reglab/statecodes huggingface dataset:
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart C - EQUIPMENT › Section 144-220-E-C-5 - Performance requirements for industrial radiography equipment

Current through 2024-29, July 17, 2024
Equipment used in industrial radiographic operations must
						meet the following minimum criteria:
A. Each radiographic exposure device, source
					 assembly or sealed source, and all associated equipment must meet the
					 requirements specified in American National Standard's Institute, N432-1980
					 "Radiological Safety for the Design and Construction of Apparatus for Gamma
					 Radiography" (published as NBS Handbook 136, issued January 1981). This
					 publication has been approved for incorporation by reference by the Director of
					 the Federal Register in accordance with
					 5 U.S.C.
					 552(a) and 1 CFR Part 51 .
B. Engineering analysis may be
					 submitted by a licensee to demonstrate the applicability of previously
					 performed testing on similar individual radiography equipment components. Upon
					 review, the Agency may find this an acceptable alternative to actual testing of
					 the component pursuant to the above referenced standard.
C. In addition to the requirements specified
					 in paragraphs E.5.A. and E.5.B. the following requirements apply to
					 radiographic exposure devices, source changers, source assemblies and sealed
					 sources.
(1) The licensee shall ensure that
					 each radiographic exposure device has attached to it a durable, legible,
					 clearly visible label bearing the:
a. Chemical
					 symbol and mass number of the radionuclide in the device;
b. Activity and the date on which this
					 activity was last measured;
c.
					 Model (or product code) and serial number of the sealed source;
d. Manufacturer's identity of the sealed
					 source; and
e. Licensee's name,
					 address, and telephone number.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart C - EQUIPMENT › Section 144-220-E-C-6 - Limits on external radiation levels from radiographic exposure devices, storage containers and source changers

Current through 2024-29, July 17, 2024
The maximum exposure rate limits for radiographic
						exposure devices, storage containers and source changers are 2 millisieverts
						(200 millirem) per hour at any exterior surface, and 0.1 millisieverts (10
						millirem) per hour at 1 meter (approximately 3 feet) from any exterior surface
						with the sealed source in the shielded position.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart C - EQUIPMENT › Section 144-220-E-C-7 - Locking of radiographic exposure devices, storage containers and source changers

Current through 2024-29, July 17, 2024
A. Each
					 radiographic exposure device must have a lock or outer locked container
					 designed to prevent unauthorized or accidental removal of the sealed source
					 from its shielded position. The exposure device and/or its container must be
					 kept locked at all times (and if a keyed lock, with the key removed at all
					 times) when not under the direct visual surveillance of a radiographer or
					 radiographer's assistant except at permanent radiographic installations as
					 stated in E.12. In addition, during radiographic operations the sealed source
					 assembly must be secured in the shielded position each time the source is
					 returned to that position.
B. Each
					 sealed source storage container and source changer must have a lock or outer
					 locked container designed to prevent unauthorized or accidental removal of the
					 sealed source from its shielded position. Storage containers and source
					 changers must be kept locked (and if a keyed lock, with the key removed at all
					 times) when containing sealed sources except when under the direct surveillance
					 of a radiographer or a radiographer's assistant.
C. Each radiographic exposure device, storage
					 container, and source changer shall be locked and the key removed from any lock
					 prior to being transported from one location and also prior to being stored at
					 a given location.
D. Locking and
					 permanent storage precautions.
(1)
					 Radiographic exposure devices, source changers, and transport containers that
					 contain sealed sources shall be secured while in storage to prevent tampering
					 or removal by unauthorized individuals.
(2) Radiographic exposure devices, source
					 changers, or transport containers that contain radioactive material may not be
					 stored in residential locations. This section does not apply to storage of
					 radioactive material in a vehicle in transit for use at temporary job sites, if
					 the licensee complies with all applicable sections of this Part and if the
					 vehicle does not constitute a permanent storage location as described in this
					 Part.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart C - EQUIPMENT › Section 144-220-E-C-8 - Radiation survey instruments

