reglab/statecodes huggingface dataset:
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5094 - Notifications and reports to individuals

Current through Vol. 24-12, July 15, 2024
Rule 94.
(1) A
					 registrant shall report radiation exposure data for an individual as specified
					 in this rule. The information reported shall include data and results obtained
					 pursuant to these rules, orders, or registration conditions, as shown in
					 records kept by the registrant pursuant to R 333.5081. A notification and
					 report shall be in writing and include all of the following:
(a) The name of the registrant, the name of
					 the individual, and the individual's unique identification number or social
					 security number.
(b) The
					 individual's exposure information.
(c) The following statement:
"This report is provided to you pursuant to Part 4 of the
						Michigan Department of Licensing and Regulatory Affairs rules entitled Ionizing
						Radiation Rules Governing the Use of Radiation Machines. You should keep this
						report for future reference."
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5096 - Presence of representatives of registrants and workers during inspection

Current through Vol. 24-12, July 15, 2024
Rule 96.
(1) A
					 registrant or an applicant for a registration shall allow the department at all
					 reasonable times, the opportunity to inspect machines, activities, facilities,
					 premises, and records under these rules.
(2) During an inspection, the registrant
					 shall allow department inspectors to consult privately with workers as
					 specified in R 333.5097. The registrant may accompany department inspectors at
					 times other than the private consultation with workers.
(3) If the workers have authorized an
					 individual to represent them during department inspections, the registrant
					 shall notify the inspectors of the authorization and shall permit the workers'
					 representative to accompany the inspectors during the inspection of physical
					 working conditions.
(4) A workers
					 representative shall be routinely engaged in work under control of the
					 registrant and shall have received instructions as specified in R
					 333.5093.
(5) If there is no
					 resulting interference with the conduct of the inspection, different
					 representatives of registrants and workers may accompany the inspectors during
					 different phases of an inspection. However, only 1 workers' representative at a
					 time may accompany the inspectors.
(6) With the approval of the registrant and
					 the workers' representative, an individual who is not routinely engaged in work
					 under control of the registrant, for example, a consultant to the registrant or
					 to the workers' representative, may accompany department inspectors during the
					 inspection of physical working conditions.
(7) Department inspectors may refuse to
					 permit accompaniment by an individual who deliberately interferes with a fair
					 and orderly inspection.
(8) Unless
					 previously authorized by the registrant, a workers representative shall not
					 have access to an area containing proprietary information.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5097 - Consultation with workers during inspections

Current through Vol. 24-12, July 15, 2024
Rule 97.
(1)
					 Department inspectors may consult privately with workers concerning matters of
					 occupational radiation protection and other matters related to applicable
					 provisions of these rules to the extent that the inspectors consider necessary
					 for the conduct of an effective and thorough inspection.
(2) During an inspection, a worker may report
					 privately to the inspectors, either orally or in writing, a past or present
					 condition that the worker believes may have contributed to or caused a
					 violation of the act, these rules, a registration condition, or unnecessary
					 exposure of an individual to radiation from machines under the registrant's
					 control. If this notice is in writing, the worker shall comply with the
					 requirements of R 333.5098(1).
(3)
					 The provisions of subrule (2) of this rule shall not be interpreted as
					 authorization to disregard instructions pursuant to R 333.5093.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5098 - Requests by workers for inspections

Current through Vol. 24-12, July 15, 2024
Rule 98.
(1) A
					 worker or a representative of workers who believes that a violation of the act,
					 these rules, or registration conditions exists or has occurred regarding
					 radiological working conditions may request an inspection of the facility by
					 the department. The request shall be in writing, describe the circumstances of
					 the perceived violation or condition, and be signed by the worker or the
					 representative of the workers. The department shall provide a copy of the
					 request to the registrant before or during the inspection. At the request of
					 the worker, the department shall protect the worker's name and the name of
					 individuals referred to in the request, except for good cause shown.
(2) If, upon receipt of a request for an
					 inspection, the department determines that the complaint meets the requirements
					 of subrule (1) of this rule, and that there are reasonable grounds to believe
					 that the alleged violation exists or has occurred, an inspection shall be made
					 as soon as practical to determine if the alleged violation exists or has
					 occurred. An inspection authorized by this rule need not be limited to matters
					 referred to in the complaint.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5100 - Inspections not warranted; informal review

