800.748.6285 HomeRequest a QuoteSamplesManufacturersProductsNewsFAQAbout UsContact IsaScale view all parts ISAscale http://www.isabellenhuette.de/en/precision-measurement/ Renowned for our expertise in the field of precision measurement in high-current applications, Isabellenhütte has been a leading global provider of advanced measuring systems for many years. Marketed under the ISAscale® brand, our precision measuring systems offer exceptionally high performance. Our commitment to achieving the ultimate in precision is combined with ease of use and a high degree of variability. Please see the ISAscale® product line below: Series: ICD-A IVT-S Sensing Functions: DC Current DC Current, Voltage and Temperature Accuracy: 0.1% 0.1% Current Range Amps: 100 - 500 100 - 2500 Operating Temp: -40C to +105C -40C to +105C Applications: Automotive/Industrial High Voltage Battery Management UPS Systems Monitoring of Li-Ion Batteries Energy Storage Systems Battery Applicatons Distinctives: Sealed six-in MCON Connector 3 Voltage Channels to Monitor Voltage of Battery Stack Compact Design CAN Interface Intermediate Circuit Voltage and Relay Function Internal CAN termination available Software Trigger and Overcurrent Functionality CAN Interface IVT-S Tutorial Video 1,000 VDC Basic Isolation Level IP Class 41 IVT-S Connector Set ICD-A ICD-A-100 ICD-A-300 ICD-A-500 Nominal Measurement Range A ±100 ±300 ±500 Power Loss W <3 <9 <9 Overcurrent Measurement Range A ±500 ±1500 ±2500 Extended Load (max. time) 5 min. A ±120 ±320 ±730 30s A ±200 ±430 ±860 10s A ±300 ±600 ±1000 1s A ±900 ±1600 ±2700 200ms A ±2000 ±3600 ±6000 Initial Accuracy %rdg ±0.1 ±0.1 ±0.1 Offset mA <±10 <±35 <±60 Linearity % of range 0.01 0.01 0.01 Noise mA (RMS) <±8 <±20 <±35 Resolution mA 1 1 1 IVT-S IVT-S - 100 IVT-S - 300 IVT-S - 500 IVT-S - 1000 IVT-S - 2500 DC Current, Voltage and Temperature Measurement Nominal Measurement Range A ±100 ±300 ±500 ±1000 ±2500 Power Loss W <3 <9 <9 <20 <32 Overcurrent Measurement Range A ±800 ±2500 ±6900 ±12200 ±48000 Extended Load (max. time) 5 min. A ±120 ±320 ±730 ±1100 ±2700 30s A ±200 ±430 ±860 ±1400 ±3200 10s A ±300 ±600 ±1000 ±2000 ±4300 1s A ±900 ±1600 ±2700 ±5500 ±11300 200ms A ±2000 ±3600 ±6000 ±12000 ±24000 Initial Accuracy %rdg ±0.1 ±0.1 ±0.1 ±0.1 ±0.1 Offset mA 8 25 75 125 500 Linearity % of range 0.01 0.01 0.01 0.01 0.01 Noise mA (RMS) 5 15 40 70 280 Resolution mA 3 10 27 47 186 IVT-S Voltage Measurement All Series IVT-S - 100 to IVT-S - 2500 Nominal Measurement Range V ±1000 Extended Range (non-linear) V ±1200 Initial Accuracy %rdg 0.1 Offset mV 100 Linearity % of range 0.01 Noise mV (rms) 60 Resolution mV 30 IVT-S Connector Set View all ISAscale parts, Request a quote or Contact us for additional information. About TESTCO | Request a Quote | Contact | Terms & Conditions TESTCO • 1290 Kifer Road • Suite 308 • Sunnyvale • CA • 94086 • USA • 800.748.6285 © Copyright 2008-2021 DistiSuite eCommerce. All Rights Reserved.