About company Products and Services | News | Vacancies | Contacts | ABS Electro » Amplifying devices » Panel devices » Manual dials (RZD) » Blocks of manual control (BRU) » Block of pointers (B-12) » Remote position indicator (DUP-M) » Regulating device (RP4-M1) Pointer block V-12 Main Products and services Ballast devices Control devices Pointer block V-12 Designed for visual monitoring of the 0-5 mA current signal and the error signal at the input of control units. Executions AT 12 Appointment visual control of the current signal (0-5) mA and the error signal at the input of the control units Variation range of the monitored DC signal 0 to 5 mA Basic error, not more than: - indicator of the current signal from the upper limit of measurement; - error signal indicator ± 4% ± 4% Weight, no more 0.2KG Overall dimensions, no more 61x81x160 mm Climatic performance UHL4.2 - for operation at temperatures from 5 to 50 ° C and relative air humidity up to 80% at t = 35 ° C and below T3 - for operation at temperatures from 10 to 55 ° C and relative air humidity up to 98% at t = 35 ° C and below Nomenclature catalog come back © 1998 - 2017 ABS ZEiM Automation Россия, 428020, г. Чебоксары, пр. И. Яковлева, 1. тел.: (8352) 305-148, (8352) 305-221, факс: (8352) 305-111 e-mail: adm@zeim.ru, sales@zeim.ru