Product: Elitech RC-5+ USB Temperature Data Logger Reusable Recorder PDF DDL
Brand: Elitech
Color: RC-5+
Elitech RC-5+ USB temperature data features a plug-and-play USB port interface. It allows for faster access to data collected in any cold chain management process. This could be in pharmaceuticals, food, life science, cooler boxes, medical cabinets, fresh food cabinets, freezer or laboratories. To achieve faster data analysis on any Elitech USB data logger, you can export data in PDF/Excel via data management software. Alternatively, you can use the multi-function LCD. For convenience and reliability, Elitech USB temperature data logger features a replaceable battery, optional mounting bracket, double button operation and double bracket. Its robust and compact design satisfies the dynamic demands in any sensitive storage and transportation system. It records real time temperature that is stored in non-volatile memory for access via the USB port. It has 0.1℃ resolution and ±0.5℃ accuracy. Specifications:  1)Temperature measuring range: -30 ℃~+70℃(-22℉~158℉); Resolution: 0.1 ℃   2)Accuracy: ±0.5 ℃(-20℃~+40℃); others, ±1 ℃   3)Temperature unit: ℃ or ℉ optional   4)Record capacity: 32000 points (MAX)   5)Record interval: 10s-24hour adjustable   6)Sensor: Internal NTC thermal resistor   7)Communication interface: USB interface  8)Power supply: CR2032 battery or power supply via USB interface, and it could upload data even if the battery is out of use   9)Battery life: in normal temperature, if the record interval sets as 15 minutes, it could be used above one year    Package Included: 1 x Data Logger 1 x CR2032 battery 1 x User Manual
REUSABLE TEMPERATURE DATA LOGGER. Wide temperature measuring range -22℉~158℉(-30°C ~+70°C). Highly temperature accuracy (±0.9℉). Perfect for the transportation and storage of pharmaceuticals, frozen food, fresh food, vegetable, fish, and so on.
AUTO PDF/EXCEL REPORT. Built-in USB port, generate PDF report automatically after connecting to PC or Android Phone, no software needed. Encryptable via PC.
LCD VISUAL DISPLAY. Press the button to check key information: current temperature value, Max or Min value since recording, Current Date, Logging points.
MULTIPLE START OPTION & ALARM SETTINGS. Start options: Press the button on the data logger; Pre-set time start or Start delay on Elitechlog software; Alarm Settings: Support up to 5 alarm settings, safeguard the valued items all over. 【Compared to Elitech RC-5】Restart by button pressing. Lower power consumption: Battery life up to 6 months. The external temperature sensor is optional. Temperature record points up to 32,000 points. IP 67 waterproof protection.
FREE ELITECHLOG SOFTWARE FOR WINDOWS & MACOS. Parameters programmed before leaving the factory: Log interval - 2 mins; Press button to start. The desired parameters also can be set on Elitechlog software: Temperature unit - ℃/℉, Log interval, Temperature alarm range, etc. 24/7 US Technician Support via Email and Phone.

Search query: usb temperature monitor
Relevance: Substitute