Login / Registration United States Home Products & Solutions Industries Services & Support Events & Expertise About Us Contact Us Turbidity Meters / Turbidity Sensors In-line, high performance turbidity meter / turbidity sensors for process control How can we assist you? Expertise Library Product Finder Service Finder Request Info Get a Quote Turbidity Meter / Turbidity Sensor Frequently Asked Questions (FAQ) What is turbidity? Turbidity is an optical characteristic which refers to the degree of clarity of a liquid. Turbidity in water is caused by individual suspended particles or colloidal matter that scatters or obstructs light transmittance: the higher the concentration of suspended particles/colloidal matter, the higher the turbidity. Such particles are normally too small to be detected by the human eye; therefore, turbidity measurement has to be done with a turbidity meter. Everyday examples of turbid liquids are: Milk - contains emulsified protein/ oil drops in water Wastewater - contains suspended particles Wheat beer - contains yeast cells Why is turbidity measurement important? In many cases, turbidity meters are used for the evaluation of the quality of water or the efficiency of a filtration process. Turbidity measurements are important indicators in many industries and applications as they not only influence the yield of the industrial process but also detect factors which are detrimental to a system. What influences turbidity measurements? As mentioned above, turbidity measurement determines the degree that suspended particles in a liquid medium, scatter light. The scattering depends on Particle concentration: Higher particle concentrations result in higher scattered light and therefore higher turbidity readings Particle shape and size: Particles smaller than 1/10 the wavelength of visible light, scatter light symmetrically. Larger particles (typically with a diameter greater than the wavelength of visible light) scatter light asymmetrically. Therefore, for measuring turbidity, the scattering angle must be considered. Light wavelength: as mentioned above, the intensity of the scattered light is dependent on particle size. In addition, the presence of color in a liquid may reduce the light being measured at the detector. As a result, the use of a suitable light wavelength must be considered for turbidity measurement. As a consequence of the three points above, it is only possible for turbidity to be used as a characteristic property of a sample if the measurement method is standardized. For example, in many brewery applications the liquid to be measured is yellowish in color and contains yeast particles. Therefore, to check for filter breakthrough, the amount of forward and side-scattered light is measured at an angle of 25o and 90o compared to the light source for quality assurance purposes. Red (650 nm) and blue (460 nm) light sources are also specified in guidelines for turbidity and color monitoring. What are the common interferences in turbidity measurement? Turbidity measurement can be adversely affected by the presence of light-absorbing substance or those that fluoresce. This interference can be minimized by using wavelengths other than those in visible light (i.e. 860 nm near-infrared). Air bubbles can interfere with turbidity measurement, so choosing an installation point where the presence of air bubbles is minimal, can be important. Windows fouling is also critical in turbidity measurement. State of-the-art technology and the ratio principle can compensate for sample color changes, aging of the lamp used as the light source and optical window fouling. What is a turbidity meter? A turbidity meter is a sensor that measures turbidity. Turbidity sensors have two things in common: A light source – e.g. a light emitting diode (LED) One or more light detectors – typically photo diodes What are the common turbidity meter measurement methods in the market? There are several types of in-line turbidity sensors that are optimized for specific measurement ranges and different applications. Forward scattered light/absorption: Turbidity sensors with this technology (e.g. METTLER TOLEDO's InPro 8300 RAMS series and InPro 8600i/D1 and /D3 series) are designed for applications with low to medium turbidity levels. Color measurement (yellowness) is also available in the InPro 8300 RAMS type TCS and COMBINE/ InPro8600i/D3. Backscattered light: These sensors (e.g. METTLER TOLEDO's InPro 8050, InPro 8100 and InPro 8200) are designed for samples with a high particle concentration of up to 250 g/l suspended solids. Depending on the application, the METTLER TOLEDO sensors are available in both stainless steel and a polysulfone body for pharmaceutical and wastewater use respectively. How do you calibrate a turbidity meter? Calibrating a turbidity meter can be done in three different ways with a transmitter. The first type is called manual calibration. This is the fastest but lowest level of calibration. The user can change the offset and the slope, and a measuring value is calculated and displayed. The second type is called multi-point calibration. This type of calibration achieves the best linearity for the process being measured. Two, three, four and five point calibrations can be performed – typically off-line with an uninstalled sensor. The third type is process or in situ calibration. This is an on-line calibration where the operator enters the process calibration menu on the transmitter and saves the current turbidity reading. The operator then takes a sample of the process liquid to measure it against a laboratory instrument in order to obtain a reference turbidity measurement. Depending on the turbidity sensor and transmitter, multiple point in situ calibrations are possible (e.g. METTLER TOLEDO M800 transmitter and backscattered light turbidity sensor). METTLER TOLEDO turbidity meter/turbidity sensors can be used in many industries such as: Biotechnology Pharmaceutical Chemical Processing Petrochemical Food and Beverage Breweries How do you select the right turbidity sensor? The turbidity sensor should be selected based on the required measurement range as well as the application. METTLER TOLEDO offers several versatile turbidity sensors that meet the requirements of many different applications, such as: Fermentation Biomass growth (optical density) Crystallization Phase separation Water in oil Filter breakthrough Activated sludge Post filtration of beer and yellowish measurement Wastewater Please contact your local METTLER TOLEDO representative for details. Overview Publications Services Get a Quote Get a Quote Get a Quote Low to Medium Turbidity Measurement (Forward/ 25⁰ / 90⁰ Scattered Light) Medium to High Turbidity Measurement (Backscatter Light Technology) RAMS Optical Product Monitor Broad measurement range from low to medium turbidity - higher productivity and quick system start-up. Broad measurement range from medium to high turbidity - quick process adaption and easy installation. Rugged and easy to use turbidity system designed for use in phase separation by turbidity/color and for product identification processes. High performance solution for breweries Fast start up with Plug and Measure thanks to Intelligent Sensor Management (ISM) Calibration data stored in the sensor Sample color monitoring (optional) Suitable for pharmaceutical, chemical or wastewater applications In situ multi-point calibration Small 12mm diameter sensor saves valuable space Pg 13.5 thread for easy installation and interface into housings Reliable solution for food & beverage industry Simple and cost effective measurement with direct 4/20 mA output Various sensor options available for special applications Display less News Product Catalog 2016 / 2017 White Papers Chemical Plants Become More Efficient Achieving PAT Goals with Process Analytics Equipment Webinars Ozone Measurement and Application Fighting Corrosion in Power Plants Is this information helpful? You are here: Home > Process Analytics > Turbidity Meters / Turbidity Sensors Products & Solutions Laboratory Balances Industrial Scales Retail Weighing Scales Rainin Pipettes and Tips Analytical Instruments Automated Reactors and In Situ Analysis Process Analytics Inline Product Inspection Transport and Logistics Services & Support Service Offerings Product Registration Technical Support Events & Expertise Expertise Library Literature: White Papers, Guides, Brochures Technical Documentation On Demand Webinars Videos Live Events Seminars & Training Trade Shows & Exhibitions Live Webinars About Us Jobs & Careers METTLER TOLEDO Business Description Innovation & Quality Global Presence Sustainability Management Investor Relations Newsroom Quick Links Contact Us Search Document Finder Product Finder Service Finder Videos © Mettler-Toledo International Inc. all rights reserved Legal, Terms, Privacy |Contact us |Site Map |Global Presence