We are here for you! +49 2602 9449-0 info@modul-technik.de Product Quickfinder MS 1054 Application area: Intensive care Product group: Media column MORE FREE SPACE THROUGH FREESTANDING PLACEMENT IN THE ROOM With our MS 1054 media column which can be installed freestanding in the room, we offer you a flexible system for all room situations in which wall-mounted supply systems are out of the question. The core of the system is a compact media column which is firmly anchored to the floor and ceiling and in which all required media accesses such as medical gases, high voltage current, data and communications technologies are integrated. The optionally available accessories make the MS 1054 a multi-purpose and flexible treatment station. Side-mounted stainless steel tubes incorporate infusion bottle holders, lighting elements and other aids. But storage consoles and drawer pull-outs can be installed at any height you want – ideal for every work situation and always exactly where they are required in the room. TECHNICAL DATA Electric specifications Nominal voltage: 230 V - 240 V / 50 Hz - continuous operation Protection class: I Protection type: IP 20 Lighting technology LED downlight (optional) Nominal voltage: 95 V – 260 V / 50 – 60 Hz Protection class: II Operating type: electronic converter (1-3,6 W) Lamp output: 1 W Protection type: IP 20 Operating pressure of medical gas technology Oxygen: 5 bar Compressed air: 5 bar Vacuum: - 0,8 bar Laughing gas 5 bar CO2: 5 bar AGSS: 5 bar General information Media current feed: From the top Optional support tubes: Side-mounted, rigid or swivel Total additional load: Max. 160 kg DOWNLOAD DATA SHEET ASSEMBLY, CLEANING, MAINTENANCE AND REPAIR Assembly Cleaning Maintenance and repair Accessories No items were found. modul technik GmbH · Rudolf-Diesel-Straße 5 · D-56410 Montabaur +49(0)2602-9449-0 +49(0)2602-9449-11 info@modul-technik.de E-MAIL TO MODUL TECHNIK Terms and conditionsDownloadsCareerCertificatesContactPrivacy policyImprint It will load necessary cookies, Google Fonts, Google Maps, OpenStreetMap, Youtube and Google Analytics. More details in our privacy policy and our imprint. Accept I disagree

We are here for you! +49 2602 9449-0 info@modul-technik.de Product Quickfinder MODUL CLASSICS WALL SUPPLY UNITS (CLASSIC) NORMAL CARE IV 1054 IV 1054 UP & IVV 1054 UP IVV 1054 MODULUX PURE PURELINE II SINUS VE-L modul technik GmbH · Rudolf-Diesel-Straße 5 · D-56410 Montabaur +49(0)2602-9449-0 +49(0)2602-9449-11 info@modul-technik.de E-MAIL TO MODUL TECHNIK Terms and conditionsDownloadsCareerCertificatesContactPrivacy policyImprint It will load necessary cookies, Google Fonts, Google Maps, OpenStreetMap, Youtube and Google Analytics. More details in our privacy policy and our imprint. Accept I disagree