Legal warning Privacy Policy 902 095 000 Start About Us Contact Equipment for Invasive and Non-Invasive Ventilation Neonatal non-invasive nCPAP ventilator LP interface for Infant Flow and Infant Flow SiPAP High Flow Systems High Frequency Oscillatory Fans (HFOV) Volumetric fans Air / Oxygen Mixers Bird CAREFUSION FANS LIST INFANT FLOW SIPAP Nasal CPAP fan (nCPAP) Noninvasive for Neonates, CareFusion brand, model Infant Flow Comprehensive SiPAP. It allows selecting different ventilatory modes adapted to the newborn: nCPAP, nCPAP / Apnea, Biphasic, Biphasic / Apnea and Biphasic Trigger. Digital touch screen for easy adjustment and monitoring of numerical parameters and pressure curve. Apnea detection in all ventilation modes by means of an impedance abdominal sensor. Monitoring of Inspiratory Pressures, Average, PEEP and Oxygen Concentration (FiO2). Integrated battery of two hours of autonomy. LP (Low Pressure) interface, as associated consumable. Download catalog LP INTERFACE FOR SIPAP LP (Low Pressure) interface , for use in conjunction with the Non-Invasive Nasal CPAP Ventilator (nCPAP) for Neonates, Infant Flow SiPAP Comprehensive. Six sizes of ARNESES depending on the head circumference of the patient. Allows ultrasound and tracing without removing or dislodging the harness. Minimizes pressure ulcers on the patient's face, due to the minimum pressure exerted in its placement. Cinco tallas de MASCARILLAS Y CANULAS NASALES, diferenciadas por código de colores. Disponibles en Tallas extra pequeña, pequeña, mediana, grande y extra grande. GENERADOR/CIRCUITO de doble jet para disminución del Trabajo Respiratorio y resistencia inspiratoria al flujo. Material de metano en extremo distal de rama espiratoria para reducción del ruido de hasta seis decibelios, permitiendo un entorno silencioso alrededor del paciente. Incorpora válvula pop-off de liberación de presión. Descargar Catálogo SENSORMEDICS 3100 - VAFO Ventiladores de Alta Frecuencia Oscilatoria (VAFO), marca CareFusion, SensorMedics 3100A y 3100B. Rango pacientes SM 3100A: prematuros-35 kg. Rango pacientes SM 3100B: 35 kg-obesidad mórbida. Diaphragm oscillator controlled by an electromagnetic coil. Adjustable amplitude, frequency and I: E ratio. It generates pressure waves at frequencies between 3-15 Hz, which are superimposed at a user-adjustable base pressure level. Active expiratory phase, minimizing entrapment intrinsic volume and PEEP. Therapeutic effects with only two controls: Oscillatory pressure amplitude (ΔP) and MAP. Changes in frequency or I: E ratio can also provide additional clinical benefits to the patient. Integrated High Flow Air-Oxygen Mixer. Download catalog Previous one two 3 Next SOFTFLOW 50 Innovative high-flow therapy that achieves an effective CO2 washing, reducing PaCO2 in patients with carbonic retention More information