Current through 2024-29, July 17, 2024
A. Each
					 licensee and registrant shall keep sufficient calibrated and operable radiation
					 survey instruments at each location where radioactive material and radiation
					 machines are present to make the radiation surveys required by this Part and by
					 Part D. Instrumentation required by this section must be capable of measuring a
					 range from 0.02 millisieverts (2 millirems) per hour through 0.01 sievert (1
					 rem) per hour.
B. The licensee or
					 registrant shall have each radiation survey instrument required under paragraph
					 A of this section calibrated:
(1) At intervals
					 not to exceed six months and after instrument servicing, except for battery
					 changes;
(2) For linear scale
					 instruments, at two points located approximately one-third and two-thirds of
					 full scale; for logarithmic scale instruments, at mid-range of each decade, and
					 at two points of at least one decade; and for digital instruments, at 3 points
					 between 0.02 and 10 millisieverts (2 and 1000 millirems) per hour; and
(3) So that an accuracy within
					 plus or minus 20 percent of the calibration source can be demonstrated at each
					 point checked.
(4) By a person
					 licensed or registered by the Agency, another Agreement State, or the NRC to
					 perform such service;
(5) At
					 energies appropriate for the licensee's or registrant's use;
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart C - EQUIPMENT › Section 144-220-E-C-9 - Leak testing and replacement of sealed sources

Current through 2024-29, July 17, 2024
A. The replacement of any sealed source
					 fastened to or contained in a radiographic exposure device and leak testing of
					 any sealed source must be performed only by persons authorized to do so by the
					 Agency, the NRC or an Agreement State.
B. The opening, repair or modification of any
					 sealed source must be performed by persons specifically authorized to do so by
					 the Agency, the NRC or an Agreement State.
C. Testing and recordkeeping requirements.
(1) Each licensee who uses a sealed source
					 shall have the source tested for leakage at intervals not to exceed six months.
					 The leak testing of the source must be performed using a method approved by the
					 Agency, the NRC or by an Agreement State. The wipe sample should be taken from
					 the nearest accessible point to the sealed source where contamination might
					 accumulate. The wipe sample must be analyzed for radioactive contamination. The
					 analysis must be capable of detecting the presence of 185 Bq (0.005 microcurie)
					 of radioactive material on the test sample and must be performed by a person
					 specifically authorized by the Agency, the NRC, or an Agreement State to
					 perform the analysis.
(2) The
					 licensee shall maintain records of the leak tests in accordance with E.29.
(3) Unless a sealed source is
					 accompanied by a certificate from the transferor, that shows that it has been
					 leak tested within six months before the transfer, it may not be used by the
					 licensee until tested for leakage. Sealed sources that are in storage and not
					 in use do not require leak testing, but must be tested before use or transfer
					 to another person if the interval of storage exceeds six months.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart C - EQUIPMENT › Section 144-220-E-C-10 - Quarterly inventory

Current through 2024-29, July 17, 2024
A. Each licensee
					 and registrant shall conduct a physical inventory at intervals not to exceed
					 three months, to account for all sources of radiation and for devices
					 containing depleted uranium received and possessed under this license.
B. Each licensee and registrant
					 shall maintain records of the inventory in accordance with E.30.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart C - EQUIPMENT › Section 144-220-E-C-11 - Inspection and maintenance of radiation machines, radiographic exposure devices, transport and storage containers, associated equipment, source changers, and survey instruments

