Products News About us Contact GasEye Cross Duct CO GasEye Cross Duct HCHO GasEye Cross Duct NH3 HT GasEye Open Path GasEye Cross Duct in-situ tunable laser gas analyzers for industrial process monitoring For in-situ detection of CO, CO2, HCl, HF, O2, HCOH, C1-C5, NH3, SO2, SO3, H2O and others directly in process streams GasEye Cross Duct CO in-situ carbon monoxide analyzer for industrial process monitoring GasEye Cross Duct HCHO in-situ formaldehyde vapor analyzer for industrial process monitoring GasEye Cross Duct NH3 HT in-situ ammonia analyzer for industrial process monitoring GasEye Open Path tunable laser analyzers for air quality monitoring GasEye Cross Duct CO - in-situ carbon monoxide analyzer for industrial process monitoring GasEye CO is a high performance in-situ carbon monoxide analyzer. Suitable for safety applications and process control. 24/7 continuous operation. No sample preparation. No zero drift. No field calibration. Low cost of ownership. Features CO ranges 0-10 ppmv /0-1000 ppmv 0-1%/0-100% Process temperature 0-1200°C Real time sensing – response time below 0.1 second High selectivity – no interference from other constituents in the gas sample, (e.g. CO2, CH4, H2O, etc.) High sensitivity – detection limit below 0.2 ppm per meter In-situ monitoring – direct in the process, no sample preparation Maintenance free – equipped with a self-calibrating feature, no field calibration necessary Robustness – IP65 enclosure, suitable for outdoor and indoor installations and harsh environments Insensitive to dust and smoke in the measured process (up to 50 g/m3) ATEX version available Example Applications Combustion control (boilers) Safety monitoring Electrofilter Protection Example Industries Power industry Chemical industry Steel industry Food industry For pricing and more information about GasEye Cross-Duct CO contact us at: sales@airoptic.pl Get the product card.pdf GasEye Cross Duct HCHO - in-situ formaldehyde vapor analyzer for industrial process monitoring GasEye HCHO is the world’s first in-situ formaldehyde vapor analyzer. Suitable for emission monitoring and process control. 24/7 continuous operation. No sample preparation. No zero drift. No field calibration. Low cost of ownership. Features HCHO ranges 0-10 ppmv / 0-1 ppmv Process temperature 0-650 °C Real time sensing – response time below 0.1 second High selectivity – no interference from other constituents in the gas sample High sensitivity – detection limit below 10 ppb per meter In-situ monitoring – reliable and representative formaldehyde measure- ments in large spaces (e.g. wood-based panel production) Maintenance free – equipped with a self-calibrating feature, no field calibration necessary Robustness – IP65 enclosire, suitable for outdoor and indoor installations and harsh environments Insensitive to dust and smoke in the measured process (up to 50 g/m3) ATEX version available TÜV Certified EN 14181:2015, EN 15267-3:2007 Example Applications Wood-based panel production Coating and varnishing Engineered wood products Medical - Veterinary Coffee roasting Painting and corrosion resistant finishes Carpet manufacture & storage Adhesives Heat treatment operations Trailer mobile home inspecting Automotive emissions For pricing and more information about GasEye Cross-Duct HCHO contact us at: sales@airoptic.pl Get the product card.pdf GasEye Cross Duct NH3 HT - in-situ ammonia analyzer for industrial process monitoring GasEye NH3 HT is a high performance in-situ ammonia analyzer. Suitable for DeNOX processes and process control. 24/7 continuous operation. No sample preparation. No zero drift. No field calibration. Low cost of ownership. Features NH3 ranges 0-15 ppmv Process temperature 250-400°C Real time sensing – response time below 0.1 second High selectivity – no interference from other constituents in the gas sample, (e.g. CO2, CH4, H2O, etc.) High sensitivity – detection limit below 0.1 ppm per meter In-situ monitoring – direct in the process, no sample preparation Maintenance free – equipped with a self-calibrating feature, no field calibration necessary Robustness – IP65 enclosire, suitable for outdoor and indoor installations and harsh environments Insensitive to dust and smoke in the measured process (up to 50 g/m3) ATEX version available Example Applications real time ammonia slip DeNOx SNCR real time ammonia slip DeNOx SCR Example Industries Power industry MWI Automotive For pricing and more information about GasEye NH3 HT contact us at: sales@airoptic.pl Get the product card.pdf GasEye Open Path - tunable laser analyzers for air quality monitoring High selectivity our measurement technology allows for sensitive and selective measurement of gas concentration down to single ppb levels In-situ monitoring measurement of gas concentration takes place directly in-situ without any need for gas sampling Real-time measurement response time of our gas analyzers is below 0,1 second Maintenance free GasEye analyzers are equipped with self-calibrating feature, minimizing cost of ownership Mid-IR ready first to integrate interband cascade laser (ICL) technology Long tradition we have a long tradition in laser spectrometer design for industrial applications Flexible design our technology allows for easy adjustment to specific customer needs ranging from extractive through harsh combustion control environments For pricing and more information about GasEye platform contact us at: sales@airoptic.pl Get the product card.pdf Copyright 2017 AIROPTIC Sp. z o.o. Site by MINIMA