**真空采血管**（**Vacutainer**）是一种真空无菌玻璃或塑料管封闭的血液收集管，管内的真空促进预定的液体量輸入。这种真空管通常用来静脉采血，也有的用来收集尿液样本。真空管管壁有时会有一层添加剂来帮助稳定和保存样本。

真空采血管由碧迪医疗的Joseph Kleiner于1947年发明 ，Vacutainer是碧迪医疗销售真空采血管的商标名。

已隱藏部分未翻譯内容，歡迎參與翻譯。
Principles
----------

Drawing blood using a vacutainer. The clear plastic sheath around the top of the tube holds the needle. The orange guard will shield the needle after it is withdrawn from the patient to protect from accidental needle sticks.
Vacutainer needle are double ended, with one side being encased in a thin rubber coating for safety. When the needle is screwed into the translucent plastic needle holder, the rubber needle is inside the holder, and the exposed needle will be inserted into the vein. When a Vacutainer tube is inserted into the holder, its rubber cap is punctured by the inner needle and the vacuum in the tube pulls blood through the needle and into the tube. The filled tube is then removed and another can be inserted and filled the same way. The amount of air evacuated from the tube predetermines how much blood will fill the tube before blood stops flowing. 

Each tube is topped with a color-coded plastic or rubber cap. Tubes often include additives that mix with the blood when collected, and the color of each tube's plastic cap indicates which additives it contains.

Blood collection tubes expire because over time the vacuum is lost and blood will not be drawn into the tube when the needle punctures the cap.

Types of tubes
--------------

A tiger top tube after centrifugation to separate blood cells from serum
Vacutainer tubes may contain additional substances that preserve blood for processing in a medical laboratory. Using the wrong tube may make the blood sample unusable for the intended purpose. These additives are typically thin film coatings applied using an ultrasonic nozzle.

The additives may include anticoagulants (EDTA, sodium citrate, heparin) or a gel with density between those of blood cells and blood plasma. Additionally, some tubes contain additives that preserve certain components of or substances within the blood, such as glucose. When a tube is centrifuged, the materials within are separated by density, with the blood cells sinking to the bottom and the plasma or serum accumulating at the top. Tubes containing gel can be easily handled and transported after centrifugation without the blood cells and serum mixing.

Vacutainer blood tubes
The meanings of the various colors are standardized across manufacturers.

The term *order of draw* refers to the sequence in which tubes should be filled. The needle which pierces the tubes can carry additives from one tube into the next, so the sequence is standardized so that any cross-contamination of additives will not affect laboratory results.

### Containers containing coagulants

* Gold or "tiger" red/black top: clot activator and gel for serum separation
* Red top (plastic, not glass): clot activator but no serum separation gel
* Orange or grey/yellow "tiger" top: thrombin, a rapid clot activator, for stat serum testing

### Containers containing anticoagulants

* Green: sodium heparin or lithium heparin used for plasma determinations in clinical chemistry (e.g. urea and electrolyte determination). Sodium heparin collection tubes are the classically preferred tube for peripheral blood or bone marrow for cytogenetic studies. Lithium heparin is considered suboptimal for cytogenetics.[來源請求]
* Light green or green/gray "tiger": for plasma determinations.
* Purple or lavender: K2 EDTA. This is a strong anticoagulant and these tubes are usually used for complete blood counts (CBC). Lavender top tubes are generally used when whole blood is needed for analysis. Can also be used for some blood bank procedures such as blood type and screen. EDTA tubes are preferred by most molecular genetics laboratories for molecular genetic studies (DNA or RNA).[來源請求]
* Grey: sodium fluoride and oxalate. Fluoride prevents enzymes in the blood from working, by preventing glycolysis so a substrate such as glucose will not be gradually used up during storage. Oxalate is an anticoagulant.
* Light blue: sodium citrate. Citrate is a reversible anticoagulant, and these tubes are used for coagulation assays.
* Dark blue: EDTA. These tubes are used for trace metal analysis.
* Black: used for Erythrocyte Sedimentation Rate (ESR).

### Other

* Red (glass): contains no additives.
* Light yellow: contains sodium polyanethol sulfonate (SPS). Used for blood culture specimens or acid-citrate-dextrose (ACD), used for blood bank studies, HLA phenotyping, and paternity testing.
* Tan (glass or plastic): contains either sodium heparin (glass) or K2EDTA (plastic). Used for lead determinations. These tubes are certified to contain no lead.

History
-------

A pile of blood collection tubesBlood collection tubes are among the basic medical supplies taken by disaster relief programs to affected areas.
### History

Vacutainer technology was developed in 1947 by Joseph Kleiner, and is currently marketed by Becton Dickinson (B-D).

The plastic tube version, known as Vacutainer PLUS, was developed at B-D in the early 1990s by E. Vogler, D. Montgomery and G. Harper amongst others of the Surface Science Group as US patents 5344611, 5326535, 5320812, 5257633 and 5246666.

Vacutainers are widely used in phlebotomy in developed countries due to safety and ease of use. Vacutainers have the advantage of being prepared with additives, allowing easy multi-tube draws, and having a lower chance of hemolysis. In developing countries, it is still common to draw blood using a syringe or syringes.

參考
--

1. 1 2  Rosenfeld L.. Clin Chem. 2000, **46** (10): 1705–14  [2015-10-04]. PMID 11017957. （原始内容存档于2011-06-07）.
2. ↑ .  [2006-02-01]. （原始内容存档于2006-02-12）.
3. ↑ .  [2006-02-01]. （原始内容存档于2005-12-16）.
4. 1 2 .  [2006-02-01]. （原始内容存档于2006-01-29）.
5. ↑   (PDF).  [2007-05-30]. （原始内容 (PDF)存档于2014-11-13）.
6. ↑ .  [2019-05-02]. （原始内容存档于2019-04-28）.
7. ↑  Ames, A.C.; Bamford, E.. Annals of Clinical Biochemistry: An International Journal of Biochemistry and Laboratory Medicine. 1975, **12** (1–6): 151–55. PMID 15637911. doi:10.1177/000456327501200136.