NEED HELP FINDING A PRODUCT? 678-730-0770 ACCOUNT LOGIN CART PRODUCTS SERVICE RESOURCES ABOUT NEWS CONTACT US Home > Magnetic Particle > Portable MPI > Magnaflux P-Series Portable Power Packs MAGNAFLUX P-SERIES PORTABLE POWER PACKS Portable Power Packs are convenient and economical inspection units that are used in a wide range of magnetic particle applications that require only moderate power. The Magnaflux Portable Power Packs are ideal for large or heavy parts, in-service mobile equipment, or job-site welds. P-1500 Datasheet P-70 Datasheet REQUEST A QUOTE Category: Portable MPI We pride ourselves in providing fast deliveries. However inventory changes rapidly and on occasion some items may not be in stock. If you require an expedited shipment, please contact us via e-mail or call our office to confirm availability of an item. *Nominal lead times vary by product. PRODUCT INFORMATION Find surface defects with AC current; switch cable connection to half wave DC (HWDC) for near-surface flaws Locate defects in any direction using prods, central conductor, coil (wrap part with 3-5 turns of cable) Infinitely variable current control Demagnetize your parts in AC mode by slowly turning the current dial to zero Built-in automatic-reset circuit breaker May be used as the main power source for the Magnaflux MAG Kit modular bench inspection system SPECIFICATIONS P-70 Length: 17.5″ (44.5 cm); P-1500 Length: 22″ (55.9 cm) P-70 Width: 9″ (22.9 cm); P-1500 Width: 9.5″ (24.1 cm) P-70 Height: 8″ (20.3 cm); P-1500 Height: 9.5″ (24.1 cm) P-70 Maximum Output*: 750 amps, AC or HWDC; P-1500 Maximum Output: 1500 amps, AC or HWDC P-70 Available Voltages: 115,230, 50 or 60 Hertz; single phase power; P-1500 Available Voltages: 230,460, 50 or 60 Hertz; single phase power P-70 Line Current: 26 amps @ 115 VAC; 13 amps @ 230 VAC; P-1500 Line Current: 42 amps @ 230 VAC; 21 amps @ 460 VAC Maximum Duty Cycle: 2 minutes ON, 2 minutes OFF P-70 Unit Weight: 35 lbs (15.9 kg); P-1500 Unit Weight: 93 lbs (42.2 kg) Note: Current ratings are based on the use of two 15-foot length 4/0 Cable RESOURCES P-1500 Datasheet P-70 Datasheet Magnetic Particle Inspection Reference Guide SIMILAR PRODUCTS GOULD-BASS MAGNE-TECH® PORTABLE MPI UNITS PARKER HI-AMP PORTABLES Home Products Service Resources About News Privacy Policy Cookie Policy Contact Us PRODUCTS Ultrasonic Eddy Current Visual Inspection Magnetic Particle Penetrant Radiography PMI / XRF Analyzers Hardness Testing Resonant PMI / XRF Analyzers Thermal Imaging Corrosion Mapping Flawed Specimens Weld Gauges Coating Thickness Crawlers Special Offers CONTACT US 5096 Bristol Industrial Way Suite A Buford, GA 30518 Sign up for our Newsletter © 2015-2021 JWJ NDT - Sitemap – Web Design by Red Clay Interactive