Current through 2024-29, July 17, 2024
A. Each
					 radiographer shall perform visual and operational checks on radiation machines,
					 survey meters, radiographic exposure devices, transport and storage containers,
					 associated equipment and source changers before use on each day the equipment
					 is to be used to ensure that the equipment is in good working condition, that
					 the sources are adequately shielded, and that required labeling is present.
					 Survey instrument operability must be performed using check sources or other
					 appropriate means. If equipment problems are found, the equipment must be
					 removed from service until repaired.
B. Each licensee and registrant shall have
					 written procedures for:
(1) Inspection and
					 routine maintenance of radiation machines, radiographic exposure devices,
					 source changers, associated equipment, transport and storage containers, and
					 survey instruments at intervals not to exceed three months or before the first
					 use thereafter to ensure the proper functioning of components important to
					 safety. Replacement components shall meet design specifications. If equipment
					 problems are found, the equipment must be removed from service until repaired.
					 All appropriate components shall be maintained in accordance with
					 manufacturers' specifications. Radiation machines, radiographic exposure
					 devices, transport containers and source changers being stored are exempted
					 from this requirement provided that each radiation machine, radiographic
					 exposure device, transport container, or source changer is inspected and
					 repaired prior to being returned to service. This inspection and maintenance
					 program shall cover, as a minimum, the items listed in Appendix B of this Part;
					 and
(2) Inspection and maintenance
					 necessary to maintain the Type B packaging used to transport radioactive
					 materials. The inspection and maintenance program must include procedures to
					 assure that Type B packages are shipped and maintained in accordance with the
					 certificate of compliance or other approval.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart C - EQUIPMENT › Section 144-220-E-C-12 - Permanent radiographic installations

Current through 2024-29, July 17, 2024
A. Each
					 entrance that is used for personnel access to the high radiation area in a
					 permanent radiographic installation must have either:
(1) An entrance control of the type described
					 in Part D.1601.A(1) that reduces the radiation level upon entry into the area,
					 or
(2) Both conspicuous visible and
					 audible warning signals to warn of the presence of radiation. The visible
					 signal must be actuated by radiation whenever the source is exposed. The
					 audible signal must be actuated, when an attempt is made to enter the
					 installation while the source is exposed.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart C - EQUIPMENT › Section 144-220-E-C-13 - Labeling, storage, and transportation

Current through 2024-29, July 17, 2024
A. The
					 licensee may not use a source changer or a container to store licensed material
					 unless the source changer or the storage container has securely attached to it
					 a durable, legible, and clearly visible label bearing the standard trefoil
					 radiation caution symbol conventional colors, i.e. magenta, purple or black on
					 a yellow background, having a minimum diameter of 25 millimeters, and the
					 wording:
CAUTION*
RADIOACTIVE MATERIAL NOTIFY CIVIL AUTHORITIES
(or "NAME OF COMPANY")
*________or "DANGER"
B. The licensee may not transport licensed
					 material unless the material is packaged, and the package is labeled, marked,
					 and accompanied with appropriate shipping papers in accordance with regulations
					 set out in Part L and 10 CFR Part 71 .
C. Locked radiographic exposure devices,
					 source changers, and storage containers must be physically secured to prevent
					 tampering or removal by unauthorized personnel. The licensee shall store
					 licensed material in a manner that will minimize danger from explosion or
					 fire.
D. The licensee shall lock
					 and physically secure the transport package containing licensed radioactive
					 material in the transporting vehicle to prevent accidental loss, tampering, or
					 unauthorized removal of the licensed material from the vehicle.
E. The licensee's name and city or town where
					 the main business office is located shall be prominently displayed with a
					 durable, clearly visible label(s) on both sides of all vehicles used to
					 transport radioactive material for temporary job site use.
F. The licensee shall ensure that each
					 radiographic exposure device has attached to it a durable, legible, clearly
					 visible label bearing the:
(a) Chemical symbol
					 and mass number of the radionuclide in the device;
(b) Activity and the date on which this
					 activity was last measured;
(c)
					 Model (or product code) and serial number of the sealed source;
(d) Manufacturer's identity of the sealed
					 source; and
(e) Licensee's name,
					 address, and telephone number.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-14 - Conducting industrial radiographic operations

