By July 11 2003, the flu-like severe acute respiratory syndrome (SARS) infected more than 8437 people worldwide, causing 813 deaths. 1 Most cases were in mainland China, 2 Hong Kong, 3 and Taiwan. In Singapore, 238 people were infected and 33 died. The disease was severe, transmissible from person to person, and has caused clusters of disease among health-care workers in particular. SARS outbreaks also happened in South-East Asia, North America, Europe, and became the first pandemic of the 21st century. As of the time of writing, 13 more people (with 1 dead) were diag-nosed with SARS in Beijing and Anhui (May 2004). It is estimated that SARS has caused a financial loss over U.S. $35 billion worldwide.
Hitherto, the diagnosis of SARS is based on clinical and epidemiologic considerations. Interestingly, the criteria for confirmed or suspected diagnosis are not perfect as yet. Fever, which is often thought essential for diagnosis, is also seen in patients with common cold; thus, it is problematic to define "suspected cases" on the basis of fever. Sending the suspected cases to hospital for specific laboratory tests is thus necessary after confirmation of the person's actual body temperature reading with usual clinical thermometer. As quoted in the Canada National Post: 4 "SARS scanners praised as placebo. Health Canada report: Cost $2-million, of the 462 000 people screened in the first month of operation, 341 had fevers, but no SARS cases, yet 'build confidence'." Current infrared (IR) systems in use at various boarder checkpoints have not been scientifically validated particularly in regards to the false negative rate. [4] [5] [6] [7] As a result they may create a false sense of security by underestimating the number of febrile (and possibly infected) individuals. The unadjusted mode threshold temperature setting in a thermal imager needs to correct for the difference between the skin and core body temperatures. It then has to take into account the effects of ambient conditions and the thermal imager's performance parameters.
Presently, there is a lack of empirical data in correlating facial skin with body temperature. Obtaining a meaningful temperature for the human body therefore requires identifying a suitable facial site that will provide reliable data across a large population since skin temperature does not solely depend on body temperature and may be affected by other physiological and environmental factors. Besides, the use of skin temperature as a proxy to the body temperature may be inaccurate at some stages of fever. When the fever starts, the body intends to raise its temperature and vasoconstriction happens to prevent a reduction heat loss through the skin. The skin temperature will, hence, be cold at that moment where the IR radiation may not reach the set threshold for the thermal scanner to pick out. As also known, a few patients who have underling diseases (such as cancer, diabetes, hypertension, or chronic obstructive pulmonary disease) or in the postoperative state would present without fever at the early stage of SARS infection. Thus, the thermal scanner and even the clinical diagnostic criteria have important limitations that might lead to false positive or negative diagnoses in the worse case. This technical note aims to discuss the effectiveness of thermal scanner for blind screening of fever subjects.
Biostatistics with regression analysis, 8 Receiver Operating Characteristics (ROC), 9 and neural networks based classification 13 is applied to analyze the blind data collected (502 without duplicate measurements, confirmed later 86 febrile and 416 healthy cases with ear thermometer) from the SARS hospital [A & E Dept., Tan Tock Seng Hospital (TTSH)] and the Civil Defense Force Academy (SCDF) in Singapore (screening ambient temperature is 25± 2°C, humidity around 60%) in which thermal imagers are used as a first line tool for the blind screening of hyperthermia. The subjects are considered febrile if his/her mean ear temperature is Ͼ = 37.7°C for adult (37.9°C for children) using Braun Thermoscan IRT 3520+. Results are drawn for the two important pieces of information: the best and yet practical region on the face to screen and guidance on optimal preset threshold temperature for the handheld radiometric IR Ther-maCAM S60 FLIR system. [10] [11] [12] [13] The focal length from the subject to scanner was fixed at 2 m and the duration of time patients must be scanned was 3 s. The detector was a focal plane array, uncooled microbolometer 320ϫ 240 pixels with a thermal sensitivity of 0.06°C at 30°C, spectral range of 7.5-13 m and measurement accuracy at ±2% of the realtime reading. 