Current through Vol. 24-12, July 15, 2024
Rule 100.
(1) If
					 the department determines, with respect to a complaint filed pursuant to R
					 333.5098, that an inspection is not warranted because there are no reasonable
					 grounds to believe that a violation exists or has occurred, the department
					 shall notify the complainant in writing of the determination. To request a
					 review of the determination the claimant shall submit a written statement of
					 position to the department director. The department director, or his or her
					 designated representative, shall send the registrant a copy of the statement by
					 certified mail, excluding, at the request of the complainant, the name of the
					 complainant. The registrant may submit an opposing written statement of
					 position to the department director. The department director, or his or her
					 designated representative, shall send the complainant a copy of the statement
					 by certified mail.
(2) At the
					 request of the complainant, the department may hold an informal conference in
					 which the complainant and the registrant may orally present their views. An
					 informal conference may also be held at the request of the registrant, but
					 disclosure of the identity of the complainant shall be made only following
					 receipt of written authorization from the complainant. After considering all
					 written and oral views presented, the department director, or designated
					 representative shall affirm, modify, or reverse the determination of the
					 department and provide the complainant and the registrant a written
					 notification of the decision. The notification shall include a discussion of
					 the basis for the decision.
(3) If,
					 upon receipt of a request for an inspection, the department determines that an
					 inspection is not warranted because the complaint does not meet the
					 requirements of R 333.5098(1), the department shall notify the complainant in
					 writing of the determination. The determination shall be without prejudice to
					 the filing of a new complaint meeting the requirements of R
					 333.5098(1).
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5101 - Employee protection

Current through Vol. 24-12, July 15, 2024
Rule 101. Employment discrimination by a registrant, or
						contractor or subcontractor of a registrant against an employee for engaging in
						protected activities under this part is prohibited.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5201 - Purpose and scope

Current through Vol. 24-12, July 15, 2024
Rule 5201.
(1)
					 This part establishes the qualification requirements of individuals engaged in
					 medical radiologic technology. Improper performance of medical radiologic
					 technology results in unnecessary exposure to machine produced ionizing
					 radiation and the unnecessary re-administration of machine produced ionizing
					 radiation. Therefore, the qualification of these individuals to apply machine
					 produced ionizing radiation to humans has a direct impact on the machine
					 produced ionizing radiation burden of the patient.
(2) Unless specifically exempt in accordance
					 with R 333.5209, a registrant shall not employ an individual to perform medical
					 radiologic technology who does not meet the requirements of this part or
					 without the expressed written approval of the department.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5202 - Definitions; A

Current through Vol. 24-12, July 15, 2024
Rule 5202.
(1)
					 "ACRRT" means American Chiropractic Registry of Radiologic
					 Technologists.
(2) "Approved
					 program" means a formal education program in the respective discipline of
					 radiography or radiation therapy that is accredited by 1 or more of the
					 following:
(a) Joint Review Committee on
					 Education in Radiologic Technology.
(b) Regional institutional accrediting
					 agencies.
(c) Conjoint Secretariat
					 of the Canadian Medical Association.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5203 - Definitions; B

Current through Vol. 24-12, July 15, 2024
Rule 5203. "Bone densitometry" means the science and art
						of applying machine produced ionizing radiation to human beings for the
						determination of site-specific bone density.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5204 - Definitions; C. to G

