We are here for you! +49 2602 9449-0 info@modul-technik.de Product Quickfinder ambient simplex Application area: Normal care Product group: Wall supply units (decorative) BEAUTIFUL FRONT WITH LOTS BEHIND! Be it the fine wood décors in the various structures, fronts in a range of bright colours or elegant patterns – ambient simplex is streets ahead in terms of taste. The connections for high and low voltage current, data and communications which we combine according to your requirements, are concealed behind the pull-down screen. Here the various lighting variants also ensure the best ambient light and cosy reading light options. Indirect room lighting can also be integrated on the top side. The outlets for medical gases are accessible from the front in this variant. EXAMPLE DECORS TECHNICAL DATA Electric specifications Nominal voltage: 230 V - 240 V / 50 Hz - continuous operation Protection class: I Protection type: IP 20 Lighting technology Nominal voltage: 230 V - 240 V / 50 Hz Protection class: I Connection type: Plug connection Connection cross-section: 1.5mm² max. Protection type: IP 20 Reading light (LED): Output ≙ 23 W Reading light (T5 fluorescent tubes): Output ≙ 24 W / 2 x 24 W Indirect lighting (LED): Output ≙ 53 W Indirect lighting (T5 fluorescent tubes): Output ≙ 2 x 39 W Night light (LED): Output ≙ 4.3 W Operating pressure of medical gas technology Oxygen: 5 bar Compressed air: 5 bar Vacuum: - 0,8 bar General information Media current feed: Rear, from top or side Décors: HPL-laminate or melamine resin coating Optimum assembly height: 1650mm (rail bottom edge) Optional light control: DALI DIM STANDARD DESIGN High-voltage current technology 6 mains sockets, brand Peha (COMPACTA) 6 potential equalisation sockets (POAG) Communications technology 1 data socket, RJ 45, 2 pcs, brand Peha (COMPACTA) 2 hollow wall sockets, Ø 68mm (e.g. for nurse call button) Lighting technology Reading light (T5 fluorescent tubes): Output ≙ 24 W Indirect lighting (T5 fluorescent tubes): Output ≙ 2 x 39 W Medical gas technology 1 gas tapping point oxygen (O²), brand modul 1 gas tapping point compressed air (Air), brand modul 1 gas tapping point vacuum (Vac), brand modul General information Rear feed Dimensions (W x H x D): 1800 mm x 200 mm x 185 mm Décor: R 4889 Bella Noce schoko (melamine resin-coated) DOWNLOAD DATA SHEET ASSEMBLY, CLEANING, MAINTENANCE AND REPAIR Assembly Cleaning Maintenance and repair Accessories AMBIENT COMPACT AMBIENT CUBE AMBIENT JUNIOR AMBIENT LIGHT COMFORT KUBION MEV 8000 MEV JUNIOR VIP 2000 modul technik GmbH · Rudolf-Diesel-Straße 5 · D-56410 Montabaur +49(0)2602-9449-0 +49(0)2602-9449-11 info@modul-technik.de E-MAIL TO MODUL TECHNIK Terms and conditionsDownloadsCareerCertificatesContactPrivacy policyImprint It will load necessary cookies, Google Fonts, Google Maps, OpenStreetMap, Youtube and Google Analytics. More details in our privacy policy and our imprint. Accept I disagree