Meny FocusParameterMonitor Home » Våra produkter » Primes Produkter » Focus & Power » FocusParameterMonitor The PRIMES FocusParameterMonitor (FPM) enables rapid determination of laser beam parameters. The measuring processes are completely controlled from the existing machine programs. Measure Laser Beams Quickly The FocusParameterMonitor (FPM) is the first system on the market that allows quick determination of the laser beam parameters in the process zone. Its advantages: It can easily be integrated into the system, and moreover the fieldbus interface (PROFINET, PROFIBUS) provides easy connection to many industrial control systems and networks. Overview At the beam inlet, an electrical closing device and a protective glass protect the measuring device from contamination. The beam coming from the laser and the processing head is directed into the device via a deflecting mirror. The beam reaches the measuring components via a beam splitter and a further deflecting mirror. The laser power is measured calorimetrically. For this purpose the laser irradiates the absorber for a defined period. With the help of the known heat capacity, the temperature increase in the test body and the irradiation period, the power is calculated. The camera-based beam analysis unit measures the beam geometry and beam position with a CCD sensor. A fieldbus interface lastly forwards the measurement data data to the system controller. An additional PC is not required. Measurements Laser Power Beam Dimensions Beam Position at: Wavelength: 1030–1090 nm Maximum Laser Power: 8 kW Measuring Period Typical: 0.3 s (for beam measurement) Operation Flexible in mounting, fully automated operation The FPM can be mounted either horizontally or vertically. Because of the risk of dirt contamination, vertical fastening with horizontal beam incidence is recommended. Its use is fully automatic. Due to connection to the machine controller, the measurement processes can be completely controlled from the existing machine programs. The fieldbus system performs this communication. Models and options For both models of the FPM, a fieldbus interface (PROFIBUS, PROFINET) can be selected. All models are fitted with a replaceable protective glass. FocusParameterMonitor Image 1:1 maximum divergence 60 mrad FocusParameterMonitor Image 2:1 maximum divergence 100 mrad An option is currently under development to allow the output of the measured light density distribution via fieldbus systems also. Download PDF KONTAKTPERSON Linda Jingnert Försäljning eftermarknad +46701168534 linda.jingnert@permanova.se Permanova Lasersystem AB Krokslätts Fabriker 30 431 37 Mölndal Växel: 031 706 1980 Fax: 031 86 46 12 Information: info@permanova.se Permanova.se Användningsområden Våra Produkter Vårt kunskapscenter Permanova som partner Om Permanova Kontakt Integritetspolicy Hitta oss i sociala kanaler Copyright © 2021 Permanova Lasersystem AB. All rights reserved. Webbplatsen använder cookies för att hjälpa oss ge dig den absolut bästa upplevelsen när du besöker oss. Läs mer om vår cookie policy. Genom att stänga detta meddelande godkänner du vår användning av dessa cookies samt att du har informerats.Cookie settingsACCEPT