Product: NOYITO USB Logic Analyzer Device with USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug
Brand: NOYITO
Description
 Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz;
 The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions;
 8-channel;
 Supports the latest version: Version 1.1.15
 
 Voltage range:
 Input voltage range: -0.5V to 5.25V
 Input Low Voltage: -0.5V to 0.8V
 Input High Voltage: 2.0V to 5.25V
 Works with 5V, 3.3V, 2.5V, 2.0V systems. May work with 1.8V but not recommended.
 Input Impedance: 1Mohm || 10pF (typical, approximate)
 Crystal: ±20ppm, 24MHz
 Error/Accuracy: pulse-width measurement: ±42ns (at 24MHz).
 
 Packing Included:
 1xLogic Analyser
 1xUSB Cable
 1xDupont Cable
 
 Download Address: 
 https://www.amazon.com/clouddrive/share/NFh0JRc8M2JpCevtrtNEl2WQ2Bblp5Uo3HnLFdsSjEA

The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions; 8-channel.

Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz;

The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions;

Input voltage range: -0.5V to 5.25V; Input Low Voltage: -0.5V to 0.8V; Input High Voltage: 2.0V to 5.25V

Input Impedance: 1Mohm || 10pF (typical, approximate); Crystal: ±20ppm, 24MHz.

Search query: usb protocol analyzer
Relevance: Exact