Meny FocusMonitor FMW+ Home » Våra produkter » Primes Produkter » Focus Measurement » FocusMonitor FMW+ The FocusMonitor FMW+ is specially designed for analyzing continuous irradiation, even at high beam powers. This universal tool for high-power radiation is the compact version of the proven, globally successful FocusMonitor FM+ with new electronics and an absorber for powers up to 1000 W. With the compact FocusMonitor FMW+, you will know the quality of the laser beam in your processing system and reliably determine the geometric dimensions of the focused laser beam as well as the focal point location in space, the beam parameter product and the beam quality factor M². Overview The FocusMonitor FMW+ is a mechanically scanning measuring system for measuring focused laser beams with diameters between 100 μm and 3 mm in tight spaces. Unlike the FocusMonitor FM+, the FMW+ does not have its own z-axis. You will therefore have to use the z-axis of the laser system to take caustic measurements. The device can be used to measure NIR as well as CO2 beams up to 1,000 W laser power. All you need to do is exchange the measuring tip and the corresponding detector. With a maximum size of 8 x 8 mm, the measurement window can be gauged with a resolution of up to 1024 x 1024 pixels. Operation With the new diagnostics tool LaserDiagnosticsSoftware (LDS), the FocusMonitor FMW+ offers considerably more functionality than ever before. In addition to faster data communication via Ethernet, it also enables the semi-automatic or manual measurement of the beam distribution as well as the determination of the beam position and beam dimensions. Accessories Changing the detector and the measuring tip is easy and can be done by the user in just a few minutes. The full FocusMonitor FMW+ system set includes the main device as well as two detectors, two measuring tips, and a practical transport box as well as other accessories. Measurement set for NIR beams Detector DFIG-PS+ For measuring NIR lasers, including photo diode with electrically variable sensitivity, wavelength range 1-1.7 μm Measuring tip FocusMonitor NIR high div For measuring high-power solid-state lasers with adjusted sensitivity, damage threshold up to 10 MW/cm2 (7500 1/min and He inert gas) Measurement set for CO2 beams Detector DFC+ With passive infrared detector for measuring CO2 lasers, wavelength range 9 – 12 μm Measuring tip FocusMonitor CO2 high power For measuring high-power CO2 lasers with adjusted sensitivity, damage threshold up to 30 MW/cm2 (7500 1/min and He inert gas) Download PDF KONTAKTPERSON Linda Jingnert Försäljning eftermarknad +46701168534 linda.jingnert@permanova.se Permanova Lasersystem AB Krokslätts Fabriker 30 431 37 Mölndal Växel: 031 706 1980 Fax: 031 86 46 12 Information: info@permanova.se Permanova.se Användningsområden Våra Produkter Vårt kunskapscenter Permanova som partner Om Permanova Kontakt Integritetspolicy Hitta oss i sociala kanaler Copyright © 2021 Permanova Lasersystem AB. All rights reserved. Webbplatsen använder cookies för att hjälpa oss ge dig den absolut bästa upplevelsen när du besöker oss. Läs mer om vår cookie policy. Genom att stänga detta meddelande godkänner du vår användning av dessa cookies samt att du har informerats.Cookie settingsACCEPT