Online Analyzers: EZ Series Analyzers EZ Series Chloride Analyzers EZ Series Chloride Analyzers Overview Downloads Accessories Gallery Help The EZ Series Online Analyzers offer multiple options to monitor Chloride in water. Typical applications are wastewater, drinking water, steam & power generation, surface water. The EZ1005 Analyzers use colorimetric analysis to measure Chloride. Standard measuring range: 1 - 10 mg/L The EZ3000 Series Chloride Analyzers use direct ISE analysis: - EZ3003; standard measuring range: 1 - 10 mg/L - EZ3004; standard measuring range: 10 - 100 mg/L - EZ3005; standard measuring range: 100 - 1000 mg/L The EZ3500 Series Chloride Analyzers use ISE measurement with standard addition: - EZ3503; standard measuring range: 1 - 10 mg/L - EZ3504; standard measuring range: 10 - 100 mg/L - EZ3505; standard measuring range: 100 - 1000 mg/L The EZ4006 Chloride Analyzers are single-parameter titrators: Standard measuring range: 25 - 500 mg/L Measuring range options: - EZ1005: Internal dilution - EZ30xx/35xx: Calibration to 25% (not EZ3003) or 50% of standard range - EZ4006: Calibration to 10%, 25% or 50% of standard range Options for all analyzers include: - Multiple stream analysis (1-8 streams) reducing cost per sampling point - Analogue and/or digital outputs for communication Some applications may require the use of a sampling/filtration or an external dilution system for sample preconditioning. For this option please check the EZ9000 Sample Preconditioning page. Use the filters in the table below to find your online Chloride analyzer. Need support or a different configuration? Please contact us. Print PDF Page Compare Selected (0) Clear Selected Part Number Description Parameter Range Outputs Output USD Price EZ4006.99001C02 Compare EZ4006 Chloride Analyzer, 1 stream, Modbus RS485 Chloride 25 - 500 mg/L Cl- Optional: 5 - 50 mg/L Cl- 5 - 125 mg/L Cl- 5 - 250 mg/L Cl- Internal dispenser dilution (max. factor 100) Modbus RS485 Optional: Active 4 - 20 mA max. 500 Ohm load, 1 to 8 outputs Modbus TCP/IP Modbus RS485 Contact Hach EZ3003.99001C02 Compare EZ3003 Chloride Analyzer, 1 stream, Modbus RS485 Chloride 1 - 10 mg/L Cl- Optional: 0.5 - 5 mg/L Cl- Modbus RS485 Optional: Active 4 - 20 mA max. 500 Ohm load, 1 to 8 outputs Modbus TCP/IP Modbus RS485 Contact Hach EZ3004.99001C02 Compare EZ3004 Chloride Analyzer, 1 stream, Modbus RS485 Chloride 10 - 100 mg/L Cl- Optional: 2.5 - 25 mg/L Cl- 5 - 50 mg/L Cl- Modbus RS485 Optional: Active 4 - 20 mA max. 500 Ohm load, 1 to 8 outputs Modbus TCP/IP Modbus RS485 Contact Hach EZ3005.99001C02 Compare EZ3005 Chloride Analyzer, 1 stream, Modbus RS485 Chloride 100 - 1,000 mg/L Cl- Optional: 25 - 250 mg/L Cl- 50 - 500 mg/L Cl- Modbus RS485 Optional: Active 4 - 20 mA max. 500 Ohm load, 1 to 8 outputs Modbus TCP/IP Modbus RS485 Contact Hach EZ3503.99001C02 Compare EZ3503 Chloride Analyzer, 1 stream, Modbus RS485 Chloride 1 - 10 mg/L Cl- Optional: 0.25 - 2.5 mg/L Cl- 0.5 - 5 mg/L Cl- Modbus RS485 Optional: Active 4 - 20 mA max. 500 Ohm load, 1 to 8 outputs Modbus TCP/IP Modbus RS485 Contact Hach EZ3504.99001C02 Compare EZ3504 Chloride Analyzer, 1 stream, Modbus RS485 Chloride 5 - 100 mg/L Cl- Optional: 1 - 25 mg/L Cl- 2.5 - 50 mg/L Cl- Modbus RS485 Optional: Active 4 - 20 mA max. 500 Ohm load, 1 to 8 outputs Modbus TCP/IP Modbus RS485 Contact Hach EZ3505.99001C02 Compare EZ3505 Chloride Analyzer, 1 stream, Modbus RS485 Chloride 50 - 1000 mg/L Cl- Optional: 10 - 250 mg/L Cl- 25 - 500 mg/L Cl- Modbus RS485 Optional: Active 4 - 20 mA max. 500 Ohm load, 1 to 8 outputs Modbus TCP/IP Modbus RS485 Contact Hach EZ1005.99001C02 Compare EZ1005 Chloride Analyzer, 1 stream, Modbus RS485 Chloride 1 - 10 mg/L Cl- Optional: 4 - 40 mg/L Cl- (with internal dilution) 8 - 80 mg/L Cl- (with internal dilution) 100 - 1000 mg/L Cl- (with internal dilution) Modbus RS485 Optional: Active 4 - 20 mA max. 500 Ohm load, 1 to 8 outputs Modbus TCP/IP Modbus RS485 Contact Hach items per page 10 | 20 | 50 page 1 of 1 Customer Service Safety Data Sheets (SDS) Certificates of Analysis Chemical Terms Chemistries & Reagents EPA Methods Parameters Water Analysis Handbook CLAROS Resources FAQs Service Support (HSO) Pick & Ship Product Security Policies and Terms Privacy Policy Do Not Sell My Info About Us About Hach Careers Email Preferences Events Training Global Distributors Contact Us Danaher Water Platform WATER QUALITY PRODUCTS FOR GOVERNMENT BUYERS Learn More Small Business © Copyright 2021 Hach | PO Box 389, Loveland, CO 80539 | 800-227-4224 | 970-669-3050