Do Slit-Lamp Shields and Face Masks Protect Ophthalmologists amidst COVID-19?
Authors: Sze Chuan Ong, Muhammad Azri Bin Razali, Lyana Shaffiee, Jun Xing Yap, Terrence Tay Yin Fei, Seng Chee Loon, Victor Koh
Journal: Ophthalmology (2020)

Severe acute respiratory syndrome coronavirus 2 (SARS-CoV-2) is transmitted primarily via respiratory droplets, contact with contaminated surfaces, or free-floating aerosols.1Burki T.K. Coronavirus in China.Lancet Respir Med. 2020; 8: 238Abstract Full Text Full Text PDF PubMed Scopus (110) Google Scholar,2van Doremalen N. Bushmaker T. Morris D.H. et al.Aerosol and surface stability of SARS-CoV-2 as compared with SARS-CoV-1.N Engl J Med. 2020; 382: 1564-1567Crossref PubMed Scopus (6785) Google Scholar The American Academy of Ophthalmology recommends the use of surgical masks and commercially available slit-lamp shields (Breath Shields; Carl Zeiss AG, Oberkochen, Germany).3American Academy of OphthalmologyImportant coronavirus updates for ophthalmologists.https://www.aao.org/headline/alert-important-coronavirus-contextDate: 2020Google Scholar However, a lack of evidence exists regarding the true efficacy of slit-lamp shields. We attempted to replicate the spread of infected aerosols and large droplets in the clinical setting of a slit-lamp examination to evaluate the efficacy of protective equipment in reducing the risk of viral transmission. This study adhered to the Declaration of Helsinki and institutional review board approval was not required. Aerosols were defined as smaller light particles that remain suspended in the air because of slowly settling velocity, whereas large droplets were defined as heavier particles that fall rapidly after a downward trajectory.4Kutter J.S. Spronken M.I. Fraaij P.L. et al.Transmission routes of respiratory viruses among humans.Curr Opin Virol. 2018; 28: 142-151Crossref PubMed Scopus (402) Google Scholar The experimental setup (Fig S1, available at www.aaojournal.org) consisted of a slit lamp (B900 Slit Lamp; Haag-Streit Holding AG, Köniz, Switzerland), a mannequin face that represented the ophthalmologist, and a spray bottle at the chin rest that represented respiratory particle production from the patient. A particle produced by the spray bottle had a peak velocity of 4.0 meters/second and a maximum horizontal distance of 2.35 m, which was comparable to particle behavior by coughing or sneezing. A high-speed camera capturing 1000 frames/second (Chronis 1.4; Kron Technologies, Inc., Burnaby, Canada) was used for video recordings (Fig 1). This process was repeated for 3 simulations: (1) no protective equipment; (2) commercially available slit-lamp breath shield installed; and (3) mask placed in front of the spray bottle (Fig S1). For simulation 3, 5 types of masks were used: an N95 respirator (N95 particulate respiratory 8210; 3M, Alexandria, MN), 3 surgical masks of different brands and bacterial filtration efficiencies ranging from 95% to 99%, and a cloth mask (bacterial filtration efficiency, 55%). The outcome measure for aerosol transmission was the number of aerosol particles in a predefined region (Fig 1, rectangle area bordered in red [31.2 × 19.6 mm]). In total, we included 26 consecutive frames (6 ms apart) from the video recordings of each simulation. Two trained graders (Y.J.X., T.T.Y.F.) independently counted the number of aerosol particles within this region, with the mean of the 2 used as the final count. A 1-way analysis of covariance test was used to compare the number of particles in this region for each simulation. To determine the risk of large droplet transmission, identical simulations were repeated with Glo Germ liquid (Glo Germ Company, Moab, UT). The slit lamp, table, and mannequin were examined under ultraviolet A light for fluorescent droplets. In simulation 1 (Fig 1A), aerosols remained suspended in the air, with the highest density anterior to the mannequin's mouth and nose. This density was reduced in simulation 2 (Fig 1B). In simulation 3 (Fig 1C), no particles could be observed for all 5 types of masks. The mean ± standard deviation number of particles in the region of interest was 42.7 ± 34.5 for simulation 1, 12.3 ± 5.7 for simulation 2, and 0.0 ± 0.0 for simulation 3 (P < 0.001; Fig S2, available at