T. +31 347 342 777 HOME WASTEWATER SAMPLING ENCLOSURES WASTEWATER SAMPLING TECHNIQUES OPTIONAL EQUIPMENT ABOUT US NEWS INLINE GUILLOTINE SAMPLER EN ISO 5667 / NEN 6600-1 The Efcon® ILS Inline Guillotine is an automatic fixed volume sampler for use on 100 % filled and pressurised effluent lines. The sampled process must be liquid, and free of air / hard solids. Designed for sampling of raw wastewater, the Efcon® Inline Guillotine sampler works with a maximum process pressure of 2.5 Bar and optional up to 5 Bar. This sampling method complies to ISO 5667-2, 5667-10 and NEN 6600-1 and is equipped with a pneumatic actuator (6-8 bar). The Inline Guillotine Sampler can be volume-, time-, or event proportional controlled via an Efcon® controller or (when ordered only the sampler hardware) via a third party PLC. Mounting fittings for the inline guillotine sampler can be supplied (option) for plastic or Stainless Steel piping. Note! Use in hazardous environment is forbidden unless mentioned in the manual and on the product. Dowload the Efcon® Inline Guillotine Sampler Datasheet (PDF) to discuss it with your colleagues via e-mail or a printed version. SERVICE Efcon® Water offers full support with annual service and on site support in The Netherlands & Belgium. Throughout Europe we support several specialised distributers. Spare parts are in stock at our distributers but can also be shipped global with UPS, DHL, Fedex etc. Downloads Service Datasheet (ENG) Service Datasheet (NED) INTERNATIONAL RESELLERS/DISTRIBUTERS At this moment Efcon® Water has multiple specialised resellers in Europe. For interest to become a member of the Efcon® family please contact us. More information? Call +31 347 342 777 or E-mail to info@avm-efcon.nl. CONTACT / EXPERT ADVICE? Call +31 347 342 777 or E-mail to info@avm-efcon.nl. EFCON INLINE GUILLOTINE SPECIFICATIONS EN ISO 5667 / NEN 6600-1 Application Influent & Effluent Advantages • Pneumatic actuator for Heavy Duty applications • Simplicity of the system • SS316 house and piston • For raw wastewater and sludge sampling • No sensors in medium giving minimum maintenance • Up to 2.5 Bar medium pressure optional up to 5 Bar Matching Enclosures • Efconomy Samplers • Industrial Samplers • Custom Made Projects Conditions • Minimum diameter of 15 mm • Main application in sampling raw wastewater • Minimum flow velocity at horizontal pipe 0.5m/sec Maintenance • Replace seals (in general every year) OVERVIEW OF SAMPLING TECHNIQUES • Go to overview of all wastewater sampling techniques OVERVIEW OF WASTEWATER SAMPLERS • Go to overview of wastewater sampler enclosures Wastewater Samplers Overview of all enclosures Efcon®omy Wastewater Samplers Efcon® Industrial Wastewater Sampler Efcon® Projects Wastewater Sampling Efcon® Carrybox Waste Water Sampler Techniques and equipment Overview of all sampling techniques Vacuum Sampler Inline Guillotine Sampler Peristaltic Sampler ILS 2W & 3W Samplers Electro Magnetic Flowmeters Special Flowmeters Wastewater Pumps & Pits Water Level Sensor About us & more Efcon® Water & AVM Efcon® Water Service Downloads International Distributors Customers Nederlands Efcon Water B.V. Hoefslag 11 4131 ME Vianen The Netherlands Phone +31 +347 342 777 Fax +31 +347 342 788 info@avm-efcon.nl Go to: contact page © 2021 Efcon® Water B.V. Terms & Conditions Contact us