reglab/statecodes huggingface dataset:
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 9 - Standards for Protection Against Radiation › Section 310:281-9-30 - Reports of theft or loss of diagnostic x-ray systems

Current through Vol. 41, No. 20, July 1, 2024
(a) Each permittee shall report to the
				Department the theft or loss of any permitted diagnostic x-ray system
				immediately after such occurrence becomes known.
(b) Each permittee who is required to make a
				report pursuant to 310:281-9-30(a) shall, within 30 days after the permittee
				learns of the loss or theft, make a report in writing to the Department,
				setting forth the following information:
(1)
				a description of the diagnostic x-ray system involved, including the kind,
				quantity, chemical, physical form, and/or model and serial numbers;
(2) a description of the circumstances under
				which the loss or theft occurred;
(3) a statement of disposition or probable
				disposition of the diagnostic x-ray system involved;
(4) radiation exposures to individuals,
				circumstances under which the exposures occurred, and the extent of possible
				hazard to persons in unrestricted areas;
(5) actions which have been taken, or will be
				taken, to recover the diagnostic x-ray system; and procedures or measures which
				have been or will be adopted to prevent a recurrence of the loss or theft of
				diagnostic x-ray system.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 9 - Standards for Protection Against Radiation › Section 310:281-9-31 - Notification of incidents

Current through Vol. 41, No. 20, July 1, 2024
(a)
Immediate notification.
Each permittee shall immediately notify
				the Department by telephone, email, or facsimile of any incident involving any
				diagnostic x-ray system used and which may have caused or threatens to cause:
(1) a dose to the whole body of any
				individual of 25 rems or more of radiation; a dose to the skin of the whole
				body of any individual of 150 rems or more of radiation;
(2) or a dose to the feet, ankles, hands, or
				forearms of any individual of 375 rems or more of radiation; or
(3) a loss of 1 working week or more of the
				operation of any facilities affected; or damage to property in excess of
				$200,000.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 9 - Standards for Protection Against Radiation › Section 310:281-9-32 - RESERVED

Current through Vol. 41, No. 20, July 1, 2024
Reserved at 8 Ok Reg
				  3503, eff 8-15-91 (emergency); Reserved at 9 Ok Reg 1643, eff
				  5-29-92
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 9 - Standards for Protection Against Radiation › Section 310:281-9-33 - Reports of overexposures and excessive levels and concentrations [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 9 - Standards for Protection Against Radiation › Section 310:281-9-34 - RESERVED

Current through Vol. 41, No. 20, July 1, 2024
Reserved at 8 Ok Reg
				  3503, eff 8-15-91 (emergency); Reserved at 9 Ok Reg 1643, eff
				  5-29-92
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 9 - Standards for Protection Against Radiation › Section 310:281-9-35 - Vacating premises

Current through Vol. 41, No. 20, July 1, 2024
Each permittee shall notify the Department of their intent
				  to relocate, store or transfer the diagnostic x-ray system 30 days before they
				  relocate, transfer or store.
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Amended at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 9 - Standards for Protection Against Radiation › Section 310:281-9-36 - Notifications and reports to individuals

Current through Vol. 41, No. 20, July 1, 2024
When a permittee is required pursuant to
				  310:281-9-31
				  to report to the Department any exposure of an individual to radiation, the
				  permittee shall also notify the individual. Such notice shall be transmitted at
				  a time not later than the transmittal to the Department.
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Amended at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 9 - Standards for Protection Against Radiation › Section 310:281-9-37 - Concentrations in water and air above natural background [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Revoked at 21 Ok Reg
				  1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 9 - Standards for Protection Against Radiation › Section 310:281-9-38 - Quantities for use with 310:281-9-12 and 310:281-9-20 [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 9 - Standards for Protection Against Radiation › Section 310:281-9-39 - Acceptable surface contamination levels [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 9 - Standards for Protection Against Radiation › Section 310:281-9-40 - Exposure to pregnant employees

