Electronic Balance Weighing Sensor Load Cell 0-1Kg Product Name: Weighing Load Cell; Rated Load; 0-1Kg ;Ranted Output; 1.0946mV/V; Lineraity: =&lt;0.005%F.S; Hysteresis; =&lt;0.005%F.S Repeatability: =&lt;0.005%F.S; Zero Balance; -/+2 %F.S; Temp Effect on Zero: =&lt;0.005%F.S. 1centigrade; Temp Effect on Zero Output: =&lt;0.005%F.S. 10centigrade: Creep: =&lt;0.005%F.S./30minu Crner Error:-/+1uV Input Impedance: 410-/+10ohm; Output Impedance: 350-/+3ohm; Insulation: &gt;5000Mohm;/50V DC Compensated Temperature Range -10centigrade~40centigrade;C Operating Temperature Range: -20centigrade~60centigrade Safe Overload: 150%F.S.; Ultimate Overload: 200%F.S; Input End: Red+, Black-; Output End:Green+, White-; Test Insrtument: E2000; Force Testing Equipment: Actual Load Excitation Voltage; 5-10V DC Total Size:4.5 x 0.9 x 0.6cm / 1.8" x 0.3" x 0.2"(L*W*T); Thead Hole Diameter:2mm / 0.079" ;Cable Length: 8.3cm / 3.3" ;Weight:6g; Package Content: 1 x Weighing Load Cell