Life Support | PO# 5308 | Pam/Bryan | [PHONE] | F41754 | 2013-09-12 00:00:00 | 10k PM | EP | CSO 130912-231 | N/A | completed |  | 2014-03-04 00:00:00 | yes | [EMAIL]
Children's Medical Center | PO# 90031180-0-9 | [NAME] | [PHONE] | IPV1C0953 | 2013-09-20 00:00:00 | OH Due: [NAME] | RKR | CSO 130919-232 | N/A | completed | OH kit 09/19/2013 | 2013-10-16 00:00:00 | yes | [EMAIL]
Children's Medical Center | PO# 1524748 | [NAME] | [PHONE] | IPV1C0554 | 2013-09-26 00:00:00 | OH Due: 09/07/2013 | RKR | CSO 130919-233 | N/A | completed | OH kit 12/03/2013 | 2013-12-27 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | WF20330 | 2013-09-27 00:00:00 | shipped for calibration 10/16 | Percuss | N/A | ISO 130920-99 | completed | RMA 1097 | 2014-02-05 00:00:00 | yes | N/A
 | returned to customer 9/30, under warranty- send to Care Fusion |  |  | F65897 | 2013-09-18 00:00:00 | Not Holding Charge Warranty MFG: 2013-02 | EP | CSO 130925-234 | N/A | did not service | 10kPM Not Due Until 2015-02 or 10,000 hours | 2013-09-30 00:00:00 | yes | [EMAIL]
 | PO# Tripp093013 |  |  | F42337 | 2013-09-18 00:00:00 | 10kPM Not holding Cahrge last 30 Minutes | EP | CSO 130925-235 | N/A | completed |  | 2013-10-04 00:00:00 | yes | [EMAIL]
 | PO# Tripp093013 |  |  | IMP11113 | 2013-09-18 00:00:00 | Annual PM past due since 02/23/2013 | RKR | CSO 130925-236 | N/A | completed | Cartridge replaced | 2013-10-04 00:00:00 | yes
 | [NAME] |  |  | F31721 | 2013-09-21 00:00:00 | 10k PM | EP | CSO 130925-237 | N/A | completed |  | 2013-09-30 00:00:00 | yes
The Care Group | PO#22680 | [NAME] | [PHONE] | F40373 | 2013-09-20 00:00:00 | 10k PM | EP | CSO 130925-238 | N/A | completed |  | 2013-10-25 00:00:00 | yes | [EMAIL]
The Care Group | PO#22680 | [NAME] | [PHONE] | F35882 | 2013-09-20 00:00:00 | 10k PM | EP | CSO 130925-239 | N/A | completed |  | 2013-10-16 00:00:00 | yes | [EMAIL]
The Care Group | PO#22680 | [NAME] | [PHONE] | F49085 | 2013-09-20 00:00:00 | 10k PM | EP | CSO 130925-240 | N/A | completed |  | 2013-10-16 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22694 RMA092713/RMA1143 | [NAME] | [PHONE] | IPVHT00157 | 2013-09-16 00:00:00 | Needs Compressor Upgrade | Percuss | CSO 130925-241 | N/A | completed | New Unit IPVHC00932 | 2014-02-25 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22694 RMA092713-1/RMA1144 | [NAME] | [PHONE] | IPVHT00161 | 2013-09-16 00:00:00 | Needs Compressor Upgrade | Percuss | CSO 130925-242 | N/A | completed | New Unit IPVHC00917 | 2014-02-25 00:00:00 | yes | [EMAIL]
The Care Group | PO#22680 | [NAME] | [PHONE] | IPVHC00667 | 2013-09-19 00:00:00 | PM | RKR | CSO 130925-243 | N/A | completed |  | 2013-10-11 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IPVHC00229 | 2013-09-23 00:00:00 | Needs Compressor Upgrade | Percuss | N/A | ISO 130925-100 | completed | RMA1464 | 2014-05-13 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11171 | 2013-09-19 00:00:00 | Annual PM Due: 07/30/2013 | RKR | N/A | ISO 130925-101 | completed |  | 2013-10-01 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11209 | 2013-09-27 00:00:00 | Annual PM Due: [NAME] | RKR | N/A | ISO 130927-102 | completed |  | 2013-09-30 00:00:00 | yes | N/A