www.aaojournal.org). Post hoc analysis showed that simulation 3 had a statistically significantly lower aerosol count than simulation 2, which in turn had a lower aerosol count than simulation 1 (P < 0.05). Hyperfluorescent areas were found on the lower half of the mannequin, slit lamp, and table for simulation 1. In simulation 2, hyperfluorescent areas were seen on the mannequin's neck, the shield, the slit lamp, and the table. In simulation 3, the hyperfluorescent area was observed only on the inner surface of the masks (Fig 1). The close proximity between the ophthalmologist and the patient increases risk of respiratory transmission of virus.4Kutter J.S. Spronken M.I. Fraaij P.L. et al.Transmission routes of respiratory viruses among humans.Curr Opin Virol. 2018; 28: 142-151Crossref PubMed Scopus (402) Google Scholar With or without the slit-lamp shield, aerosols congregated at the highest density in the region of the ophthalmologist's nose and mouth. Because SARS-CoV-2 remains viable in aerosols for hours,2van Doremalen N. Bushmaker T. Morris D.H. et al.Aerosol and surface stability of SARS-CoV-2 as compared with SARS-CoV-1.N Engl J Med. 2020; 382: 1564-1567Crossref PubMed Scopus (6785) Google Scholar a high concentration of aerosols poses an inhalation threat to ophthalmologists. Although slit-lamp shields reduced the risk of large-droplet transmission, they were limited in protection against infectious aerosols that were present on the side of the shield facing the mannequin (Fig 1B). This could be because of minor leaks at the oculars of the slit lamp where the shield is attached. Studies have demonstrated a lower efficacy of face shields against smaller aerosols because they can travel around the face shields.5Roberge R.J. Face shields for infection control: a review.J Occup Environ Hyg. 2016; 13: 235-242Crossref PubMed Scopus (132) Google Scholar When using a slit-lamp shield alone, the slit lamp, shield, and table can act as fomites for contact transmission; proper disinfection practices are needed because SARS-CoV-2 has surface stability for up to 72 hours.2van Doremalen N. Bushmaker T. Morris D.H. et al.Aerosol and surface stability of SARS-CoV-2 as compared with SARS-CoV-1.N Engl J Med. 2020; 382: 1564-1567Crossref PubMed Scopus (6785) Google Scholar Our study showed that all 5 types of face masks on patients provided the most complete protection against aerosol and droplet transmission. This could be explained because bacterial filtration efficiency was a measurement of particles filtered out at a pore size of 1.0 to 5.0 μm, which was too small to be detected by the camera. Ma et al6Ma QX, Shan H, Zhang HL, et al. Potential utilities of mask-wearing and instant hand hygiene for fighting SARS-CoV-2. J Med Virol. 2020; Mar 31. https://doi.org/10.1002/jmv.25805. Online ahead of print.Google Scholar demonstrated that surgical masks can block up to 97% of SARS-CoV-2 via aerosol transmission. However, some studies involving human participants demonstrated poor efficacy of face masks, which could be attributed to poor facial fit, leading to leakage of infectious particles. In our study, the mask provided complete coverage over the small spray nozzle, which could have accounted for a better level of protection. Limitations included technological constraints of the camera, which might have missed smaller aerosol particles. Statistical significance in reduction referred exclusively to count consistency and was not a direct estimate of reduced infection risk. The use of a spray bottle to simulate droplet or aerosol production from the patient cannot exactly replicate the actions of coughing or sneezing. Further studies are needed to interrogate the efficacy of shields and masks against viable SARS-CoV-2 in respiratory particles exceeding the minimal infectious dose. These ideally should be examined with a validated cough aerosol simulator. In conclusion, slit-lamp shields provide added protection for ophthalmologists but should be used together with other forms of personal protective equipment. All patients should wear face masks because of their efficacy in reducing aerosol and droplet transmission. Download .pdf (12.42 MB) Help with pdf files Fig S1 Download .pdf (.0 MB) Help with pdf files Fig S2