Current through Vol. 41, No. 20, July 1, 2024
If an employee of any permittee declares her pregnancy by
				  written notification, the permittee shall ensure that the DE to an embryo/fetus
				  during the entire pregnancy, due to occupational exposure of a declared
				  pregnant woman, does not exceed 0.5 rem (5 mSv).
(1) If a woman chooses not to declare
				pregnancy the occupational dose limits specified in
				310:281-9-2(a).
(2) The permittee shall make efforts to avoid
				substantial variation above a uniform monthly exposure rate to a declared
				pregnant woman so as to satisfy the limit in 310:281-9-40(a). The National
				Council on Radiation Protection and Measurement in NCRP Report No. 91
				"Recommendations on Limits for Exposure to Ionizing Radiation" (June 1, 1987)
				states that no more than 0.05 rem (0.5 mSv) to the embryo/fetus be received in
				any one month will be adopted.
(3)
				If by the time the woman declares pregnancy the permittee the DE to the
				embryo/fetus has exceeded 0.45 rem (4.5 mSv), the permittee is in compliance
				with 310:281-9-40(a), if the additional DE to the Embryo/fetus does not exceed
				0.05 rem (0.5 mSv) during the remainder of the pregnancy.
(4) The DE to an embryo/fetus shall be taken
				as the DE that is most representative of the DE to the embryo/fetus from
				external radiation. The DE from the mother's lower torso region will be used.
(5) If multiple measurements have
				been made, assignment of the DDE for the declared pregnant woman for the
				personal monitoring device that is most representative of the DE to the
				embryo/fetus shall be the DE to the embryo/fetus. Assignment of the highest DDE
				for the declared woman to the embryo/fetus is not required unless that dose is
				also the most representative DDE for the region of the embryo/fetus.
(6) If multiple measurements have not been
				made, assignment of the highest DDE for the declared pregnant woman shall be
				the embryo/fetus.
Added at 21 Ok Reg 1037,
				  eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-1 - Purpose

Current through Vol. 41, No. 20, July 1, 2024
(a) This Subchapter establishes requirements
				for the use of diagnostic x-ray systems in the healing arts and in the practice
				of veterinary medicine. All usage of such systems under this Subchapter shall
				be made by or under the supervision of an individual licensed in accordance
				with Oklahoma law to practice the healing art or veterinary medicine in which a
				diagnostic x-ray system is used.
(b) The requirements of this Subchapter are
				intended to ensure that adequate radiation protection is provided for the
				public and workers. These rules are not intended to replace calibration,
				survey, quality assurance, and/or quality control services by a radiological
				physicist or a qualified expert.
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Amended at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-1.1 - Definitions [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-2 - Accountability

Current through Vol. 41, No. 20, July 1, 2024
(a) The permittee is responsible for
				directing the operation of the diagnostic x-ray systems under the
				administrative control of the permittee. The permittee is responsible for
				assuring that the requirements of this subchapter are met in the operation of
				such diagnostic x-ray systems.
(b)
				The provisions of this subchapter are in addition to, and not in substitution
				for, other applicable provisions of these regulations.
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Amended at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-3 - Prohibitions

Current through Vol. 41, No. 20, July 1, 2024
(a) The Department may prohibit use of
				diagnostic x-ray systems, which pose significant threat or endanger public
				health and safety.
(b) Individuals
				shall not be exposed to the useful beam except for healing arts purposes and
				unless such exposure has been authorized by a licensed practitioner of the
				healing arts. This provision specifically prohibits deliberate exposure for the
				following purposes:
(1) exposure of an
				individual for training, demonstration, or other non-healing arts purposes;
				and
(2) exposure of an individual
				for the purpose of healing arts screening except as authorized by
				310:281-11-4(g).
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-4 - Diagnostic operational requirements for practitioners

Current through Vol. 41, No. 20, July 1, 2024
(a)
Technique chart.
A chart or
				manual shall be provided in the vicinity of the control panel of each non-APR
				diagnostic x-ray system, CR portable, or DR portable system which specifies,
				for all diagnostic examinations usually performed with that diagnostic x-ray
				system, the following information:
(1)
				technique factors to be utilized versus patient's anatomical size (except for
				diagnostic x-ray systems which have only automatic exposure controls);
(2) type of the film or
				film-screen combination to be used, if any;
(3) type of the grid to be used, if any; and
(4) source-to-image receptor
				distance to be used.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-5 - Radiographic entrance exposure limits

