Beamex MyBeamex English Products Solutions Services & Support Literature & Videos Training & Events About us Contact Us By Application Pressure measurement and calibration Electrical measurement and calibration Frequency measurement and calibration Temperature measurement and calibration Intrinsically safe calibrations Calibration software Pressure generation By Product Type Portable Calibrators Calibration Software Workstations Temperature blocks Accessories By Name MC6 MC5 MC4 MC2 MC5-IS MC2-IS MB FB CMX Enterprise CMX Professional CMX Light MCS200 MC5P POC6 RTS24 PGXH PGM PGV PGPH PGL EXT Beamex® MC6 advanced field calibrator and communicator Click thumbnails to zoom Beamex MC6 is an advanced, high-accuracy field calibrator and communicator. It offers calibration capabilities for pressure, temperature and various electrical signals. The MC6 also contains a full fieldbus communicator for HART, FOUNDATION Fieldbus and Profibus PA instruments. Features Applications Specifications Links and Downloads Accessories The impossible made possible: combining advanced functionality with ease-of-use Beamex MC6 is an advanced, high-accuracy field calibrator and communicator. It offers calibration capabilities for pressure, temperature and various electrical signals. The MC6 also contains a full fieldbus communicator for HART, FOUNDATION Fieldbus and Profibus PA instruments. The usability and ease-of-use are among the main features of the MC6. It has a large 5.7” color touch-screen with a multilingual user interface. The robust IP65-rated dust- and water-proof casing, ergonomic design and light weight make it an ideal measurement device for field use in various industries, such as the pharmaceutical, energy, oil and gas, food and beverage, service as well as the petrochemical and chemical industries. The MC6 is one device with five different operational modes, which means that it is fast and easy to use, and you can carry less equipment in the field. The operation modes are: Meter, Calibrator, Documenting Calibrator, Data Logger and Fieldbus Communicator. In addition, the MC6 communicates with Beamex CMX Calibration Software, enabling fully automated and paperless calibration and documentation. User interface • Large 5.7” VGA (640x480) color touch screen • Touch areas optimised for finger use (gloves/stylus) • Membrane keyboard as secondary way to use • Number / QWERTY text entry • Multilingual interface Automated procedures • Deep integration with CMX • Connection diagrams and guided procedures Robust, lightweight and ergonomic design • IP65 casing • Drop / vibration tested • Wrist / neck strap User-interface modes • Meter - a simple and easy way to measure • Calibrator - measure or generate/simulate two signals at a time. • Documenting Calibrator - perform automated, documented calibrations of process instruments • Data Logger - perform multichannel (up to 9 channels) data logging • Communicator - field communicator for HART, FOUNDATION Fieldbus and Profibus PA instruments • Settings - manage settings of the MC6 Functions • Pressure measurement • Voltage measurement • Current measurement • Frequency measurement • Pulse counting • Switch state sensing • Built-in 24 VDC loop supply • Voltage generation • Current generation • Resistance measurement • Resistance simulation • RTD measurement • RTD simulation • TC measurement • TC simulation • Frequency generation • Pulse queue generation • HART Communicator* • FOUNDATION Fieldbus Communicator* • Profibus PA Communicator* *optional features Documenting calibrator • Memory for “unlimited” number of instruments • Plant structure supported • Connection diagrams shown • Combined error limit • Custom Transfer function, Custom Calibration points • Sort, Show, Plant Structure, Find, … • Deeper integration with Beamex CMX Calibration Software Pressure modules • Up to 4 internal pressure modules (3 + B) in extended case • In slim case => Barometric module + EXT connection • Dedicated location for Barometric module • Internal relief valves (<20bar/300psi modules) • Connection for External Pressure Modules (EXT) Connections Right side: • Power • Ethernet • 2 x USB-A • USB-B Top: • Internal pressure modules (P1, P2, P3) • External Pressure Module (PX) • Second resistance/RTD channel (R2) Rear: • Barometric module Front: • TC1 (Universal), TC2 (Miniplug), connections Modularity and options • All electrical / temperature functions are included as standard Case bottom choices: • Flat (no room for internal pressure modules, only Barometer) • Extended (room for internal pressure modules) • Optional internal pressure modules Optional User Interface Modes: • Documenting Calibrator (with CMX Communication) • Data Logger • HART Communicator • FOUNDATION Fieldbus Communicator • Profibus PA Communicator • Pressure / Temperature Controller Communications Standard accessories • Accredited calibration certificate • User guide • Computer cable (USB) • Battery charger / eliminator • Internal LiPO battery pack • Test leads and clips • Appropriate pressure T-hose with internal pressure modules • CD-ROM with user manual, software tools and product information You are here:Home / Products Subscribe for e-news Subscribe for Calibration World Products By Application By Product Type By Product Name Solutions Industries Solutions Customer Case Stories Services & Support Services: Hardware Services: Software Software Support and Maintenance Programs Literature & Videos Literature Videos Corporate Magazine Training & Events Classroom Training eLearning & Videos Exhibitions & Events About Us Contact Us About Beamex Accreditations Calibration Facilities Careers Contact Form © 2014 Beamex Oy Ab. All rights reserved. Legal Notice Privacy Policy Contact Form