+7 (843) 557-39-37 +7 (843) 560-57-24 RUS | ENG +7 (843) 564-82-27 Kazan, Vosstaniya st., 100 medtech@medtech.ru Catalog Suture material Absorbable sutures Conditionally absorbable sutures Non-absorbable suture NYLON MONO POLYPROPYLENE MONO POLYVINYLIDENE FLUORIDE MONO CAPRON WOVEN CAPRON ROUND LAVSAN WOVEN POLYESTER WOVEN FLOREST-S KRUCHEN Nomenclature of manufactured needles Instruments and products for traumatology Neurosurgery instruments Plastic Surgery Instruments Vascular surgery instruments Surgical Instruments Microinstruments for ophthalmology Disposable microinstruments for ophthalmology Microinstruments for cardiovascular surgery Surgical instrument sets news 16.01.2018 Laureate of the Republican Contest "Exporter of the Year" at the end of 2017 13.06.2017 Laureates of "TechUspeh" -2016 04/05/2017 Business trip of Exporters-manufacturers of medical devices About us news Contacts Dealers Demonstration films Instrument care home Product catalog Suture material Non-absorbable suture NYLON MONO NYLON MONO Nylon ( N) Analogs: POLYAMIDE, KAPRON MONO; ETHILON, MONOSOF, DAFILON Description: Non-absorbable synthetic monofilament suture material. Composition: aliphatic polyamide Color: blue or black Coverage: none Applications: skin sutures, general, plastic surgery, ophthalmology, microsurgery, microvascular surgery. stitching of bronchi, aponeurosis Resorption : does not dissolve, but has a loss of thread strength while in tissues, up to 10% within 1-1.5 years, over time it is encapsulated by connective tissue Tissue reaction : negligible Functional properties: Absence of capillarity and wicking. The thread is softer than polypropylene with a minimum "shape memory", provides reliable knitting and good knot security. Smooth passage through fabrics and no pulling effect on fabrics. Excellent visibility in the operating field. The thread has high tensile strength, smooth surface. Resistant to infections Contraindications: Due to a loss of tensile strength of approximately 10% within 1 year, the Nylon surgical suture should not be used in surgical procedures requiring long-term maintenance of the strength of the suture and should not be used in any cases where permanent suture strength is required, for example, in fixing a vascular prosthesis Adverse reactions: temporary edema and erythema around the wound, the thread is reactogenic, the tissues may react to it with slight inflammation Sterilization: gas method - ethylene oxide (EO) Range: it is possible to manufacture from 10/0 USP to 0 USP inclusive with needles and without needles Условия хранения: в упаковке изготовителя при температуре от +5 до +40°С, влажности не выше 80%, вдали от источников тепла, в месте, защищенном от влаги и прямых солнечных лучей Срок годности: гарантийный срок годности – 3 или 5 лет (в зависимости от типа упаковки) со дня стерилизации при соблюдении условий транспортирования и хранения Утилизация: использованные иглы и остатки шовного материала помещать в предназначенные для них контейнеры Внимание: информация на сайте носит ознакомительных характер и не может использоваться для включения в техническое задание. Для получения технического задания просим отправлять запрос по адресу: medtech@medtech.ru © 2017 OOO PTO "MEDTECHNIKA" Production of medical instruments, suture material. Kazan, Vosstaniya st., 100 medtech@medtech.ru +7 (843) 557-39-37 +7 (843) 560-57-24 +7 (843) 564-82-27