reglab/statecodes huggingface dataset:
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:10 - General safety provisions to protect persons from radiation exposures

Current through Register Vol. 51, page 10, July 15, 2024
The licensee shall be responsible for directing the operation
				  of any X ray system under the licensee's administrative control. The licensee
				  or the licensee's agent shall provide that:
(1) No X ray system may be operated for
				diagnostic purpose unless the system meets the provisions of this
				chapter;
(2) Any person who is
				operating the X ray system is adequately instructed in the safe operating
				procedures and competent in the safe use of the equipment commensurate with the
				size, scope, and nature of the service. Any such person shall be instructed and
				demonstrate competence in subjects outlined in §
									44:03:01:14.01;
(3) Procedures and auxiliary equipment
				utilized produce images of good diagnostic quality with minimum patient
				radiation exposure;
(4) Digital
				imaging equipment is consistent with the diagnostic objective of the
				examination;
(5) The film screen
				combinations shall be the fastest speed consistent with the diagnostic
				objective of the examination;
(6) A
				technique chart or manual is located in the vicinity of the control panel of
				each machine that specifies, for all diagnostic examinations performed with
				that system, the following:
(a) The technique
				factors to be used that are specific to a patient's anatomical part, size, or
				age (for pediatrics), except for any system that has only automatic X ray
				exposure controls;
(b) The type of
				film-screen combination to be used;
(c) The type of grid to be used, if
				any;
(d) The SID to be used, except
				for dental and all other fixed SID radiographic equipment;
(e) The type and placement of patient
				shielding to be used;
(f) The
				routine views for all procedures done with each machine; and
(g) For mammography, an indication of
				kVp/target/filter combination;
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:10.01 - Requirements for personal protective devices

Current through Register Vol. 51, page 10, July 15, 2024
Special personal protective devices shall be used to protect
				  eyes, skin, bone, and certain organs from unnecessary radiation exposure when
				  possible These protective devices must be readily accessible and in good
				  working condition.
A gonad shield of not less than 0.5 millimeters lead
				  equivalent material must be used for human patients who have not passed the
				  reproductive age during radiographic procedure in which the gonads are in the
				  useful beam, unless the shield would interfere with the diagnostic
				  procedure.
Protective equipment shall be checked annually for defects to
				  assure reliability and integrity. A record of this test shall be made and
				  maintained. If such defect is found, equipment shall be replaced or removed
				  from service until repaired.
Mechanical-holding devices shall be used when the technique
				  permits. The individual holder shall be protected and no individual may be used
				  routinely to hold the image receptor or patient. Written safety procedures
				  shall indicate the requirements for selecting an individual to be a holder and
				  the procedure the holder shall follow.
General Authority:
SDCL
					 34-21-4.1,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-15,
					 34-21-25.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:10.02 - Operator protection requirements

Current through Register Vol. 51, page 10, July 15, 2024
Operators of fixed medical radiographic units shall be within
				  a shielded area large enough to provide protection from unattenuated direct
				  scatter or stray radiation originating from the table or wall-mounted image
				  receptor. The booth walls shall be permanently fixed barriers of at least two
				  meters (seven feet) high.
Operators of dental radiographic units must comply with
				  §
									44:03:01:08.06.
A lead glass patient-viewing window, mirrors, closed circuit
				  television, or an equivalent system must be available to permit the operator to
				  continuously observe the patient during exposure. If a patient-viewing window
				  is used, it must be a minimum of one square foot and must be located at least
				  eighteen inches from the edge of the control booth for any new construction and
				  any renovation, addition or change in space use of existing facilities. The
				  exposure switch must be permanently mounted so that it cannot be conveniently
				  operated outside the shielded area.
For mobile and portable X ray systems to be used less than
				  one week in the same location, the control must be positioned so that the
				  operator is at least six feet away from the tube housing and the patient during
				  an exposure and is not exposed to greater than two millirems (0.02
				  millisievert) in any one hour.
For mobile and portable X ray systems to be used more than
				  one week in the same location, the operator must be provided with a movable
				  protective barrier at least 6.5 feet high, thirty inches wide, and a lead glass
				  viewing window.
General Authority:
SDCL
					 34-21-4.1,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-15.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:10.03 - Reports of incidents involving radiation sources

