PRODUCTS SERVICES APPLICATIONS ABOUT US CONTACT US Wireless, easy-to-use, comfortable EEG Headsets for healthcare professionals. OverviewFeaturesApplicationsSpecifications & ManualsSupport The Stat X-Series mobile EEG systems offer healthcare professionals cost-effective and practical solutions for on-call and portable EEG in minutes. The headset design enable mobility during recordings, while also minimizing the artifacts and noise typically associated with traditional, wired EEG systems. Standard software and SDK are included to enable the development of applications with the potential to add diagnostic or treatment value. Features 9 channels of high-quality EEG plus one optional ECG, EOG, EMG, ECG channel 20 channels of high-quality EEG plus one optional ECG, EOG, EMG, ECG channel 10 minute rapid setup Automated wireless impedance checks ESU-MC Bluetooth wireless range up to 10 meters 8+ hour rechargeable battery life On-board accelerometer to quantify head movement and position Data quality monitoring and feedback Sizing to fit adolescent (age 6) throughadult heads Multiple options for data synchronization Provides files viewable in Persyst® and other recognized software Application The Stat X-Series System(s) is FDA cleared prescription use or use in IRB approved studies. The system can be used in the home, at healthcare facility, or clinical research environment on patients ages 6 and older to acquire EEG, EOG, ECG, and EMG signals. The Stat X-Series system only acquires and displays physiological signals, no claims are being made for analysis of the acquired signals with respect to the analyses’ accuracy, precision, and reliability. Those interested in developing non-medical EEG applications should consider the B-Alert research EEG systems and their suite of related software. Software Each X-Series system includes acquisition software with functionality including: Acquisition, storage and display of raw data signals Capability for manual event marking and annotation of files Integration with third party event markers using the External Syncing Unit (ESU) Automated delivery of resting state tasks eye open and eyes closed Files saved in European Data Format (EDF) Basic SDK to facilitate programming access to all software functions Upgrades: B-Alert Live is a visual EEG software solution that combines real-time and off-line processing and analysis tools. LabX is a highly flexible and comprehensive offline software suite of analysis tools for all X-Series applications. The ESU, or External Synching unit, is used to integrate signals and markers from third party sources or software into the .edf file with millisecond precision. Contact us for more information. Case Studies EEG-Based Biomarkers for HIV-Associated Neurocognitive Disorder (HAND) Dr. Thomas Marcotte with UCSD’s HIV Neurobehavioral Research Program used the B-Alert systems and the Alertness and Memory Profiler (AMP) to characterize neurocognitive decline in HIV patients. AMP-based EEG metrics were also shown to generalize in predicting performance driving errors on the road for HIV patients and healthy controls. Poster and publication are available upon request. Identify Neurophysiological Biomarkers of Distress As part of a DARPA funded project to detect and analyze psychological signals, the Advanced Brain Monitoring research team developed a customized, autobiographical stress elicitation tool. B-Alert X24 was applied to collect neurophysiological data from healthy and PTSD-diagnosed subjects under stress, and provided the basis for significant early findings. Specifications Channels up to 20 EEG plus one optional ECG channel Sampling rate Sampling rate 256 samples/second - all channels Dynamic Range ±1,000 µV Resolution 16 bit Input impedance 500 M, typical Common mode rejection ration -115 dB, typical Bandpass characteristics 0.1 Hz HPF Firmware, 67Hz LPF Hardware Noise Noise 3 μV peak-to-peak, typical Head movement/position Angles obtained with 3-axis 12-bit accelerometer RF Band Bluetooth 2.4 to 2.48 GHz (ISM band), latency < 340ms Transmission mode Bluetooth SPP 2.0- via USB Dongle or External Syncing Unit Data transmission range ~ 10 meters, line of sight with onboard antenna Expand Manuals X-Series Stat X24 Hardware Technical Manual X-Series Stat X10 Hardware Technical Manual X-Series Stat Software Technical Manual Related Products / Services B-Alert X-Series Clinical Trials Research Services Get in touch CONTACT US Products Newsletter Signup Sleep Profiler Sleep Profiler PSG2 B-Alert X-Series EEG Stat X-Series Wireless EEG Night Shift Sleep Positioner Apnea Guard PoSAS Resources Email Applications Publications Privacy Policy Terms of Use Career Opportunities 2237 Faraday Ave, Suite 100 Carlsbad, CA, 92008 Phone: (866) 677-2737 Fax: (760) 476-362 © 2020 Advanced Brain Monitoring, Inc. All Rights Reserved