About News Contact Careers Europe, Middle East & Africa Terumo Global Products EventsLearning Search HomeMedical fieldPeripheral InterventionAngio-Seal™ Vascular Closure Device VIP Angio-Seal™ VIP Vascular Closure Device Description Characteristics Preparation Procedure Clinical evidence General specifications Item specifications Description Simple.Proven. Fast Sealing. Angio-seal is a medical device indicated for use in closing and reducing time to hemostasis at the femoral arterial puncture site in patients who have undergone diagnostic angiography procedures or interventional procedures (minimal artery diameter of 4mm). The 6F Angio-Seal device is for closure after usage of a 6 French or smaller procedural sheath, while the 8F Angio-Seal device is for closure after usage of an 8 French or smaller procedural sheath. Characteristics Insertion Sheath Arteriotomy Locator 6F (2.0 mm) – 0.035 in. (0.89 mm) Guidewire with J-Straightener or 8F (2.7 mm) – 0.038 in. (0.96 mm) Guidewire with J-Straightener Angio-Seal consists of 3 bioabsorbable components: anchor, collagen, and suture 10 units per box Single use, one year shelf life after sterilization Sterilized by gamma radiation. Do not re-sterilize Available in select markets Storage, packaging and disposal: Should be stored in a cool location (between 10°C and 25 °C) Contains resorbable materials that degrade by exposure to heat and moisture; therefore, the device may not be resterilized Sterile and non-pyrogenic in unopened and undamaged package Dispose of contaminated units, components, and packaging materials utilizing standard hospital procedures and universal precautions for biohazardous waste. Preparation Video instructions Procedure Step by step instructions 1Locate the artery Insert the Angio-Seal locator and sheath system over the wire until blood flows through the drip hole in the locator Withdraw and re-insert to confirm correct location into the vessel Remove the locator and wire, leaving the Angio-Seal sheath in place 2Set the anchor Insert the Angio-Seal device into the sheath, click into place and click back to deploy and set the anchor Next Clinical evidence Case report with Angio-Seal VIP Case report with Angio-Seal VIP performed by Carsten W.K.P. Arnoldussen, M.D. See all clinical evidence General specifications Content Vascular closure device Insertion sheath Arteriotomy locator 70 cm guidewire with "J" straightener Units per box 10 Item specifications Item reference French size Guidewire diameter (in) 610132 6 0.035 610133 8 0.038 Enlarge table Do you need more info about this product? Our sales representatives are here for you. Contact us Related products Previous Radifocus® Obturator Accessory Glidesheath Slender® Introducer Progreat® Micro Catheter System NaviCross® Support Catheter AZUR® CX Peripheral Coil System Next 1 2 3 4 5 6 Didn't find your product? Consult our products in our Product Index Terumo Europe Products A-Z Sitemap Privacy and cookies notice Terms of use Menu Products About Events News Contact Careers Contact us For information on products or services available, for technical support, or to report a complaint. Contact us Follow us © 2021 Terumo Europe NV Europe, Middle East & Africa Are you a healthcare professional? This webpage can include promotional content regarding one or several products of Terumo Europe, or some procedures concerning the use or implantation of such products. This webpage is exclusively intended for healthcare professionals and is in no event directed to the general public For Pharmaceutical Professionals interested in our B2B products portfolio, please visit Terumo Pharmaceutical Solutions. No Yes ✕ Functional cookies are necessary for the use of this website and cannot be refused. Other cookies are only placed if you choose to do so. For further information please read our cookies notice. Accept all cookies Manage cookies