CRYOGENIC FLUIDS AND CRYOGENIC TECHNOLOGY +7 (495) 374 69 52 the main About company Products and services news Make an order Contacts CRYOGENIC INSTRUMENTS ANALYTICAL EQUIPMENT MEASURING EQUIPMENT CRYOGENIC EQUIPMENT Cryogenic vessels and containers Cryogenic refrigerators Nitrogen liquefiers Helium liquefiers, helium collection systems Cryogenic pipeline Cryogenic fittings Nitrogen batchers Overflow devices VACUUM EQUIPMENT CONSUMABLES DEVELOPMENT AND PRODUCTION Accessories: Dewars Overflow equipment See also: Nitrogen liquefiers Print a page Helium liquefiers, liquid helium vapor condensers, helium collection systems. CryoMech PT Reliquefier helium condensers are designed to recover evaporated liquid helium from flooded cryogenic systems such as superconducting magnets, incl. UIS, squid magnetometers, NMR spectrometers, etc. Helium condensers are fully automated and operate in a continuous mode. Allows to work with traditional liquid cryogenic systems without intermittent refrigerant filling. Recondensers of liquid helium vapors do not have their own storage capacity; liquefaction takes place directly into the filler neck of the cryogenic system. They are presented in two models with a capacity of 10 and 15 liters of liquid helium per day. Model 410 415 Productivity, l / day - in liquefaction mode - in recondensation mode 10 - 15 27 Time to reach operating mode, hours <4 <4 Power consumption, kW 7.2 9.2 Weight, kg 34 40 The operating time before the first service is at least 20,000 hours. CryoMech LHeP compact helium liquefiers are designed to liquefy gaseous helium coming from flooded cryogenic systems, from a helium network or cylinders. Liquefaction and accumulation of liquid helium occurs in a storage tank. The productivity of helium liquefiers is 15-60 liters per day. The gel liquefier can be equipped with a system for collecting and purifying gaseous helium of high or low pressure. Model LHeP15 LHeP22 LHeP40 LHeP60 Productivity, liters per day 15 22 40 60 Accumulating capacity, l 60 150 n / a 500 Time to reach operating mode * , hours <24 <36 n / a <36 Power consumption, kW 7.2 9.2 n / a 27.6 Weight, kg 338 463 n / a 1228 * When switched on on a warm Dewar vessel. Operating time until the first service is at least 20,000 hours. Helium liquefiers based on a cryogenic turbo expander manufactured by Linde Kryotechnik are designed to liquefy gaseous helium from a helium network or cylinders. Model L70 L140 L280 Unit productivity without precooling with liquid nitrogen, l / h 20-35 45-70 100-145 Unit capacity with liquid nitrogen precooling, l / h 40-70 90-140 200-290 Accumulating capacity, l the choice the choice the choice Helium collection system the choice the choice the choice Системы сбора газообразного гелия различных конфигураций: 1. Нормального давления на базе резинового мешка; 2. Среднего давления 10-15атм с накопительной емкостью от 500л; 3. Высокого давления 50-150атм Возможно изготовление системы сбора гелия под имеющуюся инфраструктуру с учетом специфики работы лаборатории. Компания ООО “Криотрейд” поставит гелиевые станции, системы сбора гелия, выполнит необходимые монтажные, пуско-наладочные и сервисные работы на территории России и СНГ. Сделать заказ Карта сайта Поиск по сайту Контакты (с) 2008-2014 г. ООО "КРИОТРЕЙД" Адрес: 123098, г. Москва, пл. Академика Курчатова, д. 1 Телефон/факс: +7 (495) 374 69 52, +7 (499) 196 60 95, эл. почта: sales@cryotrade.ru