Current through Register Vol. 51, page 10, July 15, 2024
Any radiation incident considered a potential hazard shall be
				  reported to the department by the licensee within 24 hours by telephone or fax
				  at the number shown on the license during normal business hours or the first
				  workday following a holiday or weekend.
General Authority:
SDCL
					 34-21-4.1,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-15,
					 34-21-30.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:10.04 - Requirements of X ray film processing and darkroom

Current through Register Vol. 51, page 10, July 15, 2024
Each installation using a radiographic X ray system and using
				  radiographic film shall have available suitable equipment for handling and
				  processing radiographic film in accordance with the following:
(1) Any manual processing tank shall be
				constructed of mechanically rigid corrosion resistant material. The temperature
				of solutions in a tank shall be maintained within the range of 60 to 80 degrees
				Fahrenheit, inclusive. Film shall be developed in accordance with the
				time-temperature relationship recommended by the film manufacturer. A
				thermometer and timer shall be utilized to indicate the actual temperature of
				the developer and signal the passage of a preset time appropriate to the
				developing time required.
(2)
				Automatic processing of the film shall be in accordance with the
				time-temperature relationships recommended by the film and processor
				manufacturers. A control program to maintain the automatic processor operating
				parameters according to manufacturer's recommendations must be used;
(3) Film storage shall be provided and the
				film stored according to manufacturer's recommendations as to protection of
				radiation, heat, humidity, and storage position;
(4) Safelighting in the film
				processing-loading area shall be with the filter, bulb wattage, and distances
				recommended by the film manufacturer for film emulsion used by the facility.
				The safelighting shall prevent a pre-exposed film from increasing in density
				greater than 0.10 when exposed for two minutes with the safelights
				on;
(5) Pass boxes, if provided,
				shall be so constructed as to exclude light from the darkroom when cassettes
				are placed in or removed from the boxes, and shall incorporate adequate
				shielding from stray radiation to prevent exposure of undeveloped
				film.
General Authority:
SDCL
					 34-21-4.1,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-15.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:10.05 - Quality assurance program requirements

Current through Register Vol. 51, page 10, July 15, 2024
The licensee shall have a written, on-going quality assurance
				  program specific to the equipment and procedures that are performed in the
				  facility to ensure consistent high-quality images with minimum patient
				  exposure. The tests performed for quality control purposes shall be included in
				  a log containing acceptability limits, results of tests, date, initials of
				  operator or testing individual, and corrective action taken, if needed. Tests
				  for film processing shall include temperature, chemical replacement, processor
				  operating parameters, and darkroom fog, and be performed on a routine basis.
				  Any quality control test done on diagnostic tubes shall be done annually and
				  include SID accuracy, X ray and light field alignment, X ray and bucky
				  alignment, and collimator dial accuracy. All dental intraoral, panoramic,
				  tomography, and machines that have fixed SID and collimator are excluded from
				  SID accuracy, X ray and light field alignment, X ray and bucky alignment, and
				  collimator dial accuracy.
A qualified expert shall perform measurement of the
				  reference air kerma rate on medical fluoroscopic equipment for both typical and
				  maximum values. The measurements shall be made annually or after any
				  maintenance of the system which might affect the radiation exposure rate and
				  the results shall be posted where any operator may have ready access to them
				  while using the fluoroscopic equipment. Results of the measurements shall
				  include:
(1) The roentgen per minute
				or milliGray per minute;
(2) The
				technique factors used to determine such results;
(3) The name of the person performing the
				measurements; and
(4) The date the
				measurements were performed.
General Authority:
SDCL
					 34-21-4.1,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-15.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:11.01 - Shielding plan review prior to installation of radiation facilities or equipment

