MASTECH MY6243 Portable Digital LC Meter Capacitance Meter Capacitor Inductance Meter Tester The digital LC meter gives a direct reading of inductance and capacitance on a 3 1/2 digits LCD display. Six ranges from 2nF to 200uF and four ranges from 2mH to 2H give precision readings , which included virtually all capacitance and inductance used in electronic engineering laboratory, production, service shops and schools. Its battery operation, light weight and small size make it a truly portable instrument. Features: 21mm LCD display provided max. indication of 1999 6 anges from 2nF to 200uF 4 ranges from 2mH to 2H High accuracy in measuring Zero value automatically Dual-slope integration A/D converter Overload indication of "1" Safety desighed test probe Specifications: CAP: 2n/20n/200nF 1% ; 2/20/200F 2% Inductance: 2mH/20mH/200mH 2% ; 2H 5% General Characteristics: Size: 91mm x 189mm x 31.5mm Weight: Approx. 300g (battery included) Power: 1 * 6F22 (9V) Battery (not included) Package Size: 24 * 17 * 6.5cm / 9.5 * 6.8 * 2.5in Package eight: 661g / 1.5lb