Menü ausklappen HomeProductsLeak DetectionLeak DetectorsMultipurposeASM 340ASM 340, 200/240 V, European mains cable, configurable I/O interface board Similar Image ASM 340, 200/240 V, European mains cable, configurable I/O interface board Order number: JSVA00A2MH9A Powerful rotary vane pump with 15 m3/h backing pump capacity 2.5 l/s helium pumping speed 200/240 V operating voltage, European power cable Customizable and detachable operator interface Integrated SD card for data processing Fully configurable I/O interface (37 pins D-Sub) to fit to customer needs Wired or wireless remote control and sniffer probes available as accessories Request product Compare product Remember product Technical data Dimensions Accessories Downloads Backing pump capacity 15 m³/h | 8.83 cfm | 250 l/min Connection flange (in) DN 25 ISO-KF Detectable gases 4He, 3He, H2 Dimensions (L x W x H) 547 x 375 x 393 mm | 21.54 x 14.76 x 15.47 inch I/O interfaces Configurable 37 pins I/O Interface: Connection, device side Configurable 37 pins I/O Minimum detectable leakage rate for helium (sniffing leak detection) 5 · 10-10 Pa m3/s | 3.75 · 10-9 Torr l/s | 5 · 10-9 mbar l/s Minimum detectable leakage rate for helium (vacuum leak detection) 5 · 10-13 Pa m³/s Operating temperature (sniffing test) 0-35 °C | 32-95 °F | 273-308 K Operating temperature (vacuum test) 0-45 °C | 32-113 °F | 273-318 K Power consumption max. 850 W Protection degree IP20 Pumping speed for He 2.5 l/s Sound power 54 dB(A) Sound pressure level 54 dB(A) Start-up time (20°C) without calibration ~ 3 min Supply 200 – 240 V, 50/60 Hz Test method Vacuum and sniffing leak detection Weight 56 kg | 123.46 lb Adapters (electrical) Adapter Cables 37 pin D-Sub/25 pin D-Sub adaptor cable A333758 Compatibility modules Compatibility module, I/O HLT 5xxx 122742 Profibus Profibus board for ASM 340 122253 Cables Cables to connect an external gauge to an ASM 340 equipped with 37 pin I/O board Cable for external measurement unit, 3 m A333746 Cable for external measurement unit, 10 m A333747 Cable for external measurement unit, 20 m A333748 Mains cables / Power cords Mains cable, 2.5 m, Switzerland 103718 Mains cable, 2.5 m, U.K. 104411 Mains cable, 2 m, Italy 104758 Connections Bypass options Bypass option with Euro-style safety plug PT 445 411 -T Exhaust connectors Exhaust connector for external oil mist separator - DN 25 ISO-KF 122405 Control Devices Remote controls Remote control, leak rate in mbar l/s, legend in English 106688 Remote control, leak rate in Pa m 3 /s, legend in English 108880 Remote control RC 10, wired/wireless, for ASM and ASI leak detectors 124193 Protection Inlet filter, 5 µm, DN 25/25 ISO-KF 105844 Sniffing/Spraying Helium pistols/ spray guns Helium spray gun "Elite-Kit", spray gun with additional accessories in a compact case 109951 Helium spray gun, standard 112535 Sniffer probes LP 503, sniffer probe with standard tip, 3 m BG449207-T LP 505, sniffer probe with standard tip, 5 m BG449208-T LP 510, sniffer probe with standard tip, 10 m BG449209-T Sniffer probe, 5 m hose length, rigid 9 cm nozzle SNC1E1T1 Sniffer probe, 10 m hose length, rigid 9 cm nozzle SNC2E1T1 Transport Transport cart, 2 wheels, for ASM 340 122570 File Type Language Date Filesize Data sheet PDF EN 11-23-2021 ~ 613.9 KB further downloads in Chinese Czech Dutch English French German Italian Korean Polish Portuguese Russian Spanish Swedish File Type Language Date Filesize ASM 340 - Maintenance Instruction (121762MEN_05) PDF EN 06-22-2021 ~ 4.4 MB Operating instructions ASM 340 - Operating (121762OEN_06) PDF EN 01-12-2021 ~ 12.2 MB Operating instructions ASM 340 - Short Guide (121762CEN_04) PDF EN 11-30-2015 ~ 729.1 KB Operating instructions RS232 Communication Protocol - Detectors 3G - Operating (122215OEN_05) PDF EN 11-30-2015 ~ 1.1 MB Operating instructions RS232 Communication Protocol - Detectors 3G - Memo (122215CEN_03) PDF EN 11-30-2015 ~ 102.3 KB Operating instructions 37 pin I/O Interface - Operating (123894OEN_03) PDF EN 06-25-2018 ~ 2.1 MB Operating instructions 37 pin I/O Interface - Installation (123895IEN_03) PDF EN 06-12-2018 ~ 1.1 MB Your local contact Pfeiffer Vacuum Benelux B.V. Newtonweg 11 4104 BK Culemborg Netherlands T +31 345 478 400 F +31 345 531 076 write message Newsletter Sitemap Data Protection Legal Notice General Terms and Conditions Newsletter Literature Ordering Downloads Contact Group Press Investor Relations Careers © 2021 Pfeiffer Vacuum GmbH