Product: KAIWEETS Digital Clamp Meter T-RMS 6000 Counts, Multimeter Voltage Tester Auto-ranging, Measures Current Voltage Temperature Capacitance Resistance Diodes Continuity Duty-Cycle (AC Clamp Meter)
Brand: KAIWEETS
Key Features: 1. T-RMS True RMS AC voltage and current for accurate measurements on non-linear signals. 2. 6000 Counts The maximum measurable value is 5999, which is enough for industrial use. It's a professional clamp meter with high accuracy. 3. Auto Power Off No operation in 15 minutes, the Meter will beep 3 times, indicating for Aoto Power Off. 4. LCD Display The easy-to-read LCD display gives you clear measurements. 5.Safty Rating CAT III 600 V safety rating. Specifications: Color: Black & Red AC Current (A): 60A/600A ±(2.5%+5) AC Voltage (V): 6V/60V/600V ±(1.0%+3) DC Voltage (V): 600mV/6V/60V/600V ±(0.5%+3)  Resistance (Ω): 600Ω/6kΩ/60kΩ/600kΩ/6MΩ/60MΩ ±(1.0％+3） Capacitance (F): 10nF/100nF/1μF/10μF/100μF/1mF/10mF/100mF ±(4.0％+3） Frequency (Hz): 10Hz~10MHZ ±(0.5%+5) Temperature: -20℃-1000℃/-4℉-1832℉ NCV: Yes LPF: Yes Low-z: Yes Max/Min: Yes Data Hold: Yes Diode Test: Yes Flashlight: Yes LCD Backlight: Yes Continuity Buzzer: Yes Over Range Indication: Yes Low Battery Indication: Yes Power: 2 x AAA 1.5V Battery (Not Included) Weight :187g/6.55oz Size:193x73x34mm/7.6x2.87x1.34in Package: 1 x Digital Clamp Meter 1 x Temperature Probe 2 x Test Leads (A Pair) 2 x AAA 1.5V Batteries 1 x Soft Carrying Case 1 x User Manual
True-RMS clamp meter: This voltage meter can accurately measure AC Current, AC/DC Voltage, Frequency or Duty Cycle, Resistance, Capacitance and also provides Diode, Continuity and Temperature tests. True RMS for accurate measurements on non-linear loads. Easy to use for labs factories and household.
LowZ & LPF: Low input impedance helps prevent false readings due to ghost voltage. Low Pass Filter (LPF) for accurate measurement of variable frequency drive signals when measuring motors and transformer. To measure the current you need to clamp the meter around one of the wires and not the whole power cord.
NCV Defection Function: Non-contact voltage detection effectively check electrical status through measuring EM field intensity without touching any wires; The NCV button lies on the right side. Simply place your thumb on it and push.
Multi-function: Two-color backlit LCD Display - When the voltage > 80V, the current > 3A, the LCD backlit screen turns red to warn you of high voltage/ LED Flashlight / Temperature Measurement for liquids and air conditioning ports included K-Type Thermocouple / Low Battery Indication / Auto power off after 15 minutes inactivity.
Safety Service: Safety standard IEC 61010-1, CAT III 600V. Double insulation. 36-Month product service provided. 7/24 Customer Service.

Search query: voltage meter
Relevance: Exact