Current through Vol. 41, No. 20, July 1, 2024
The in-air exposure determined for the technique used for
				  the specified average human adult patient thickness for routine medical
				  radiography cannot exceed the entrance exposure or dose limits shown below.
(1) Entrance exposure and dose limits
				for the following radiographic technique are:
(A) for a 23 cm P/A chest has the exposure of
				2 mR and the exposure/dose limit of 30 mR;
(B) for a 23 cm P/A grid chest has the
				exposure of 25 mR and the exposure/dose limit of 35 mR;
(C) for a 23 cm abdomen(KUB) has the exposure
				of 500 mR and the exposure/dose limit of 600 mR;
(D) for a 23 cm lumbar spine (A/P) has the
				exposure of 500 mR and the exposure/dose limit of 800 mR;
(E) for a 13 cm cervical spine (AP) has the
				exposure of 125 mR and the exposure/dose limit of 200 mR;
(F) for a 15 cm skull (lateral) has the
				exposure of 150 mR and the exposure/dose limit of 250 mR;
(G) dental intraoral exams have a exposure of
				00 mR and the exposure/dose limit of 600 mR; and
(H) for computed tomography exposures for the
				following techniques are:
(i) for a 16 cm
				adult head the exposure is 600 mR;
(ii) for a 32 cm adult abdomen the exposure
				is 3500 mR; and
(iii) for a 16 cm
				pediatric abdomen the exposure is 2500 mR.
(iv) These are CTDI w dose measurements as
				recommended by the American College of Radiology in their voluntary
				certification of CT system.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-6 - Additional requirements for use of diagnostic x-ray systems in the healing arts of medicine, podiatry, and chiropractic

Current through Vol. 41, No. 20, July 1, 2024
(a)
Viewing system.
Windows,
				mirrors, closed circuit television, or an equivalent system shall be provided
				to permit operator to continuously observe the patient during
				irradiation.
(b)
Control
				panel.
Each x-ray control shall be located in such a way as to meet the
				following requirements:
(1) Stationary
				diagnostic x-ray systems are required to have the x-ray control permanently
				mounted in a permanent control booth or station so that the operator is
				required to remain in that area during the entire exposure and not be exposed
				to greater than 2 millirems in any 1 hour. See section
				310:281-11-20
				for measurement protocol to determine compliance.
(2) Mobile and portable diagnostic x-ray
				systems shall meet the requirements of a stationary diagnostic x-ray system
				when used for greater than 7 consecutive working days in the same
				location.
(3) For mobile and
				portable diagnostic x-ray systems, single event exposures and configuration,
				the x-ray control shall be positioned so that the operator is at least 6 feet
				away from the tube housing and the patient during an exposure and is not
				exposed to greater than 2 millirems in any 1 hour.
(4) The operator shall be able to maintain
				visual and aural contact with the patient.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-7 - Additional operational controls for fluoroscopic x- ray systems and spot-film devices

Current through Vol. 41, No. 20, July 1, 2024
(a) Fluoroscopic
				configuration, including fluoroscopic table designs when combined with
				operating and safety procedures, shall not permit any portion of any
				individual's body, except the head, neck, and extremities, to be exposed to
				scattered radiation emanating from above or below the tabletop unless the
				radiation has passed through not less than a total of 0.25 millimeter lead
				equivalent material. The material may be, but is not limited to, drapes,
				self-supporting curtains, or viewing shields, in addition to any lead
				equivalency provided by a protective apron.
(b) Where sterile fields or special
				procedures prohibit the use of normal protective barriers or drapes,
				310:281-11-7(a) does not apply, if all of the following conditions are met:
(1) All persons, except the patient, in the
				room where fluoroscopy is performed, shall wear protective aprons which provide
				a shielding equivalent of 0.5 millimeter of lead.
(2) The fluoroscopist and all other personnel
				in the room, except the patient, shall have appropriate personnel monitoring
				devices.
(3) The fluoroscopic field
				size shall be reduced to the absolute minimum required for the procedure being
				performed (area of clinical interest).
(4) Operating and safety procedures shall
				reflect the above conditions, and fluoroscopy personnel shall exhibit awareness
				of situations requiring the use and/or nonuse of the protective drapes.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-8 - Additional operational controls for CT x-ray systems

Current through Vol. 41, No. 20, July 1, 2024
(a) The CT x-ray system shall not be operated
				except by an individual who has been specifically trained in its operation.
(b) The control panel must include
				the following information for the system's operation and evaluation:
(1) dates of the latest evaluation and spot
				checks, and the location within the facility where the results of those tests
				may be obtained;
(2) the results of
				at least the most recent spot checks conducted on the system; and
(3) the distance in millimeters between the
				tomographic plane and the reference plane, if a reference plane is utilized.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-9 - Additional operational controls for mammography systems

