Voltage: from 300 Volts to 200 000 Volts Power: from 150 J/s to 1 500 Joule / second Size (rack - depth): 3U - 480 mm / 3U - 600mm / 5U - 600mm Applications (examples): Sterilization, capacitor charging (tests), High energy physic application, Lasers systems CCR 15kV-1.5kJ 0 to 15 kV, 1.5 kJ/s Non-contractual photo Overview Description Options Electrical specifications Physical description Compact & industrial design Low stored energy Voltage & current outputs continuously adjustable from 0 to 100% Voltage & current setting & monitoring from front panel or remote interface Customized voltage & current* with no extra cost High reliability & efficiency High precision regulation Total protection against arc, overload, short circuit & over temperature Automatic regulation crossover Over current limitation Remote, inhibit & interlock functions 3 ½ digit display with polarity display Double resonance technology (ZVC) Safety switch key Air cooling 2 years warranty Front panel Rear panel * You can choose your own full scales (Vmax and Imax) even if these values are not available in standard. For example, you can get a 3750 V generator based on 750 J/s model. In this case, you will get an output voltage adjustable from 0 to 3750 V and an output current from 0 to 0.4A. High voltage CCR models are designed to charge high voltage capacitors and Marx generators They operate as constant current sources. The numerous advantages of compactness, robustness, reliability, and saving offered with the double resonance technology are already available on these models which constitute the low end range. Five series of chargers are offered: 150J/s, 300J/s, 600J/s, 750J/s and 1.5kJ/s. These chargers are all based on a 3U and 5U height 19’’ rack presentation. Using a modular design, our products integrate the latest technology components. Above 20KV, HV blocks are in resin. Reversibility and floating outputs Non instrumented front panel Automatic inhibit when discharging RS232, GPIB, Ethernet and Profibus interfaces Optic fiber transmission kit with RS232 only LABVIEW run time RS232 or GPIB or Ethernet Relay interface 24V DC Insulated remote interface Emergency stop switch Repetition frenquecy > 10 Hz. No extra cost Adaptation customer to remote interface Multichannel control unit up to 6 chargers Zero floating Tropicalization Transportable “all terrain” container Specific AC or DC mains power inputs OEM Design on request Remote front panel Any special requests can be considered ** Possible restrictions when multiple options (please contact our sales department for more details or check our compatibility matrix) or Compatibility matrix Output Voltage and Output Current Both Voltage Output and Current Output are continuously adjustable from 0 to 100 % (full scale) In local mode: by using 10 turns potentiometers resolution 0.05% In remote mode: by an external 0 to 10 V Maximum frequency In standard: 10 Hz On request: up to 1 kHz depending on the application Reproducibility* (pulse to pulse voltage variation) For frequency ≤ 10 Hz: ± 0.1 % better on request For ± 10% mains voltage: ± 0.1 % Current regulation * Load Regulation: ± 0.05 % of full current (for 0 – 100 % load) Line Regulation: ± 