My Account Log In To HOME PRODUCTS SUPPORT NEWS COMPANY CONTACT 0 TRS-6100 Automated Refraction System The TRS-6100 Automated Refraction System is comprised of a control panel with a large 10.4″ color LCD touch screen and an intuitive, ergonomic keyboard for simple one-hand operation. The refractor forehead rest has a range of 25±2 mm to accommodate more patients. With programmed monocular fogging, this reduces accommodation and allows more realistic refractions with both eyes open. The tilt screen and angled refractor head allows realistic near testing. Your most frequently used charts can be saved on a ‘favorites’ page for quick and easy access. REQUEST QUOTE SUBJECTIVE REFRACTORS / TRS-6100 OVERVIEW FEATURES SPECS REVIEWS DOWNLOADS REQUEST QUOTE Your most frequently used charts can be saved on a ‘favorites’ page for quick and easy access Increase Patient Flow & Revenue By streamlining the flow of data from pre-test instruments, simplifying the exam process for the patient, linking charts to coordinate with selected tests, and speeding up the final verification by the patient, significant time can be saved over the entire refraction process. Saving minutes per patient throughout the day allows for increased time spent with patients discussing their care, or additional patients in the schedule. Measurable Range -29D to +26.75D Cylinder Range 0.00D to ±8.75D Wide Visual Field 40 Degrees Contrast Test Programmed monocular fogging reduces accommodation & allows more realistic refractions with both eyes open. Accommodate More Patients The refractor forehead rest has a range of 25±2 mm to accommodate more patients. OPD-Scan III Summary OPD Wavefront maps can be displayed for more effective diagnostics and patient education. Remote Touch Screen Display Large 10.4″ color LCD touch screen with an intuitive, ergonomic keyboard for simple one-hand operation. Tilt Screen for Near Testing The tilt screen and angled refractor head allows realistic near testing. TRS-6100 AUTOMATED REFRACTION VIDEOS Product Specs Measurement Range Spherical Power -29.00 to +26.75 D (0.12, 0.25, 0.5, 0.75, 1.00, 1.25, 1.50, 1.75, 2.00, 2.25, 2.50, 2.75, 3.00 D increments) Cylindrical Power 0.00 to ±8.75 D (0.25/1.00/2.00/3.00 D increments) Cylinder Axis 0 to 180° (1, 5, 15° increments) Pupillary Distance Distance PD: 48 to 80 mm Near PD (35 cm): 50 to 74 mm Full convergence Distance PD: 54 to 80 mm Prism 0.00 to 20.00Δ (0.1/0.5/2Δ increments) Refractor Head Auxiliary lenses Cross cylinder: ±0.25D, ±0.50D, ±0.25 D auto cross Occluder: Available Pinhole: 2.0 mm in diameter Color filter: Right red, Left green Polarizing filter: Right 135° / Left 45°, Right 45° / Left 135° Fixed cross cylinder: ±0.50 D (fixed with the Axis set at 90º) Spherical lens for retinoscope: 0D, +1.5D, +2.0 D (selectable by setting) Maddox rod (red): Right horizontal, Left vertical Dispersion prism (fixed): Right 6ΔBU, Left 10ΔBI Dispersion prism for binocular balance: Right 3-10ΔBD, Left 3-10ΔBU Dispersion prism for horizontal phoria: Right 3-10ΔBU, Left 3-10ΔBD Dispersion prism for vertical phoria: Right 5-15ΔBI, Left 5-15ΔBI Binocular open fogging: 0.00 to +9.00 D Near Gaze Distance 350 to 700 mm (can be changed in 50 mm increments) Field of Vision 40° (VD = 12mm) 39° (VD = 13.75mm) Pupillary Distance Distance PD: 48 to 80 mm Near PD (35 cm): 50 to 74 mm Full convergence Distance PD: 54 to 80 mm Forehead Rest Adjustment 25±2 mm Vertex Distance Scale 12, 13.75, 16, 18, 20 mm Horizontal Level Adjustment ±2.5° Control Box Touch Screen 10.4-inch color LCD Dimensions & Mass Refractor Head 16.06 x 4.21 x 10.91” / 7.1 lbs Control Box 10.23 x 9.06 x 8.15” / 4.6 lbs Relay Box 7.44 x 8.7 x 2.87 / 3.1 lbs Printer 3.98 x 3.39 x 4.76 / 1.3 bs General Info Printer Built-in IC Reader/Writer Built-in Power Supply 100 to 240 V AC 50/60 Hz Power Consumption Max. 90 VAC TRS SUCCESS STORIES Jerome Hernandez, OD Miami, FL Clearly in the era of COVID-19, the way we practice eye care has been impacted in ways we never saw coming just a few months ago. "Social distancing" is now a phrase we see applied to all aspects of how we interact with others outside of our ho... Read More Irvin Borish, OD ...Interpreting refractive data - not gathering it - will remain the cornerstone of Optometry. The future of the profession rests on its ability to adopt more of a medical model toward gathering patient data. Quantitative measures can be done o... Read More Tim Couch, Operations Director Atlanta, GA The Marco Total Refraction System has been a wonderful addition to our practice. It has helped us with clinical flow and efficiencies; improving glasses and contact prescription accuracy while decreasing the time to do glare testing etc. Our su... Read More The TRS-6100 Digital Refractor is comprised of a control panel with a large 10.4″ color LCD touch screen and an intuitive, ergonomic keyboard for simple one-hand operation TRS IN PUBLICATIONS TRS DIGITAL REFRACTOR TRS-5100 Streamlines Refractions – TRS – OO 2014 TRS-5100 Total Refraction System – Seegers – EP 2014 What is the JCC, and why do we use it? – Corwin – OP 2018 How to Approach the Duochrome Test – Corwin – OP 2018 OPTOMETRIC MANAGEMENT WOMEN IN OPTOMETRY TRS-6100 Automated Refraction System One-hand Operation Measurable Range -29D to +26.75D Cyl 0.00D to ±8.75D Portable Control Pad 10.4″ Color Touch Screen Minimizes Repetitive Stress Injuries REQUEST QUOTEExplore Epic Workstations Related Products EPIC-6100 Refraction Workstation All components of the refraction process are combined for peak efficiency in the EPIC-6100. There are Multiple ARK/NCT/OPD/LM configuration options av ... Deluxe Instrument Stand The new Marco Deluxe Stand combines a fresh, new updated appearance while embracing our traditional standards of performance, function and durability. ... CONTACT MARCO Marco | 11825 Central Parkway, Jacksonville, FL 32224 Toll Free: 1 (800) 874-5274 Local: 904-642-9330 Fax: 904-642-9338 CONNECT WITH US Important Warranty Information & Important Support Expiration Notices Marco Instruments are warrantied by Marco Ophthalmic, Inc. against defective materials and workmanship under normal use for a period of one year from the date of original purchaser invoice (An authorized distributor is not an original purchaser.) For more details and information LEARN MORE MARCO NEWSLETTER © 2019 - Marco - All rights reserved. Site Map Standard Terms & Conditions Service Agreement Terms & Conditions Privacy Policy