Call +44 (0) 1726 879800 Contact About News Meet The Team Document Library Applications Measurements Products Services Home » IL55BV2 Suspended Solids and Sludge Density Sensor IL55BV2 Suspended Solids and Sludge Density Sensor THIS PRODUCT IS OBSOLETE This product is no longer in production – please see SoliTechw² IL for new requirements Inline Sensor for sludge density, suspended solids and turbidity monitoring The IL55BV2 sensor provides reliable and repeatable, continuous in-line measurement of sludge density, suspended solids and turbidity. The optical element has a sapphire window, this offers excellent resistance to fouling and damage caused by the abrasion from flow particles. The sensor is suitable for a variety of applications including the auto de-sludging of primary and final settlement tanks, and reliable sludge density measurement ensures that downstream treatment processes can operate at their optimum level. For specialist advice call: +44 (0)1726 879800 ONLINE ENQUIRY View Related Product Documents Overview General Specification Measurement Specification Ordering Information & Spare Parts The sensor uses infrared back-scatter at 960nm, the light is transmitted and received through a lens that is optically split. The infrared measurement principle allows the sensor to work in applications where entrained air prevents the use of ultrasonic sensors. The sensor is installed through a mounting boss onto any pipeline that is larger than 65 mm diameter. The sensing element is positioned flush with the inside wall of the pipe, ensuring that the optical surface is kept clean by the flow passing the sensor and making sure the sensor does not cause an obstruction to the flow. When detecting turbidity the sensor can give immediate warning of a filter failure or of contamination of a process by suspended particles. The sensor is also well suited to the monitoring of sludge at tanker intake points, giving an excellent indicator of the strength of the incoming liquor allowing both process adjustments and the charges to be applied to the tanker operator. 7200 Monitor The 7200 Monitor combines with the Partech range of Suspended Solids and Turbidity Sensors More Details Turbi-Tech 2000HR Suspended Solids Sensor The Turbi-Tech 2000HR Sensor has been specifically designed for use in wastewater treatment systems where high levels of suspended solids are desirable in the aeration phase More Details Turbi-Tech 2000LA Suspended Solids Sensor Self Cleaning sensor designed for use in aeration systems monitoring Suspended Solids also known as Mixed Liquor Suspended Solids or Activated Sludge More Details Testimonial “At Kier Services we deliver approved ‘Partech’ Water Industry instrumentation for our clients and find them very professional and supportive with their services along with a personal approach. A pleasure to do business with” Warren Burson Kier June 9, 2014 Sign up for News I consent to my email address being used by Partech to send me news and updates. Follow us News Paul Deacon becomes Partech’s Operations Manager How do I know when my Partech sensor requires calibration? COVID-19 Safe Partech Training FAQ: How does Turbidity & Suspended Solids differ? Our Multi Skilled Sales Engineers at work Twitter Sorry, no Tweets were found. Top About Us Services Latest News Contact Us Terms & Conditions We'll assume you're ok with this, but you can opt-out if you wish.Accept Read More