EXILE Technologies Aquana Quantum Contract Manufacturing Investor Relations Careers COMPANY CAPABILITIES PRODUCTS NEWS SHOP CONTACT SENSORS GS-ONE LF GS-One LF Geophone PRODUCT DETAILS The GS-ONE LF 4.5Hz & 5Hz geophones’ improved sensitivity makes these single-element vibration monitoring systems’ designs compatible with the latest 24-bit data acquisition systems’ capabilities to record the extended dynamic range and bandwidth. FEATURES Moving mass: 25.2g (0.889 ounces) Maximum coil excursion p-p: Less than 3mm (0.120 inches) Diameter is 30.5 mm (1.2 inches) Height less terminals is 40.7 mm (1.6 inches) Weight is 131.00g (4.60 ounces) Operation and Storage Temperature Range -40 to 80°C Operation and Storage Temperature Range -40 to 176°F VERTICAL MECHANICAL SPECIFICATIONS Moving mass 25.2 g 0.889 oz Maximum coil excursion p-p 3.05 mm 0.120 in. Diameter 30.5 mm 1.2 in. Height 40.6 mm 1.6 in. Weight 131 g 4.62 oz Operating and storage temperature range –40°C to +80°C –40°F to +176°F ELECTRICAL SPECIFICATIONS All parameters are specified at 25°C in the vertical position unless otherwise stated. Open-circuit sensitivity 2.55 V/in/s (100.4 V/m/s) +10%, –10% Frequency 4.5 Hz ±0.75 Hz Spurious frequency 120 Hz typical Distortion at vertical 0.25% Typical measured at 12 Hz with 0.7 in/s p-p Open-circuit damping 0.42 ±0.10 Coil resistance 2450 Ω ±5% Sensitivity with 20 kΩ load 2.27 V/in/s (89.4 V/m/s) typical Damping with 20 kΩ load 0.735 typical Tilt angle when coil hits end stop 24° from vertical Typical HORIZONTAL MECHANICAL SPECIFICATIONS Moving mass 25.2 g 0.889 oz Maximum coil excursion p-p 3.05 mm 0.120 in. Diameter 30.5 mm 1.2 in. Height 40.6 mm 1.6 in. Weight 131 g 4.62 oz Operating and storage temperature range –40°C to +80°C –40°F to +176°F ELECTRICAL SPECIFICATIONS All parameters are specified at 25°C in the horizontal position unless otherwise stated. Open-circuit sensitivity 2.55 V/in/s (100.4 V/m/s) +10%, –10% Frequency 4.5 Hz ±0.75 Hz Spurious frequency 120 Hz typical Distortion at horizontal 0.25% Typical measured at 12 Hz with 0.7 in/s p-p Open-circuit damping 0.42 ± 0.10 Coil resistance 2450 Ω ±5% Sensitivity with 20 kΩ load 2.27 V/in/s (89.4 V/m/s) typical Damping with 20 kΩ load 0.735 typical Tilt angle when coil hits end stop 5° from horizontal SPECIFICATIONS (4.5 HZ) Vertical MECHANICAL SPECIFICATIONS Moving mass 25.2 g 0.889 oz Maximum coil excursion p-p 3.05 mm 0.120 in. Diameter 30.5 mm 1.2 in. Height 40.6 mm 1.6 in. Weight 131 g 4.62 oz Operating and storage temperature range –40°C to +80°C –40°F to +176°F ELECTRICAL SPECIFICATIONS All parameters are specified at 25°C in the vertical position unless otherwise stated. Open-circuit sensitivity 2.55 V/in/s (100.4 V/m/s) +10%, –10% Frequency 4.5 Hz ±0.75 Hz Spurious frequency 120 Hz typical Distortion at vertical 0.25% typical measured at 12 Hz with 0.7 in/s p-p Open-circuit damping 0.42 ±0.10 Sensitivity with 20 kΩ load 2.27 V/in/s (89.4 V/m/s) typical Damping with 20 kΩ load 0.735 typical Tilt angle when coil hits end stop 24° from vertical typical HORIZONTAL MECHANICAL SPECIFICATIONS Moving Mass 25.2 g 0.889 oz Maximum coil excursion p-p 3.05 mm 0.120 in. Diameter 30.5 mm 1.2 in. Height 40.6 mm 1.6 in. Weight 131 g 4.62 oz Operating and Storage Temperature Range –40°C to +80°C –40°F to +176°F ELECTRICAL SPECIFICATIONS All parameters are specified at 25°C in the horizontal position unless otherwise stated. Open-circuit sensitivity 2.55 V/in/s (100.4 V/m/s) +10%, –10% Frequency 4.5 Hz ±0.75 Hz Spurious frequency 120 Hz typical Distortion at horizontal 0.25% typicalmeasured at 12 Hz with 0.7 in/s p-p Open-circuit damping 0.42 ±0.10 Coil resistance 2450 Ω ±5% Sensitivity with 20 kΩ load 2.27 V/in/s (89.4 V/m/s) typical Damping with 20 kΩ load 0.735 typical Tilt angle when coil hit end stop 5° typical from horizontal CONTACT US TODAY RESOURCES Information Request Related Products GS-32CT GS-30CT MP-25 HS-1 3C SeisMonitor GS-1 Seismometer GS-1 3C SeisMonitor GS-One LF GS-ONE GS-14 Omni 2400 SMC-1850 Omni-X LT GS-20DX GS-20DM GS-20DH MP-26 Hydrophone MP-24 HS-1 GET IN TOUCH WITH US PRODUCTS Land Exploration Marine Exploration Downhole Smart Meter Connectivity Sensors COMPANY Values & Ethics HSE Quality Customer & Supplier Relations CAPABILITIES NEWS Events Announcements SHOP CONTACT EXILE Technologies Contract Manufacturing Quantum Investor Relations Careers © 2020 Geospace Technologies Privacy Policy Terms of Use Sitemap Design & development by versacreative.com