Products Catalogue Company Area Media News Contact Expo English Italiano BUNKER / SAFETY PERIMETERS PRODUCTS TEST BENCHES FOR VALVES Horizontal Horizontal with 0° reaction column disposal Horizontal with 30° reaction column disposal Horizontal with 45° reaction column disposal Vertical Tiltable Multiple Stations Water immersion GAS PSV PRV PVSV safety valves Cryogenic & Helium microleakage test benches TEST BENCHES FOR PIPES MOBILE WORKSHOP TEST BENCHES FOR ACTUATORS SPECIAL APPLICATIONS SKID Automatic Semi-automatic Manual ACCESSORIES BUNKER / SAFETY PERIMETERS PRODUCTS FILTERING CONCRATE MODULAR BUNKER BLOCKS Interlocking concrete blocks that can be used to assemble a safety perimeter around a testing area. The access door is... CUSTOMIZED BUNKER PROTECTIONS A control system will rules the access to thedangerous area, including specific behaviour in case of mandatory “Visual inspection” required... SANDWICH STEEL/WOOD MODULAR PROTECTION PANEL Sandwich panel STEEL / WOOD. Ballistic calculation according EN1522-FB1/FB2/FB6. Inspection windows in bullets proof crystal according EN1063-BR1/BR4/BR6. LIGHT MODULAR PROTECTION PANEL Light perimeter protection with steel web welded together to support structure. Test benches for valves, pipes & actuators FollowFollowFollow think’ PC PROGETTI S.r.l. a single shareholder company Via Artigianato, 3 - 22069 Rovellasca (CO) VAT IT 03110580135 Head Office and assembly workshop: Phone +390296749415 / +390236726776 | mail Technical service: mail Credits| Privacy | Legal ©2021 All Rights Reserved

BPA-250K Category: TEST BENCHES FOR ACTUATORS Homepage › Test benches for actuators › BPA-250K STATIC TORQUE PEAK MEASURE AND ENDURANCE DYNAMIC TEST Scotch YOKE ¼ turn actuator test bench, able to perform TORQUE test in static positions and dynamic endurance full load simulation. It is controlled by SKMM-100 pressurization skid and test data are collected by TestREC6.6.0- ACT control software. *Standard version is described. Test benches can be customized in any technical aspect in accord to client peferences.(Patent Pending) 🔍 NOMINAL TORQUE 250.000 Nm WORKING RANGE 10 – 100% TORQUE MEASUREMENT Indirect (loading cell) ANGLE RANGE 0.0° - 90.0° ZERO ADJUSTMENT -5.0° / +5.0° BIDIRECTIONAL REACTION FORCE INCLUDED STATIC TORQUE MEASURING ANGLE RES. 0,1° STATIC TORQUE FIXED POINT (PIN BLOCK) Available as option GENERAL ACCURACY 0.5% F.S. DYNAMIC TEST SIMULATION Dynamic brake ROTATION SPEED 0,1 – 2,0 °/sec (Other on request) DIMENSIONS (MECH) 2700 (L) x 2600 (D) x 1600(H) RELATED PRODUCTS BPA-40K BPA-400K BPA-130K BPA-10K SHARE Test benches for valves, pipes & actuators FollowFollowFollow think’ PC PROGETTI S.r.l. a single shareholder company Via Artigianato, 3 - 22069 Rovellasca (CO) VAT IT 03110580135 Head Office and assembly workshop: Phone +390296749415 / +390236726776 | mail Technical service: mail Credits| Privacy | Legal ©2021 All Rights Reserved