About us Products News Certifications & Honors Innovations & Commitments Contact us +36 1 280 6900 mail vpn_key BABYLIFE KLA-145L Fluorescent or LED type Phototherapy Unit PROPERTIES keyboard_arrow_down TECHNICAL DATA keyboard_arrow_down GALLERY keyboard_arrow_down Featured benefits Mobile floorstand device Heavy duty stand with 4 castors (2 lockable) 3.2" touchscreen colour LCD Working & total hour counter & clock Adjustable treatment time & intensity (25/50/75/100%) 6 pcs fluorescent bulb or LED version Stand with two fix height Horizontal radiating fix head Properties Version M Version MA Version MB Version MU Two-position stand with fix height and horizontal radiating head Technical data Space requirement: 500x1000x1500 mm Height of radiating head: Versions M and MB: 980 or 1380 mm, Versions MA and MU: between 1260-1660 mm Lifetime of fluorescent bulb: min. 2000 working hours (bulb) min. 50 000 working hours (LED) Range of radiating wavelength: 400/500 nm Operation distance: 300/400 mm Size of radiated area: 220x420/250 x 450 mm Effective radiation intensity: (Ebi): > 15 mW/sq.cm/nm at 300 mm (6 bulb), > 35 mW/sq.cm/nm at 300 mm (LED) Luminous emittance at 40 cm: at least 6200 lux Type of bulb: LightTech LTC 18W/2 G 11 (blue), PHILIPS PL-L 4P 18W/830 (white) Type of LED: MLED 15 Electric power: 108 W (6 bulb)/100 W (LED) Mains voltage: 230V/50Hz+/-10% (230V/60Hz or 110V/50-60Hz on request) Weight: 24 kg (standard version) Additional features: Skin temperature sensor (optional) Irradiance meter(optional) Display of hospital's name or other text (optional) Eye protection mask (optional) Gallery 1953 - 2021 © MEDICOR Zrt. | Privacy Policy