Current through Vol. 24-12, July 15, 2024
Rule 5204.
(1)
					 "CAMRT" means Canadian Association of Medical Radiation
					 Technologists.
(2) "CBRPA" means
					 Certification Board for Radiology Practitioner Assistants.
(3) "CCI" means Cardiovascular Credentialing
					 International.
(4) "Chiropractic
					 radiography" means the science and art of applying machine produced ionizing
					 radiation to human beings for diagnostic evaluation of skeletal
					 anatomy.
(5) "Continuing education
					 activity" means a learning activity that is planned, organized, and
					 administered to enhance the professional knowledge and skills underlying
					 professional performance that a medical radiologic technologist uses to provide
					 services for patients, the public, or the medical profession. To qualify as
					 continuing education, the activity must be planned and organized to provide
					 sufficient depth and scope of a subject area.
(6) "CE credit" or continuing education
					 credit means a unit of measurement for continuing education activities. One CE
					 credit is awarded for 1 contact hour, which is 50 minutes. Activities longer
					 than 1 hour are assigned whole or partial credits based on the 50-minute
					 hour.
(7) "Direct supervision"
					 means the required individual must be present in at least an adjacent area and
					 immediately available to furnish assistance and direction throughout the
					 procedure, and is responsible for the control of quality, radiation safety
					 protection, and technical aspects of the application of radiation to human
					 beings for diagnostic, therapeutic, or research purposes.
(8) "General supervision" means the procedure
					 is furnished under the overall direction and control of a licensed practitioner
					 whose presence is not required during the performance of the procedure., and is
					 responsible for the control of quality, radiation safety protection, and
					 technical aspects of the application of radiation to human beings for
					 diagnostic, therapeutic, or research purposes.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5205 - Definitions; L

Current through Vol. 24-12, July 15, 2024
Rule 5205.
(1)
					 "Licensed practitioner" means a health practitioner licensed under article 15
					 of the public health code, 1978 PA 368, MCL
					 333.16101 to
					 333.18838.
(2) "Limited diagnostic radiographer" means
					 an individual, other than a licensed practitioner, who, while under the general
					 supervision of a licensed practitioner, performs limited diagnostic
					 radiography.
(3) "Limited
					 diagnostic radiography" means the science and art of applying machine produced
					 ionizing radiation to human beings for limited diagnostic purposes.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5206 - Definitions; M

Current through Vol. 24-12, July 15, 2024
Rule 5206.
(1)
					 "Medical radiologic technologist" means an individual, other than a licensed
					 practitioner, who, while under the general supervision of a licensed
					 practitioner performs medical x-radiation procedures involving the application
					 of machine produced ionizing radiation to human beings for diagnostic,
					 therapeutic, and research purposes.
(2) "Medical radiologic technology" means the
					 science and art of performing medical x-radiation procedures involving the
					 application of machine produced ionizing radiation to human beings for
					 diagnostic, therapeutic, and research purposes. The specialized disciplines of
					 medical radiologic technology are medical radiography, radiation therapy
					 technology, chiropractic radiography, limited diagnostic radiography, and
					 radiologist assistant.
(3) "Medical
					 radiographer" means an individual, other than a licensed practitioner, who,
					 while under the general supervision of a licensed practitioner, applies machine
					 produced ionizing radiation to a part of the human body.
(4) "Medical radiography" means the science
					 and art of applying machine produced ionizing radiation to human beings for
					 diagnostic, and research purposes.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5207 - Definitions; P

Current through Vol. 24-12, July 15, 2024
Rule 5207.
(1)
					 "Personal supervision" means the required individual i s in attendance in the
					 room during the performance of a procedure, and is responsible for the control
					 of quality, radiation safety and protection, and technical aspects of the
					 application of radiation to human beings for diagnostic, therapeutic, or
					 research purposes.
(2) "Physician's
					 assistant" means an individual who is licensed as a physician's assistant under
					 part 170 or part 175 of the public health code, 1978 PA 368, MCL
					 333.17001 to
					 333.17097 and
					 333.17501 to
					 333.17556.
(3) "Podiatric" means a radiographic
					 examination of the toes, foot, ankle, calcaneus, and distal tibia/fibula but
					 does not include the knee joint.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5208 - Definitions; R

Current through Vol. 24-12, July 15, 2024
Rule 5208.
(1)
					 "Radiation therapy technology" means the science and art of applying radiation
					 emitted from x-ray machines or particle accelerators to human beings for
					 therapeutic purposes.
(2)
					 "Radiologist assistant" means an individual, other than a licensed
					 practitioner, who as a medical radiographer with advanced-level training and
					 certification, performs a variety of activities under the direct, general, or
					 personal supervision of a radiologist, certified by the American Board of
					 Radiology (ABR), the American Osteopathic Board of Radiology (AOBR), or Royal
					 College of Physicians and Surgeons of Canada (RCPSC), in the areas of patient
					 care, patient management, clinical imaging, and interventional procedures. The
					 radiologist assistant shall not interpret images, make diagnoses, or prescribe
					 medications or therapies.
(3)
					 "Radiology" means the branch of medicine that deals with the study and
					 application of imaging technology to diagnose and treat disease.
(4) "RCEEM" or "recognized continuing
					 education evaluation mechanism" means a mechanism used by the ARRT for
					 evaluating the content, quality, and integrity of an educational activity. The
					 evaluation shall include a review of education objectives, content selection,
					 faculty qualifications, and educational methods and materials.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5209 - Exemptions

