DS2-1 Digital Force Gauge with Output, Cap. 18 Oz / 500 G / 5 N, 110V Measure compression and tension force Accuracy: 0.2% Full scale 1 LSD Capacity(Resolution): 18.00 (0.01 ozf) / 500.0 (0.1 gf) / 5.000 (0.001 N) Includes NIST Traceable Calibration Certificate Includes 6 standard measuring attachments, AC Adapter/Charger & Case ozf, gf, and Newton units are push-button selectable Real time and peak mode (selectable) Display: 4 digit, 12 mm high Display Update Rate: 10 times per second Reversible Display Overload capacity: 200% Full scale, display flashes beyond 105% full scale Data Processing Speed: 1,000 data/second (30 data/second rate selectable) Outputs: RS-232, Mitutoyo Digimatic and 1 VDC analog output Operating Temperature Range: 32 to 100 F (0 - 40 C) Overload alarm, low battery display indicator Power supply: Rechargeable NiMH battery pack or AC adapter (included) Setpoints: Programmable high/low setpoints with LCD indicators Housing Dimensions: 215 x 65 x 51 mm (L x W x H) Kit Contents: All DS2 gauges are sold in a complete kit with gauge, hard plastic carrying case, a set of six measuring adapters and AC adapter/charger. RS-232C, Digimatic and analog outputs 2 Year Warranty Ergonomic, heavy-duty metal construction110V AC Adapter Charger