Search query: a fib monitors

Product: KardiaMobile Single-Lead Personal EKG Monitor | FDA-Cleared | Detects AFib
Brand: AliveCor
Color: Grey
Relevance: Exact
EKG ON THE GO: Take unlimited medical-grade EKGs anytime, anywhere. No subscription required.
EKG HISTORY: Store your EKGs on your phone, and email to your doctor with the press of a button.16 bit resolution
TRUSTED BY DOCTORS: FDA-Cleared to detect Atrial Fibrillation, Bradycardia, Tachycardia or Normal Heart Rhythm in 30 seconds.
EASY TO USE: Simply place your fingers on the sensors—no wires, patches or gels required.
Works with most smartphones & tablets. See compatible devices below. Not recommended for use with pacemakers and ICDs and does not detect heart attack

Product: Pevor Portable Rechargeable 12 inch-Resolution Color Screen 6 Parameter Monitors
Brand: Pevor
Relevance: Exact
Specification:
E--G:
Lead Mode:5 Leads ( R, L, F, N, C or RA, LA, LL, RL,V)
Lead selection:I, II, III, avR, avL, avF, V,
Waveform:2 ch
Lead mode:3 Leads ( R, L, F or RA, LA, LL)
Lead selection:I, II, III,
Waveform:1 ch
Gain:´2.5mm/mV, ´5.0mm/mV, ´10mm/mV, ´20mm/mV, auto
HR and Alarm
Range:
Adult:15 ~ 300 bpm
Neo/Ped:15 ~ 350 bpm
Accuracy:± 1% or ± 1bpm,which great
Resolution:1 bpm
Sensitivity:> 200 (uV P-P)
Differential Input Impedance:> 5 MΩ

RESPARATION:

Method:Impedance between R-F(RA-LL)
Differential Input Impedance:>2.5 MΩ
Measuring Impedance Range:0.3~5.0Ω
Base line Impedance Range:0 – 2.5 KΩ
Bandwidth:0.3 ~ 2.5 Hz
R-esp. Rate
Measuring and Alarm Range
Adult:0 ~ 120 rpm
Neo/Ped:0 ~ 150 rpm
Resolution:1 rpm
Accuracy:±2 rpm
Apean Alarm:10 ~ 40 S

Pulse Rate:

Measuring and Alarm Range:0~254bpm
Resolution:1bpm
Accuracy:±2bpm
I.5TEMP measurement:
Channel:1
Measuring and Alarm Range:0 ~ 50 °C
Resolution:0.1°C
Accuracy:±0.1°C
Actualization interval: about 1 Sec.
Average Time Constant: < 10 Sec.
Sp(0)2:

Measuring Range:0 ~ 100 %
Alarm Range:0 ~ 100 %
Resolution:1 %
Accuracy: 70% ~ 100% ±2 %; 0% ~ 69% unspecified
Actualization interval:about 1 Sec.
Alarm Delay:10 Sec.

About N(1)BP:

Method:Oscillometric
Mode:Manual, Auto, STAT
Measuring Interval in AUTO Mode: 1, 2, 3, 4, 5, 10, 15, 30, 60, 90, 120, 180, 240,480 (Min)
Measuring Period in STAT Mode:5 Min
Pulse Rate Range:40 ~ 240 bpm
Alarm Type:SYS, DIA, MEAN
12 inch large LCD screen for convenient to read. Dr-ug solubility calculation function, ArrhythmiA analysis function.
High quality and easy to operate, comes with manual. The back of the machine is designed with a storage box.
Can be arbitrarily set the alarm limit, automatic sound and light alarm.
Built-in lithium battery, AC and DC, can provide long-term monitoring after power failure.
100%GUARANTEE-We provide a 30days no reason refund policy, make you buy with confidence! Order now!