Current through Vol. 24-12, July 15, 2024
Rule 5209.
(1)
					 Nothing in this part shall be construed to limit or affect in any respect, the
					 medical practice of individuals properly licensed under article 15 of the
					 public health code, 1978 PA 368, MCL
					 333.16101 to
					 333.18838 with respect to their
					 professions and scope of practice.
(2) The following individuals are exempt from
					 the requirements of this part:
(a) A student
					 enrolled in an approved program applicable to his or her profession who, as a
					 part of his or her course of study, applies machine produced ionizing radiation
					 to human beings while under the direct supervision of a licensed practitioner
					 or medical radiologic technologist who meets the requirements of this
					 part.
(b) An individual employed as
					 a dental assistant or dental hygienist who performs radiography under the
					 general supervision of a licensed practitioner.
(c) A nurse, technician, or other assistant
					 who, under the general supervision of a licensed practitioner, performs bone
					 densitometry.
(d) A medical
					 radiologic technologist performing mammography who meets the qualification
					 requirements of R 333.5630.
(e) A
					 computed tomography (CT) operator who meets the qualification requirements of R
					 333.5705.
(f) A nuclear medicine
					 technologist who, under the supervision of an authorized user, utilizes sealed
					 and unsealed radioactive materials for diagnostic, treatment, and research
					 purposes.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5210 - Categories and types of qualification

Current through Vol. 24-12, July 15, 2024
Rule 5210.
(1)
					 The department recognizes all of the following categories in the practice of
					 medical radiologic technology:
(a) Medical
					 radiography.
(b) Radiation therapy
					 technology.
(c) Radiologist
					 assistant.
(d) Chiropractic
					 radiography.
(e) Limited diagnostic
					 radiography.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5211 - Credentialing requirements

Current through Vol. 24-12, July 15, 2024
Rule 5211. Individuals who seek to qualify for active
						status in medical radiologic technology shall possess the appropriate
						credential or credentials as specified below or equivalent:
(a) Medical radiography:
(i) ARRT - radiography (R).
(ii) CAMRT - registered technologist,
					 radiation technology (RTR).
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5212 - Limited diagnostic radiography requirements

Current through Vol. 24-12, July 15, 2024
Rule 5212.
(1)
					 Individuals who perform limited diagnostic radiography shall pass an approved
					 program or obtain a minimum of 40 hours of training relevant to the radiologic
					 science within the limited scope of practice. This training must include both
					 clinical and didactic components.
(2) The following general categories must be
					 included in limited diagnostic radiography training or program:
(a) Fundamentals of healthcare.
(b) Medical terminology.
(c) Patient care and management.
(d) Human anatomy and physiology.
(e) Imaging production and
					 evaluation.
(f) Imaging equipment
					 and radiation production.
(g)
					 Radiation protection and radiobiology.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 4 - NOTICES, INSTRUCTIONS AND REPORTS TO WORKERS; INSPECTIONS FOR USERS OF RADIATION MACHINES › Section R. 333.5213 - Continuing education requirements