Current through Vol. 41, No. 20, July 1, 2024
Incorporation by reference: Adopted reference. Except as
				  otherwise specifically provided, the performance of mammography shall conform
				  to standards as described in "Quality Mammography Standards" Code of Federal
				  Regulations (CFR) 21 CFR Parts 900, et. seq. effective April 28, 1999, as
				  amended, promulgated pursuant to be the federal "Mammography Quality Standards
				  Act." Final Rule as published in Federal Register of October 28, 1997 and
				  corrected in Federal register of November 10, 1997.
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Amended at 10 Ok
				  Reg 73, eff 10-5-92 (emergency); Amended at 10 Ok Reg 1695, eff 6-1-93; Amended
				  at 21 Ok Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-10 - Additional operational controls for dental radiographic systems

Current through Vol. 41, No. 20, July 1, 2024
(a) Film holding devices shall be used except
				in individual cases in which the practitioner has determined that such holding
				devices are contraindicated. Written safety procedures required by
				310:281-11-4(b)
				state the criteria under which the exception may apply.
(b) The tube housing support shall be
				constructed and adjusted so that the tube housing cannot drift from its set
				position during an exposure. Neither the tube housing nor support housing shall
				be hand-held during an exposure.
(c) The operator shall stand at least 6 feet
				from the useful beam or behind a protective barrier. Where a protective barrier
				is utilized, a viewing system shall be provided.
(d) Individuals who operate only dental
				radiographic systems are exempted from the personnel monitoring requirements of
				310:281-9-12.
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff
				  5-29-92
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-11 - Additional operational controls for veterinary x- ray systems

Current through Vol. 41, No. 20, July 1, 2024
(a) In no case shall an individual hold the
				x-ray tube during any radiographic exposure.
(b) Unless required to restrain an animal,
				the operator shall stand at least 6 feet away from the useful beam and the
				animal during radiographic exposures.
(c) No individual, other than the operator,
				shall be in the x-ray room or area while exposures are being made unless such
				individual's assistance is required.
(d) When an animal must be held in position
				during radiography, mechanical supporting or restraining devices may be used
				when technique permits.
(1) If the animal
				must be held by an individual, that individual shall be protected with
				appropriate shielding devices, such as protective gloves and apron, and the
				person shall be so positioned that no part of his/her body except hands and
				arms will be struck by the useful beam.
(2) The exposure of any individual who holds
				animals shall be monitored by film badge, TLD, or other appropriate device.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-12 - General requirements for all human diagnostic x-ray systems

Current through Vol. 41, No. 20, July 1, 2024
(a)
Battery charge indicator.
On
				battery-powered x-ray generators, visual means shall be provided on the control
				panel to indicate whether the battery is in a state of charge adequate for
				proper operation.
(b)
Leakage
				radiation from the diagnostic x-ray tube assembly.
The leakage radiation
				from the diagnostic x-ray tube assembly measured at a distance of 1 meter in
				any direction from the x-ray tube assembly shall not exceed 100 milliroentgens
				in 1 hour when the x-ray tube is operated at its leakage technique factors.
				Measurement is averaged over an area of 100 square centimeters with no linear
				dimension greater than 20 centimeters.
(c)
Radiation from components other
				than the diagnostic x-ray tube assembly.
The radiation emitted by a
				component other than the diagnostic x-ray tube assembly cannot exceed 2
				milliroentgens in 1 hour at 5 centimeters from any accessible surface of the
				component when it is operated in an assembled x-ray system under any conditions
				for which it was designed. Measurement is averaged over an area of 100 square
				centimeters with no linear dimension greater than 20 centimeters.
(d)
Beam quality.
(1)
Half-value layer (HVL).
(A) The half-value layer of the useful beam
				for a given x-ray tube potential shall not be less than the values shown below.
				If it is necessary to determine such half-value layer at an x-ray tube
				potential which is not listed in the following table, linear interpolation may
				be made. Half-value layer measurements shall be made at no less than 90 percent
				maximum rated potential or 80 kVp, whichever is lower for the unit.
(i) the half-value layer for the selected
				tube potential of 30 kVp is 0.3 mm of aluminum; (terminology and units listed
				below are the same listed here);
(ii) 40 kVp is 0.4
(iii) 50 kVp is 0.5
(iv) 51 kVp is 1.2
(v) 60 kVp is 1.3
(vi) 70 kVp is 1.5
(vii) 71 kVp is 2.1
(viii) 80 kVp is 2.3
(ix) 90 kVp is 2.5
(x) 100 kVp is 2.7
(xi) 110 kVp is 3.0
(xii) 120 kVp is 3.2
(xiii) 130 kVp is 3.5
(xiv) 140 kVp is 3.8 and
(xv) 150 kVp is 4.1
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-13 - Fluoroscopic x-ray systems and spot-film devices except radiation therapy simulators

