ISO 9001 en / de / kz / ua 04-05 October Course BP-302 "Practice of Lean Manufacturing" The course is recommended for senior managers, as well as leading technical specialists responsible for the reliability of the MRO system at the enterprise October 18-20 TOP-105: Tribodiagnostics. Analysis of oils and lubricants A course for specialists responsible for ensuring the reliability of machines and equipment, as well as diagnostics and identification of the causes of the development of defects 194044, St. Petersburg, st. Pig- iron, 40 (499) 4033707 (812) 3350085, E-mail: info@baltech.ru the main Products The educational center Technical service Articles Certificates Contacts Proton-SPP Centering Thermal imagers Pyrometers Induction heaters Balancing Balancing machines Vibration diagnostics BALTECH VP-3470 Vibration analyzer CSI 2125 Vibration analyzer CSI 2130 Vibration analyzer CSI 2140 Vibration monitoring CSI 9420 Vibrometers Stationary systems Bearing test stand Proton-SPP GOST R 52859-2007 GOSTs Oil analysis Tool print version make a homepage - Vibration diagnostics - Bearing test stand - Proton-SPP DOWNLOAD BROCHURE PROTON-SPP BEARING DIAGNOSTICS Download the PROTON-SPP-I instruction Download the PROTON-SPP-II instruction Stands "PROTON-SPP" are intended for the correct detection of rolling bearing defects (new and used with an inner diameter, depending on the modification of the stand 07-50mm, 40-150mm) before their installation during the repair of industrial equipment (incoming inspection), as well as bearing diagnostics can be performed. The design of the stand was based on the measurement technique (MVI) of vibration of rotating rolling bearings with a cylindrical bore and propagating to the following bearings, incl. with one and two shields and seals: single-row and double-row deep groove ball bearings; single-row and double-row angular contact ball bearings; radial roller with short cylindrical rollers; tapered roller bearings; roller radial spherical single-row and double-row; spherical angular contact roller bearings. The MVI used in the design of the stand was approved by the head institute of the bearing industry VNIPP (Moscow) on July 01, 2004. under the number MVI VNIPP.002-04 and is intended for mandatory use in the bearing industry and use at consumer enterprises and other users of rolling bearings. The development of the bearing quality control stand was carried out in cooperation with the specialists of the bearing plants. The bearing inspection stand is certified and entered into the State Register of Measuring Instruments. These stands have an international certificate IORS: 2010 and compliance with ISO 15242-1,2: 2004. Operators working at any incoming inspection stands must be certified at the training center of Baltech LLC according to the PU-201 course "Reliability of bearing assemblies". Purpose: Automatic diagnostics of bearings of the technical condition of rolling bearings of various modifications. Registration and analysis of vibration signals in order to identify the features of the technical condition of rolling bearings. Keeping a database of rolling bearings, selection of similar, in terms of geometric dimensions, bearings for replacement. Specifications Bore diameter of controlled bearings Proton-SPP-II (Proton-SPP) 40 - 150 mm (07 - 50 mm) The number of sensors used to record information from the bearing under study 1 vibration sensor Working range of measurements (according to MVI) 25-10,000 Hz RMS measurement amplitude range of vibration velocity, dB 40-110 Signal to noise ratio, dB not less than 50 Rotation speed according to standard 1800/900 rpm The time required for diagnostics of one bearing, excluding the time for mounting and dismounting the bearing on the machine (depending on the signal registration parameters and the type of computer) 5-60 sec (optional 0.6 sec) Pneumatic system pressure, atm 4-6 The type of computer used not worse than Pentium IV Power consumption, kW no more than 1.5 Stand weight 600.0 ± 10.0 kg dimensions 1015x950x1473 mm Back Home T / f (812) 335-00-85 All rights reserved. When using the site materials reference is obligatory