Dell Children's Hospital Seton/Trimedx | Warranty Purchase Date: 11/21/2012 | Rebecca/Dell Childrens | [PHONE] | IPV1C2022 | 2013-09-30 00:00:00 | Would not adjust pressure | Percuss | CSO 130925-245 | N/A | completed | RMA 1098 | 2013-11-21 00:00:00 | yes | [EMAIL]
 |  |  |  | IPVHC00501 | 2013-09-27 00:00:00 | PM | RKR | N/A | ISO 130927-103 | completed |  | 2013-10-04 00:00:00 | yes | N/A
 |  |  |  | IPVHT00202 | 2013-09-27 00:00:00 | Needs Compressor Upgrade | Percuss | N/A | ISO 130927-104 | completed | RMA1465 | 2014-05-13 00:00:00 | yes | N/A
[NAME] | PO# Tripp111913 | [NAME] | [PHONE] | C48526 | 2013-09-30 00:00:00 | 10k PM | EP | CSO 131001-246 | N/A | completed |  | 2013-11-25 00:00:00 | yes | [EMAIL]
[NAME] | PO# Tripp111913 | [NAME] | [PHONE] | F56005 | 2013-09-30 00:00:00 | 10k PM Due: 2014-04 or 10,000 hours | EP | CSO 131001-247 | N/A | completed |  | 2013-11-27 00:00:00 | yes | [EMAIL]
[NAME] | PO#6106 | [NAME] | 210-696-2626x1213 | D01328 | 2013-09-30 00:00:00 | 10k PM | EP | CSO 131002-248 | N/A | completed |  | 2013-12-23 00:00:00 | yes | [EMAIL]
[NAME] | [NAME] | [NAME] | 210-696-2626x1213 | C14906 | 2013-09-30 00:00:00 | 30k PM | EP | CSO 131002-249 | N/A | completed |  | 2013-11-07 00:00:00 | yes | [EMAIL]
[NAME] | PO#6106 | [NAME] | 210-696-2626x1213 | C14907 | 2013-09-30 00:00:00 | 30k PM | EP | CSO 131002-250 | N/A | completed |  | 2013-12-13 00:00:00 | yes | [EMAIL]
[NAME] | PO#6106 | [NAME] | 210-696-2626x1213 | C07212 | 2013-09-30 00:00:00 | 30k PM | EP | CSO 131002-251 | N/A | completed |  | 2013-12-17 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | PO# 39583 | [NAME] | [PHONE] | IPVHC00368 | 2013-10-03 00:00:00 | Not Percussing | RKR | CSO 131007-252 | N/A | completed | Last PM Date: 01/24/2013 | 2013-10-11 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | PO# 39465 | [NAME] | [PHONE] | IPVHC00093 | 2013-10-03 00:00:00 | Making rattling sound percussion low | RKR | CSO 131007-253 | N/A | completed | Last PM Date: 08/08/2013 | 2013-10-11 00:00:00 | yes | [EMAIL]
Peak Medical/Med One | Warranty Purchase Date: 09/27/2013 | [NAME] | [PHONE] | IPV1C2206 | 2013-10-04 00:00:00 | Failed on [NAME] pressure display | Percuss | CSO 131007-254 | N/A | completed | RMA 1099 | 2013-11-21 00:00:00 | yes | [EMAIL]
American Medical Equipment | Pam Fish | Pam Fish | [PHONE] | C12778 | 2013-10-04 00:00:00 | Turbine has high pitch noise | EP | CSO 131007-255 | N/A | completed | Turbine under warranty | 2013-10-24 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP11026 | 2013-10-09 00:00:00 | PM | RKR | N/A | ISO 131009-105 | completed |  | 2013-10-22 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10861 | 2013-10-09 00:00:00 | OH Due: 01/15/2013 | RKR | N/A | ISO 131009-106 | completed | OH kit 10/10/2013 | 2013-10-24 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11079 | 2013-10-10 00:00:00 | PM Unit [NAME]'s | RKR | N/A | ISO 131010-107 | completed |  | 2013-10-15 00:00:00 | yes | N/A
