Spreadsheet: 75492d1276071040-consolidate-4-worksheets-into-1-bom-for-forum
Sheet: looming

 | [NAME] of Materials | MTTS | CPAP box electric
ELECTRONICS:
Author: | [NAME] | Date initial: | 2010-05-19 00:00:00
 |  | Updated: | 2010-06-08 00:00:00
Model version:
Note: | Refer to wiring diagram:  WRI 02 cpap wiring_2.0.pdf
Sequence | Designator | Type no. | Value | Footprint | Description | Nr. required |  | price 1pc (US$) | Order code: | Note | Reference:
1 | PCB1 | MAINHW-LH091125 | version 2.2 |  | Mainboard | 1 |  |  | PCB: MAINBOAR-V2
2 | PCB2 | DISPLHW-LH100518 | version 2.0 |  | Display | 1 |  |  | PCB: DISPLAY-V2
3 | PCB3 | EXPHW-LH091130 | version 1.1 |  | Mainboard expansion interface | 1 |  |  | PCB: INTERFACE
4 | PCB4, PCB5 | HEATHW-091125 | version 1.0 |  | Buckheater | 2 |  |  | PCB: BUCKHEATER
5 | POWERSUPPLY | NES-100-12 | Meanwell Taiwan |  | 12V/8.5A powersupply | 1 |  |  | MW: NES-100-12
6 | button1,button2, [NAME] |  | key1,key2,key3 | frontpanel mount |  | 3 |  |  | CG: PUSHBUTTON-FP
7 | wire1 | 6 wires + insulator | 30cm |  | 2*red, 2*yellow, 2*green | 1 |  |  | CG: 4STRANDWIRE | connect from keys to display
8 | ribbon1 | 10pin | 10cm | header mount | serial | 1 |  |  | HT: RIBBON-10P | connect from display to mainboard
9 | ribbon2 | 16pin | 20cm | header mount | bargraph | 1 |  |  | HT: RIBBON-16P | connect from display to mainboard
10 | wire2 | 1 wire | 20cm |  | red | 1 |  |  | CG: WIRE-RED-D2 | connect from display to expansion interface
11 | wire2 | 1 wire | 20cm |  | black | 1 |  |  | CG: WIRE-BLACK-D2 | connect from display to expansion interface
12 | [NAME] | 1 wire | 20cm |  | red | 1 |  |  | CG: WIRE-RED-D2 | connect from powersupply to expansion interface
13 | [NAME] | 1 wire | 20cm |  | black | 1 |  |  | CG: WIRE-BLACK-D2 | connect from powersupply to expansion interface
14 | EURO | 3pin socket male |  | frontpanel mount | 220V in | 1 |  |  | CG: EURO-SOCKET
15 | VN 220V | 3pin socket female |  | frontpanel mount | 220V out | 1 |  |  | CG: LIOA-SOCKET
16 | THERMFUSE |  | 220V/4A | frontpanel mount | thermal fuse | 1 |  |  | CG: TERMFUSE
17 | ROCKERSWITCH | dual pole | 220V/10A | frontpanel mount |  | 1 |  |  | CG: ROCKERSW
18 | [NAME], Nwire3 | 1 wire | 10cm |  | blue | 3 |  |  | CG: WIRE-BLUE-D2 | 220V Neutral
19 | Lwire1,Lwire2, Lwire3 | 1 wire | 10cm |  | [NAME] | 3 |  |  | CG: WIRE-BROWN-D2 | 220V L1
20 | Ewire1,Ewire2, [NAME] | 1 wire | 10cm |  | yellow/green | 3 |  |  | CG: WIRE-YELLOW-D2 | 220V Earth
21 | Cableshoes | o-ring |  |  |  | 5 |  |  | CG: CABLESHOE-O | connect wires to powersupply
22 | Cableshoes | slide |  |  |  | 4 |  | 0.18838 | [NAME]: A27821CT-ND | connect wires to rockerswitch
23 | Cableshoes | o-ring |  |  |  | 1 |  |  | CG: CABLESHOE-O | connect earthwire to chassis
24 | connector | 7pin socket male |  | frontpanel mount |  | 1 |  |  | CG: CONN-7PIN-MF | tubeset connect
25 | wire4 | 1 wire | 30cm |  | red | 2 |  |  | CG: WIRE-RED-D2 | connect from connector to heaterboard tubeset
26 | wire5 | 1 wire | 30cm |  | blue | 2 |  |  | CG: WIRE-BLUE-D2 | connect from connector to heaterboard humid
27 | toroid | ferrite 3cm diameter |  |  |  | 1 |  |  | CG: FERRITETOROID | 5 turns [NAME]
28 | wire6, wire7 | 1 wire | 30cm |  | blue | 2 |  |  | CG: WIRE-BLUE-D1 | connect from connector to expansion interface
29 | wire8, [NAME] | 1 wire | 30cm |  | yellow | 2 |  |  | CG: WIRE-YELLOW-D1 | connect from connector to expansion interface
30 | humidifier heater | 8 ohm |  |  |  | 1 |  |  | MTTS: HUMHEATER
31 | tempsensor1, [NAME] | DS18B20 |  | TO92 |  | 2 |  | 2.28 | [NAME]: DS18B20+-ND
32 | EUROCABLE |  |  |  |  | 1 |  |  | MW: EUROCABLE