Home Services Products References Market Experience Intellectual Property Contact Home » Panacea P200 Panacea P200 Panacea P200 The Panacea P200 is the most popular in BESST’s Panacea family. With a 1.75” outer diameter, the P200 is suitable for use in 2” and larger wells. It is available in Delrin plastic and in 316 stainless steel models depending on your application and budget. The P200 also features a 60 micron alumina based ceramic filter as well as a redundant screen for thorough filtering, ensuring that no sediment becomes lodged in the valves. This advanced filter system keeps the pump operational for long-term use with little to no maintenance. Delrin P200 (PU-P-200-1000) Stainless Steel P200 (PU-P-200-2000) Body Construction Delrin Plastic 316 Stainless Steel Length (in) 23.5 23.5 Outer Diameter (in) 1.75 1.75 Weight (lbs) 2 9 Maximum Pressure (PSI) 200 1500 Maximum Depth (ft bgs) 400 3000 Minimum Submergence (ft) 6' with Volume Booster 6' with Volume Booster Internal Volume (mL) 200 200 Standard Filter Pore Size (µm) 60, others available 60, others available Standard Tubing OD (in) 1/4, others available 1/4, others available Seal Material Buna-N, others available Buna-N, others available Gas Compatibility Nitrogen, Compressed Air Nitrogen, Compressed Air ZIST Compatibility 2" and 4", requires adapter 2" and 4", requires adapter PDF Downloads Product Overview Panacea P200 Delrin Guide · © 2018 All rights reserved BESST, Inc. ·

Home Services Products References Market Experience Intellectual Property Contact Home » In-Situ Headspace Test In-Situ Headspace Test For more information on the In-Situ Headspace Test, Click HERE · © 2018 All rights reserved BESST, Inc. ·