Events News Investors Contact PATIENTS PRACTICE RESEARCH ABOUT Home / Practice / The PSCS / Technical Specs TECHNICAL SPECS WEIGHT Refrigeration unit 79.4 lbs (36 kg) Caps between 28 oz (795 g) and 31.7 oz (900 g), includes coolant and cover DIMENSIONS Height 25.2 in (640 mm) x Width 12.6 in (320 mm) x Depth 16.5 in (420 mm) Extended height 64.9 in (1650 mm) POWER SUPPLY 100-120 v or 230 v (50/60 Hz) single phase 850 VA All electrical circuits earthed and protected with fused system All fuses are Bussmann quick-blow fuses—10 A at 250 Vac with F-speed CONDITION OF USE Temperature: 50ºF to 86ºF (10ºC to 30ºC) Atmospheric pressure: 70-106 kPa Relative humidity: 30% to 75% TEMPERATURE CONTROL Electronic thermostat with touch-screen controller Coolant flow rate and temperature continuously monitored in line ALARMS Visual and audible for high temperature conditions Loss of coolant flow INLET PRESSURE OF COOLANT FLOW Pressure into cap – 17 psi Flow rate of coolant 0.3-1.7 fluid oz/s (10-50 ml/s) REFRIGERATION – Hermetically sealed unit. CFC-free R134A refrigerant COOLANT – Orbis C. See data sheet section 10 DEVICE CLASSIFICATION CE marked in accordance with annex V of the Medical Devices Directive 93/42/ EEC for a Class IIa Device. Body No. CE 0473 ELECTRICAL CLASSIFICATION BS EN 60601-1: 2006 + IEC60601-1-2006 for medical electrical equipment, BS EN 60601-1-2: 2007 + IEC60601-1-2006 electromagnetic compatibility Class 1 type BF Product specifications are subject to continued evaluation and improvements and subject to change without prior notification. PRACTICE The PSCS Technical Specs The Procedure How Scalp Cooling works Scalp Cooling FAQs Clinical Efficacy Paxman USA Hub CEO Safety Considerations SIGN UP FOR UPDATES PATIENTS Events News Investors Contact PRACTICE PSCS Procedure Clinical Efficacy Media Pack RESEARCH The Science Research Partners Global Studies USA Study ABOUT Our Story Join Us Distributors Global Partners © Paxman 2020 Site by The Engine Room Privacy & CookiesReferences and Data

Contact English Investor Paxman Scalp Cooling INVESTOR Financial information The stock Press releases Newsletter Financial calendar Insiders Corporate governance Subscribe Get continuous information from the company via email. Select the types of messages you want to subscribe to by filling in the checkbox for each type. Press releases Reports Annual reports Other news Subscribe To subscribe to this, you need to accept our general terms and conditions for the purpose of GDPR. Approve Paxman Scalp Coolers International House Penistone Road Fenay Bridge Huddersfield West Yorkshire UK HD8 0LE name * Email * request * VISIT PAXMANCOOLING.COM Paxman Scalp Cooling Clinical Efficacy Clinical Trials Global Partners About Paxman Locations Contact ADDRESS Paxman Coolers Limited International House Penistone Road Fenay Bridge Huddersfield West Yorkshire UK HD8 0LE Paxman AB (publ) Pirgatan 13 SE-374 35 KARLSHAMN Sweden IR CONTACT Emelie Gustafsson, CFO Tel +46 (0) 70 - 454 37 91 Email: emelie .gustafsson @ paxman.se © Paxman 2021. Privacy & Cookies Site map