Spreadsheet: 63008d1263984927-import-external-data-doesnt-import-imported
Sheet: Sheet1

F1 | F2 | F3 | F4 | F5 | F6 | F7 | F8 | F9 | [NAME] | F11
 | SERVICE EQUIPMENT
 | update date: |  |  |  |  | remember to add remaining 2008 deliveries |  |  |  | J
 | Updated by: | Trang
Unit | Serial # | Location |  |  |  | 2008 Status |  | 2008 Follow-up action |  | 2009 Status
CPAP | KMC050302 | Ha Nam Provincial Hospital | 2005-07-01 00:00:00 | 1 | 2008-11-04 00:00:00 | not warm | 2008-12-31 00:00:00 |  | 2009-03-05 00:00:00 | ok
CPAP |  | Ha Nam Provincial Hospital | 2005-07-01 00:00:00 | 2 | 2008-11-04 00:00:00 | sensor | 2008-12-31 00:00:00 | Change Sensor | 2009-03-05 00:00:00 | ok
CPAP |  | Quang Ninh Provincial Hospital | 2005-07-01 00:00:00 | 3 | 2008-11-04 00:00:00 | unstable display temperature |  |  | 2009-03-05 00:00:00 | ok
CPAP | KMC060401 | Son La Provincial Hospital | 2005-07-01 00:00:00 | 4 | 2008-11-04 00:00:00 | sensor | 2008-11-25 00:00:00 | change sensor | 2009-03-05 00:00:00 | ok
CPAP | [NAME] | Son La Provincial Hospital | 2005-07-01 00:00:00 | 5 | 2008-11-04 00:00:00 | sensor | 2008-11-25 00:00:00 | change sensor | 2009-03-05 00:00:00 | ok
CPAP | KMC60403 | Son La Provincial Hospital | 2005-07-01 00:00:00 | 6 | 2008-11-04 00:00:00 | sensor | 2008-11-25 00:00:00 | change sensor | 2009-03-05 00:00:00 | ok
Photo | KMPT060201 | Son La Provincial Hospital | 2005-07-01 00:00:00 | 7 | 2008-11-04 00:00:00 | bulb | 2008-11-05 00:00:00 | 5 bulb | 2009-03-05 00:00:00 | ok
Photo | KMPT060202 | Son La Provincial Hospital | 2005-07-01 00:00:00 | 8 | 2008-11-04 00:00:00 | bulb | 2008-11-05 00:00:00 | 5 bulb | 2009-03-05 00:00:00 | ok
Warmer | KMW060201 | Son La Provincial Hospital | 2005-07-01 00:00:00 | 9 | 2008-11-04 00:00:00 | bulb | 2008-11-05 00:00:00 | change bulb | 2009-03-05 00:00:00 | ok
Warmer | [NAME] | Son La Provincial Hospital | 2005-07-01 00:00:00 | 10 | 2008-11-04 00:00:00 | not warm |  |  | 2009-03-05 00:00:00 | ok
CPAP | KMC050301 | C Hosptal | 2005-07-01 00:00:00 | 11 |  | Display not working |  |  | 2009-01-19 00:00:00 | sensor string fail
CPAP | [NAME] | C Hosptal | 2005-07-01 00:00:00 | 12 |  | Frame Bending |  |  | 2009-01-19 00:00:00 | sensor string fail
CPAP | KMC060303 | [NAME] | 2005-07-01 00:00:00 | 13 |  | [NAME] |  |  | 2009-01-19 00:00:00 | sensor fail
CPAP | No 2 | [NAME] | 2005-07-01 00:00:00 | 14 |  | [NAME] |  |  | 2009-01-19 00:00:00 | transformer broken down
CPAP | KMC060302 | Postal Service Hospital 1 | 2005-07-01 00:00:00 | 15 |  | LED low power
CPAP |  | Son La Provincial Hospital | 2005-07-01 00:00:00 | 16 |  |  |  |  | 2009-01-14 00:00:00 | CPAP washbasin
