Product: ETCR ETCR2000A+ Clamp On Digital Ground Resistance Meter Tester 0.01-200Ω
Brand: ETCR

The lead time is about 3 business days, Fedex/DHL shipping time is 5-7 days. 
Specifications:
 
 Function:Ground resistance test, Loop resistance test 
 Power Supply:6VDC 4 PCS of LR6 lkaline batteries 
 Resistance Range:0.01-200Ω 
 Resistance Resolution:0.001Ω 
 Highest Resistance Accuracy:±1%±0.01Ω 
 Data memory:99 Sets 
 Alarm Threshold Setting Range:Resistance: 1-199Ω 
 Alarm Sound and Light:“Beep-beep-beep” alarm tone, Alarm Symbol Flashing, Press AL key to turn on, turn off Alarm or set Alarm Threshold. 
 Interference Signal Recognition:“Beep-beep-beep” tone, NOISE symbol blinking 
 Working Temperature and Humidity:-20C-55C20%RH-90%RH 
 LCD Screen:4 Digits LCD display, Screen L×W: 47mm×28.5mm 
 Clamp Size: 65mm×32mm 
 Clamp Opening Extent: 32mm 
 Tester Weight:1160g (With batteries) 
 Clamp Dimension: L×W×T:285mm×85mm×56mm 
 Protection Level:Dual Insulation 
 Structural Feature:Clamp CT 
 Shifting:Automatic shifting 
 External magnetic field:<40A/m 
 External electrical field:<1V/m 
 Single measurement time:0.5 second 
 Resistance measurement frequency:>1KHz 
 Measured current frequency:50/60Hz Automatic measurement 
 Low Battery Indication:Low battery symbol displays when battery is lower then 5.3V 
 Memory Full Indication:MEM symbol blinks when all 99 sets memory is full 
 Over Range Indication:Out of range indication “OLΩ”, "L0.01” or “OL A” 
 Auto Power-off:5 minutes after Boot up 
 Power Consumption:Max 50mA either in booting up or normally working, Continuously working for 30 hours
Self-test in 0.5 second, boot instantly
0.01-200Ω, Wider range
With beeps and NOISE symbol blinking. Strong anti-interference ability.
Maximum 50mA, Longer battery life
With fastening screws, Secure, Reliable, Easy battery replacement.

Search query: earth ground meter
Relevance: Substitute