Antenna Test & Calibration We promote experience. Antenna Systems Solutions offers a comprehensive antenna testing service tailored for your needs. All our measurements are carried in out in a type approved accredited anechoic chamber in Madrid (Spain) according to the ISO17025* standard. Spherical Near Field Planar Near Field Cylindrical Near Field Compact Range Please contact the sales office for a quotation and availability CONTACT Highlights & Facilities The antenna test ranges consists of three anechoic radiofrequency chambers: Spherical near field antenna measurements Compact range and a planar-cylindrical-spherical near field system in order to measure large antennas and RCS The two chambers share two complete radiofrequency and control systems, placed in two contiguous rooms. The anechoic chambers are equipped with air conditioning systems, with filters class 10000 (clean area), in order to measure on board satellite antennas. From 0.8 to 60 GHz Dimensions up to 2.5 meters (Satellite antennas, earth stations, cellular telephony base stations, active & passive arrays) Radar Cross Section Measurement of Radar Cross Section of materials from 6 to 40 GHz Radomes Measurement of radomes and similar structures for antennas from 0.8 to 40 GHz Our chambers Spherical near field antenna measurements & Compact range and a planar-cylindrical-spherical near field system in order to measure large antennas and RCS CHAMBER 1 CHAMBER 2a CHAMBER 2b Spherical Near-field Test Set-up The main characteristics of the first chamber are: Dimensions: 7.3 x 4.3 x 4.3 m. 18” and 12” pyramidal RAM to measure directly at frequencies above 3 GHz. (from 1.5 GHz with FFT gating techniques). AUT positioner system: roll over azimuth, able to support antennas up to 50 Kg. and 2m in diameter. Roll positioner is on a longitudinal table in order to allocate the antenna over the center of the measurement sphere. Probe positioner system: polarisation (roll) over an xyz manual positioner in order to align the probe’s z-axis to the centre of the sphere. Alignment system is made with levels, mirrors and an autocollimating theodolite. Rotary joints over all positioners. Measurement probes: axial corrugated conical horns of high polarisation purity. Standard gain horns: Self design to cover the complete frequency band. Frequency measurement range: 1.5 – 40GHz. Uncertainty for gain measurement: +/-0.164 dB for far field measurement procedure and +/-0.142 dB for near field measurement procedure (2σ). Uncertainty for CP pattern measurement (-30 dB level): +/-0.278 dB for far field measurement procedure and +/-0.204 dB for near field measurement procedure (2σ). Uncertainty for XP pattern measurement (-30 dB level): +/-0.643 dB for far field measurement procedure and +/-0.615 dB for near field measurement procedure (2σ). System supplier and producer of antenna metrology systems Products & Services Components Systems Software Calibration & Alignment Maintance Teaching Terms, Conditions & Warranty Customer Terms & Conditions Supplier Terms & Conditions Warranty Contact Info +34 942 05 04 85 info-es@asysol.com Isabel Torres 9, 39011 Santander, SPAIN 08:00 AM - 06:00 PM All rights reserved © 2021 | Antenna Systems Solutions S.L. | Cookies Policy | Legal Warning | Data protection policy