We are here for you! +49 2602 9449-0 info@modul-technik.de Product Quickfinder GENERAL TERMS AND CONDITIONS The general terms and conditions of modul technik GmbH apply. We would be happy to send you our terms and conditions on request. Download GTC modul technik GmbH · Rudolf-Diesel-Straße 5 · D-56410 Montabaur +49(0)2602-9449-0 +49(0)2602-9449-11 info@modul-technik.de E-MAIL TO MODUL TECHNIK Terms and conditionsDownloadsCareerCertificatesContactPrivacy policyImprint It will load necessary cookies, Google Fonts, Google Maps, OpenStreetMap, Youtube and Google Analytics. More details in our privacy policy and our imprint. Accept I disagree

We are here for you! +49 2602 9449-0 info@modul-technik.de Product Quickfinder moduversa HF Application area: Operating rooms Product group: Pendant systems OR FEATHERLIGHT AND ALWAYS ON YOUR SIDE Whether you are sitting or standing, moduversa HF is always by your side and offers optimum access in every position to all media connections and apparatus on the rotatable BSP 2500 supply head. The positionable system has infinite horizontal and vertical adjustments and with an arm length of 1 metre can adapt to every room size and work situation. The maximum arm load capacity of 180 kg is sufficient for most medical applications. The maintenance-free electromagnetic brake system and the smooth swing arm enable precise positioning in both a vertical and horizontal direction. And here the spring can be set precisely for the respective load. LED signal lamps on the braking system make accurate operation easier. Like all moduversa products, moduversa HF can be designed as a single or tandem system. TECHNICAL DATA Electric specifications Nominal voltage: 115 V - 240 V / 50-60 Hz - continuous operation Protection class: I Protection type: IP 20 Operating pressure of medical gas technology Oxygen: 5 bar Compressed air: 5 bar Compressed air: 8 bar Vacuum: - 0,8 bar Laughing gas 5 bar CO2: 5 bar AGSS: 5 bar Air-Motor: 8 bar General information Media current feed: From the top in the support arm Braking system: Electromagnetic (optional compressed air) Comfort option: Braking system with BrakeGuide*(optical feedback) Total additional load of supply unit: Project-related Height-adjustment: Spring force * BrakeGuide cannot be combined with compressed air brake Support arm length 1000 mm Max. additional load support arm 180 kg DOWNLOAD DATA SHEET ASSEMBLY, CLEANING, MAINTENANCE AND REPAIR Assembly Cleaning Maintenance and repair Accessories DVE 5500 MODUFILIO MODUVERSA MODUVERSA HF PLUS MODUVERSA HM MODUVERSA HM PLUS MODUVERSA PLUS modul technik GmbH · Rudolf-Diesel-Straße 5 · D-56410 Montabaur +49(0)2602-9449-0 +49(0)2602-9449-11 info@modul-technik.de E-MAIL TO MODUL TECHNIK Terms and conditionsDownloadsCareerCertificatesContactPrivacy policyImprint It will load necessary cookies, Google Fonts, Google Maps, OpenStreetMap, Youtube and Google Analytics. More details in our privacy policy and our imprint. Accept I disagree