Open navigation Open language menu Open search APOLLO MR3T High precision room laser for positioning in an MRI environment PREVIOUS NEXT Key figures 0.5 millimeter line width 0 degrees of freedom 0 mounting options 0 laser colors High precision room lasers for a magnetic environment APOLLO MR3T lasers have barely ferromagnetic resonance. Hence they are convenient for a magnetic environment such as know with an MRI or MR LINAC room. Not only for positioning of patients but also for precice placement of phantoms for quality assurance room lasers are an optimal solution. The exact positioning improves the safeness which is a crucial factor in radiation therapy. Thanks to their high quality, room lasers from LAP, remain in use and highly precise for an above-average number of years. Features Precision Even at a distance of 4 meters, the maximum laser line width is 0.5 mm to max. 1 mm. The line straightness within the line shows barely any deviations, even at a distance of 3 meters. You can always be sure of the precision of our lasers. Visibility The different red, green, and blue diode colors, the optical room laser components are designed and manufactured with extreme accuracy. Plane parallel glass with anti-reflective coating prevents any interference with the laser beam. The brightness distribution remains constant at all times. Benefit from optimal visibility for precise patient positioning. Stability A high-quality aluminum housing reliably protects the fine mechanics and optical components inside. You don't need to fear damage even from gentle impacts. Highly shatterproof glass stands up to all the rigors of clinical daily life. Flexibility Adapt the room laser design to the circumstances of your space. Direct wall mounting or on a freestanding support are both possible. In addition, the room laser can be inclined up to 45°. This gives you flexibility even in tight spaces. MR-conditional The components are magnet-safe and can be used without impact on the imaging. Hence they are perfectly suitable for a magnetic environment. Simple adjustment Laser adjustment is convenient via remote control. Fast and smooth laser position and focus adjustment save you time for QA and treatment. Configuration options of APOLLO MR3T APOLLO MR3T The APOLLO system has three (optionally four) lasers for the coronal, sagittal and transversal body planes. You can choose the laser color, line and/or cross laser, and mounting option, all adapted to your individual room situation. An optional fourth laser prevents shadowing from the LINAC gantry. APOLLO MR3T bridge APOLLO laser bridge consists of three fixed cross hair laser modules for projecting red, green or blue crosses to mark the patient in all three body planes. Technical data APOLLO MR3T Laser color (typical wave length) red (638 nm), green (520 nm), blue (450 nm) Laser class 2 Line width up to 4 m distance < 0.5 mm (blue), < 1 mm (red, green) Line length at 3 m distance > 3 m Power supply 100 ... 240 V AC Internal voltage 24 V DC Operating Temperature 15 … 30 °C Dimensions (H × W × D) 221 × 110 × 100 mm (8.7 × 4.3 × 3.9”) Weight 2.6 kg Technical data APOLLO MR3T bridge Dimensions customized Width: 2594-5000 mm (102.0‘‘-196.9‘‘), Height: 2300-2800 mm (90.6‘‘-110.0‘‘) Weight approx. 100 kg Laser color (typical wave length) red (638 nm), green (520 nm), blue (450 nm) Laser class 2 Laser adjustment Remote controlled Line width up to 4 m distance < 0.5 mm (blue), < 1 mm (red, green) Line length at 3 m distance > 3 m Power supply 100 … 240 V AC, 50 … 60 Hz Scope of delivery Laser system, remote control, cable set Download PDF 2 MB APOLLO MR3T data sheet PDF 733 KB APOLLO MR3T Bridge data sheet CONTACT US! INDUSTRIES PRODUCTS SERVICE COMPANY CAREER LEGAL Visit us on XING Visit us on LinkedIn Visit us on Youtube Names of other companies and products displayed on this website can be trademarks or registered trademarks which do not belong to LAP, but to their respective owners. © 2021 LAP GmbH Laser Applikationen / Imprint / Privacy policy / Sitemap / Cookie preferences