Given a passage classify if the passage has a definite objective/aim/goal or not. Output '1' if the passage has a defininte objective/aim/goal and output '0' if the passage does not have a definite objective/aim/goal.

We previously localized protease-activated receptor 2 (PAR-2) on human spermatozoa and demonstrated that activation of PAR-2 by the mast cell (MC) product tryptase inhibits sperm motility. Importantly, tryptase-secreting MCs are encountered in the male and female genital tract, implying that MC-spermatozoa interactions may be as yet unrecognized factors affecting sperm fertilizing ability. In order to elucidate how tryptase via activation of PAR-2 acts in human spermatozoa, we studied intracellular signal transduction events.', 'Impairment of sperm motility by tryptase was not dependent on the presence of extracellular Ca2+ and tryptase did not alter intracellular Ca2+ levels. Pre-incubation with pertussis toxin (PTX) failed to prevent tryptase effects on sperm motility. Western blot analyses revealed that tryptase increased phosphorylation of the mitogen-activated protein kinases (MAPK) ERK1/2, an effect which was blocked by the MAPK pathway inhibitor PD98059. Pre-treatment of spermatozoa with this inhibitor also blocked the inhibtion of sperm motility evoked by tryptase.
0

The cost effectiveness of preoperative localization in cases of primary hyperpara-thyroidism has not been established. We analyzed the potential savings in operative time after localization with technetium 99m (99mTc) sestamibi scan.', 'Thirty-three patients had localization of a solitary adenoma with 99mTc-sestamibi. Measurement was made of the time required for adenomectomy, unilateral neck exploration (UNE), unilateral neck exploration and confirmation of one contralateral parathyroid gland (UNEC), or bilateral neck exploration (BNE).', 'The total operative time in minutes was 76.4 +/- 18.8 for adenomectomy; 87.5 +/- 20.4 for UNE; 105.6 +/- 25.0 for UNEC; and 117.9 +/- 26.7 for BNE. The time difference was significant between adenomectomy versus UNE, UNEC, and BNE. There were also significant time differences between UNE versus UNEC and BNE.
0

Experimentally measured pullout forces for stent grafts (SGs) are used in clinical discussions and as reference values in bench studies and computer simulations. Previous values of these forces are available from studies in which the SG was pulled out in the straight caudal direction. However, clinical and numerical studies have suggested that displacement forces acting on SGs are directed more anteriorly. The objective of this study was to measure pullout forces as a function of angulation and to test the hypothesis that pullout forces decrease with increasing angulation.', 'Six different SGs (Bolton Treovance, Cook Zenith Flex, Cook Zenith LP, Medtronic Endurant, Medtronic Talent, and Vascutek Anaconda) were deployed in fresh bovine aortas, then pulled out by an electronic motor at 1 mm/s, while tension force was measured continuously with a digital load cell. The SG off-axis angulation was changed from 0 to 90 degrees in increments of 10 degrees. The test system was submerged in a custom-built temperature-controlled saline bath at 37C. At least three tests were performed for each device at each angle (with the exception of the Cook Zenith Flex, which experienced plastic deformation of its barbs after a single test per device). Each aortic specimen was used only once and then discarded. Hand-sutured graft anastomoses were also tested at 0 degrees to provide a reference value.', 'A total of 374 pullout tests were performed for the SGs and anastomoses. Sixty-four tests were excluded because of failure of the aorta or apparatus before device pullout. The remaining 310 tests showed pullout forces that demonstrated a decrease in the average pullout force for all six devices from 0 to 90 degrees (Bolton Treovance from 39.3 N to 23.9 N; Cook Zenith Flex from 59.8 N to 48.9 N; Cook Zenith LP from 50.3 N to 41.8 N; Medtronic Endurant from 29.9 N to 25.8 N; Medtronic Talent from 6.0 N to 5.5 N; and Vascutek Anaconda from 37.0 N to 30.3 N). For reference, the mean pullout force for the hand-sutured anastomoses was 63 N.
1