Current through Register Vol. 51, page 10, July 15, 2024
Prior to construction, the floor plans, shielding
				  specifications, and equipment arrangement of any new installation or any
				  modification of existing installations utilizing ionizing radiation machines
				  shall be submitted to the department for review and approval. The plans must
				  show:
(1) The normal location of the
				system's radiation port, the general direction of the useful beam, the location
				of any windows and doors or other openings, the location of the operator's
				booth, and the location of the control panel;
(2) The structural composition and thickness
				or lead equivalent of all walls, doors, partitions, floor, and ceiling of the
				room concerned;
(3) The dimensions
				of any room concerned;
(4) The type
				of occupancy of any adjacent areas inclusive of space above and below the room
				concerned. If there is an exterior wall, the plans must show distance to the
				closest area where it is likely that individuals may be present;
(5) The make and model of the equipment, the
				maximum technique factors, and the energy waveform; and
(6) The type of examinations or treatments
				which will be performed with the equipment and the anticipated workload of the
				system in mA-minutes per week.
The department may require the applicant to utilize the
				  services of a qualified physicist to determine the shielding requirements prior
				  to the plan review and approval. The approval of such plans may not preclude
				  the requirement for additional modifications should a subsequent change of
				  operating conditions create the possibility of an individual receiving a dose
				  in excess of the limits.
General Authority:
SDCL
					 34-21-4.1,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-17.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:11.02 - Installation requirements of all radiation equipment

Current through Register Vol. 51, page 10, July 15, 2024
No person may make, sell, lease, transfer, lend, assemble, or
				  install any radiation machine or any supplies used in connection with such a
				  machine unless such supplies and equipment meet the requirements of this
				  chapter when placed in operation.
General Authority:
SDCL
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-17.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:11.03 - Notification of installation of radiation equipment

Current through Register Vol. 51, page 10, July 15, 2024
Any assembler who installs radiation machines in this state
				  shall report to the department within fifteen days the following
				  information:
(1) The name and address
				of the persons who have received these machines;
(2) The manufacturer, model, and serial
				number of each radiation machine and major components transferred;
(3) The date of transfer of each radiation
				machine; and
(4) The date of
				installation of each radiation machine.
In the case of diagnostic X ray systems which contain
				  certified components, the licensee must submit a copy of the assembler's
				  report, Form FDA 2579 (6/95), prepared in compliance with requirements of the
				  Federal Diagnostic X Ray Standard, 21 C.F.R. 1030.30(d), as of April 1, 2022,
				  to the department within fifteen days following completion of the
				  assembly.
For the purposes of this section, assembler means any person,
				  including the owner of the X ray system or the owner's employee or agency,
				  engaged in the business of assembling, replacing, or installing one or more
				  components into an X ray system or subsystem that is subsequently used to
				  provide professional or commercial services.
General Authority:
SDCL
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-17.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:11.04 - Licensee requirements after installation of radiation equipment

Current through Register Vol. 51, page 10, July 15, 2024
After completion of construction and installation of
				  radiation equipment, the licensee shall maintain for inspection:
(1) Model and serial number of all major
				components and user's manuals for those components;
(2) Records of surveys, calibrations,
				maintenance, and modifications performed on the X ray systems;
(3) The maximum rated technique factors of
				each machine with tube rating and cooling curves charts; and
(4) Scale drawing of the room in which a
				stationary radiation machine system is located with such drawing indicating the
				use of areas adjacent to the room and an estimation of the extent of occupancy
				by an individual in such areas; the results of a survey for radiation levels
				present at the operator's position and at pertinent points outside the room;
				and the type and thickness of materials, or lead equivalency, in each
				protective barrier used in the room.
For the purpose of this section, cooling curve is the
				  geographical relationship between heat units stored and cooling
				  time.
General Authority:
SDCL
					 34-1-17,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-17.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:11.05 - Installation, maintenance, and operation of radiation equipment

Current through Register Vol. 51, page 10, July 15, 2024
Equipment shall be installed, maintained, and operated
				  according to the manufacturer's specifications.
General Authority:
SDCL
					 34-1-17,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-17.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:12.01 - Radiation producing equipment calibration

