Search query: hvac gauge manifold digital

Product: BELEY Refrigeration Digital Manifold HVAC System Gauge Set with Temperature Clips and Hoses Specifications:
Pressure test: gauge pressure
Pressure test unit: Kpa; Mpa; bar; inHg; PSI.
Pressure test range: 0 Kpa – 6000 Kpa
Pressure test resolution: 1 Kpa
Pressure test accuracy: +/- 0.5 %(FS)+ 5dgt
Pressure overload limit: 10000 Kpa (10 Mpa; 100 bar;)
Vacuum test: relative vacuum
Vacuum test unit: Kpa; Mpa; bar; inHg; PSI.
Vacuum test range: -101 Kpa – 0 Kpa
Vacuum test resolution: 1 Kpa
Temperature test unit: °C (Celsius), °F (Fahrenheit)
Temperature test range: -40°C–150°C (-40°F–302°F)
Temperature test resolution:
0.1°C (-40°C–99.9°C), 1°C (100°C–150°C)
0.1°F (-40°F–99.9°F), 1°F (100°F–302°F)
Temperature test accuracy: +/- 0.5 °C + 2dgt +/- 0.9 °F + 2dgt
Built-in 89 kinds of refrigerant NIST: According to American NIST standard
Power Supply: 4 X 1.5V (SIZE.AA / LR6) Battery not included
Dimensions: 170*110*50mm weight: 950g

Compatible with Refrigerant System: R11 R13 R113 R114 R115 R116 R12 R123 R124 R125 R1270 R134A R14 R141B R142B R143A R152A R170 R21 R218 R22 R227EA R23 R236EA R245CA R245FA R290 R32 R401A R401B R401C R402A R402B R403A R403B R404A R404B R405A R406A R407A R407B R407C R407D R407E R408A R409A R41 R410A R410B R411A R411B R412A R413A R414A R414B R415A R415B R416A R417A R418A R419A R420A R421A R421B R422A R422B R422C R422D R423A R424A R425A R426A R427A R428A R50 R500 R501 R502 R503 R504 R507A R508A R508B R509A R600 R600A R717 R744(Co2) R1234

Package Includes:
1x HVAC Digital Manifold Gauge
1x User Manual (in English)
2x Temperature Clips
3x Test Pipes
1X Suitcase
(Battery not included)
MUST HAVE -- The intelligent electronic refrigerant group instrument is an auxiliary instrument for the installation, testing and maintenance of refrigeration equipment such as air conditioners, fridge and cold storage. HVAC System Gauge Set has double pressure test, dual temperature test, digital readout, multi-unit switching, multi-mode function and built-in refrigerant database.
SUPER CONVENIENT TOOL -- The instrument can measure double pressure at the same time, as well as dual temperature measurement, with automatic multi-unit pressure conversion, automatic conversion of temperature Celsius / Fahrenheit, to facilitate different needs.
WHAT YOU CAN GET MORE -- Built-in 89 kinds of refrigerant pressure-evaporation temperature database, also calculate the subcooling superheat, to facilitate direct reading of operating process data. Also it tests percentage of vacuum measurement, pressure leak measurement, leak time speed record. It is deserved to have this multi-functional, accurate and simply operated digital manifold let you do the job right.
PREMIUM DESIGN -- Large-size liquid crystal display, LCD backlight support, data display clear and easy to read, convenient light operation. Long-life valve switches, 1/4-inch standard interface design to ensure that the instrument's durability and versatility.
HIGH-PRECISION AND WARRANTY -- The instrument uses high-strength engineering plastic and flexible non-slip silicone design. The whole machine is solid and comfortable to hold. Built-in 32-bit digital processing unit and high-precision data acquisition unit, high data and stability. 3-7days Fast Shipment via DHL. We provide 30day money-back and 24month warranty ensure high-quality after-sales service. BELEY