Unitversity of Texas Medical Branch | UTMB2197 | Dustin | [PHONE] | IPVC0969 | 2013-10-16 00:00:00 | PM Gauge not working correctly | RKR | CSO 131011-256 | N/A | completed |  | 2014-05-23 00:00:00 | yes | [EMAIL]
Unitversity of Texas Medical Branch | UTMB2197 | Dustin | [PHONE] | IPVC0973 | 2013-10-16 00:00:00 | Repair Gauge pressure does not change | RKR | CSO 131011-257 | N/A | completed | A50067-C-REPLC & SN# [NAME]: 11/15/2013 | 2014-05-22 00:00:00 | yes | [EMAIL]
Unitversity of Texas Medical Branch | STILL UNDER WARRANTY | Dustin | [PHONE] | IPVC0993 | 2013-10-16 00:00:00 | Warranty until 06/24/2014 shuts off in use | RKR | CSO 131011-258 | N/A | completed |  | 2013-11-12 00:00:00 | yes | [EMAIL]
Good Samaritan Hospital  (California) | Device was never service customer wants unit shipped back 12/6/2013 | [NAME] | [PHONE] | IPVC1004 | 2013-10-15 00:00:00 | OH due 04/04/2014 case & O2 upgrade | RKR | CSO 131011-259 | N/A | completed | OH kit w/ case: 11/02/2013 Received: 12/02/2013 | 2013-12-06 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | C11633 | 2013-10-10 00:00:00 | 10k PM Mother Board | EP | N/A | ISO 131014-108 | completed |  | 2013-11-21 00:00:00 | yes | N/A
IME | IME | IME | IME | [NAME] | 2013-10-22 00:00:00 | 30k PM Analog Board | EP | N/A | ISO 131022-109 | completed |  | 2013-11-22 00:00:00 | yes | N/A
US Med Equip | PO#85584AH | [NAME] | [PHONE] | IPV1C1722 | 2013-10-25 00:00:00 | PM Percussion not adjust needs calibration | RKR | CSO 131024-260 | N/A | completed |  | 2013-11-12 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | [NAME] | 2013-10-28 00:00:00 | 30k PM | EP | N/A | ISO 131028-110 | completed |  | 2013-10-30 00:00:00 | yes | N/A
IME | IME | IME | IME | C11661 | 2013-10-28 00:00:00 | 30k PM | EP | N/A | ISO 131028-111 | completed |  | 2013-11-01 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11152 | 2013-10-28 00:00:00 | Annual PM | RKR | N/A | ISO 131028-112 | completed |  | 2013-10-30 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP1940067 | 2013-10-28 00:00:00 | Annual PM | RKR | N/A | ISO 131028-113 | completed |  | 2013-11-04 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10850 | 2013-10-28 00:00:00 | OH Due: 01/15/2013 | RKR | N/A | ISO 131028-114 | completed | OH Kit 10/29/2013 | 2013-11-18 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10994 | 2013-10-28 00:00:00 | OH Due: 11/19/2013 | RKR | N/A | ISO 131028-115 | completed | OH kit 10/29/2013 | 2013-11-19 00:00:00 | yes | N/A
[NAME] | PO#UHS-9CXJSR 11/25/13 | [NAME] | [PHONE] | IPV1C0283 | 2013-10-31 00:00:00 | OH Due: 11/16/2012 No Pressure & Rate too low | RKR | CSO 131029-261 | N/A | completed | OH kit 11/02/2013 Received: 12/02/2013 | 2014-01-27 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10996 | 2013-10-29 00:00:00 | OH Due: 12/13/2013 | RKR | N/A | ISO 131029-116 | completed | OH kit 11/01/2013 | 2013-12-19 00:00:00 | yes | N/A
IME | IME | IME | IME | C11643 | 2013-10-30 00:00:00 | 10k PM Due: 11-18-2013 or 32,197.3 | RKR | N/A | ISO 131030-117 | completed |  | 2013-11-01 00:00:00 | yes | N/A
