+39 031 790005 info@ennebielettronica.com Mappa Home Electronic Manufacturing Electromedical Devices Electronic Devices DNA and RNA Analyzer DNA sequences analysis has never been so easy. GeneOne is the device able to detect fragments of DNA and RNA for analysis in the laboratory or on site CONTACT US It is a compact and flexible system that allowing the analysis of polymerase chain reaction (PCR) characterized by a high cost / performance ratio. It is an apparatus capable of detecting DNA and RNA fragments for analysis in the laboratory or on site. The device DNA and RNA Finder Ennebi Electronics is used in the study, research and identification of pathogen agents (virus/bacterials) and contamination, and in the generic analysis of detection of DNA and RNA sequence of the samples. The product consists of a thermo cycler and a measuring system combined in a single portable and solid equipment. The total absence of moving parts reduces consumption and allows use even in the toughest conditions. Its operation is fully configurable and the thermal profiles are imported and exported with a USB support. The graphical touchscreen interface simplifies the user interaction and walkthrough the user in all the procedure. Technical specifications Thermal cycler for PCR and fluorescence detector lend point to low consumption. Supports standard configuration and isothermal thermal cycler cells thermoelectric temperature control with repeatability of 0.1 ° C. Maximum temperature 95 ° C. Minimum ambient temperature: 5 ° C Heat exchanger sink active high efficiency, heated lid anti-condense Support for standard tubes 4.5ul Fluorescence detector with optical filters, system excitation and detection of the issue modular (1/2 wavelengths 490nm-535nm) Control system TFT 4.3″ tousch screen GUI control for viewing machine status and configuration Operating voltage: 230V-100W for the version with 16 tubes, 12Vdc auxiliary User Input: Touch screen TFT LCD Thermal cycler to TEC element Detection temperature in two places, the plateau of 16 tubes Absolute accuracy +/- 0.5 degree Optical Unit Technical characteristics: Optical filters and focusing tubes of pipe measurement Detection of Current illuminator Features Control firmware able to be upgraded USB port for connection via PC Able to be expanded at two wavelengths Full bridge Peltier Management with PWM regulation Languages selection © 2016 Ennebi Elettronica S.r.l. - Tutti i diritti sono riservati P.Iva 02513320966 - Sede legale e operativa: Via Papa Giovanni XXIII 13/15, 22060 Novedrate (CO) REA: 293924 - PEC: ennebielettronicasrl@pec.it Capitale sociale: 10.400€ interamente versato Privacy Policy Questo sito memorizza le preferenze dell'utente ed utilizza tecnologie che consentono di effettuare analisi statistica in forma anonima delle pagine e del comportamento dei visitatori. Gli utenti possono tuttavia disabilitare le funzionalità non necessarie. Ok Dettagli