Current through Vol. 41, No. 20, July 1, 2024
(a)
Limitation of useful beam.
(1)
Primary barrier.
(A) The
				fluoroscopic imaging assembly shall be provided with a primary protective
				barrier which intercepts the entire cross section of the useful beam at any
				source-to-image receptor distance (SID).
(B) The x-ray tube used for fluoroscopy shall
				not produce x-rays unless the barrier is in position to intercept the useful
				beam and the imaging device is in place and operable.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-14 - Radiographic diagnostic x-ray systems

Current through Vol. 41, No. 20, July 1, 2024
(a)
Beam limitation.
This Subsection applies to radiographic systems
				used in the healing arts of medicine, chiropractic, and podiatry. It does not
				apply to fluoroscopic, dental, veterinary, or computed tomography systems.
(1)
General purpose stationary, mobile
				and portable diagnostic x-ray systems.
(A) The tube housing shall be of diagnostic
				type.
(B) Adjustable collimators,
				with the same degree of protection as is required of the housing, shall be
				provided to restrict the useful beam to the area of clinical interest. They
				shall provide for independent, stepless adjustment of at least two dimensions
				of the x- ray field.
(C) A method
				shall be provided for visually defining the perimeter of the x-ray field. The
				total misalignment of the edges of the visually defined field with the
				respective edges of the x-ray field along either the length or width of the
				visually defined field shall not exceed 2 percent of the distance from the
				source to the center of the visually defined field when the surface upon which
				it appears is perpendicular to the axis of the x-ray beam.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-15 - Computed tomography x-ray systems

Current through Vol. 41, No. 20, July 1, 2024
(a)
System requirements.
(1)
Tomographic plane indication and alignment.
(A) For any single tomogram system, means
				shall be provided to permit visual determination of the tomographic plane or a
				reference plane offset from the tomographic plane.
(B) For any multiple tomogram system, means
				shall be provided to permit visual determination of the location of a reference
				plane. The reference plane can be offset from the location of the tomographic
				planes.
(C) If a device using a
				light source is used to satisfy 310:281-11-15(a)(1)(A) or (B), the light source
				shall provide illumination levels sufficient to permit visual determination of
				the location of the tomographic plane or reference plane under ambient light
				conditions of up to 500 lux.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-16 - Dental radiographic systems General requirements

Current through Vol. 41, No. 20, July 1, 2024
(1)
Timers.
Means shall be
				provided to terminate the exposure at a preset time interval, preset product of
				current and time, a preset number of pulses, or a preset radiation exposure to
				the image receptor. In addition, it shall not be possible to make an exposure
				when the timer is set to a "zero" or "off" position if either position is
				provided.
(2)
Reproducibility.
With a timer setting of 0.3 second or less, the
				average exposure period (T) shall be greater than or equal to 5 times the
				maximum exposure period (T max) minus the minimum exposure period (T min) when
				4 timer tests are performed:
T
>
5(
T
max -
T
min)
(3)
X-Ray control.
(A) An x-ray control shall be incorporated
				into each dental x-ray system such that an exposure can be terminated by the
				operator at any time, except for exposures of 0.5 second or less.
(B) The exposure switch shall be of the
				dead-man type.
(C) Each x-ray
				control shall be located in such a way as to permit the operator to remain in
				an area of less than 2 millirems in any 1 hour during the entire exposure.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-17 - Veterinary x-ray systems

Current through Vol. 41, No. 20, July 1, 2024
(a)
Systems.
(1) The system shall
				display the selected exposure factors before the exposure begins.
(A) The requirement of 310:281-11-17(a)(1)(A)
				may be met by permanent markings on equipment having fixed technique factors.
				Indication of technique factors shall be visible from the operator's position
				except in the case of spot films made by the fluoroscopist.
(B) The x-ray control shall provide visual
				indication of the production of x-rays observable at or from the operator's
				protected position whenever x-rays are produced.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-18 - Therapeutic x-ray systems of less than 1 MeV [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-19 - X-ray and electron therapy systems with energies of 1 MeV and above [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-20 - Determining skin entrance exposures