Current through Register Vol. 51, page 10, July 15, 2024
The licensee shall provide that calibrations are performed on
				  a diagnostic radiographic system if that system does not meet the minimum
				  performance criteria specified in §§
									44:03:01:08.02 to 44:03:01:08.07,
				  inclusive, and if there is any change or replacement of components that could
				  cause a change in the radiation output of that system.
The calibration may not exceed three months after any change
				  or replacement of components that could cause a change in the radiation output.
				  The calibration of the radiation output of the X ray system shall be performed
				  by or under the direction of a qualified expert. Calibration of the radiation
				  output of an X ray system shall be performed with a calibrated dosimetry
				  system. Any X ray machine shall be surveyed by the department of a qualified
				  expert every three years to verify the X ray machine meets the standards of
				  this chapter. Any X ray machine not meeting the standards of this chapter must
				  be calibrated. Any computed tomography system shall be surveyed by a medical
				  physicist prior to clinical use and on an annual basis.
General Authority:
SDCL
					 34-1-17,
					 34-21-4.1,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-15.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:12.02 - Surveys of radiation producing facilities and radiation equipment

Current through Register Vol. 51, page 10, July 15, 2024
Any new radiation producing facility and any existing
				  radiation producing facility shall have a radiation protection survey made by a
				  qualified expert or the department. The survey shall be done after any change
				  in the facility or equipment that might cause a significant increase in
				  radiation hazard. The licensee shall obtain a written report of the survey from
				  a qualified expert. The licensee shall transmit a copy of the report to the
				  department within thirty days of receipt of the report. The survey and report
				  shall indicate all instances where the installation, in the opinion of the
				  qualified expert or department, is in violation of the
				  regulations.
General Authority:
SDCL
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-15.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:14.01 - Operator requirements for radiation producing equipment

Current through Register Vol. 51, page 10, July 15, 2024
Any person who is certified or registered by the American
				  Registry of Radiologic Technologists, licensed by another state, or who has
				  documented eighty hours of orientation and training in the operation of
				  radiation producing equipment by a qualified instructor may operate radiation
				  producing equipment, excluding diagnostic computed tomography equipment. The
				  instructor must verify competency for each procedure the operator will be
				  performing. If an actual patient is not available, the procedure may be
				  simulated. For the purposes of complying with the provisions of electronic
				  health records certification criteria established pursuant to 45 C.F.R. 495.6,
				  in effect on September 4, 2014, a radiologic technologist certified and
				  registered by the American Registry of Radiologic Technologists or licensed by
				  another state is considered to be a licensed health care professional. Dental
				  radiographers shall have a minimum of sixteen hours of training. Chiropractic
				  radiographers shall have a minimum of thirty-six hours of training.
General Authority:
SDCL
					 34-21-4.1,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-15.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:14.02 - Operator training requirements for radiation producing equipment - Diagnostic computed tomography excluded

Current through Register Vol. 51, page 10, July 15, 2024
A qualified instructor must do all training for operators of
				  radiation producing equipment. Documentation of the training must include the
				  dates, instructor, and subjects covered. Continuing education credits would
				  qualify as part of the training. An individual must have documented training
				  for the operation of radiation producing equipment in:
(1) Fundamentals of radiation safety must
				cover characteristics of radiation, units of radiation measurement, hazards of
				exposure to radiation, levels of radiation from sources, and methods of
				controlling radiation dose;
(2)
				Familiarization with equipment must cover identification of controls, function
				of each control, how each control affects technique chart, and utilization of
				technique chart;
(3) Film
				processing must cover film speed as related to patient exposure, film
				processing with
(4) Instruction on
				computed radiography or digital radiography must cover exposure index
				values;
(5) Anatomy and positioning
				relative to scope of practice, including patient preparation, and correct
				method for performing procedures; and
(6) The requirement of federal and state
				regulations pertinent to the services offered.
General Authority:
SDCL
					 34-21-4.1,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-14.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:14.03 - Operator continuing education requirements