IME | IME | IME | IME | F25481 | 2013-10-30 00:00:00 | VCDW Patient Death | CF | N/A | ISO 131030-118 | completed |  | 2013-11-18 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11210 | 2013-10-30 00:00:00 | Annual PM | RKR | N/A | ISO 131030-119 | completed |  | 2013-11-05 00:00:00 | yes | N/A
The Care Group | PO#22697 | [NAME] | [PHONE] | C23708 | 2013-10-16 00:00:00 | 10k PM | EP | CSO 131101-262 | N/A | completed |  | 2014-01-03 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22688 | [NAME] | [PHONE] | C15811 | 2013-10-16 00:00:00 | 30k PM | EP | CSO 131101-263 | N/A | completed |  | 2013-11-15 00:00:00 | yes | [EMAIL]
The Care Group | UNIT NOT SERVICED CUSTOMER DID NOT WANT TO PAY FOR INFESTATION | [NAME] | [PHONE] | B04924 | 2013-10-16 00:00:00 | 30k PM INFESTATION | EP | CSO 131101-264 | N/A | completed | ON HOLD Per Customer | 2013-12-10 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22694 | [NAME] | [PHONE] | IPVHT00205 | 2013-10-16 00:00:00 | PM | RKR | CSO 131101-265 | N/A | completed |  | 2013-12-04 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#020114KFPM | [NAME] | [PHONE] | F39642 | 2013-10-17 00:00:00 | 10k PM | EP | CSO 131101-266 | N/A | completed |  | 2014-02-26 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#061414KFPM9256 | [NAME] | [PHONE] | F49243 | 2013-10-17 00:00:00 | 10k PM | EP | CSO 131101-267 | N/A | completed |  | 2015-01-20 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | NOT SERVICED | [NAME] | [PHONE] | F49241 | 2013-10-17 00:00:00 | NOT Serviced 10kpm/solenoid mount Cancelled by Customer | EP | CSO 131101-268 | N/A | completed |  | 2015-01-12 00:00:00 | yes | [EMAIL]
Ms.[NAME]/[NAME] of Disability Services | NOT SERVICED |  |  | U199800051 | 2013-10-17 00:00:00 | OH/Upgrade to [NAME]: F00012 | Percuss | CSO 131101-269 | N/A | completed | Denial of Service | 2022-04-06 00:00:00 | yes | N/A
IME | IME | IME | IME | [NAME] | 2013-11-06 00:00:00 | 10k PM | EP | N/A | ISO 131106-120 | completed |  | 2013-11-20 00:00:00 | yes | N/A
IME | IME | IME | IME | F50224 | 2013-11-06 00:00:00 | 10k PM | EP | N/A | ISO 131106-121 | completed |  | 2016-08-24 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10899 | 2013-11-06 00:00:00 | OH Due: [NAME] | RKR | N/A | ISO 131106-122 | completed | OH kit 11/07/2013 | 2014-01-10 00:00:00 | yes | N/A
Partsource/Baylor University Medical Center | PO# PRTS1607960 | [NAME] | [PHONE] | IMP10680 | 2013-11-07 00:00:00 | Repair unit dropped replace cartridge | RKR | CSO 131107-270 | N/A | completed |  | 2013-12-06 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10836 | 2013-11-13 00:00:00 | PM | RKR | N/A | ISO 131113-123 | completed |  | 2013-11-14 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11217 | 2013-11-13 00:00:00 | PM | RKR | N/A | ISO 131113-124 | completed |  | 2013-11-15 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10900 | 2013-11-13 00:00:00 | OH Due: [NAME] | RKR | N/A | ISO 131113-125 | completed | OH kit 11/13/2013 | 2014-01-16 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10937 | 2013-11-13 00:00:00 | OH Due: 07/02/2013 (broken lens cover) | RKR | N/A | ISO 131113-126 | completed | OH kit 11/13/2013 | 2014-01-20 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11005 | 2013-11-13 00:00:00 | OH Due: 12/15/2013 | RKR | N/A | ISO 131113-127 | completed | OH kit 11/13/2013 | 2014-01-21 00:00:00 | yes | N/A
