+358 (0) 75 75 808 60tilaukset@timik.fi Yhteystiedot Career Tuotteet Huolto ja tuki Seminaarit ja kongressit Valiant Ultra 450 Home / Valiant Ultra 450 Valiant Ultra 450 – Multifunctional robust and reliable treadmill Ultra large treadmill for various applications. By choosing accessories and options the treadmill can be configured to your specific application. The standard configuration is as follows: Belt surface: 450 x 300 cm Speed: 0 - 40 km/h Start-up speed and speed steps: 0,1 km/h Positive elevation: 0 - 15.5% Negative elevation: -15.5 - 0 % Safety: fallstop, emergency stop button and safety lanyard Display: programmable 7" Control Unit showing most important ergometry parameters Handrails: adjustable and removable Connectivity: to be controlled by external ECG- and Pulmonary devices and by a Lode Software Network (LEM or LCRM) Maximum patient weight: 225 kg The functionality of this treadmill can be enhanced considerably by using it with the Lode Ergometry Manager Software (#955900). A USB A-B cable for service purposes as well as connecting to ECG and pulmonary testing devices will be standard delivered with the product. To connect older ECG and pulmonary testing devices with RS232 or other connectors you need a special interface cable that can be ordered separately. To connect LEM or LCRM you need a special interface cable that can be ordered under part number 930930. Small adjustment steps The speed of the treadmill can be adjusted in the smallest steps of only 0.1 km/h! Optional integrated BPM and SpO2 The treadmill can be extended with a stable and reliable blood pressure module and SpO2 measurement. Positive Elevation The positive elevation of the treadmill is standard 25%. This slope is enough for protocols used for the normal and healthy test subjects. Other elevations are available on request. Standard Safety Lanyard The product comes with a standard safety lanyard to enable automatic emergency stop Robust design The product is designed to withstand continuous heavy use by subjects in most weights and sizes. Versatile Interfacing Various interface protocols guarantee perfect communication with all commonly known stress ECG and spirometry equipment Service friendly belt treadmill Lode treadmills are almost maintenance free. Because of the unique impregnated belt, no belt-greasing and no bearing-greasing are required. The belt is centered automatically without additional adjustment. The only annual maintenance that has to be done is checking the running and electrical parts and the velocity calibration of the treadmill. Closed loop motor control This way of motor control enables (full brake at zero speed, load and angle independent speed and rigid speed control at all speeds, typically at interest when using steep angles, low speeds and/or high Test Subject weight). Workload Comfort Connectivity Dimensions Power Requirements Standards and Safety Environmental conditions Download Valiant Ultra 450 product sheet >>> TIETOA MEISTÄ Timik Medical Oy on Pohjoismainen terveysteknologia-alan yritys, joka on keskittynyt hoidollisiin ja diagnosoiviin terveydenhuollon laitteisiin, sekä tarvikkeisiin. Timik Medical on keskittynyt toiminnassaan myös urheilulääketieteessä, kuntoilussa ja kuntoutuksessa käytettäviin laitteisiin. Olemme omistautuneita saattamaan uusimmat hyvinvointiteknologian laitteet niiden ihmisten saataville, joiden jokapäiväiseen elämään vakava sairaus vaikuttaa. Timik Medical Oy:n yksityisyydensuojailmoitus Timik Medical Oy Innopark 2 Vankanlähde 7 13100 HÄMEENLINNA Tel: +358 (0) 75 75 808 60 info@timik.fi Timik Group AB Hammarbacken 4 A 191 49 SOLLENTUNA Tel: +46 (0)8-35 15 00 Org.nr 556968-6529 Search for: © 2021 Timik - Theme by Puro Tuotteet Huolto ja tuki Seminaarit ja kongressit Yhteystiedot