Product: Digital Insulation Resistance Tester DCV ACV Meter Megohmmeter Meter VC60B+ with Unit Symbol Display
Brand: Vici
Color: Dark gray-yellow
Insulation resistance text range:
 0~200MΩ (250V±10% test voltage, Resolution: 100kΩ, Basic accuracy: ±(4.0%+2))
 0~200MΩ (500V±10% test voltage, Resolution: 100kΩ, Basic accuracy: ±(4.0%+2))
 0~2GΩ (1000V±10% test voltage, Resolution: 1MΩ, Basic accuracy: ±(4.0%+2))
 Resistance text range: 0~2kΩ (Basic accuracy: ±(1.0%+5), Resolution: 1Ω)
 DCV text range: 0~1000V (Basic accuracy: ±(1.0%+5), Resolution: 1V)
 ACV text range: 0~750V (Basic accuracy: ±(1.5%+5), Resolution: 1V)
 
 
 VC60B+ Special function:
 Continuity alarm Less than 50Ω±10Ω
 Low battery indication Less than approx.7.5V 
 Data hold √
 LCD backlight Approx.10Sec.
 Function protection √ 
 Input impedance 10MΩ
 Sampling rate 3 times/Sec.
 AC frequency response 40-400Hz
 Operation mode Manual range
 Max.display 1999
 LCD size 2.95 x 2inch
 Power source 1.5V AA battery x6pcs, or external DC 9V power source(Not include battery)
 Power consumption Quiescent current approx.3mA
 
 General characteristic:
 Color: Dark gray-yellow
 Meter size: 6.9(L)*4.6(W)*2.5(D)
 Standard accessory: Test lead, shoulder strap, operation manual
 
 Package include:
 
 1pc x VICHY VC60B+ multimeter
 1pc x Measurement Cables with Clips (1 pair)
 1pc x Operation Manual
 1pc x Shoulder strap

It is an accurate, reliable insulation tester. Easy to use.
Design of large LCD display and unit symbol display. Unit symbol display, easy to use.
It has Strong anti-magnetic and anti-jamming function.
Full function protection, anti high voltage circuit design. High voltage and short alarm input.
Dual Alarm: sound and light. When a shortcut resistance occurred the display returns to zero to prevent burning underground.

Search query: megohmmeter
Relevance: Exact