0 Career Contact Language Products Industries Support Downloads About PR ASK FOR A QUOTE 5337D 2-wire transmitter with HART protocol RTD, TC, Ohm, and bipolar mV input 2 analog inputs and 5 device variables with status available HART protocol revision selectable from HART 5 or HART 7 Hardware assessed for use in SIL applications Mounting in hazardous gas and dust area Overview Technical details Accessories Application Linearized temperature measurement with TC and RTD sensors e.g. Pt100 and Ni100. HART communication and 4...20 mA analog PV output for individual, difference or average temperature measurement of up to two RTD or TC input sensors. Conversion of linear resistance to a standard analog current signal, e.g from valves or Ohmic level sensors. Amplification of bipolar mV signals to standard 4...20 mA current signals. Up to 63 transmitters (HART 7) can be connected in a multidrop communication setup. Technical characteristics HART protocol revision can be changed by user configuration to either HART 5 or HART 7 protocol. The HART 7 protocol offers: ∙ Long Tag numbers of up to 32 characters. ∙ Enhanced Burst Mode and Event notification with time stamping. ∙ Device variable and status mapping to any dynamic variable PV, SV, TV or QV. ∙ Process signal trend measurement with logs and summary data. ∙ Automatic event notification with time stamps. ∙ Command aggregation for higher communication efficiency. 5337D is designed according to strict safety requirements and is therefore suitable for applications in SIL installations. Continuous check of vital stored data. Meeting the NAMUR NE 21 recommendations, the 5337 HART transmitter ensures top measurement performance in harsh EMC environments. Additionally, the 5337D meets NAMUR NE43 and NE89 recommendations. Mounting / installation For DIN form B sensor head mounting. Configuration via standard HART communication interfaces or by PR 5909 Loop Link. Manuals 5337V103_UK.pdf 5337V102_UK.pdf 5337V100_UK.pdf Data sheet 5337D.pdf ATEX certificate 5337D_issue10.pdf 5337D_issue09.pdf IECEx certificate 5337DIEC.pdf Declaration of Conformity 5337DoC_101_UK.pdf Safety note SN5300-3_107_1616.pdf SN5300_106_1509.pdf Russian manual 5337V102_RU.pdf 5337V103_RU.pdf CSA certificate 533x_633x_CSA.pdf FM certificate DNV marine certificate DNV_2231_41xx_51xx_53xx.pdf FMEDA report FMEDA_5337_R026V1R0.pdf FMEDA_5337_R026V3R1.pdf FMEDA_5337_R026V3R2.pdf SIL declaration 5335_6335_5337_6337_7501_SIL2_declaration_V1.pdf HART certificate 5337_HART.pdf INMETRO certificate INMETRO_5335_5337_issue1.pdf INMETRO_5335_5337_issue0.pdf EAC certificate EAC_Ex_2014.pdf Environmental Conditions Operating temperature -40°C to +85°C Calibration temperature 20...28°C Relative humidity < 95% RH (non-cond.) Protection degree (encl./terminal) IP68 / IP00 Mechanical specifications Dimensions Ø 44 x 20.2 mm Weight approx. 50 g Wire size 1 x 1.5 mm2 stranded wire Screw terminal torque 0.4 Nm Vibration IEC 60068-2-6 2...25 Hz ±1.6 mm 25...100 Hz ±4 g Common specifications Supply Supply voltage 8.0...30 VDC Isolation voltage Isolation voltage, test / working 1.5 kVAC / 50 VAC Response time Response time (programmable) 1...60 s Voltage drop 8.0 VDC Programming Loop Link & HART® Signal / noise ratio > 60 dB Accuracy Better than 0.05% of selected range Signal dynamics, input 22 bit Signal dynamics, output 16 bit EMC immunity influence < ±0.1% of span Extended EMC immunity: NAMUR NE 21, A criterion, burst < ±1% of span Input specifications Common input specifications Max. offset 50% of selected max. value RTD input RTD type Pt50, Pt100, Pt200, Pt500, Pt1000, Ni50, Ni100, Ni120, Ni1000 Cable resistance per wire (max.) 5 Ω (up to 50 Ω per wire is possible with reduced measurement accuracy) Sensor current Nom. 0.2 mA TC input Thermocouple type B, E, J, K, L, N, R, S, T, U, W3, W5, LR Cold junction compensation (CJC) Constant, internal or external via a Pt100 or Ni100 sensor Voltage input Measurement range -800...+800 mV Min. measurement range (span) 2.5 mV Input resistance 10 MΩ Output specifications Current output Signal range 4…20 mA Min. signal range 16 mA Load (@ current output) ≤ (Vsupply - 8) / 0.023 [Ω] Sensor error indication Programmable 3.5…23 mA NAMUR NE 43 Upscale/Downscale 23 mA / 3.5 mA Common output specifications Updating time 440 ms HART protocol revisions HART 5 and HART 7 Observed authority requirements EMC 2014/30/EU Approvals ATEX 2014/34/EU KEMA 03ATEX1537 IECEx KEM 10.0083X FM FM17US0013X CSA 1125003 INMETRO NCC 12.0844 X EAC TR-CU 020/2011 EAC Ex TR-CU 012/2011 RU C-DK.GB08.V.00410 DNV-GL Marine Stand. f. Certific. No. 2.4 SIL Hardware assessed for use in SIL applications 5909 Loop Link communications interface 7009 10-turn potentiometer, 200 Ω 7011 Dial for 10-turn potentiometer 7012 1-turn potentiometer, 1 kΩ 7020 Knob for 1-turn potentiometer 7020A Knob for 10-turn potentiometer 7028 10-turn potentiometer, 2 kΩ 7430B Pt100 cable sensor 7430C Pt100 cable sensor 7440 Thermowell for Pt100 sensor 8421 DIN rail fitting FEATURED LINKS HIGHLIGHTS - PRODUCTS Temperature 5337 temperature transmitter I.S. interfaces 4104 universal signal transmitter Multifunctional 3117 bipolar isolated converter Communication 3118 isolated converter/splitter Isolation 9116 universal converter (SIL) Displays 4511 communication enabler CONTACT US PR electronics P: +45 86 37 26 77 Email/ contact form GLOBAL OVERVIEW NEWSLETTER Signup to our newsletter and get all the latest news from us. Select country ©PR electronics 2013. All rights reserved. | Sitemap | Disclaimer | Privacy | Cookies