+39 031 790005 info@ennebielettronica.com Mappa Home Electronic Manufacturing Electromedical Devices Electronic Devices Medical Power Supply Systems Important reference point for the design, manufacturing sale of electronic and electro-medical equipment since 1995 CONTATTACI Ennebi Elettronics develops a series of systems and customized solutions for power application in the medical field The three main types of medical power supplies proposed by Ennebi Elettronica are: DC/DC Converters DC/AC Inverters Main AC network power supplies Depending on the type of customer need they may have Single or double insulation with or without galvanic isolation Input protected by fuse Working temperature: normal range (0 to 50 ° C) or customized for other temperatures CE certificated for electrical equipment and UNI CEI EN ISO 13485: 2012 Medical All the power supply system are Compliant with Medical Safety standards and EMC specifications for medical applications. Ennebi Elettronics can rely on a team of engineers specialized in the design, development and customization of Power Medical Systems. The following table shows the characteristics of a POWER SUPPLY NETWORK for MEDICAL LASER DIODE. It is the model SB115 / 230 / 24V / 10A Input voltage 115/230 Vac +/- 15% a 50/60Hz Output 24Vdc – 10Amp Input technical characteristics Worldwide input range with active PFC. Inrush current Limited – on circuit stamped single fuse or externally (to implement). EMI / EMC with leakage current to GND in line with the regulations for electrical safety of electrical equipment EN60601 Heatsink Protection over 90 ° +/- 5 ° with hysteresis. Able to work in parallel with other power supplies. Output technical characteristics Insulation in line with the regulations for electrical safety for electro-medical equipment EN60601 Overload Protection (Inom + 10%) Continuous Short-circuit protection Optional O’ring diode I/O – signal Led for Power on and output voltage regulator trimmer Working temperature Range from -10°C to +40° Storage Temperature Range from -20°C to +80°C Dimensions 220mm x 70mm x 85 mm Reference standards EN 60601-1(1990) + A1(1993)+A2(1996)+A13(1996)EN 60601-2-22(1996) (Par.10) CEI EN 60825-1(2003), [EN 60825-1 (1994) + A11(1996) + A1(2002) + A2(2001)], EN 60601-1-2(2001) © 2016 Ennebi Elettronica S.r.l. - Tutti i diritti sono riservati P.Iva 02513320966 - Sede legale e operativa: Via Papa Giovanni XXIII 13/15, 22060 Novedrate (CO) REA: 293924 - PEC: ennebielettronicasrl@pec.it Capitale sociale: 10.400€ interamente versato Privacy Policy Questo sito memorizza le preferenze dell'utente ed utilizza tecnologie che consentono di effettuare analisi statistica in forma anonima delle pagine e del comportamento dei visitatori. Gli utenti possono tuttavia disabilitare le funzionalità non necessarie. Ok Dettagli