CPAP |  | Thanh Hoa Pediatric Hospital | 2005-07-01 00:00:00 | 17 |  |  |  |  | 2009-03-24 00:00:00 | sensor broken
CPAP | KMC060304 | [NAME] | 2005-07-01 00:00:00 | 18 |  |  |  |  | 2009-01-19 00:00:00 | air blender broken down
CPAP | KMC060403 | [NAME] | 2005-07-01 00:00:00 | 19 |  |  |  |  | 2009-01-09 00:00:00 | cover of air blender broken down
CPAP | KMC060405 | [NAME] | 2005-07-01 00:00:00 | 20 |  |  |  |  | 2009-01-09 00:00:00 | transformer broken down
CPAP | KMC060407 | [NAME] | 2005-07-01 00:00:00 | 21 |  |  |  |  | 2009-01-09 00:00:00 | Sensor string fail
CPAP | KMC060409 | [NAME] | 2005-07-01 00:00:00 | 22 |  |  |  |  | 2009-01-09 00:00:00 | transformer broken down
CPAP | KMC050302 | C Hosptal | 2006-07-01 00:00:00 | 23 |  |  |  |  | 2009-01-19 00:00:00 | sensor string fail
CPAP | KMC060301 | [NAME] | 2006-07-01 00:00:00 | 24 |  |  |  |  | 2009-01-19 00:00:00 | sensor fail
CPAP | No 1 | [NAME] | 2006-07-01 00:00:00 | 25 |  |  |  |  | 2009-01-19 00:00:00 | sensor fail
CPAP | KMC060301 | Postal Service Hospital 1 | 2006-07-01 00:00:00 | 26
CPAP |  | Son La Provincial Hospital | 2006-07-01 00:00:00 | 27 |  |  |  |  | 2009-01-14 00:00:00 | CPAP washbasin
Warmer |  | Ha Nam Provincial Hospital | 2006-07-01 00:00:00 | 28 |  |  |  |  | 2009-02-17 00:00:00 | sensor string fail
CPAP | KMC060301 | Bac Giang Provincial Hospital | 2006-07-01 00:00:00 | 29 |  |  |  |  | 2009-03-05 00:00:00 | sensor string fail
CPAP | KMC060302 | Bac Giang Provincial Hospital | 2006-07-01 00:00:00 | 30 |  |  |  |  | 2009-03-05 00:00:00 | sensor string fail
CPAP |  | C Hosptal | 2006-07-01 00:00:00 | 31 |  | sensor | 2008-12-19 00:00:00 | Change sensor | 2009-03-05 00:00:00 | ok
Warmer |  | C Hosptal | 2006-07-01 00:00:00 | 32 |  | sensor | 2008-12-19 00:00:00 | Change sensor | 2009-03-05 00:00:00 | ok
CPAP | [NAME] | [NAME] | 2006-07-01 00:00:00 | 33 |  |  |  |  | 2009-01-09 00:00:00 | plug 4 broken down
CPAP | KMC060404 | [NAME] | 2006-07-01 00:00:00 | 34 |  |  |  |  | 2009-01-09 00:00:00 | sensor string plug 4 fail
CPAP | KMC060406 | [NAME] | 2006-07-01 00:00:00 | 35 |  |  |  |  | 2009-01-09 00:00:00 | transformer broken down
CPAP | [NAME] | [NAME] | 2006-07-01 00:00:00 | 36 |  |  |  |  | 2009-01-09 00:00:00 | transformer broken down
CPAP | KMC060410 | [NAME] | 2006-07-01 00:00:00 | 37 |  |  |  |  | 2009-01-09 00:00:00 | Lose plug of pump & air blender broken
CPAP | [NAME] | NZ Health Trust | 2007-04-28 00:00:00 | 38
CPAP | KMC0604998 | NZ Health Trust | 2007-04-28 00:00:00 | 39
CPAP | KMC0604999 | NZ Health Trust | 2007-04-28 00:00:00 | 40
CPAP | [NAME] | Lao Cai Provincial Hospital | 2007-05-21 00:00:00 | 41 | 2008-11-10 00:00:00 | ok |  |  | 2009-03-18 00:00:00 | ok