We are here for you! +49 2602 9449-0 info@modul-technik.de Product Quickfinder ambient cube Application area: Normal care Product group: Wall supply units (decorative) THE BEAUTY OF SIMPLICITY Who says functional supply solutions in hospitals must be sterile and boring? ambient cube proves the opposite by combining modern design with high-quality materials and functional technology to create a real eye catcher. Take a look for yourself. As you can see, you see nothing! That is apart from fine wood surfaces, since all the electrical installation elements disappear out of view underneath the module. As is the case with all our products, ambient cube can be furnished according to your specific wishes and equipped with a wide range of lighting options. You can of course also coordinate housing colours and décors individually to your own room design and to the other furniture. EXAMPLE DECORS TECHNICAL DATA Electric specifications Nominal voltage: 230 V - 240 V / 50 Hz - continuous operation Protection class: I Protection type: IP 20 Lighting technology Nominal voltage: 230 V - 240 V / 50 Hz Protection class: I Connection type: Plug connection Connection cross-section: 1.5mm² max. Protection type: IP 20 Reading light (LED): Output ≙ 23 W Reading light (T5 fluorescent tubes): Output ≙ 24 W / 2 x 24 W Indirect lighting (LED): Output ≙ 53 W Indirect lighting (T5 fluorescent tubes): Output ≙ 2 x 39 W Night light (LED): Output ≙ 4.3 W Operating pressure of medical gas technology Oxygen: 5 bar Compressed air: 5 bar Vacuum: - 0,8 bar General information Media current feed: Rear, from top or side Décors: HPL-laminate or melamine resin coating Optimum assembly height: 1650mm (rail bottom edge) Optional light control: DALI DIM STANDARD DESIGN High-voltage current technology 6 mains sockets, brand Peha (COMPACTA) 6 potential equalisation sockets (POAG) Communications technology 1 data socket, RJ 45, 2 pcs, brand Peha (COMPACTA) 2 hollow wall sockets, Ø 68mm (e.g. for nurse call button) Lighting technology Reading light (T5 fluorescent tubes): Output ≙ 24 W Indirect lighting (T5 fluorescent tubes): Output ≙ 2 x 39 W Medical gas technology 1 gas tapping point oxygen (O²), brand modul 1 gas tapping point compressed air (Air), brand modul 1 gas tapping point vacuum (Vac), brand modul General information Rear feed Dimensions (W x H x D): 1800 mm x 200 mm x 205 mm Décor: R 4889 Bella Noce schoko (melamine resin-coated) DOWNLOAD DATA SHEET ASSEMBLY, CLEANING, MAINTENANCE AND REPAIR Assembly Cleaning Maintenance and repair Accessories AMBIENT COMPACT AMBIENT JUNIOR AMBIENT LIGHT AMBIENT SIMPLEX COMFORT KUBION MEV 8000 MEV JUNIOR VIP 2000 modul technik GmbH · Rudolf-Diesel-Straße 5 · D-56410 Montabaur +49(0)2602-9449-0 +49(0)2602-9449-11 info@modul-technik.de E-MAIL TO MODUL TECHNIK Terms and conditionsDownloadsCareerCertificatesContactPrivacy policyImprint It will load necessary cookies, Google Fonts, Google Maps, OpenStreetMap, Youtube and Google Analytics. More details in our privacy policy and our imprint. Accept I disagree