Current through Vol. 41, No. 20, July 1, 2024
The following is the method for measuring and calculating
				  patient entrance skin exposures in diagnostic x-ray examinations.
(1) Have the operator set the x-ray tube at
				the source to image receptor distance (SID) routinely used, and adjust the
				collimation as though a patient was in the beam. Measure the distance from the
				x-ray source to the surface against which the patient rests. Subtract the
				thickness of the patient part to obtain the source to skin distance (SSD).
(2) Place the ionization chamber
				in the center of the x-ray field and measure the source to chamber distance
				(SCD). Use of a test stand (such as the CDRH stand) to position the chamber at
				a reasonable distance from the table will reduce backscatter contribution.
				Placing the chamber at the actual SSD will accomplish this and allow direct
				reading of the ESE.
(3) Record the
				routine technical factors used by the facility for the standard patient
				thickness (Table l) and make exposures utilizing these techniques.
(4) For phototimed procedures a phantom must
				be utilized to control the exposure time and achieve an accurate exposure
				estimate. When utilizing a phantom, the measuring chamber should be positioned
				in the beam between the phantom and the x-ray tube (but not placed over an
				active photocell), and should be located approximately nine inches from the
				phantom to reduce backscatter contribution.
(5) Calculate the entrance skin exposure as
				follows if a direct result was not obtained:
ESE
=
				(
Raw mR)
x (
SCD
) 
				2 / (
SSD
) 
				2
(6) Compare the results of these measurements
				for the speed of the imaging system used by the facility with the ESE guides
				published by the Conference of Radiation Control Program Directors, Publication
				88-5 "Average Patient Exposure Guides - 1988", or a similarly derived group of
				values, (allowing for the speed of the imaging system used by the facility).
(7) There are many different
				techniques for measuring ESE, which may result in significant differences in
				measured values. Factors that can cause variations include instrument
				calibration, backscatter, collimation, estimation of focal spot location,
				choice of phantom, and location of chamber in the primary beam.
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Amended at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 11 - Use of X-Rays in the Healing Arts and Veterinary Medicine › Section 310:281-11-21 - Additional information required for healing arts screening

Current through Vol. 41, No. 20, July 1, 2024
Persons requesting that the Department approve a healing arts
				  screening program shall submit the following information and evaluation:
(a)
Administrative controls.
(1) The name and address of the applicant
				and, where applicable, the names and addresses of agents within
				Oklahoma.
(2) The diseases or
				conditions for which the x-ray examinations are to be used in
				diagnosis.
(3) A detailed
				description of the x-ray examinations proposed in the screening
				program.
(4) A description of the
				population to be examined in the screening program(i.e., age, sex, physical
				condition.)
(5) An evaluation of
				any known alternate methods not involving radiation which could achieve the
				goals of the screening program and why these methods are not being
				utilized.
(6) All screening exams
				must have an order from an Oklahoma licensed medical practitioner. The exam
				order must be within the scope of the practitioner's license. The only
				exceptions to this are screening mammography and bone density exams.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 13 - Radiation Safety Requirements for Analytical X-Ray Equipment [Revoked] › Section 310:281-13-1 - Purpose [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 13 - Radiation Safety Requirements for Analytical X-Ray Equipment [Revoked] › Section 310:281-13-1.1 - Definitions [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 13 - Radiation Safety Requirements for Analytical X-Ray Equipment [Revoked] › Section 310:281-13-2 - Equipment requirements [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 13 - Radiation Safety Requirements for Analytical X-Ray Equipment [Revoked] › Section 310:281-13-3 - Area requirements [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 13 - Radiation Safety Requirements for Analytical X-Ray Equipment [Revoked] › Section 310:281-13-4 - Operating requirements [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 13 - Radiation Safety Requirements for Analytical X-Ray Equipment [Revoked] › Section 310:281-13-5 - Personnel requirements [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 9 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-1 - Purpose [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-2 - Registration requirements [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-3 - General requirements for registration of particle accelerators [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-4 - Additional requirements for registration of human use particle accelerators [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-5 - [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Reserved at 8 Ok Reg
				  3503, eff 8-15-91 (emergency); Reserved at 9 Ok Reg 1643, eff 5-29-92; Revoked
				  at 21 Ok Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-6 - Limitations [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-7 - Shielding and safety design requirements [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-8 - Particle accelerator controls and interlock systems [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-9 - Warning devices [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-10 - Operating procedures [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-11 - Radiation monitoring requirements [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 15 - Radiation Safety Requirements for Particle Accelerators [Revoked] › Section 310:281-15-12 - Ventilation systems [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				  eff 8-15-91 (emergency); Added at 9 Ok Reg 1643, eff 5-29-92; Revoked at 21 Ok
				  Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Subchapter 17 - Use of Teleradiology in Healing Arts › Section 310:281-17-1 - Incorporation by reference