14 The average temperature of the skin surface was measured from the field of view of the thermal imager with an appropriate adjustment for skin emissivity. Human skin emissivity may vary from site to site ranging from 0.94 to 0.99 (0.98 was used here). The following spots were logged and analyzed from the frontal profiles: forehead; eye region; average cheeks; nose; mouth (closed); and average temples. The images were further processed to obtain readings from both frontal and side profiles from some of the subjects. Figures 1 and 2 present an example of the processed thermal images of both frontal and side (left) profiles from the same subject. The following spots were logged from the subjects with frontal and side profiles: forehead; eye region; average cheeks; nose; mouth (closed); average temple; side face; ears; and side temple. Figure 3 further illustrates an example of temperature profiles from a temperature operation using thermal imager with temperature reading. Reproducibility of both the instrument and physiological assumptions was established by comparing paired left-right readings of the temples and cheeks. 12 Figure 4 shows an example of the regression line obtained with skin temperatures extracted from the near inner eye range. Table I indicates that the thermal imagers hold much promise for mass screening when the readings from the front averaged temples region have good correlation with the ear temperature, followed by maximum temperature in near-eye and forehead regions. Figure 5 presents the ROC curve obtained and Table II is the ROC report. A randomly selected individual from the fever group has a test value larger than that for a randomly selected individual from the normal group in 90.7% of the time. The ROC test can distinguish between the normal and febrile groups and an optimum threshold temperature (marked as * in Table II) for the thermal imager, to detect an adult aural temperature of 37.7°C and above, is 34.6°C and this allows sensitivity of 90.7% and specificity of 75.8%. (In fact, by drawing a vertical line in Fig. 4 at skin temp= 34.6°C and a horizontal line at body temp= 37.7°C, one can quickly assess the sensitivity and specificity that matches the prediction from ROC with Med-Calc. Note that there are 3ϫ as many data points in the lower left box than in the top left box due to overlapping of operating points for normal temperature range.) This setting is dependent on the values of sensitivity and specificity (both to be high) from the ROC analysis results. Any temperature readings that exceed this setting will trigger off the alarm and there is need to verify the fact.
In the single layer perception backpropagation neural network (NN) experiment, 15 different parameters are tested at each time to choose the best parameter for multilayer perception evaluation in order to achieve a good score with different numbers of hidden neurons. In Kohonen network, the 10ϫ 10 Kohonen Layer has a better performance ͑Ͼ97.3% ͒ to classify the classes. Increasing the number of hidden neurons will improve the performance but the score will descend if the number of hidden neurons is exceeded. Thus, from the NN results, 13 it shows that such type of neural network 15 can be used as clinical decision support system for mass screening of febrile subjects. The current data are only valid for the groups on which it was taken in the indoor control environment, and that further work will be necessary to verify whether this approach yields reasonable sensitivity and specificity for actual massscreening tests in airports where factors such as environmental factors (if without aerobridge), 10 the physiological site offset, 11 and the performance characteristics of thermal imager 12 are crucial to warrant the most accurate and reliable screening operation. Such tests will have to be carefully planned and conducted, since the number of febrile subjects walking through airports, for example, is probably quite low. This step is thus essential, however, if we are to rely on IR imaging for SARS screening, since there is already general criticism that the method produces a false sense of security.
Generally, different people do not always measure success in the same way. In the case of SARS, thermography probably did play a positive role during the SARS crisis, as far as psychological warfare (political) was concerned. The lesson is that we could have done a much better job of actually finding SARS cases if the methodology used had been correct implemented.
In all, SARS is an evil affecting all humanity, not just some country or region. The mortality of SARS patients is considerable (4%-10%). Early detection and recognition of patients with severe involvement, and timely prevention and management, are critical in reducing mortality should there be any future outbreak. To prevent the spread of the infectious agent in SARS across the country boarder effectively, one should thus emphasize the importance of international collaboration as an effective strategy to control emerging diseases in people, particularly for those diseases with the potential to become pandemic such as SARS and avian-flu in which fever is one of the main symptom, etc. 
The author would like to express his appreciation to members of the Technical Reference Committee on Thermal Imagers under Medical Technology Standards Division by SPRING, Dr. K. H. Pwee, Director of CSTA, Ministry of Health, Dr. G. J. L. Kaw, Consultant Radiologist, TTSH of National Health Group, Singapore for sharing of their views and interests on "Thermal Imagers for Fever Screening-Selection, Usage and Testing." a)