Current through 2024-29, July 17, 2024
A. Whenever radiography is performed at a
					 location other than a permanent radiographic installation, the qualified
					 radiographer must be accompanied by at least one other qualified radiographer
					 or qualified radiographer assistant. This additional qualified individual shall
					 observe the operations and be capable of providing immediate assistance to
					 prevent unauthorized entry. Radiography may not be performed if only one
					 qualified individual is present.
B.
					 All radiographic operations conducted at locations of use authorized on the
					 license must be conducted in a permanent radiographic installation, unless
					 specifically authorized by the Agency.
C. A licensee may conduct lay-barge, offshore
					 platform, or underwater radiography only if procedures have been approved by
					 the Agency.
D. Whenever radiography
					 is performed at a location other than a permanent radiographic installation,
					 the radiographer must notify the requesting person's radiation safety officer
					 or the person responsible for safety matters for those that do not have a
					 radiation safety officer, before taking any licensed material onto the
					 location.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-15 - Radiation safety officer for industrial radiography

Current through 2024-29, July 17, 2024
The RSO shall ensure that radiation safety activities are
						being performed in accordance with approved procedures and requirements in the
						daily operation of the licensee's or registrant's program.
A. The minimum qualifications, training, and
					 experience for RSO's for industrial radiography are as follows:
(1) Possession of a high school diploma or a
					 certificate of high school equivalency based on the GED test.
(2) Completion of the training and testing
					 requirements of E.16;
(3) Two years
					 of documented experience including knowledge of industrial radiographic
					 operations; and
(4) Formal training
					 in the establishment and maintenance of a radiation protection
					 program.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-16 - Requirements for qualifications of radiographic personnel

Current through 2024-29, July 17, 2024
A. Radiographer assistant. No licensee or
					 registrant shall permit any individual to act as a radiographer assistant until
					 the individual possesses the original or a copy of an Agency-issued radiography
					 assistant status card or certification ID card.
(1) To obtain an Agency-issued radiography
					 assistant status card, the licensee, registrant, or the individual must
					 document to the Agency on Agency Form HHE-851 or equivalent that such
					 individual has successfully completed a course of at least 40 hours on the
					 applicable subjects outlined in E Appendix A. The course must be one accepted
					 by the Agency, another agreement state, or the NRC.
(2) The radiographer assistant must carry a
					 copy of the completed Agency Form HHE-851 listed above, in the interim period
					 after submitting documentation to the Agency and before receiving a
					 radiographer assistant status card. The copy of the completed HHE-851 that was
					 submitted to the Agency may be used in lieu of the radiographer assistant
					 status card for a period of 60 days from the date recorded by the radiographer
					 assistant on the documentation.
(3)
					 The individual shall notify the Agency by telephone, telegram, telefacsimile,
					 electronic media transmission, or in writing of the need for a replacement
					 radiographer assistant status card. The individual shall carry a copy of
					 documentation of the request while performing industrial radiographic
					 operations until a replacement radiographer assistant status card is received
					 from the Agency.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-17 - Operating and emergency procedures

Current through 2024-29, July 17, 2024
Operating and emergency procedures must include, as a
						minimum, instructions as outlined in Appendix C of this Part.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-18 - Personnel monitoring

Current through 2024-29, July 17, 2024
A. The licensee or
					 registrant may not permit any individual to act as a radiographer or
					 radiographer assistant unless, at all times during radiographic operations,
					 each individual wears, on the trunk of the body, a direct reading dosimeter or
					 an electronic personal dosimeter, an operating alarm ratemeter, and a personnel
					 dosimeter. At permanent radiography installations where other appropriate
					 alarming or warning devices are in routine use, the wearing of an alarming
					 ratemeter is not required.
(1) Pocket
					 dosimeters must have a range from zero to 2 millisieverts (200 milliroentgens)
					 and must be recharged at the start of each work shift. Electronic personal
					 dosimeters may only be used in place of ion-chamber pocket
					 dosimeters.
(2) Each personnel
					 dosimeter must be assigned to and worn by only one individual.
(3) Film badges must be replaced at least
					 monthly. and all other personnel dosimeters that require replacement must be
					 replaced at least quarterly. All personnel dosimeters must be evaluated at
					 least quarterly or promptly after replacement, whichever is more frequent.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-19 - Radiation surveys

