Call +44 (0) 1726 879800 Contact About News Meet The Team Document Library Applications Measurements Products Services Home » TurbiTechw² HR Suspended Solids Sensor TurbiTechw² HR Suspended Solids Sensor Self Cleaning Sensor for Suspended Solids and Turbidity Monitoring The TurbiTechw² HR Sensor has been specifically designed for use in wastewater treatment systems where high levels of suspended solids are desirable in the aeration phase. Membrane Bio Reactors (MBR) typically run with MLSS values in the region of 8,000 to 14,000 mg/l. High MLSS concentrations promote numerous process benefits, including stable operation, complete nitrification, and reduced biosolids production reducing biological volume requirements (and associated footprint) to only 20-30 percent of conventional biological processes. For specialist advice call: +44 (0)1726 879800 ONLINE ENQUIRY View Related Product Documents Overview General Specification Measurement Specification Ordering Information & Spare Parts The TurbiTechw² HR sensor is suitable for monitoring solids in higher ranges than the standard LA version of the sensor due to it’s shorter path length. In addition the sensor incorporates a self cleaning mechanism that ensures the optical surfaces are kept clean at all times, the cleaning system is designed to avoid problems with ragging and does not smear the optical system. The cleaning system ensures that manual intervention on a routine basis is not required, the sensor should simply be checked as part of general site housekeeping. The large optical surface and sample volume combine to ensure that the sensor is providing information that is reliable and representative of the solids present in the process. Deposits of fats and grease on the sensing area do not prevent the sensor from measuring unlike sensors with small optical surfaces The TurbiTechw² HR Sensor uses infrared light at 860 nm and operates using the Light Attenuation Principle. The cleaning mechanism uses a polyurethane wiper seal, sealed by 2 Nitrile ‘H’ Rings that finish the cleaning process. The Nitrile seals can be exchanged for Viton if the process media dictates. The cleaning process is automatically initiated by the 7300w² Monitor at a user determined frequency. The cleaning process takes only 90 seconds, which means that the sensor is available for 99.5% of the time with a 6 hourly cleaning cycle. 7300w² Monitor The core product in the WaterWatch² range - the 7300w² Monitor provides all the user interface and system configuration More Details TurbiTechw² LA Suspended Solids Sensor Self Cleaning sensor designed for use in aeration systems monitoring Suspended Solids also known as Mixed Liquor Suspended Solids or Activated Sludge More Details TurbiTechw² LS Suspended Solids and Turbidity Sensor Self Cleaning Suspended Solids or Turbidity sensor designed for discharge and inlet monitoring More Details 750w² Portable Monitor The 750w² Monitor provides, quick and accurate readings of Suspended Solids, Sludge Blanket Level, and Turbidity. More Details Testimonial Quote from one of our Technical Support Engineers 'Managed to update to latest V1 firmware on the TurbiTech, customer was very happy that they had better resolution readings by having a free firmware upgrade. The engineer Shaun also asked me to pass on his thanks to everyone at Partech especially Victor, as he said the customer service Partech provide is far superior to any other supplier he deals with' UK Water Company June 29, 2018 Sign up for News I consent to my email address being used by Partech to send me news and updates. Follow us News Paul Deacon becomes Partech’s Operations Manager How do I know when my Partech sensor requires calibration? COVID-19 Safe Partech Training FAQ: How does Turbidity & Suspended Solids differ? Our Multi Skilled Sales Engineers at work Twitter Sorry, no Tweets were found. Top About Us Services Latest News Contact Us Terms & Conditions We'll assume you're ok with this, but you can opt-out if you wish.Accept Read More