Contact Join us Your First Light Who are we ? Products Downloads News Applications Software Find a distributor PRODUCTS > OCAM²S OCAM²S FAST LOW LIGHT LOW NOISE ELECTRONIC SHUTTERED SCIENTIFIC THEORY OF OPERATION OCAM²S is a high speed low noise camera with an integrated electronic shutter able to run at 2067 FPS with subelectron readout noise. To achieve this performance, OCAM²S uses the E2V CCD 219 EMCCD, an 8 output split frame transfert CCD. To minimize smearing, the CCD219 high speed metal buttressed clock lines are driven by OCAM²S at a speed as high as 7 Mlines/s transferring each frame in the store section in only 12 microseconds. The Embedded Electronic Shutter in OCAM²S is able to deliver an arbitrary number of integration pulses that can be shorter than 1 µs with a precision better than 50ns. OCAM²S also offers an extremely low latency : 43μs between exposure and first pixel availability. APPLICATIONS ASTRONOMY LIFE SCIENCES / RESEARCH EXO-PLANETS RESEARCH ASTRONOMICAL OBSERVATIONS ADAPTIVE OPTICS SHARE SEND BY E MAIL WATCH OUR VIDEOS MAIN FEATURES Embedded electronic shutter <1µs < 1 electron read out noise up to 2067 FPS full frame 3700 FPS in 2×2 binning mode 95% peak qe typical (deep depletion silicon) ultra low latency camera link tm full interface : 43 µs OTHER FEATURES 240 x 240 pixels state of the art EMCCD Wide 24 µm pixels 14 bits precision A/D converter Cooled operation for low dark current Integrated cooling temperature controller Fully sealed resistant aluminum body with low thermal gradient Clock & trigger input / output for synchronous operation CUSTOMISATION Contact us for more information DOCUMENTATION OCAM²S – DATASHEET OCAM² – USER MANUAL – 20210727 READ PUBLICATION OCAM2S: an integral shutter ultrafast and low noise wavefront sensor camera for laser guide stars adaptive optics systems OCAM2S: an integral shutter ultrafast and low noise wavefront sensor camera for laser guide stars adaptive optics systems READ ALL PUBLICATIONS TYPICAL PERFORMANCES INTERFACE DRAWING WAVEFRONT SENSOR OPTION Test measurement Result Unit Minimal Pulse Width <1 μs Extinction ratio >1000 N/A Shutter Resolution <0.05 μs Mean readout noise at 2067 FPS and multiplication gain ~600 0.3 e- Quantization 14 bits Dark signal at 2067 FPS at -45°C <0.01 e- pixel-1 frame-1 Detector Operating Temperature -45 °C Peak Quantum Efficiency at 650nm 95 % Linearity at gain x1000 from 10 e- to 150 e- <3 % Linearity at gain x1 from 15,000 e- to 150,000 e- <3 % Image Full Well capacity at gain x1, 2067 FPS 100 000 e- Parallel CTE at gain x1, 2067 FPS >0.99995 N/A Serial CTE at gain x1, 2067 FPS >0.99994 N/A CONTACT FIND A DISTRIBUTOR TESTIMONIAL "FLI provides the turnkey high frame rate high sensitivity cameras systems for our instrument, enabling fast and easy integration of some of the most advanced detector technologies." Dr. Olivier Guyon, Astronomer at National Astronomical Observatory of Japan – SUBARU Telescope, SCExAO Instrument FIRST LIGHT VISION SUPPORTS PLEORA 2021 Nov 12 First Light Vision now supports PLEORA GigE Vision 2.0 connectivity for Camera Link Full cameras + VISIT FIRST LIGHT AT VISION 2021 IN STUTTGART! 2021 Sep 14 Visit our booth 10G58 from October 5 to 7, 2021! + HAPPY BIRTHDAY FIRST LIGHT IMAGING! 2021 Jul 27 First Light Imaging is 10 years old! + READ ALL NEWS Applications Astronomy Life Sciences / Research Industry Surveillance / Safety Art Inspection News FIRST LIGHT VISION SUPPORTS PLEORA MARITIME SURVEILLANCE WITH C-RED 2 Newsletter Email You agree to your personal data being stored and used to receive newsletters from First Light Imaging. © Copyright 2021 FIRST LIGHT - Legal Notice - FLI SAS General Sales Conditions - FLI CORP Terms and Conditions - Confidentiality Policy - Design by Kaiman