0.05 % of full load (for Mains Voltage ± 10 %) Voltage holding: 0.25% p-p of full voltage better on request Protections: Against short circuit & HV arc to ground Shutdown on over temperature & open interlock Stored energy: 0.2 J/ kJ Stability (after one hour warm-up) 0.01 % / hour, 0.03 % / 8 hours of operating with constant load and ambient temperature Temperature coefficient: 100 ppm/°C Operating temperature: from 0 to 40°C Efficiency: > 90 % full load Air Cooling by air : Dust filter fans on front panel Easily removable filters for cleaning Mains Voltage: Up to 600J/s: 170V to 255 VAC 47 - 63 Hz Single Phase + Earth Power factor correction Input ≥0.98 From 750J/s: 400 VAC ±10% 47 – 63 Hz 3 Phases + Earth Power factor: ≥0.98 Inrush current: limited to full power operating current Calibration: with probes periodically approved by Authorized Measurement Laboratories CE Certification * For fixed polarity CCR 150J/s Series CCR 300 J/s Series CCR 600J/s Series CCR 750J/s Series CCR 1.5kJ/s Series Voltage Current Model Current Model Current Model Current Model Current Model 300 V ON REQUEST 700 V 429 mA CCR700V-150J 857 mA CCR700V-300J 1.71 A CCR700V-600J 2.14 A CCR700V-750J 4.29 A CCR700V-1.5KJ 1 kV 300 mA CCR1KV-150J 600 mA CCR1KV-300J 1.2 A CCR1KV-600J 1.50 A CCR1KV-750J 3.00 A CCR1KV-1.5KJ 2.5 kV 120 mA CCR2.5KV-150J 240 mA CCR2.5KV-300J 480 mA CCR2.5KV-600J 600 mA CCR2.5KV-750J 1.20 A CCR2.5KV-1.5KJ 5 kV 60.0 mA CCR5KV-150J 120 mA CCR5KV-300J 240 mA CCR5KV-600J 300 mA CCR5KV-750J 600 mA CCR5KV-1.5KJ 10 kV 30.0 mA CCR10KV-150J 60.0 mA CCR10KV-300J 120 mA CCR10KV-600J 150 mA CCR10KV-750J 300 mA CCR10KV-1.5KJ 15 kV 20.0 mA CCR15KV-150J 40.0 mA CCR15KV-300J 80.0 mA CCR15KV-600J 100 mA CCR15KV-750J 200 mA CCR15KV-1.5KJ 20 kV 15.0 mA CCR20KV-150J 30.0 mA CCR20KV-300J 60.0 mA CCR20KV-600J 75.0 mA CCR20KV-750J 150 mA CCR20KV-1.5KJ 30 kV 10.0 mA CCR30KV-150J 20.0 mA CCR30KV-300J 40.0 mA CCR30KV-600J 50.0 mA CCR30KV-750J 100 mA CCR30KV-1.5KJ 40 kV 7.50 mA CCR40KV-150J 15.0 mA CCR40KV-300J 30.0 mA CCR40KV-600J 37.5 mA CCR40KV-750J 75 mA CCR40KV-1.5KJ 50 kV 6.00 mA CCR50KV-150J 12.0 mA CCR50KV-300J 24.0 mA CCR50KV-600J 30.0 mA CCR50KV-750J 60 mA CCR50KV-1.5KJ 60 kV 5.00 mA CCR60KV-150J 10.0 mA CCR60KV-300J 20.0 mA CCR60KV-600J 25.0 mA CCR60KV-750J 50 mA CCR60KV-1.5KJ 80 kV 3.75 mA CCR80KV-150J 7.50 mA CCR80KV-300J 15.0 mA CCR80KV-600J 18.7 mA CCR80KV-750J 37.5 mA CCR80KV-1.5KJ 100 kV 3.00 mA CCR100KV-150J 6.00 mA CCR100KV-300J 12.0 mA CCR100KV-600J ON REQUEST 120 kV 2.50 mA CCR120KV-150J 5.00 mA CCR120KV-300J ON REQUEST 150 kV 2.00 mA CCR150KV-150J 4.00 mA CCR150KV-300J 200 kV ON REQUEST For other/higher voltage, please Polarity can be positive, negative, floating (option) or reversible (option) Floating or reversibility options can modify the size of the unit Example of reference : CCR 15KV-150J Model: CCR model (capacitor charger) Voltage: Adjustable from 0 to 15 kV Power: 150 J/s charging power 3U - 19" rack H133xW483xD480 mm H133xW483xD600 mm 5U - 19" rack H222xW483xD600 Home Products Options Technical informations News Jobs About Technix Contact © 2013 TECHNIX reproduction forbidden - 6, rue Eugène Dupuis 94000 CRÉTEIL FRANCE - Tel: +33 (0)1 56 71 28 60 - EULA - Sitemap