Digital 12 Channel 12 Lead Electrocardiograph ECG EKG Machine Interpretation PC Software
Introduction ECG1200G is an electrocardiograph which can collect 12-lead ECG signal simultaneously and print ECG waveform with thermal printing system. It features in, recording and displaying ECG waveform in AUTO/Manual mode, measuring and diagnosing ECG waveform parameters automatically, prompting for “Lead off” and “Lack of paper”, multi-language interface, AC/DC, selecting rhythm lead, printing the trend chart and histogram of P-R interval that convenient to observe abnormal heart rhythm, case database management, button operation and touch screen function, etc. Function 1)Sync collection for 12-lead ECG, a sampling accuracy of 24-bit can get high-quality ECG waveform via power frequency filter, baseline filter and EMG filter of ECG signal. 2)3/6/12-lead ECG waveform and other information such as print mode, sensitivity, paper speed, filter can be displayed in one screen, convenient for contrastive diagnosis. 3)Multiply printing modes and formats, including auto 12x1, auto 6x2+1 (rhythm lead), auto 6x2, auto 3x4+2 (rhythm lead), rhythm 12, rhythm 10, rhythm 8, rhythm 6, manual, etc. Trend chart and histogram of P-R interval can be printed, waveform length printed can be adjusted, and with time print function, which meets different requirements. 4)Paper speed: 5mm/s, 6.25mm/s, 10mm/s, 12.5mm/s, 25mm/s, 50mm/s. 5)With the functions of auto-analysis and auto-diagnosis for routine ECG parameters, provide measurement results and auto-diagnosis conclusion for HR, P-R interval, P Duration, QRS Duration, Q-T interval, Q-Tc, P Axis, QRS Axis, T Axis, R(V5), S(V1), R(V5)+S(V1), etc., which reduces the doctor’s burden. 6)Built-in large capacity memory, convenient for case review and statistic. 7)Multi-language(Chinese, English, Turkish, Portuguese, German, Russian, Kazakh and other languages) interface and report. 8)In optimal DC state, up to 10-hour standby time, continuous print more than 3-hour, which meets the requirements of visiting a patient at home and body examination. Performance Input mode: floating and defibrillation protection Frequency response: Rated input amplitude Input frequency and waveform Relative output response 1.0 0.67Hz~40Hz, Sine wave ±10%a 0.5 40Hz~100Hz, Sine wave +10 %, -30 %a 0.25 100Hz~150Hz, Sine wave +10 %, -30 %a 0.5 150 Hz ~ 500 Hz, Sine wave +10 %, -100 %a 1.5 ≤1Hz,200ms, Triangle wave +0 %, -10 %b a relative to 10Hz b relative to 200 ms Time constant:≥ 3.2s CMRR: >105dB Filter: power frequency(AC50/60 Hz), EMG(25 Hz/35 Hz (-3 dB)), baseline drift filter Recording way: thermal printing system Specification of recording paper: 210 mm(W)×20 m(L) high-speed thermal paper Time base selection(paper speed): 12.5 mm/s, 25 mm/s, 50 mm/s, error: ±5% Gain control(sensitivity): 5, 10, 20 mm/mV, accuracy is ±2%; Standard sensitivity: 10 mm/mV±0.2 mm/mV Recording mode: auto mode, rhythm mode and manual mode Measurement parameters: HR, P-R interval, P Duration, QRS Duration, T Duration, Q-T interval, Q-Tc, P Axis, QRS Axis, T Axis, R(V5), S(V1), R(V5)+S(V1) Safety classification: Class I, type CF defibrillation-proof applied part Polarization resistance voltage: ±610mV Noise level: ≤12 µVp-p ECG signal input sampling frequency: 32 kHz Waveform data processing sampling frequency: 1 kHz Sampling accuracy: 24-bit The minimum detection signal: 10 Hz, 20 µV(peak-peak value) deflected sinusoidal signal can be detected Pacing detection channel: standard II Accuracy of input signal: ±5% Amplitude quantization: ≤5µV/LSB Interchannel time deviation: <100 µs Power supply: AC: 100V~240V(50/60Hz) DC: 14.8V/3500mAh rechargeable lithium battery Degree of protection against ingress of liquid: IPX0 Working mode: continuous working Accessories Standard configuration: A lead cable A limb electrode A chest electrode A thermal recording paper A power cord An earth wire A user manual Physical characteristic Working environment Temperature: 5℃~40℃ Relative humidity: 25%~95%(non-condensing) Atmospheric pressure: 700 hPa~1060 hPa Transport and storage environment Temperature: -20 ℃～+55 ℃ Relative humidity: ≤95% Atmospheric pressure: 500 hPa~1060 hPa Dimension: 340mm(L) × 320mm(W) × 85mm(H) Net weight: 3.2Kg Certification SFDA, CMC, CE, FDA.
Brand: MediaEclat
Category: Business & Industrial > Medical > Medical Equipment > Vital Signs Monitors > ECG Monitors