IME | IME | SOLD | SOLD | C16057 | 2013-11-13 00:00:00 | NOTED SOLD: 12/23/2019 | EP | N/A | ISO 131113-128 | completed |  | 2019-12-23 00:00:00 | RKR | N/A
IME | IME | SOLD | SOLD | C16058 | 2013-11-13 00:00:00 | NOTED SOLD: 12/23/2019 | EP | N/A | ISO 131113-129 | completed |  | 2019-12-23 00:00:00 | RKR | N/A
IME | IME | IME | IME | F50228 | 2013-11-18 00:00:00 | 10k PM | EP | N/A | ISO 131121-130 | completed |  | 2014-12-17 00:00:00 | yes | N/A
IME | IME | IME | IME | F55639 | 2013-11-18 00:00:00 | 10k PM | EP | N/A | ISO 131121-131 | completed |  | 2014-12-08 00:00:00 | yes | N/A
[NAME] | PO#[PHONE] | [NAME] | [PHONE] | IPV1C1566 | 2013-12-05 00:00:00 | OH Due: 11/19/2013 | RKR | CSO 131122-271 | N/A | completed | OH kit 12/12/2013 | 2014-02-14 00:00:00 | yes | [EMAIL]
[NAME] | PO#[PHONE] | [NAME] | [PHONE] | VDR420030007 | 2013-12-05 00:00:00 | OH Due: 11/29/2013 | RKR | CSO 131122-272 | N/A | completed | OH kit 12/12/2013 | 2014-04-04 00:00:00 | yes | [EMAIL]
IME/ Wave Healthcare | IME | IME | IME | [NAME] | 2013-11-18 00:00:00 | Annual PM | RKR | CSO 131125-273 | N/A | completed |  | 2014-04-10 00:00:00 | yes | N/A
The Care Group | PO# 22696 | [NAME] | [PHONE] | B04606 | 2013-11-18 00:00:00 | 10k PM | EP | CSO 131125-274 | N/A | completed |  | 2013-12-04 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22696 | [NAME] | [PHONE] | B05031 | 2013-11-18 00:00:00 | 30k PM | EP | CSO 131125-275 | N/A | completed |  | 2013-12-04 00:00:00 | yes | [EMAIL]
The Care Group | PO# 22696 | [NAME] | [PHONE] | B05039 | 2013-11-18 00:00:00 | 10k PM | EP | CSO 131125-276 | N/A | completed |  | 2013-12-04 00:00:00 | yes | [EMAIL]
The Care Group | PO#22697 | [NAME] | [PHONE] | B27365 | 2013-11-18 00:00:00 | 10k PM | EP | CSO 131125-277 | N/A | completed |  | 2014-01-03 00:00:00 | yes | [EMAIL]
The Care Group | PO#22697 | [NAME] | [PHONE] | B27366 | 2013-11-18 00:00:00 | 30k PM INFESTATION | EP | CSO 131125-278 | N/A | completed |  | 2014-02-06 00:00:00 | yes | [EMAIL]
The Care Group | PO#22697 | [NAME] | [PHONE] | F53908 | 2013-11-18 00:00:00 | 10k PM | EP | CSO 131125-279 | N/A | completed |  | 2014-01-08 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#061514KFPM9258 | [NAME] | [PHONE] | F00528 | 2013-11-20 00:00:00 | 30k PM | EP | CSO 131125-280 | N/A | completed |  | 2014-06-18 00:00:00 | yes | [EMAIL]
S.TX.Monitoring | PO#120313KFPM | [NAME] | [PHONE] | F39643 | 2013-11-20 00:00:00 | 10k PM | EP | CSO 131125-281 | N/A | completed |  | 2013-12-13 00:00:00 | yes | [EMAIL]
[NAME] | PO#TRIPP112613 | [NAME] | [PHONE] | C05237 | 2013-11-21 00:00:00 | 10k PM | EP | CSO 131125-282 | N/A | completed |  | 2013-12-27 00:00:00 | yes | [EMAIL]
[NAME] | PO#TRIPP112613 | [NAME] | [PHONE] | C47053 | 2013-11-21 00:00:00 | 10k PM | EP | CSO 131125-283 | N/A | completed |  | 2013-12-26 00:00:00 | yes | [EMAIL]
[NAME] | PO#Don Boswell | [NAME] | [PHONE] | C48522 | 2013-11-22 00:00:00 | 10k PM | EP | CSO 131125-284 | N/A | completed |  | 2014-02-06 00:00:00 | yes | [EMAIL]
Apple Homecare | PO# 10339 | [NAME] | [PHONE] | IPVHC00388 | 2013-11-27 00:00:00 | PM | RKR | CSO 131126-285 | N/A | completed |  | 2013-12-13 00:00:00 | yes | [EMAIL]
Covenant Health System | PO# CM-112133 | [NAME] | [PHONE] | IPV1C1488 | 2013-11-26 00:00:00 | OH Due: 06/04/2013 | RKR | CSO 131204-286 | N/A | completed | OH kit 12/04/2013 | 2014-04-17 00:00:00 | yes | [EMAIL]
Parkland Hospital | PO#637502 | [NAME] | [PHONE] | VDR4980018 | 2013-11-27 00:00:00 | OH Due: 05/13/2012 | RKR | CSO 131204-287 | N/A | completed | OH kit 12/04/2013 | 2014-04-22 00:00:00 | yes | mailto:[EMAIL]
Parkland Hospital | PO#637502 | [NAME] | [PHONE] | VDR40214 | 2013-11-27 00:00:00 | OH Due: 01/19/2013 | RKR | CSO 131204-288 | N/A | completed | OH kit 12/04/2013 | 2014-04-28 00:00:00 | yes | mailto:[EMAIL]
Parkland Hospital | PO#637502 | [NAME] | [PHONE] | [NAME] | 2013-11-27 00:00:00 | PM Due: [NAME] | RKR | CSO 131204-289 | N/A | completed |  | 2013-12-09 00:00:00 | yes | mailto:[EMAIL]
Parkland Hospital | PO#637502 | [NAME] | [PHONE] | VDR4980020 | 2013-11-27 00:00:00 | PM Due: 11/19/2013 | RKR | CSO 131204-290 | N/A | completed |  | 2013-12-09 00:00:00 | yes | mailto:[EMAIL]
Medical Plus Supplies | PO#42334 | [NAME] | [PHONE] | IPVHC00391 | 2013-11-27 00:00:00 | PM Due: 10/06/2012 Neb Switch & Fan Gaurd | RKR | CSO 131204-291 | N/A | completed |  | 2013-12-27 00:00:00 | yes | [EMAIL]
Medical Plus Supplies | PO#42403 RMA1240 | [NAME] | [PHONE] | IPVHT00117 | 2013-11-27 00:00:00 | Needs Compressor Upgrade | Percuss | CSO 131204-292 | N/A | completed | New Unit IPVHC00910 | 2014-01-21 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | F55656 | 2013-12-05 00:00:00 | 10k PM Due: 2014-03 or 10,000 usage hours | EP | N/A | ISO 131205-132 | completed |  | 2014-09-15 00:00:00 | yes | N/A
[NAME] | PO#[PHONE] | [NAME] | [PHONE] | WF20100 | 2013-12-05 00:00:00 | Calibration | Percuss | CSO 131211-293 | N/A | completed | RMA1259 | 2014-01-30 00:00:00 | yes | [EMAIL]
Dell Children's Hospital Seton/Trimedx | PO# 300490000008890 | [NAME] | [PHONE] | IPV1C0230 | 2013-12-05 00:00:00 | OH Due: 12/31/2007 | RKR | CSO 131211-294 | N/A | completed | OH kit 12/12/2013 | 2014-02-17 00:00:00 | yes | [EMAIL]
Dell Children's Hospital Seton/Trimedx | PO# 300490000008890 | [NAME] | [PHONE] | IPV2C0355 | 2013-12-05 00:00:00 | PM Due: 11/21/2013 | RKR | CSO 131211-295 | N/A | completed |  | 2014-01-15 00:00:00 | yes | [EMAIL]
The Care Group | PO#22699 | [NAME] | [PHONE] | F50039 | 2013-12-05 00:00:00 | 10k PM | EP | CSO 131211-296 | N/A | completed |  | 2014-01-10 00:00:00 | yes | [EMAIL]
The Care Group | PO#22700 | [NAME] | [PHONE] | F50698 | 2013-12-05 00:00:00 | 10k PM | EP | CSO 131211-297 | N/A | completed |  | 2014-01-15 00:00:00 | yes | [EMAIL]
The Care Group | PO#22702 | [NAME] | [PHONE] | F50701 | 2013-12-05 00:00:00 | 10k PM | EP | CSO 131211-298 | N/A | completed |  | 2014-01-27 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10864 | 2013-12-11 00:00:00 | OH Due: 01/15/2013 | RKR | N/A | ISO 131212-133 | completed | OH kit 12/12/2013 | 2013-12-30 00:00:00 | yes | N/A
Dell Children's Hospital Seton/Trimedx | TMX574181 | Mohammed | [PHONE] | IPV1C1265 | 2013-12-19 00:00:00 | OH | RKR | CSO 131219-302 | N/A | completed | OH kit 12/26/2013 | 2014-02-25 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IPVHC00374 | 2013-12-13 00:00:00 | PM Due: 07/05/2012 | RKR | N/A | ISO 131216-134 | completed |  | 2014-01-22 00:00:00 | yes | N/A
[NAME] | PO#joseph122013 | [NAME] | 210-696-2626x1213 | C06692 | 2013-12-13 00:00:00 | Internal Battery Malfunction | EP | CSO 131216-299 | N/A | completed |  | 2013-12-23 00:00:00 | yes | [EMAIL]
[NAME] | PO#joseph122013 | [NAME] | 210-696-2626x1213 | C07214 | 2013-12-13 00:00:00 | 10k PM | EP | CSO 131216-300 | N/A | completed |  | 2014-01-31 00:00:00 | yes | [EMAIL]
[NAME] | PO#joseph122013 | [NAME] | 210-696-2626x1213 | F50892 | 2013-12-17 00:00:00 | 10k PM | EP | CSO 131217-301 | N/A | completed |  | 2014-01-15 00:00:00 | yes | [EMAIL]
Children's Medical Center | PO# 1525914 | [NAME] | [PHONE] | VDR40186 | 2014-01-03 00:00:00 | PM Due: 10/23/2013 | RKR | CSO 131219-303 | N/A | completed |  | 2014-01-23 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP11046 | 2013-12-20 00:00:00 | PM Due: 05/12/2012 | RKR | N/A | ISO 131220-135 | completed |  | 2013-12-30 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10895 | 2013-12-23 00:00:00 | OH Due: 03/30/2013 | RKR | N/A | ISO 131223-136 | completed | [NAME] 12/23/2013 | 2014-04-17 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP11042 | 2013-12-23 00:00:00 | PM Due: 10/30/2013 | RKR | N/A | ISO 131223-137 | completed |  | 2013-12-30 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10469 | 2013-12-23 00:00:00 | OH Due: [NAME] | RKR | N/A | ISO 131223-138 | completed | OH kit 12/26/2013 | 2014-05-07 00:00:00 | yes | N/A
The Care Group | PO#22704 | [NAME] | [PHONE] | IPVHC00204 | 2013-12-19 00:00:00 | PM Due: 03/13/2013 | RKR | CSO 131226-304 | N/A | completed |  | 2014-01-08 00:00:00 | yes | [EMAIL]
The Care Group | PO#22705 | [NAME] | [PHONE] | [NAME] | 2013-12-19 00:00:00 | 10k PM | EP | CSO 131226-305 | N/A | completed |  | 2014-01-29 00:00:00 | yes | [EMAIL]
The Care Group | PO#1614 | [NAME] | [PHONE] | B04895 | 2013-12-19 00:00:00 | 30k PM | EP | CSO 131226-306 | N/A | completed |  | 2014-01-30 00:00:00 | yes | [EMAIL]
The Care Group | PO#22707 | [NAME] | [PHONE] | F50695 | 2013-12-19 00:00:00 | 10k PM | EP | CSO 131226-307 | N/A | completed |  | 2014-01-22 00:00:00 | yes | [EMAIL]
The Care Group | PO#22708 | [NAME] | [PHONE] | F57065 | 2013-12-19 00:00:00 | 10k PM | EP | CSO 131226-308 | N/A | completed |  | 2014-01-27 00:00:00 | yes | [EMAIL]
IME | IME | IME | IME | IMP10838 | 2013-12-26 00:00:00 | PM Due: 02/05/2014 A99517 replaced | RKR | N/A | ISO 131226-139 | completed |  | 2013-12-31 00:00:00 | yes | N/A
IME | IME | IME | IME | IMP10859 | 2013-12-12 00:00:00 | OH Due: 01/15/2013 | RKR | N/A | ISO 131226-140 | completed | OH kit 12/11/2013 | 2014-04-09 00:00:00 | yes | N/A