Home News Servizi Prodotti Partners Contatti P 60 Pressure ranges -0,95...60 bar Pressure medium Air Possible areas of application SIKA test pumps and pressure generators can be used everywhere, including on site in workshops, test and measurement rooms as well as laboratories. They cover a broad spectrum of industries with diverse applications. Assembly and commissioning Manufacturing and production Maintenance and service Quality assurance and test equipment monitoring Repair Description Practical The test pumps and pressure generators are designed to enable the direct connection of all pressure systems to be tested using adapters. The test sample is easily connected using the rugged industrial hose with integrated quick coupling and supplied adapters. The reference is fitted directly at the top of the pump using a positioning adapter. The required test pressure is initially generated using the handles and then adjusted precisely with the fine adjustment valve. As a result, the pressure on both instruments is the same. The pressure relief valve allows continuous pressure reduction and ensures accurate and easy testing, even with decreasing pressure. In the simplest case, the pressure is indicated by an analogue pressure gauge. An easy to read digital pressure gauge or hand-held instrument can also be used. The accuracy or adjustment of the pressure measuring device being tested can be checked by comparing the indicated reference value with the measured value for the device under test. Mobile and independent SIKA test pumps and pressure generators are ideal for mobile use. Their low weight and compact design make them easy to transport directly to the measurement site. The instruments can be used immediately and do not require an additional power supply. There is no need to take along nitrogen bottles or connect the equipment to a compressed air network. Wear-free manual pressure generation is simple and easy, regardless of ambient temperature and orientation. OEM version and full version Depending on the model a matching pressure hose is part of the basic configuration of the OEM version of the test pump. The hydraulic hoses are fitted with a self-sealing quick coupling. Inch, conical or metric adapters for all commonly used connection threads are available in the full version. A matching seal kit is also included with the pump. All of the equipment is held in a carrying case with a foam-rubber insert. Technical Features OEM version Pressure medium Air Dimensions Approx 240 x 170 x 50 mm Weight Approx 1.1 kg Pressure ranges Negative pressure -0.95 bar Positive pressure 60 bar Connections Reference G¼ Test sample G¼ with quick coupling and pressure hose (1m) Full version Adapter kit Chrome-plated brass Gasket kit Teflon® Seals and O-rings Dimensions Approx. 450 x 370 x 110 mm Weight Approx. 4.2 kg Documentazione Datasheet 1.204 KB Prodotti Anemometri a filo caldo Calibratori di pressione portatili Calibratori di pressione da laboratorio Calibratori di segnale portatili Calibratori di segnale da laboratorio Calibratori di temperatura Calibratori di umidità Celle di carico Indicatori digitali Manometri digitali Misure di comfort Pompe manuali generazione pressione vuoto Pressure and Temperature scanner Pressure scanner miniaturizzati Sensori di livello Sensori di pressione Software Termometri Digitali Torsiometri Trasmettitori rotanti di segnali elettrici Supporto | Contattaci Web Design by AvanthaDesign.com | Copyright © Danetech S.r.l. All rights reserved.