Current through 2024-29, July 17, 2024
The licensee shall:
A. Not conduct a radiographic operation
					 unless calibrated and operable radiation survey instrumentation as described in
					 E.8 is available and used at each site where radiographic exposures are
					 made.
B. Use a survey instrument
					 that meet the requirements of E.8.A.-C. of this Part, to conduct a physical
					 survey of the entire circumference of the radiographic exposure device and the
					 guide tube after each exposure when approaching the device or the guide tube.
					 The survey must determine that the sealed source has returned to its shielded
					 position before exchanging films, repositioning the exposure head, or
					 dismantling equipment.
C. Conduct a
					 survey of the radiographic exposure device with a calibrated radiation survey
					 instrument any time the source is exchanged and whenever a radiographic
					 exposure device is placed in a storage area (as defined in E.3) to ensure that
					 the sealed source is in its shielded position.
D. Post potential radiation areas where
					 industrial radiographic operations are to be performed in accordance with E.22,
					 based on estimated dose rates, before industrial radiographic operations begin.
					 An area survey shall be performed during the first radiographic exposure to
					 confirm that E.22 requirements have been met and that unrestricted areas do not
					 have radiation levels in excess of the limits specified in Part D.
E. Meet the requirements of paragraph D of
					 this section, each time re-establishment of the restricted area is required.
					 The requirements of E.19.E do not apply to pipeline industrial radiographic
					 operations when the conditions of exposure including, but not limited to, the
					 radiographic exposure device, duration of exposure, source strength, pipe size,
					 and pipe thickness remain constant.
F. Perform a lock-out survey, in which all
					 accessible surfaces of the radiographic exposure device or source changer are
					 surveyed.
G. Perform surveys on
					 storage containers to ensure that the source is shielded and the radiation
					 levels do not exceed the limits specified in Part D. These surveys shall be
					 performed initially with the maximum amount of radioactive material present in
					 the storage location and thereafter at the time of the quarterly inventory and
					 whenever storage conditions change.
H. Perform a survey meeting the requirements
					 of E.19.B, on the radiographic exposure device and the source changer after
					 every sealed source exchange.
I.
					 Maintain records of the surveys required by E.19 in accordance with Subpart
					 E.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-20 - Requirements for underwater, offshore platform, and lay-barge radiography

Current through 2024-29, July 17, 2024
A. Underwater, offshore platform, and/or
					 lay-barge radiography shall not be performed unless specifically authorized in
					 a license issued by the Agency in accordance with E.4 of this Part.
B. In addition to the other requirements of
					 this section, the following requirements apply to the performance of offshore
					 platform or lay-barge radiography.
(1)
					 Cobalt-60 sources with activities in excess of 20 curies (740 GBq)(nominal) and
					 iridium-192 sources with activities in excess of 100 curies (3.7 TBq)(nominal)
					 shall not be used in the performance of offshore platform or lay-barge
					 radiography.
(2) Collimators shall
					 be used for all industrial radiographic operations performed on offshore
					 platforms or lay-barges.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-21 - Surveillance

Current through 2024-29, July 17, 2024
A. During each radiographic operation the
					 radiographic personnel, shall maintain continuous direct visual surveillance of
					 the operation to protect against unauthorized entry into a high radiation area,
					 as defined in Part A.2.A, except at permanent radiographic installations where
					 all entryways are locked and the requirements of E.12 are met.
B. Radiographic exposure devices shall not be
					 left unattended except when in storage or physically secured against
					 unauthorized removal or tampering.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-22 - Posting

