ITALIANO ENGLISH FRANÇAIS Home > Armchairs > Arcadia X View the gallery >> ARMCHAIRS Arcadia X Any position at any height. The large angular excursions allow the armchair to assume the configuration in which the backrest, seat and legs are arranged on an ideally horizontal plane. Steel base with a thickness of 15 mm, treated with antioxidant and covered with an ABS casing with an embossed stain-resistant finish. Support structure, entirely in steel treated against oxidation, mounted on a 350 ° rotating element with Torlon® balls, covered with high-strength ABS + PMMA casings. Fasteners: all in high-strength stainless steel. Rotation movement of the structure on the base for 350 °. Swing brake with progressive effect via pedal lever. Vertical pantograph movement with a minimum seat height of 490mm, ergonomically favorable to the approach and departure of the patient, and, with a maximum height of 850 mm. Nominal lifting load 220 kg. Backrest inclination movement 75 ° (-15 °, -90 °). Longitudinal inclination movement (trendelenbourg) 32 °. Spreadable leggings with angular amplitude of 30 ° + 30 °, mounted on telescopic slides with travel up to mm. 300, tiltable and vertically liftable with an angular width of 90 °, perfectly vertical to the ground in the rest position. maximum height reachable at the end 1,340 mm. Coaxial armrests with variable height with respect to the seat, in synchrony with the inclination of the backrest. The armrests - seat parallelism remains constant in all operating conditions. For operational needs of free space, the coaxial lift arranges the vertical positioning of the armrests in correspondence with the backrest. All the movements of the chair are activated with the Single Touch function : Double command of vertical movement Double control of the backrest inclination Double command of the longitudinal inclination (trendelenbourg) Commands for automatic recall of memories (return to "0" and two customizable by the operator). When reaching a stored position, all movements occur in synchrony, ensuring a quick response to the command. Download the technical sheet Upholstery colors Two-color examples SMD STERN PODIA S.M.D. Srl progetta e produce dispositivi medici per la podologia e attrezzature per il laboratorio podologico. I prodotti con marchio Stern Podia rappresentano il fulcro della produzione S.M.D. Tutti i prodotti Stern Podia sono progettati e prodotti in Italia, nel rispetto dei più alti standard qualitativi e normativi. Via Bruno Buozzi, 41 40057 Cadriano Granarolo dell'Emilia (BO) Tel. 051.766143 Fax. 051.766093 smdpodos@smdmed.com HOME CONTACTS PRIVACY CREDITS SMD srl VAT number 02195391202 CF 02195391202 - Share capital euro 50,000.00 fully paid-up to the Bologna Company Reg. 02195391202 REA n.419949 The information contained in this site is intended exclusively for operators in the sector, therefore not aimed at a mass audience. The images and descriptions are shown purely as an indication. SMD srl reserves the right to modify the equipment at any time without notice. Contact your local distributor.