3 D differential phase contrast microscopy

We demonstrate 3D phase and absorption recovery from partially coherent intensity images captured with a programmable LED array source. Images are captured through-focus with four different illumination patterns. Using first Born and weak object approximations (WOA), a linear 3D differential phase contrast (DPC) model is derived. The partially coherent transfer functions relate the sample’s complex refractive index distribution to intensity measurements at varying defocus. Volumetric reconstruction is achieved by a global FFT-based method, without an intermediate 2D phase retrieval step. Because the illumination is spatially partially coherent, the transverse resolution of the reconstructed field achieves twice the NA of coherent systems and improved axial resolution.  (2014). 11. P. Cloetens, W. Ludwig, J. Baruchel, D. V. Dyck, J. V. