Spreadsheet: 28423d1222696163-copy-page-info-multiple-times-in-same-page-new-cal-sheets
Sheet: CAL SHEET

INSTRUMENT MAKE AND MODEL |  |  |  |  | [NAME] |  |  |  | SERIAL NO.:
 |  | CALIBRATION GASES - MIXTURES / METHANE IN AIR
(1)          0 PPM Concentration: |  |  |  | <1 |  | Cylinder No.: | 2451862Y |  | Shelf Life: | 2010-03-24 00:00:00
(2)      500 PPM Concentration: |  |  |  | 495 |  | Cylinder No.: | EB0010622 |  | Shelf Life: | 2009-07-15 00:00:00
(3)   1,000 PPM Concentration: |  |  |  | 959 |  | Cylinder No.: | 436822 |  | Shelf Life: | 2010-07-10 00:00:00
 | LEAK DEFINITION GAS CONCENTRATION MIXTURES / METHANE IN AIR
2,000 PPM Concentration: |  |  |  | 1952 |  | Cylinder No.: | K951511 |  | Shelf Life: | 2010-07-10 00:00:00
10,000 PPM Concentration: |  |  |  | 9986 |  | Cylinder No.: | CC250586 |  | Shelf Life: | 2008-12-19 00:00:00
To begin this procedure, 1) remove the TVA from its case, 2) turn the TVA power switch "ON", allow the TVA to self test, 3) press  the
"CONTROL" button -turn pump on by pressing #1, 4) press the  "CONTROL" button -ignite the burner assembly by pressing #3, allow the
instrument to warm up for at least thirty (30) minutes, then complete the following steps:
A.  At main menu select set-up by pressing #2.
B.  At set-up menu select calibration by pressing #1.
C.  At calibration menu select zero by pressing #3.
[NAME] gas sample to the inlet probe, press the "ENTER" button twice.
E.  From Calibration menu, select span gas by pressing #4. Use the arrow up/down to select span gas concentration.
F.  Apply selected PPM gas concentration to sample inlet probe, press the "ENTER" button twice. Selected PPM
concentration should be accepted.
G.  Repeat steps E and F for second calibration concentration.
[NAME] to main menu and select "RUN" by pressing #1.
I.   Introduce Zero Air and any calibration gases for leak definitions applicable to the equipment being monitored at the
indicated site and on the date documented on this form.  Record the reading below.
TIME | ZERO AIR INSTRUMENT READING |  | LEAK DEFINITION |  | CALIBRATION GAS CONCENTRATION |  | -0.1 | +10% | INSTRUMENT READING (must be within +/-10% of cal gas concentration)
 |  |  | 0 |  | < 1 PPM |  | 0 | 10
 |  |  | 500 |  | 495 |  | 446 | 544
 |  |  | 1000 |  | 959 |  | 864 | 1054
 |  |  | 2000 |  | 1952 |  | 1757 | 2147
 |  |  | 10000 |  | 9986 |  | 8988 | 10984
 |  |  |  | DRIFT ASSESSMENT (500 ppm CAL GAS)
 |  |  | % Accuracy=Instrument Reading/Calibration gas Concentration x 100)
 |  |  |  |  | Instrument Reading: |  |  |  | % Accuracy: | 0
 |  |  |  |  | Instrument Reading: |  |  |  | % Accuracy: | 0
 |  |  |  |  | Instrument Reading: |  |  |  | % Accuracy: | 0
(If the % accuracy is less than 90%, all valves monitored since the last drift assessment with a reading greater than 100 ppm and pumps with
a reading of greater than 500 ppm must be monitored.)
I hereby certify that this VOC detector was calibrated based on the specifications outlined in Reference MethoD 21, CFR 40  Part 60 and
applicable LDAR regulations.
CALIBRATION PERFORMED BY (SIGNATURE) |  |  |  |  |  |  |  |  | DATE: | 2008-10-01 00:00:00
ADDITIONAL OPERATORS (PRINT ONLY)
NAME: |  |  |  |  |  |  | TECH ID: | #N/A
NAME: |  |  |  |  |  |  | TECH ID: | #N/A
NAME: |  |  |  |  |  |  | TECH ID: | #N/A