About us Products News Certifications & Honors Innovations & Commitments Contact us +36 1 280 6900 mail vpn_key BabyLife KLA-145LT Tunnel Type LED Phototherapy Unit TECHNICAL DATA keyboard_arrow_down Featured benefits A 360-degree superintensive phototherapy device Even light distribution Reduced treatment time of hyperbilirubinemia Mobile floor stand device Heavy-duty stand with 4 castors (2 lockable) 100 % sliding out cradle with net Rails on both side Working & total hour counter & clock 5" touchscreen color LCD with lock Air and skin temperature sensor Adjustable height of the upper tunnel part Adjustable irradiance level The upper and lower part of the tunnel operation can be selected Observation windows on the front and both side Displayed information: date & time, patient information, treatment duration, the distance between upper & lower part, intensity adjustment, skin temperature, alarms, Fahnrenheit/Celsius converter, language, trend Adjustable treatment time & intensity (0 - 100%) 16 pcs LED tube Technical data The lifetime of the LED bulbs min. 50 000 working hours Range of radiating wavelength: 400/500 nm Effective radiation intensity: (Ebi): lower part 65 uW/sq.cm/nm, upper part 65 uW/sq.cm/nm Electric power: 300 W Mains voltage: 110 - 230 V ± 10%, 50/60 Hz Weight: 65 kg (standard version) Additional features: SpO2 sensor (optional) Display of hospital's name or other text (optional) Irradiance meter (optional) Eye protection mask (optional) 1953 - 2021 © MEDICOR Zrt. | Privacy Policy

About us Products News Certifications & Honors Innovations & Commitments Contact us +36 1 280 6900 mail vpn_key Certifications and Honors MEDICOR Recognitions Certifications Honors More about MEDICOR Products News Contact us 1953 - 2021 © MEDICOR Zrt. | Privacy Policy