Alpha pascal Products and services Library Events Company Contacts Mon- Fri , 9: 00-18: 00 today is a day off ← Ctrl K-9 and K-25 GUSK Ctrl → " RSL and RPG " media separators Installed on devices GUSK and MP RSL oil / water Designed for verifications and calibrations of measuring instruments that do not allow the ingress of oil, including oxygen version. Does not require an additional source of pressure generation. RPG liquid / gas Designed for verifications and calibrations of measuring instruments that do not allow ingress of liquids, including food and oxygen versions. Requires power supply connection: compressed air cylinder or compressor. Benefits Expands the range of applications of pressure generation devices available at the enterprise without purchasing new ones. The design of the devices is unified and, unlike analogs, has a maximum separation range of media up to 25 MPa. They do not cause additional non-systemic measurement errors due to the absence of a mechanical seal in the diaphragm design. Main characteristics RSL RPG Pressure creation range, MPa 0 ... 25 0 ... 25 Verifiable measuring instruments. PC one one Weight, kg five 8 Shared environments oil water liquid gas Documentation Datasheet PDF 0.327 MB RPG User Guide PDF 0.207 MB RSL User Guide PDF 0.200 MB Pneumohydraulic RPG and diaphragmless RSL Standard delivery by additional order 1.medium separator 2. a set of fittings for connecting the calibrated measuring instruments with threads: M20 × 1.5 M12 × 1.5 G½ G¼ 1 piece 2 pcs 1 piece 1 piece 3. The set of sealing rings 4. The set of SPARE PARTS 5. passport → For RPG ONLY 6. шланг для подключения внешнего источника давления Габариты разделителя и упаковки РПГ диаметр117 мм длина235 мм ширина175 мм высота160 мм Схематично РГБ диаметр107 мм высота180 мм Схематично Фотографии и видео 6 фото © 1997—2014 Альфапаскаль Контактная информация Эл. почта: info@alfapascal.ru +7 (351) 724-74-50 (09:00–18:00)