Current through Vol. 41, No. 20, July 1, 2024
(a)
Adopted reference.
Except as otherwise specifically provided, the
				performance of teleradiology shall conform to standards as described in the
				2002 revised edition of the "ACR Technical Standard for Teleradiology" as
				defined in the American College of Radiology Practice Guidelines and Technical
				Standards.
(b)
Exceptions.
(1) Section III,
				Qualifications of Personnel, "In all cases this means a licensed and/or
				registered radiologic technologist, radiation therapist, nuclear medicine
				technologist or sonographer is not incorporated by reference.
(2) Section III,B,1 "Certified by the
				appropriate registry and/or possess unrestricted state licensure" is not
				incorporated by reference.
(3)
				Section V, Licensing, Credentialing, And Liability, "Physicians who provide the
				official interpretation of images transmitted by teleradiology should maintain
				licensure as may be required for provision of radiologic service at both the
				transmitting and receiving sites" is not incorporated by reference.
(4) Doctors of Veterinary Medicine are
				exempted from this subchapter.
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 281 - Medical Marijuana Regulations [Revoked] › Appendix B

Current through Vol. 41, No. 20, July 1, 2024
Added at 8 Ok Reg 3503,
				eff 8-15-91 (emergency) ; Added at 9 Ok Reg 1643, eff 5-29-92; Revoked and
				reenacted at 21 Ok Reg 1037, eff 5-13-04
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 282 - Laboratory Certification Standards [Revoked] › Subchapter 1 - General Provisions [Revoked] › Section 310:282-1-1 - Purpose [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 9 Ok Reg 3129,
				  eff 7-1-92 (emergency); Added at 10 Ok Reg 1699, eff 6-1-93; Revoked at 12 Ok
				  Reg 2325, eff 6-26-95
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 282 - Laboratory Certification Standards [Revoked] › Subchapter 1 - General Provisions [Revoked] › Section 310:282-1-2 - Definitions [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 9 Ok Reg 3129,
				  eff 7-1-92 (emergency); Added at 10 Ok Reg 1699, eff 6-1-93; Revoked at 12 Ok
				  Reg 2325, eff 6-26-95
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 282 - Laboratory Certification Standards [Revoked] › Subchapter 3 - Standards for Certification [Revoked] › Section 310:282-3-1 - Certification process [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 9 Ok Reg 3129,
				  eff 7-1-92 (emergency); Added at 10 Ok Reg 1699, eff 6-1-93; Revoked at 12 Ok
				  Reg 2325, eff 6-26-95
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 282 - Laboratory Certification Standards [Revoked] › Subchapter 3 - Standards for Certification [Revoked] › Section 310:282-3-2 - Provisional certification [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 9 Ok Reg 3129,
				  eff 7-1-92 (emergency); Added at 10 Ok Reg 1699, eff 6-1-93; Revoked at 12 Ok
				  Reg 2325, eff 6-26-95
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 282 - Laboratory Certification Standards [Revoked] › Subchapter 3 - Standards for Certification [Revoked] › Section 310:282-3-3 - Performance evaluation [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 9 Ok Reg 3129,
				  eff 7-1-92 (emergency); Added at 10 Ok Reg 1699, eff 6-1-93; Revoked at 12 Ok
				  Reg 2325, eff 6-26-95
Oklahoma Administrative Code › Title 310 - Oklahoma State Department of Health › Chapter 282 - Laboratory Certification Standards [Revoked] › Subchapter 3 - Standards for Certification [Revoked] › Section 310:282-3-4 - Quality assurance [REVOKED]

Current through Vol. 41, No. 20, July 1, 2024
Added at 9 Ok Reg 3129,
				  eff 7-1-92 (emergency); Added at 10 Ok Reg 1699, eff 6-1-93; Revoked at 12 Ok
				  Reg 2325, eff 6-26-95