Current through Vol. 24-12, July 15, 2024
Rule 5213.
(1)
					 The required number of CE credits for limited diagnostic radiography is 12 CE
					 credits.
(2) CE credits required by
					 subrule (1) of this rule must be completed within 24 months after the effective
					 date of these rules and every 24 months after their initial
					 completion.
(3) The options for
					 meeting CE requirements are any of the following:
(a) Activities approved by an RCEEM. Among
					 the requirements for qualification as an RCEEM, an organization shall be
					 national in scope, non-profit, radiology based, and willing to evaluate CE
					 activity developed by a medical radiologic technologist within a given
					 discipline. Organizations with current RCEEM status include:
(i) American College of Radiology.
(ii) American Healthcare Radiology
					 Administrators.
(iii) American
					 Institute of Ultrasound in Medicine.
(iv) American Roentgen Ray Society.
(v) American Society of Nuclear
					 Cardiology.
(vi) American Society
					 of Radiologic Technologists.
(vii)
					 Association of Vascular and Interventional Radiographers.
(viii) Canadian Association of Medical
					 Radiation Technologists.
(ix)
					 Medical Dosimetrist Certification Board.
(x) Radiological Society of North
					 America.
(xi) Society of Diagnostic
					 Medical Sonography.
(xii) Section
					 for Magnetic Resonance Technologist of the International Society for Magnetic
					 Resonance in Medicine.
(xiii)
					 Society of Nuclear Medicine Technologist Section.
(xiv) Society of Vascular
					 Ultrasound.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › Section R. 333.5281 - Purpose and scope

Current through Vol. 24-12, July 15, 2024
Rule 281.
(1)
					 This part establishes radiation safety requirements for persons utilizing
					 radiation machines for industrial radiography and a classification system for
					 industrial radiographic installations and use.
(2) This part applies to all registrants who
					 use radiation machines for industrial radiography. Nothing in this part applies
					 to the use of radiation machines by a health practitioner licensed under
					 article 15 of the act.
(3) In
					 addition to the requirements of this part, all registrants are subject to the
					 applicable provisions of the other parts of these rules.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › Section R. 333.5282 - Definitions

Current through Vol. 24-12, July 15, 2024
Rule 282. As used in this part:
(a) "Industrial radiography" means the
					 examination of the macroscopic structure of materials by nondestructive methods
					 utilizing radiation machines.
(b)
					 "Installation" means a location, having boundaries specified by the registrant,
					 where for a period of more than 30 days 1 or more radiation machines are used.
					 A part of a building, an entire building, a plant, or plant site may be
					 designated as an installation.
(c)
					 "Radiographer" means an individual who performs or who, in attendance at the
					 site where radiation machines are being used, personally supervises class D
					 radiographic operations and who is responsible to the registrant for assuring
					 compliance with the requirements of these rules and all registration
					 conditions.
(d) "Radiographer's
					 assistant" means an individual who, under the personal supervision of a
					 radiographer, uses radiation machines or survey instruments in class D
					 radiographic operations.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › CLASSIFICATION › Section R. 333.5293 - Class enumeration

Current through Vol. 24-12, July 15, 2024
Rule 293.
(1)
						For registration and approval purposes, industrial radiographic installations
						shall be classified as class AA, class A, class B, or class C.
(2) For the purpose of registering and
						approving industrial radiography and radiation machines intended for limited
						use at temporary job site locations, this use shall be classified as a class D
						operation.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › CLASSIFICATION › Section R. 333.5296 - Class A installations

Current through Vol. 24-12, July 15, 2024
Rule 296.
(1)
						Class A installations shall comply with all requirements of R 333.5294 except
						for a permissible dose rate of 0.07 millisievert per hour (7 mrem/h) at any
						accessible external point.
(2) A
						personnel radiation dosimeter shall be permanently assigned to each
						occupationally exposed individual. This monitoring shall be continuous during
						employment as a radiation worker.
(3) Personnel exposure records shall be kept
						on permanent available file at the facility where the exposure occurs for
						inspection by the department.
(4)
						Class A approval permits unlimited use at maximum capacity.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › CLASSIFICATION › Section R. 333.5297 - Class B installations

Current through Vol. 24-12, July 15, 2024
Rule 297.
(1)
						Class B installations shall comply with all requirements of R
						333.5296.
(2) Radiation machine
						current and potential controls shall be mechanically or electrically limited so
						as not to exceed the normal operating conditions as specified by the registrant
						at the time of application for registration.
(3) Class B approval permits unlimited use
						under normal operating conditions as specified by subrule (2) of this
						rule.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › CLASSIFICATION › Section R. 333.5298 - Class C installations

