Home Companies Production program Products Jobs & careers Contact Downloads Home > Products > KM 2000 accessories KM 2000 accessories Hydrocarbon measurement Sample gas extraction Carborundum filter with protective cap, gas sampling tube with shut-off valve and test gas valve Executions: Material V4A (1.4571) a.) With insulation b.) Without insulation a) b) Test adapter - for ITR 498 sensor (PVC) - for GTR 196 sensor (PVC) - for ADOS 592 TOX sensor (PVC) - for GTR 196 and ADOS 592 TOX sensor (aluminum) Mounting stud for assembling the extraction tube Executions: - Material V4A (1.4571) - Material St 37 - with single flange DN50 / PN6 - 50 mm long Explosion protection - Explosion protection for sample gas lines - Type Wt-BAM-tested Condensate separator - Condensate separator with mounting plate - 175 x 260 x 125 mm (WxHxD) Fine dust filter with filter sleeve Cutting ring screw connection on both sides for intermediate sample gas line installation R 1/4 "or 8 mm cutting ring screw connection Gas flow chart a) Messgasentnahme b) Montagestutzen mit Einfachflansch c) Beheizte Gasentnahme 1) Messgaseingang 2) Prüfgaseingang 3) Vorfilter bzw. Ausgleichsfilter 4) Messgaspumpe 5) Strömungsregler 6) Strömungsmesser 7) Strömungswächter 8) Heizspirale 9) Katalysatorkammer 10) Vergleichsmess-Stelle 11) Mess-Stelle 12) Reaktionskammer 13) Messverstärker 14) Grenzwertmelder 1–4 15) Messwertintegrator 16) Linienschreiber 17) Konzentrationsanzeiger 18) Heizmanschette 19) Temperaturregler 20) Widerstandsthermometer 21) Gasausgang Beheizte Entnahmeleitung Technische Daten: Länge: 1 m – 100 m Operating temperature: 65 ° C ... 120 ° C Nominal size: 4… 8 Nominal voltage: 230 V = Outer diameter: 42 mm Hose end caps: with strain relief and kink protection in PA 6 or silicone External protection: Polyamide 6 corrugated hose, flame retardant, halogen-free Temperatures: -40 ° C… +150 ° C, briefly> 150 ° C optional accessories: Temperature controller Test gas cylinder - Zero point gas: synthetic air - N-butane in synthetic air (calibration gas) - for non-aggressive gases - for hydrogen sulfide (in stainless steel) Contents: 10 l Filling pressure: at least 100 bar Pressure reducer Pressure reducer for connection to test gas bottles Form: max. 200 bar Outlet pressure: 0.1 bar - 3 bar Material: Brass Product selection: Evaluation units and analyzers FlexADOS 914 FlexADOS 914 LON® FlexADOS 914 CP FlexADOS 914 MED GW 399 MWS 906 MWS 903 MWS 897 Biogas 401 Biogas 905 KM 2000 CnHm EM RG 399 Gas sensors and transmitters GTR 210 GTR 210 MED GTR 196 TOX 914 LON® TOX 592 LCTR 903 Filter Guard 206 software Log & View software equipment KM 2000 accessories Gas warning system accessories Product downloads KM 2000 CnHm EM accessories Language : German Germany) English (United Kingdom) English (United States) Français (France) Español (España, Alfabetización Internacional) Português (Brazil) Русский (Россия) ^ To the top Successful for over 100 years About ADOS GmbH Since 1900 we have been developing portable gas measuring devices and stationary gas measuring systems for almost all industrial areas of application, which measure and detect a multitude of hazardous substances in the smallest concentrations with highly sensitive sensors and thus trigger a timely gas warning for people in case of danger. imprint data protection General conditions of sale Contact Jobs Sitemap ADOS GmbH, 2021