You will be given a definition of a task first, then some input of the task.
Given a passage with a context and an answer, construct a question on the basis of the information present in the passage. Construct the question in such a way that (i) it is unambiguous, (ii) it is answerable from the passage, (iii) its answer is unique (iv) it answer uses text phrases from the passage. Avoid creating questions that (i) can't be answered correctly without actually understanding the passage and (ii) uses the same words or phrases given in the passage.

Context: The characteristic ease of use, rapid time to result, and low cost of malaria rapid diagnostic tests (RDTs) promote their widespread use at the point-of-care for malaria detection and surveillance. However, in many settings, the success of malaria elimination campaigns depends on point-of-care diagnostics with greater sensitivity than currently available RDTs. To address this need, a sample preparation method was developed to deliver more biomarkers onto a malaria RDT by concentrating the biomarker from blood sample volumes that are too large to be directly applied to a lateral flow strip.', 'In this design, Ni-NTA-functionalized magnetic beads captured the Plasmodium falciparum biomarker HRPII from a P. falciparum D6 culture spiked blood sample. This transfer of magnetic beads to the RDT was facilitated by an inexpensive 3D-printed apparatus that aligned the sample tube with the sample deposition pad and a magnet beneath the RDT. Biomarkers were released from the bead surface onto the lateral flow strip using imidazole-spiked running buffer. Kinetics of HRPII binding to the Ni-NTA beads as a function of blood sample volume were explored prior to determining the effect of the proposed method on the limit of detection of Paracheck RDTs.', 'More than 80% of HRPII biomarkers were extracted from blood sample volumes ranging from 25 to 250L. The time required to reach 80% binding ranged from 5 to 60min, depending on sample volume. Using 250L of blood and a 30-min biomarker binding time, the limit of detection of the Paracheck Pf RDT brand was improved by 21-fold, resulting in a limit of detection below 1 parasite/L.
Answer: This approach has the sensitivity and simplicity required to assist in malaria elimination campaigns in settings with limited access to clinical and laboratory resources.
Output:
Does direct transfer of HRPII-magnetic bead complexes to malaria rapid diagnostic tests significantly improve test sensitivity?