Current through Vol. 24-12, July 15, 2024
Rule 298.
(1)
						Class C installations shall comply with all requirements of R 333.5296 except
						for a permissible dose rate of 0.5 millisievert (50 mrem) per hour at any
						accessible external point.
(2) The
						maximum weekly exposure time of radiation machines within the enclosure shall
						be established by the department under the conditions specified by the
						registrant at the time of application.
(3) Warning signs shall be posted in those
						areas outside the enclosure in which the radiation dose rate at any accessible
						external point exceeds 0.02 millisievert per hour (2 mrem/h). The dose rate
						shall be measured with the radiation machine tube placed at position of nearest
						source-wall distance that is radiographically usable under conditions of
						maximum radiation output permitted by the design or limited operating
						characteristics of the radiation machine.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › CLASSIFICATION › Section R. 333.5299 - Class D operations

Current through Vol. 24-12, July 15, 2024
Rule 299.
(1)
						Industrial radiography conducted under conditions not meeting the provisions
						and requirements of R 333.5294 to R 333.5298 shall be classified as class D
						operations and shall not be operated longer than 30 days unless written
						authorization is granted by the department.
(2) Written authorization may be granted by
						the department for class D operations longer than 30 days but not longer than 6
						months when an undue and unnecessary hardship may result from the 30-day
						limitation. Written request by the registrant for this authorization is
						required and shall describe the hardship involved as well as provide written
						assurance of compliance with the requirements of these rules for class D
						operations.
(3) Notwithstanding
						subrules (1) and (2) of this rule, a registrant routinely engaged in providing
						industrial radiography services with mobile or portable radiation machines at
						temporary job site locations may conduct such class D operations without time
						limitation subject to all of the following conditions:
(a) The registrant shall give written notice
						to the department at least 2 working days before starting radiographic work at
						a job site. The notice shall include the radiographer's name; the machine
						registration number; the nature, duration, and scope of use; and the exact
						location of each job site. If the 2 work-day period would impose an undue
						hardship on the registrant, upon application to the department, he or she may
						arrange for other notification to comply with the intent of this
						requirement.
(b) These class D
						operations shall be limited to locations and circumstances which cannot meet
						the provisions and requirements of permanent installation classification
						without undue and unnecessary hardship.
(c) A copy of written operating and emergency
						procedures shall be filed with and approved by the department.
(d) Upon reasonable notice from the
						department, the registrant shall submit to the department or otherwise make
						available copies of specific records pertaining to radiographic operations and
						personnel conducting these operations within this state.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › REQUIREMENTS FOR CLASS D RADIOGRAPHIC OPERATIONS › Section R. 333.5302 - Operating and emergency procedures

Current through Vol. 24-12, July 15, 2024
Rule 302. A registrant's written operating and
						  emergency procedures for class D radiographic operations shall include
						  instructions in at least the following:
(a) The use of radiation machines to be
						employed such that an individual is not likely to be exposed to radiation doses
						in excess of the limits established in R 333.5057 to R 333.5061.
(b) Methods and occasions for conducting
						radiation surveys.
(c) Methods for
						controlling access to radiographic areas.
(d) Personnel monitoring and the use of
						individual monitoring devices.
(e)
						Transportation to field locations, including packing of radiation machines in
						the vehicles, posting of vehicles, and control of radiation machines during
						transportation.
(f) Minimizing
						exposure of persons in the event of an accident.
(g) Procedure for notifying proper persons in
						the event of an accident.
(h)
						Maintenance of records.
(i)
						Inspection and maintenance of radiation machines.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › REQUIREMENTS FOR CLASS D RADIOGRAPHIC OPERATIONS › Section R. 333.5305 - Security

Current through Vol. 24-12, July 15, 2024
Rule 305.
(1)
						During each class D radiographic operation, the radiographer or radiographer's
						assistant shall maintain a direct surveillance of the operation to protect
						against unauthorized entry into a high radiation area, except where the high
						radiation area is equipped with interlocks as described in R 333.5294(5), or
						where the high radiation area is locked to protect against unauthorized or
						accidental entry.
(2) A
						radiographer or radiographer's assistant shall not perform or permit class D
						radiographic operation unless all individuals present in or entering the
						resulting radiation area are wearing individual monitoring devices.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › REQUIREMENTS FOR CLASS D RADIOGRAPHIC OPERATIONS › Section R. 333.5306 - Radiation surveys and survey records

