In this task, you are given an abstract of article. Your task is to generate title for this article. Preferred titles are under thirty words.
One example: Alterations of the architecture of cerebral white matter in the developing human brain can affect cortical development and result in functional disabilities. A line scan diffusion-weighted magnetic resonance imaging (MRI) sequence with diffusion tensor analysis was applied to measure the apparent diffusion coefficient, to calculate relative anisotropy, and to delineate three-dimensional fiber architecture in cerebral white matter in preterm (n = 17) and full-term infants (n = 7). To assess effects of prematurity on cerebral white matter development, early gestation preterm infants (n = 10) were studied a second time at term. In the central white matter the mean apparent diffusion coefficient at 28 wk was high, 1.8 microm2/ms, and decreased toward term to 1.2 microm2/ms. In the posterior limb of the internal capsule, the mean apparent diffusion coefficients at both times were similar (1.2 versus 1.1 microm2/ms). Relative anisotropy was higher the closer birth was to term with greater absolute values in the internal capsule than in the central white matter. Preterm infants at term showed higher mean diffusion coefficients in the central white matter (1.4 +/- 0.24 versus 1.15 +/- 0.09 microm2/ms, p = 0.016) and lower relative anisotropy in both areas compared with full-term infants (white matter, 10.9 +/- 0.6 versus 22.9 +/- 3.0%, p = 0.001; internal capsule, 24.0 +/- 4.44 versus 33.1 +/- 0.6% p = 0.006). Nonmyelinated fibers in the corpus callosum were visible by diffusion tensor MRI as early as 28 wk; full-term and preterm infants at term showed marked differences in white matter fiber organization. The data indicate that quantitative assessment of water diffusion by diffusion tensor MRI provides insight into microstructural development in cerebral white matter in living infants.
Solution is here: Microstructural development of human newborn cerebral white matter assessed in vivo by diffusion tensor magnetic resonance imaging.
Explanation: This title is appropriate for the given abstract because the focus of this abstract is cerebral white matter assesment.

Now, solve this: CONTEXT Health care applications of autoidentification technologies, such as radio frequency identification (RFID), have been proposed to improve patient safety and also the tracking and tracing of medical equipment. However, electromagnetic interference (EMI) by RFID on medical devices has never been reported.
OBJECTIVE To assess and classify incidents of EMI by RFID on critical care equipment.
DESIGN AND SETTING Without a patient being connected, EMI by 2 RFID systems (active 125 kHz and passive 868 MHz) was assessed under controlled conditions during May 2006, in the proximity of 41 medical devices (in 17 categories, 22 different manufacturers) at the Academic Medical Centre, University of Amsterdam, Amsterdam, The Netherlands. Assessment took place according to an international test protocol. Incidents of EMI were classified according to a critical care adverse events scale as hazardous, significant, or light.
RESULTS In 123 EMI tests (3 per medical device), RFID induced 34 EMI incidents: 22 were classified as hazardous, 2 as significant, and 10 as light. The passive 868-MHz RFID signal induced a higher number of incidents (26 incidents in 41 EMI tests; 63%) compared with the active 125-kHz RFID signal (8 incidents in 41 EMI tests; 20%); difference 44% (95% confidence interval, 27%-53%; P < .001). The passive 868-MHz RFID signal induced EMI in 26 medical devices, including 8 that were also affected by the active 125-kHz RFID signal (26 in 41 devices; 63%). The median distance between the RFID reader and the medical device in all EMI incidents was 30 cm (range, 0.1-600 cm).
CONCLUSIONS In a controlled nonclinical setting, RFID induced potentially hazardous incidents in medical devices. Implementation of RFID in the critical care environment should require on-site EMI tests and updates of international standards.
Solution:
Electromagnetic interference from radio frequency identification inducing potentially hazardous incidents in critical care medical equipment.