CVE-2021-44464
Assigner: icscert
Published: 2022-01-21T18:17:42
State: PUBLISHED
affected: product: Vigilant Software Suite (Mastermed Dashboard)
vendor: Fresenius Kabi
versions: lessThan: 2.0.1.3
status: affected
version: unspecified
versionType: custom
credits: value: Julian Suleder (ERNW Research GmbH), Nils Emmerich (ERNW Research GmbH), Raphael Pavlidis (ERNW Research GmbH), and Dr. Oliver Matula (ERNW Enno Rey Netzwerke GmbH) reported these vulnerabilities to the German Federal Office for Information Security (BSI) in the context of the BSI project ManiMed (Medical Device Manipulation Project).
descriptions: value: Vigilant Software Suite (Mastermed Dashboard) version 2.0.1.3 contains service credentials likely to be common across all instances. An attacker in possession of the password may gain privileges on all installations of this software.
metrics: cvssV3_1: attackComplexity: LOW
attackVector: NETWORK
availabilityImpact: LOW
baseScore: 6.3
baseSeverity: MEDIUM
confidentialityImpact: LOW
integrityImpact: LOW
privilegesRequired: LOW
scope: UNCHANGED
userInteraction: NONE
vectorString: CVSS:3.1/AV:N/AC:L/PR:L/UI:N/S:U/C:L/I:L/A:L
version: 3.1
problemTypes: descriptions: cweId: CWE-798
description: CWE-798 Use of Hard-coded Credentials
type: CWE
solutions: value: Fresenius Kabi has created new versions to address these vulnerabilities:
•	Link+ v3.0 (D16 or later)
•	VSS v1.0.3 (or later)
•	Agilia Connect Pumps Wifi Module (D29 or later)
•	Agilia Connect Partner v3.3.2 (or later) 
Fresenius Kabi has initiated communication on this topic in April 2021 with customers to inform them about availability of the new versions in corresponding countries. Contact Fresenius Kabi online or by phone at 1-800-333-6925 for more information on updating to the newest version.
Fresenius Kabi also identified that early Link+ devices (approximatively 1200 devices) would need hardware change to support D16 or later firmware. Until those devices can be replaced in customers’ installation, Fresenius Kabi recommends users to rely on CISA recommendation. 

source: advisory: ICSMA-21-355-01
discovery: EXTERNAL
title: Fresenius Kabi Agilia Connect Infusion System hard coded credentials
workarounds: value: CISA recommends users take defensive measures to minimize the risk of exploitation of these vulnerabilities. Specifically, users should:
•	Minimize network exposure for all control system devices and/or systems, and ensure they are not accessible from the Internet.
•	Locate control system networks and remote devices behind firewalls and isolate them from the business network.
•	When remote access is required, use secure methods, such as virtual private networks (VPNs), recognizing VPNs may have vulnerabilities and should be updated to the most current version available. Also recognize VPN is only as secure as its connected devices.

x_generator: engine: Vulnogram 0.0.9