DPC - DIFFERENTIAL PRESSURE CONTROL New: connectivity for new IoT applications. Description The DPC Differential Pressure Control measures the differential pressure in different ranges between 0 and up to 10 kPa. The integrated pressure sensors are appropriate for non-aggressive gases. Typical uses-cases are air-filtration- and air-conditioning-units as well as clean-room and medicine technics. Another application is the measurement of the volume flow with an installed aperture. The base unit already delivers features such as 3-digit LED display, analogue output signal (4-20 mA), 2-point-regulator and three individually programmable relays switches. Because of the modular concept the device is quickly customisable to allow individual configurations. Available options are listed below. Standard Features Differential pressure measurement in various pressure ranges (e.g. +10 kPa) 3-digit LED display Scaled analog output signal (4...20 mA) 2-point-regulator with programmable relay 2 alarm programmable relays switches (min, max) programmable alarm waiting time 24 VDC power supply CE conformity More Options Hardware Individual print Jack-connector for standard power supply customer-specific n-pole M12-plug NEW: IoT module for Ethernet and WLAN connection Software Modbus-connection for digital I/Os Programmable analog output signal (differential pressure, volume flow and PID) Function of the relays switch (2-point-regulation or flushing) MQTT protocoll (only in combination with IoT module hardware option) Applications Air-filtration and air-conditioning Medicine applications Clean room control Contact Contact person: Dr. Martin Neumann Mail: request@ed-chemnitz.de Phone: +49 371 524 59 0 CREATE ROUTE show in google maps +49 (0) 371 524 59-0 contact us Xing Twitter LinkedIn EDC Electronic Design Chemnitz GmbH Technologie-Campus 4 09126 Chemnitz AGB Impressum Datenschutzbestimmungen We use cookies to improve your experience on our site. To find out more, read our privacy policy I agree