Current through Vol. 24-12, July 15, 2024
Rule 306.
(1)
						A class D radiographic operation shall not be conducted unless calibrated and
						operable radiation survey instrumentation, as described in R 333.5307, is
						available and used at each site where radiographic exposures are
						made.
(2) For class D radiographic
						operations, a physical radiation survey shall be conducted to determine that
						the radiation machine is off before each entry into the radiographic exposure
						area.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › REQUIREMENTS FOR CLASS D RADIOGRAPHIC OPERATIONS › Section R. 333.5307 - Radiation survey instruments

Current through Vol. 24-12, July 15, 2024
Rule 307.
(1)
						A registrant shall maintain calibrated and operable radiation survey
						instruments to make physical radiation surveys as required by R 333.5306 and R
						333.5063. A radiation survey instrument shall be calibrated at least annually
						and after each instrument servicing. A record of the calibration shall be
						maintained for examination by the department.
(2) Instrumentation required by this rule
						shall have a range such that 0.02 millisievert per hour (2 mrem/h) through 0.01
						sievert per hour (1 rem/h) can be measured and shall be capable of measuring
						radiation of the energies and at the dose rates to be encountered.
(3) During repair or calibration of survey
						instruments required by this rule, spare operable and calibrated instruments
						shall be provided or class D radiographic operations shall be terminated
						pursuant to R 333.5306(1).
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › REQUIREMENTS FOR CLASS D RADIOGRAPHIC OPERATIONS › Section R. 333.5308 - Utilization logs

Current through Vol. 24-12, July 15, 2024
Rule 308. A registrant shall maintain for inspection by
						  the department current logs, which contain all of the following information for
						  each radiation machine:
(a) The
						machine registration number.
(b)
						The identity of the radiographer to whom the machine is assigned.
(c) The locations where used and dates of
						use.
(d) The signature or initials
						of the individual certifying each entry.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 6 - INDUSTRIAL RADIOGRAPHIC OPERATIONS AND INSTALLATIONS › REQUIREMENTS FOR CLASS D RADIOGRAPHIC OPERATIONS › Section R. 333.5309 - Limitations for radiographers and radiographer assistants

Current through Vol. 24-12, July 15, 2024
Rule 309.
(1)
						A registrant shall not permit an individual to act as a radiographer until all
						of the following have been met:
(a) The
						individual was instructed in the subjects outlined in table 309 and has
						demonstrated understanding of these subjects.
TABLE 309
Instruction of radiographers
I
Fundamentals of Radiation Safety
A. Characteristics of x-radiation
B. Units of radiation dose
C. Hazards of excessive radiation
									 exposure
D. Levels of radiation from radiation
									 machines
E. Methods of controlling radiation
									 dose
1. Working time
2. Working distances
3. Shielding
II
Radiation Detection Instrumentation to be
									 Used
A. Use of radiation survey instruments
1. Operation
2. Calibration
3. Limitations
B. Survey techniques
C. Use of individual monitoring
									 devices
III
Radiographic Equipment to be Used
A. Operation and control of x-ray
									 equipment
IV
The Requirements of Pertinent State
									 Regulations
V
The Registrant's Written Operating and
									 Emergency Procedures
VI
Registration Conditions
(b) The individual received copies of and
						instruction in the rules contained in this part and the applicable provisions
						sections of R 333.5051 to R 333.5089, registration conditions, and the
						registrant's operating and emergency procedures, and has demonstrated an
						understanding of these subjects.
(c) The individual demonstrated competence to
						use the radiation machine and survey instruments which will be employed in his
						or her assignment.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 7 - MEDICAL X-RAY INSTALLATIONS › Section R. 333.5311 - Purpose and scope

Current through Vol. 24-12, July 15, 2024
Rule 311.
(1)
					 This part establishes requirements governing the use of x-radiation in
					 medicine, osteopathy, chiropractic, and podiatry.
(2) This part applies to all registrants who
					 use x-radiation as a health practitioner or on behalf of a health practitioner
					 licensed under article 15 of the act for the intentional exposure of
					 humans.
(3) In addition to the
					 requirements of this part, all registrants who use x-radiation as a health
					 practitioner or on behalf of a health practitioner licensed under article 15 of
					 the act are subject to all applicable provisions of these rules.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 7 - MEDICAL X-RAY INSTALLATIONS › THERAPEUTIC MACHINES OPERATED ABOVE 85 KVP › Section R. 333.5315 - Enclosures

