Open navigation Open language menu Open search ASTOR Precise and manually adjustable room lasers for patient alignment at LINAC isocenter Key figures 0 units per year 0 degrees of freedom 0 mounting options 0 laser colors Cost-conscious positioning precision Our ASTOR room lasers allow precisely reproducible positioning of the patient on the linear accelerator. The finest laser lines, with a max. 1 mm width, project the isocenter for highly precise patient positioning. Room lasers are also a fundamental element of phantom alignment for quality assurance. The ASTOR system has three (optionally four) lasers for alignment with skin markings. You can choose laser color, design variant, and line or cross lasers, all individually adapted to your room situation and workflow. A fourth laser can be used to prevent potential shadowing from the gantry. Ultra-precise laser lines in red or green are made possible by state-of-the-art precision mechanics and optics. ASTOR room lasers are reduced to the maximum, so even a modest budget can achieve the highest degree of precision. Features of ASTOR room laser Precise Even at a distance of 4 meters, the maximum laser line width is 1 mm. The line straightness within the line shows barely any deviations, even at a distance of 3 meters. You can always be sure of the precision of our lasers. Optimally visible The optical room laser components are designed and manufactured with extreme accuracy. Plane parallel glass with anti-reflective coating prevents any interference with the red or green laser beam. The brightness distribution remains constant at all times. Benefit from optimal visibility for precise patient positioning. Robust A high-quality aluminum housing reliably protects the fine mechanics and optical components inside. You don't need to fear damage even from gentle impacts. Highly shatterproof glass stands up to all the rigors of clinical daily life. Flexible Adapt the room laser design to the circumstances of your space. Direct wall mounting or on a freestanding floor support are both possible. In addition, the room laser can be inclined up to 45°. This gives you flexibility even in tight spaces. Resistant to radiation Scattered radiation occurs in environments surrounding linear accelerators. The components in our room lasers have been tested under extreme exposure conditions and optimized for these conditions. The many years of availability without compromising on quality is an exceptional track record that speaks for itself. Durable The quality of our products is also reflected in their service life – and in our support. Should you need any service, we are quick to respond and offer the right solution for you. This is how we work for a sustainable partnership. Technical data Laser color (typical wave length) red (635 nm), green (520 nm) Laser class 2 Line width up to 4 m distance < 1 mm Line length at 3 m distance > 3 m Power supply 100 ... 240 V AC Internal voltage 24 V DC Operating temperature 15 … 30 °C Dimensions (H × W × D) 182 × 86 × 90 mm Weight 1.4 kg Downloads PDF 1 MB ASTOR Brochure PDF 1,017 KB ASTOR Data sheet CONTACT US! INDUSTRIES PRODUCTS SERVICE COMPANY CAREER LEGAL Visit us on XING Visit us on LinkedIn Visit us on Youtube Names of other companies and products displayed on this website can be trademarks or registered trademarks which do not belong to LAP, but to their respective owners. © 2021 LAP GmbH Laser Applikationen / Imprint / Privacy policy / Sitemap / Cookie preferences