Current through 2024-29, July 17, 2024
A. All areas in which industrial radiography
					 is being performed must be conspicuously posted as required by D.28. Exceptions
					 listed in D.29 do not apply to industrial radiographic operations.
B. Whenever practicable, ropes and/or
					 barriers shall be used in addition to appropriate signs to designate areas in
					 accordance with Part D.28 and to help prevent unauthorized entry.
C. During pipeline industrial radiographic
					 operations, sufficient radiation signs and other barriers shall be posted to
					 prevent unmonitored individuals from entering the area in accordance with Part
					 D.28.
D. In lieu of the
					 requirements of E.22.A, a restricted area may be established in accordance with
					 Part D and be posted in accordance with E.22.A, for example, both signs may be
					 posted at the same location at the boundary of the restricted area.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-23 - Reciprocity

Current through 2024-29, July 17, 2024
A. All reciprocal recognition of licenses or
					 certificates of registration by the Agency will be granted in accordance with
					 Parts C & E of this rule.
B.
					 Reciprocal recognition by the Agency of an individual radiographer
					 certification will be granted provided that:
(1) The individual holds a valid
					 certification in the appropriate category and class issued by a certifying
					 entity, as defined in this Part;
(2) The requirements and procedures of the
					 certifying entity issuing the certification afford the same or comparable
					 certification standards as those afforded by this Part; and
(3) The individual submits a legible copy of
					 the certification to the Agency prior to entry into Maine.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-24 - Radiation safety requirements for the use of radiation machines

Current through 2024-29, July 17, 2024
A. Locking of radiation machines. The control
					 panel of each radiation machine shall be equipped with a locking device that
					 will prevent the unauthorized use of an x-ray system or the accidental
					 production of radiation. The radiation machine shall be kept locked and the key
					 removed at all times except when under the direct visual surveillance of a
					 radiographer.
B. Permanent storage
					 precautions for the use of radiation machines. Radiation machines shall be
					 secured, while in storage to prevent tampering or removal by unauthorized
					 individuals.
C. Requirements for
					 radiation machines used in industrial radiographic operations.
(1) Equipment used in industrial radiographic
					 operations involving radiation machines manufactured after October 1, 1987,
					 shall be certified at the time of manufacture to meet the criteria set forth by
					 ANSI N537-1976, except accelerators used in industrial radiography.
(2) The registrant's name and city or town
					 where the main business office is located shall be prominently displayed with a
					 durable, legible, clearly visible label(s) on both sides of all vehicles used
					 to transport radiation machines for temporary job site use.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart D - RADIATION SAFETY REQUIREMENTS › Section 144-220-E-D-25 - Reserved

Current through 2024-29, July 17, 2024
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-26 - Records of the specific license for industrial radiography

Current through 2024-29, July 17, 2024
Each licensee shall maintain a copy of its license,
						license conditions, documents incorporated by reference, and amendments to each
						of these items until superseded by new documents approved by the Agency, or
						until the Agency terminates the license.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-27 - Records of receipt, transfer, and disposal of sealed sources

Current through 2024-29, July 17, 2024
A. Each licensee and registrant shall
					 maintain records showing the receipts, transfers, and disposal of sources of
					 radiation and devices using DU for shielding and retain each record for three
					 years after it is made.
B. The
					 records must include the date of receipt, transfer, or disposal, the name of
					 the individual making the record, radionuclide, number of becquerels (curies)
					 or mass (for DU); and manufacturer, model, and serial number of each source of
					 radiation and/or device, as appropriate.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-28 - Records of radiation survey instruments

Current through 2024-29, July 17, 2024
Each licensee and registrant shall maintain records of
						the calibrations of its radiation survey instruments that are required under
						E.8 and retain each record for three years after the calibration
						date.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-29 - Records of leak testing of sealed sources and devices containing depleted uranium

