Contact Service About Supplies Products Home Exceptional mini C-arm imaging begins here Designed around your needs, TAU 1512 introduces the next generation of mini C-arms. Exclusive dose-saving features, advanced Orthotouch interface, and improved articulation and arm design are all packaged in a small and light-weight form factor you expect from an Orthoscan mini C-arm. Get the quality and performance you desire in a package to meet any budget. DOWNLOAD BROCHURE REQUEST QUOTE Anatomy revealed See more on-screen content with a resolution of 1920 x 1200 in bright detail on a hi-bright 24″ monitor for an immersive work experience. Powered by Orthotouch 2.0 — the advanced touchscreen interface — with color interface, touch support, anatomical selections, and pediatric software. Expect exceptional diagnostic imagery with a 12″ x 9.5″ live image to see fine details of anatomy with vivid clarity. 24" DIAGNOSTIC MONITOR Orthotouch 2.0 ADVANCED TOUCHSCREEN INTERFACE 12″ x 9.5″ LIVE IMAGE Imaging evolved Orthoscan TAU 1512 is engineered from the inside out for durability and mobility in a light-weight frame. Surgeon driven design means navigating restrictive rooms and hallways while compact mode allows for simple storage and transportation. A mini C-arm that works with you 1. C-ARM DESIGN | DESIGNFocus on your case with an increased orbital rotation of 160°, larger arc depth of 20”, and improved umbilical cable. Effortlessly maneuver the C-arm to attain preferred views without stressing your patient’s anatomy. 2. BILATERAL CONTROLS | New backlit controls are easier to see with additional buttons for upgraded functionality. 3. LED LIGHTS | Insufficient lighting during a case can make it challenging to see anatomy clearly. Unique to the Orthoscan TAU family, surgical LED lights provide additional illumination on the anatomy. Eliminate shadows cast from the C-arm and the OR so you can work with certainty. Intelligent Dose Reduction At Orthoscan, we understand your concerns about radiation exposure and the demand for high-quality images. That’s why the TAU family includes cutting-edge Intelligent Dose Reduction that provides the best in diagnostic image quality while reducing exposure to you and your patients. up to 66% CUMULATIVE DOSE REDUCTION* OPTIMIZED DOSE FILTER NEXT GENERATION CMOS THE ONLY MINI C-ARM AVAILABLE WITH PEDIATRIC INDICATION While Orthoscan TAU is a low dose device by design, it is also the only mini C-arm designed for pediatric use . By engaging the Optimized Dose Filter, Orthoscan TAU can reduce dose significantly in children and adults. * Using Intelligent Dose Reduction (with Cu filter) when compared to Orthoscan 1000-0004. In continuous mode, users will not attain the stated levels of cumulative dose reduction possible with that offered in pulsed fluoroscopy. In clinical practice, the use of IDR may reduce patient dose depending. In clinical practice, the use of IDR may reduce patient dose depending on the clinical task, patient size, anatomical location, and clinical practice. A consultation with a radiologist and a physicist should be made to determine the appropriate dose to obtain diagnostic image quality for the particular clinical task. Surgeon driven design IMPROVED BILATERAL CONTROLS New backlit controls are easier to see with additional functionality. UPGRADED SERVICEABILITY Redesigned from the inside out to reduce service maintenance. THREE-WAY BRAKE CONTROL Maneuver your system with ease down hallways, around corners, and in the operating room. USER-FRIENDLY CONNECTIVITY Easy access to the power button and I/O ports directly on top of the chassis. INTEGRATED CABLE PUSHERS Protects power cords and other cables from being damaged. ENHANCED SECURITY Keep your equipment and data safe in the modern healthcare environment. ON-SCREEN HELP Description of common system functions available directly on-screen. What’s in a name? TAU is the name of a mathematical ratio commonly found in nature. It’s often referred to as the Golden ratio. This ratio can be used to create aesthetically pleasing, natural-looking compositions in designs. The TAU family was designed using this principle. Specifications US Metric MECHANICAL Free space 13.8″ Weight 475lb Height 48″ Footprint (W x L) 29″ x 33″ Orbital rotation (over rotation) 160° Pivot rotation 430° Lateral rotation (wig-wag) 320 ° Vertical range 26.5″ Arc depth 20″ Distance to cabinet 58″ Distance to wheel base 44″ Keyboard handles ✓ DETECTOR Type CMOS detector Detector resolution 2 k x 1.5 k Detector size 15cm x 12cm Useful array 15cm x 12cm Pixel spacing 75 microns Dose rate AKR / DAP MONITOR Display size 24″ LCD Primary “live” image 12″ x 9.5″ Dual reference image 5″ x 4″ Brightness 575 cd/m2 Resolution 1920 x 1200 Interface DVI Arm horizontal reach 26″ Arm vertical travel 14″ Arm vertical height 19″ Arm lateral rotation 216° + 370° + 200° Touchscreen ✓ HDMI (external monitor) ✓ Built-in DICOM calibration ✓ Color support ✓ ELECTRICAL Input power 90-253 VAC @ 47-63 Hz Sterile field controls Bilateral Multifunction wireless foot switch ✓ Laser alignment ✓ Surgical LED lights ✓ USB 2.0 port ✓ EMI filter ✓ X-RAY GENERATOR Focal spot 42.5 microns kV range 40 – 78 kVp mA range 0.040 – 0.160 mA Operating mode Continuous Collimator Fixed (normal, mag) Pediatric filter ✓ Magnification mode ✓ Documentation Wireless communication Capable SSD capacity 64GB Image capacity 26,000 Video capacity 14.4min Cine loop frame rate 30 fps Adaptive noise suppression Automatic Manual noise suppression 3 modes Cine loop export ✓ Edge enhancement ✓ Post process brightness / contrast ✓ Printer ✓ DICOM 3.0 compliant Storage SCU Worklist SCU Print SCU MPPS SCU DoseSR SCU Query SCU Storage Commitment BRAKES & WHEELS Braking system Lock, release, directional Ground clearance 4″ PERFORMANCE Start-up time 60 sec Shutdown time 20 sec Software Software Orthotouch™ 2.0 Operating system Windows 10 Pediatric workflow support ✓ On-screen measurements ✓ Ready for the next step? Request a quote PRODUCT BROCHURES TAU 1512 Brochure – US (2.0 MB) TAU Family Portfolio - US (3.1MB) Discover our other products UPGRADE YOUR EXPECTATIONS DISCOVER PORTABLE DIAGNOSTIC IMAGING DISCOVER VERSATILE DESIGN MEETS PROVEN C-ARM TECHNOLOGY DISCOVER The global leader in mini C-arm imaging. We are committed to setting new standards with unmatched quality, innovation, and service. Orthoscan Inc 14555 N 82nd St Scottsdale AZ 85260 United States Tel: 480.503.8010 Fax: 480.503.8011 Home Products Supplies About Service Contact Visit our global site 2021 Orthoscan, Inc. All rights reserved. Cookies Privacy Terms My account We use cookies on our website to give you the most relevant experience by remembering your preferences and repeat visits. By clicking “Accept”, you consent to the use of ALL the cookies. Do not sell my personal information. Cookie Settings Accept