Current through Register Vol. 51, page 10, July 15, 2024
Any operator of a radiation producing equipment shall have
				  eight hours of documented continuing education over a two-year period
				  on:
(1) Radiation safety;
(2) Equipment operation;
(3) Image and film processing and archiving;
				and
(4) Emergency
				procedures;
(5) Anatomy;
(6) Positioning of film and body
				parts;
(7) Orientation or training
				in new developed procedures;
(8)
				Infection control; or
(9) Rules
				pertinent to the services offered.
Continuing education must be related to imaging competencies
				  performed by the operator. Practitioners of the healing arts and any employee
				  of a dental or chiropractic facility are excluded from the continuing education
				  requirement.
General Authority:
SDCL
					 34-21-4.1,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-14.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:14.04 - Operator requirements for computed tomography equipment

Current through Register Vol. 51, page 10, July 15, 2024
Diagnostic computed tomography equipment shall be operated
				  by personnel who are certified and registered by the American Registry of
				  Radiologic Technologists or certified and registered in CT by the Nuclear
				  Medicine Technologist Certification Board. This does not apply to personnel
				  using computed tomography equipment for attenuation correction or radiation
				  therapy planning.
General Authority:
SDCL
					 34-21-4.1,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-4.1,
					 34-21-14.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:15 - Disposal of radioactive materials

Current through Register Vol. 51, page 10, July 15, 2024
No licensee may dispose of any medical radioactive material
				  without prior written approval of the department unless under license of the
				  Nuclear Regulatory Commission. Data regarding the potential radium or other
				  radiation hazard are required.
General Authority:
SDCL
					 34-21-15.
Law Implemented:
SDCL
					 34-21-31.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:16 - Loss or theft of radioactive material

Current through Register Vol. 51, page 10, July 15, 2024
Each licensee shall report by telephone fax, or electronic
				  mail and followed up with a written report to the department the loss or theft
				  of any source of radiation immediately after the occurrence becomes
				  known.
General Authority:
SDCL
					 34-1-17,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-11.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:03 - Radiation safety › Chapter 44:03:01 - Radiation control › Section 44:03:01:17 - Amendment, suspension, or revocation of license

Current through Register Vol. 51, page 10, July 15, 2024
Any amendment to a license shall be made by issuing a new
				  license. Any suspension or revocation proceeding, in accordance with SDCL
				  chapter 1-26, can be initiated only after exposure surveys are conducted by the
				  department and a serious hazard to public health and safety is
				  determined.
General Authority:
SDCL
					 34-1-17,
					 34-21-15.
Law Implemented:
SDCL
					 34-21-19.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:01

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-7,
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:02

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:03

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:04

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:05

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:06

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:07

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:08

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:10

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:11

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
Scope and standards of nursing practice -- Basic role,
					 §
											20:48:04:01.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:12

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:13

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:14

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
Clinical laboratory services,
					 §44:04:10:01.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:15

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:16

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:16 - Ambulatory surgery center standards, Repealed › Section 44:04:16:17

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:19 - Adult foster care, Repealed › Section 44:04:19:01

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:19 - Adult foster care, Repealed › Section 44:04:19:02

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:19 - Adult foster care, Repealed › Section 44:04:19:03

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
"Food Guide Pyramid," 1996, USDA's Center for Nutrition
					 Policy and Promotion, United States Department of Agriculture, 1120
					 20th Street, NW, Suite 200, North Lobby, Washington,
					 DC 20036-3475. Phone: 1-202-606 -8000.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:19 - Adult foster care, Repealed › Section 44:04:19:04

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:19 - Adult foster care, Repealed › Section 44:04:19:05

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:19 - Adult foster care, Repealed › Section 44:04:19:06

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:19 - Adult foster care, Repealed › Section 44:04:19:07

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:19 - Adult foster care, Repealed › Section 44:04:19:08

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:19 - Adult foster care, Repealed › Section 44:04:19:09

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:19 - Adult foster care, Repealed › Section 44:04:19:10