Current through 2024-29, July 17, 2024
Each licensee shall maintain records of leak test results
						for sealed sources and for devices containing DU the results must be stated in
						units of becquerels (microcuries). The licensee shall retain each record for
						three years after it is made or until the source in storage is
						removed.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-30 - Records of quarterly inventory

Current through 2024-29, July 17, 2024
A. Each
					 licensee and registrant shall maintain records of the quarterly inventory of
					 sources of radiation and devices containing depleted uranium required by E.10
					 and retain each record for three years after the date of the
					 inventory.
B. The record must
					 include the date of the inventory, name of the individual making the inventory
					 record, radionuclide, number of becquerels (curies) or mass (for DU) in each in
					 device, location of source of radiation, and manufacturer, model and serial
					 number of each source of radiation, as appropriate.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-31 - Utilization logs

Current through 2024-29, July 17, 2024
A. Each licensee
					 and registrant shall maintain logs showing for each sealed source the following
					 information:
(1) A description, including the
					 make, model and serial number, of the radiographic exposure device, radiation
					 machine, or transport and storage container in which the sealed source is
					 located.
(2) The identity and
					 signature of the radiographer to whom assigned; and
(3) The location(s) where used and date(s) of
					 use, including the date(s) removed and returned to storage.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-32 - Records of inspection and maintenance of radiation machines, radiographic exposure devices, transport and storage containers, associated equipment, source changers, and survey instruments

Current through 2024-29, July 17, 2024
A. Each licensee and registrant shall
					 maintain records specified in E.11 of equipment problems found in daily checks
					 and quarterly inspections of radiographic exposure devices, transport and
					 storage containers, associated equipment, source changers, survey instruments,
					 and radiation machines; and retain each record for three years after it is
					 made.
B. The record must include
					 the date of check or inspection, name of inspector, equipment involved, any
					 problems found, and what repair or maintenance, if any, was done.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-33 - Records of alarm system and entrance control tests at permanent radiographic installations

Current through 2024-29, July 17, 2024
Each licensee and registrant shall maintain records of
						alarm system and entrance control device tests required under E.12 and retain
						each record for three years after it is made.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-34 - Records of training and certification

Current through 2024-29, July 17, 2024
Each licensee and registrant shall maintain the following
						records (training and certification) for three years after the record is
						made:
A. Records of training of each
					 radiographer, and each radiographer's assistant. The record must include
					 radiographer certification documents and verification of certification status,
					 copies of written tests, dates of oral and practical examinations, as required
					 in E.16.D and names of individuals conducting and receiving the oral and
					 practical examinations. A copy of the radiographer assistant status card will
					 satisfy the documentation requirements of E.16.A and a certification ID card
					 will satisfy the documentation requirements of E.16.B; and
B. Records of annual refresher safety
					 training and semiannual inspections audits of job performance for each
					 radiographer and each radiographer's assistant. The records must list the
					 topics discussed during the refresher safety training, the dates the annual
					 refresher safety training was conducted, and names of the instructors and
					 attendees. For audits of job performance, the records must also include a list
					 showing the items checked and any non-compliances observed by the
					 RSO.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-35 - Copies of operating procedures, emergency procedures, and internal audit requirements

Current through 2024-29, July 17, 2024
A. Each licensee
					 and registrant shall maintain a copy of current operating and emergency
					 procedures until the Agency terminates the license or certificate of
					 registration. Superseded material must be retained for three years after the
					 change is made.
B. Records of the
					 internal audit requirements for the use of radiation machines and the use of
					 sealed sources shall be retained for three years from the date of the
					 audit.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-36 - Records of personnel monitoring procedures

Current through 2024-29, July 17, 2024
Each licensee and registrant shall maintain the following
						exposure records specified in E.19:
A.
					 Direct reading dosimeter or electronic personal dosimeter readings and yearly
					 operability checks for three years after the record is made.
B. Records of alarm ratemeter calibrations
					 for three years after the record is made.
C. Reports received from the individual
					 monitoring device processor until the Agency terminates the license or
					 certificate of registration.
D.
					 Records of estimates of exposure as a result of: off-scale personal direct
					 reading dosimeters, or lost or damaged personnel dosimeters until the Agency
					 terminates the license.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-37 - Records of radiation surveys

