APPLICATION FIELDS OUR PRODUCTS ABOUT US +49 911 376691-0 EN BlueEye 220 - 380 nm THE SOLUTION FOR LIFE SCIENCES AND MEDICAL TECHNOLOGY UV hyperspectral camera with a wavelength range from 220 to 380 nm The BlueEye is a pushbroom HSI camera that expands the common spectral range to include UV and covers UV-A, UV-B and UV-C. This makes it especially suitable for applications in biotechnology, forensics and medical technology. With the possibility of using absorption or fluorescence as a measurement signal, the camera opens up new fields of application that go beyond NIR and VIS technologies. For example, it can detect invisible contamination on surfaces and is suitable for ingredient analysis and cellular diagnostics. Do you have questions or are you interested in this product? Contact us, we will be happy to advise you! Key Features 220 … 380 nm wave length range back illuminated sCMOS sensor USB 3.1 interface different cooling options available 40 fps @ 2048 x 2048 pixels (max. sensor resolution with passiv cooling) free of periodical calibration of optical path components excellent optical imaging performance designed without moving mechanical parts Food processing industry Mechanical cultivation and harvesting methods as well as global transport and supply chains present the entire food processing industry with new challenges. HSI cameras from INNO-SPEC help with quality assurance: For example, they can track down foreign bodies, measure the composition of ingredients, or find spoiled items. TECHNICAL SPECIFICATIONS Spectograph Spectral range typ. typ. 220 to 380 nm Dispersion ~0.1 nm / px Smile < 160 µm Keystone < 20 µm F/# 2.4 Standard slit-width 80 µm Electronics Sensor Back-illuminated CMOS Sensor pixels 2048 x 2048 Active area (spatial x spectral Pixel) typ.1845 x 2048 Width 6.5 µm x 6.5 µm Bit depth 16 bit (2 x 12 bit ADC @ low & high gain) Frame rate ~ 40 fps full frame Data interface USB 3.1 Power supply USB 3.1 Typ C, < 4.5 W Sensor cooling Passive Operation Conditions Temperature (operating) +10 °C to +40 °C < 80% rel. humidity, non-condensing Temperature (storage) -10 °C to +60 °C Mechanics Dimensions l x w x h 140 x 70 x 165 mm Weight < 1.3 kg Lens mount Standard C-mount Florian Bolland Sales & Customer Care florian.bolland@inno-spec.de +49 911 376691-46 +49 159 0402 9112 We are happy to help! inno-spec GmbH Sigmundstraße 220 Eingang B7 90431 Nürnberg info@inno-spec.de +49 911 376691 - 0 +49 911 376691- 10 Privacy Policy | Imprint SEND All rights reserved Wir verwenden Cookies, um unsere Website und unseren Service zu optimieren. Cookie-Richtlinie - Impressum Alle akzeptieren verwerfen Vorlieben