Login » Home Products Service News and press About us Downloads Contact ScanLaf ScanVac ScanCool ScanSpeed ScanConstruct Fortuna Sterile Cabinets Jupiter Class 1 Cabinets Mars Class 2 Cabinets Orion Class 2 type B2 Cabinets Minerva Titan/Titan PCR table top sterile cabinet Atlas table top Class 1 Cabinet Sirius Class 2 & antibiotic & Cytotoxic handling cabinet VacSafe Pumps ScanSpeed MiniVac Evaporators ScanSpeed MaxiVac Evaporators ScanVac CoolSafe Freeze Drying CoolSafe Superior, floor standing Snowbird ChillSafe Cooling & Cryobaths Micro Centrifuge Range Low Speed centrifuges Multipurpose Centrifuge Range Organisation Board of directors Management team The story af LaboGene Mission statement Sales & business development Supply chain Products ScanLaf ScanVac VacSafe Pumps ScanSpeed MiniVac Evaporators ScanSpeed MaxiVac Evaporators ScanVac CoolSafe Freeze Drying CoolSafe Superior, floor standing ScanCool ScanSpeed ScanConstruct VacSafe Pumps VacSafe 15 portable energy-saving vacuum/aspiration pumps with low pressure and high performance. Eco friendly operation- traps and contains all vapours and condensates Maintenance Free with no oils or membranes Corrosion resistance and strong cleanable design Noiseless, portable, compact design Capacity of 2x10 l/min or 1x 15 l/min with 30mbar @ 24°C & 7mbar @ 2°C Low power consumption of 100 watts Low operational costs and no expensive spares Ideal applications are rotary evaporation, gel dryers, vacuum concentration, Elisa washers, vacuum desiccators, distillation, vacuum ovens, Buchner flask filtration, etc. Principle of Operation The VacSafe 15 vacuum/aspiration pump works on the the “Venturi” principle, similar to the old, traditional water jet pump, A high pressure water jet is produced by the pump and allowed to expand into a larger volume. This expansion thereby creates a low pressure which in turn produces a vacuum on the pump side inlet. The speed of the water produced by the jet and the water volume expansion are the determining factors in the air removal capacity The water temperature determines the ultimate vacuum achieved, the lower the temperature the lower the vacuum. Advantage of using the VacSafe 15. The VacSafe 15 only uses 4 litres of water and any solvents or acids that are expelled via the vacuum are collected in the water reservoir and are prevented from being exhausted into the laboratory environs. The contaminated water can then be disposed of safely and replaced with fresh water. It has been calculated that operating the VacSafe for 1 day (8hours) can safe 5tons of water compared to the Water Jet pump. Water Temperature Pressure Achieved 40° 73mbar 30° 43mbar 24° 30mbar 10° 14mbar 5° 10mbar 2° 7mbar Contact LaboGene You are always welcome to contact LaboGene by phone (+45) 3940 2566 Name: * E-mail: * Telephone: * Message: * Copyright 2015 LaboGene™. All rights reserved | LaboGene ApS | Industrivej 6-8 | DK-3540 Lynge Terms & Conditions