Current through 2024-29, July 17, 2024
Each licensee and registrant shall maintain a record of
						surveys required in E.19 and E.25.E for three years after it is
						made.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-38 - Form of records

Current through 2024-29, July 17, 2024
Each record required by this Part must be legible
						throughout the specified retention period. The record may be the original or a
						reproduced copy or a microfilm provided that the copy or microfilm is
						authenticated by the authorized personnel and that the microfilm is capable of
						reproducing a clear copy throughout the required period. The record may also be
						stored in electronic media with the capability for producing legible, accurate,
						and complete records during the required retention period. Records, such as
						letters, drawings, and specifications, must include all pertinent information,
						such as stamps, initials, and signatures. The licensee or registrant shall
						maintain adequate safeguards against tampering with and loss of
						records.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart E - RECORD KEEPING REQUIREMENTS › Section 144-220-E-E-39 - Location of documents and records

Current through 2024-29, July 17, 2024
A. Each
					 licensee and registrant shall maintain copies of records required by this Part
					 and other applicable Parts of this rule.
B. Each licensee and registrant shall also
					 maintain copies of the following documents and records sufficient to
					 demonstrate compliance at each applicable field station;
(1) The license or certificate of
					 registration authorizing the use of sources of radiation;
(2) A copy of Parts D, E, and J of the Agency
					 regulations;
(3) Utilization
					 records for each radiographic exposure device or radiation machine dispatched
					 from that location.
(4) Records of
					 equipment problems identified in daily checks of equipment,
(5) Records of alarm system and entrance
					 control checks, if applicable;
(6)
					 Records of direct reading dosimeters such as pocket dosimeter and/or electronic
					 personal dosimeter readings, if applicable;
(7) Operating and emergency
					 procedures;
(8) Evidence of the
					 latest calibration of the radiation survey instruments in use at the
					 site;
(9) Evidence of the latest
					 calibrations of alarm rate meters and operability checks of pocket dosimeters
					 and/or electronic personal dosimeters;
(10) Latest radiation survey records;
(11) The shipping papers for the
					 transportation of radioactive materials; and
(12) When operating under reciprocity; a copy
					 of the NRC or Agreement State License or certificate of registration
					 authorizing the use of sources of radiation.
Code of Maine Rules › 10 - DEPARTMENT OF HEALTH AND HUMAN SERVICES › 144 - DEPARTMENT OF HEALTH AND HUMAN SERVICES - GENERAL › Chapter 220 - Rules Relating to Radiation Protection › Part E - RADIATION SAFETY REQUIREMENTS FOR INDUSTRIAL RADIOGRAPHIC OPERATIONS › Subpart F - NOTIFICATIONS › Section 144-220-E-F-40 - Notifications

Current through 2024-29, July 17, 2024
A. In addition to the reporting requirements
					 specified under other sections of this Part and other applicable Parts of this
					 rule, each licensee shall provide a written report to the Agency; Radiation
					 Control Program, Maine Center for Disease Control and Prevention, Department of
					 Health and Human Services, 11 State House Station, Augusta, Maine 04333-0011,
					 within 30 days of the occurrence of any of the following incidents involving
					 radiographic equipment:
(1) Unintentional
					 disconnection of the source assembly from the control cable;
(2) Inability to retract the source assembly
					 to its fully shielded position and secure it in this position;
(3) Failure of any component (critical to
					 safe operation of the device) to properly perform its intended
					 function;
(4) An indicator on a
					 radiation machine fails to show that radiation is being produced;
(5) An exposure switch on a radiation machine
					 fails to terminate production of radiation when turned to the off position;
					 or
(6) A safety interlock fails to
					 terminate x-ray production.