Current through Vol. 24-12, July 15, 2024
Rule 315.
(1)
						An enclosure shall be a permanent part of the building or equipment. Portable
						protective barriers shall not be used for permanent installations.
(2) The degree of protection required for an
						enclosure shall be determined by the workload, use and occupancy factors, and
						the tube potential, tube current, mechanical movement, and distance. The design
						shall be subject to approval by the department.
(3) All wall, ceiling, and floor areas that
						can be irradiated by the useful beam plus an additional area extending at least
						30 centimeters (1 foot) beyond shall be provided with a primary protective
						barrier.
(4) For equipment capable
						of operating above 150 kVp, the control station shall be outside of the therapy
						room.
(5) The enclosure shall be so
						constructed that individuals may at all times be able to escape from
						within.
(6) If the radiation
						exposure rate within the therapy room is so high that an individual who is
						accidentally in the treatment room when the machine is turned "ON" may receive
						as much as 1250 mR exposure during the time required to reach an access door,
						special cut-off, or panic buttons shall be required. When pressed, these
						buttons, operable by open hand at appropriate positions about the treatment
						room, shall cause the irradiation to be terminated.
(7) Effective means shall be provided to
						prevent access to the treatment room during exposure. For equipment capable of
						operating above 150 kVp, each access door to the treatment room shall be
						provided with a fail-safe interlock. The interlock system shall be so designed
						that the failure of any 1 component cannot jeopardize the safety of the system,
						such as the use of series connected double switch assemblies at access doors,
						and dual interlock relays. If an access door is opened when the machine is
						"ON", the interlock shall cause termination or reduction of exposure as
						specified in R 333.5312(14).
(8)
						Red warning signal lights, energized only when the useful beam in "ON", shall
						be located on the control panel and near each entrance to the therapy room.
						Under conditions as specified in subrule (6) of this rule a visible signal
						shall also be located within the therapy room. Depending upon control panel and
						door locations, a single warning signal light may be sufficient to comply with
						this subrule.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 7 - MEDICAL X-RAY INSTALLATIONS › THERAPEUTIC MACHINES OPERATED AT OR BELOW 85 KVP › Section R. 333.5321 - X-ray equipment

Current through Vol. 24-12, July 15, 2024
Rule 321.
(1)
						The x-ray equipment shall comply with the requirements of R 333.5312, excluding
						subrules (11), (14), and (16).
(2)
						Maximum potential shall be mechanically or electronically limited to 85
						kVp.
(3) A contact therapy machine
						shall meet the additional requirement that the leakage radiation at 5
						centimeters (2 inches) from the surface of the tube housing shall not exceed
						0.1 R/h. As used in this subrule, "contact therapy machine" means an x-ray
						therapy machine designed for source to skin treatment distances of 5
						centimeters or less at tube potentials in the range of 20 to 50 kVp.
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 7 - MEDICAL X-RAY INSTALLATIONS › THERAPEUTIC MACHINES OPERATED AT OR BELOW 85 KVP › Section R. 333.5322 - Enclosures

Current through Vol. 24-12, July 15, 2024
Rule 322. An enclosure shall comply with the
						  requirements of R 333.5315(1) and (2).
Michigan Administrative Code › Department - Labor and Economic Opportunity › MIOSHA › X-Ray Machine Registration › Ionizing Radiation Rules Governing the Use of Radiation Machines › Part 7 - MEDICAL X-RAY INSTALLATIONS › THERAPEUTIC MACHINES OPERATED AT OR BELOW 85 KVP › Section R. 333.5323 - Conditions of operation

Current through Vol. 24-12, July 15, 2024
Rule 323.
(1)
						Operation shall comply with the requirements of R 333.5317.
(2) If the x-ray tube of a contact therapy
						machine as defined in R 333.5321(3) is hand held during irradiation, the
						operator shall wear protective gloves and a protective apron. When practical, a
						cap of at least 0.5 millimeter lead equivalence should cover the aperture
						window of the tube housing of the apparatus when not being used. Because the
						exposure rate at the surface of the window of contact therapy and beryllium
						window machines may be more than 10,000 roentgens per minute, extreme
						precautions shall be taken to prevent accidental exposure to the useful
						beam.