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:01

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-1-17,
					 34-12-13.
Law Implemented:
SDCL
					 34-12-7.
Rules of General Applicability, ch 44:04:01, for
					 information regarding license requirements for health care facilities by the
					 Department of Health; Alcohol and Drug Abuse, art 46:05, for the operational
					 and care standards of programs accredited by the Department of Human
					 Services.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:02

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-1-17,
					 34-12-13.
Law Implemented:
SDCL
					 34-12-7.
NFPA 101 Life Safety Code,
2000 edition,
					 National Fire Protection Association. Copies may be obtained from the National
					 Fire Protection Association, P.O. Box 9101, Quincy, MA 02269-9101. Phone
					 1-800-344 -3555. Cost: $53.50.
International Building Code
, 2000 edition.
					 Copies may be obtained from International Conference of Building Officials,
					 5360 South Workman Mill Road, Whittier, California 90601-2298. Phone: (562)
					 699-0541. Cost: $69.55.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:03

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:04

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:05

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:06

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:07

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:08

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:09

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:10

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:11

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:12

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:13

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:14

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:15

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:16

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:17

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:18

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:19

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:20

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:21

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:22

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:23

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:24

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:25

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:04 - Medical facilities › Chapter 44:04:20 - Inpatient chemical dependency treatment facility, Repealed › Section 44:04:20:26

Current through Register Vol. 51, page 10, July 15, 2024
General Authority:
SDCL
					 34-12-13.
Law Implemented:
SDCL
					 34-12-13.
South Dakota Administrative Rules › Title 44 - Department of Health › Article 44:05 - Ambulance Operation › Chapter 44:05:01 - Definitions › Section 44:05:01:01 - Definitions

Current through Register Vol. 51, page 10, July 15, 2024
Words defined in SDCL
				  34-11-2 shall have the same
				  meaning when used in this article. In addition, terms used in this article
				  mean:
(1) "Ambulance service
				director," a person in charge of the day-to-day management of a ground or air
				ambulance service:
(2)
				"Attendants," ambulance personnel who have completed a course of instruction in
				emergency care approved by the department pursuant to SDCL
				34-11-6 and who are responsible
				for the rendering of direct patient care to the sick or injured:
(3) "C.F.R.," Code of Federal
				Regulations;
(4) "Department." the
				South Dakota Department of Health:
(5) "Emergency medical responder (EMR)," a
				person who has completed an emergency medical responder course approved by the
				department and who has successfully demonstrated entry-level competency through
				the NREMT and is certified by the department;
(6) "Emergency medical technician (EMT)." a
				person who has completed an emergency medical technician course approved by the
				department and who has successfully demonstrated entry-level competency through
				the NREMT and is certified by the department;
(7) "EVOC." an emergency vehicle operator
				course approved by the department consisting of a curriculum of didactic
				instruction and actual driving, developed to improve emergency vehicle driving
				skills;
(8) "Gross incompetence."
				general or severe lack, of the ability, skill, or knowledge reasonably
				necessary to serve as an ambulance driver or attendant:
(9) "Licensee," a person or entity that has
				obtained and is in possession of a currently valid license to operate a ground
				or air ambulance service in this state;
(10) "Medical air transport," the transport
				of a patient by a licensed air ambulance
(11) "Medical director," a person licensed to
				practice medicine in this state under SDCL 36-4 who is responsible for
				providing medical supervision and direction to a licensed ambulance
				service;
(12) "National Registry of
				Emergency Medical Technicians (NREMT)," the National EMS Certification Agency
				responsible for validating entry-level competency of pre-hospital
				providers;
(13) "Ground ambulance."
				a motor vehicle operated over public roadways under the jurisdiction of this
				state that is in compliance with chapter 44:05:04.
(14) "Response." the response of an ambulance
				vehicle and personnel to an emergency for the purpose of rendering medical care
				or transportation, or both?, to someone ill or injured, including cancelled
				calls, no transports and standby events where medical care may be
				rendered.
General Authority:
SDCL
					 34-11-5.
					 34-11-6.1.
Law Implemented:
SDCL
					 34-11-6.1.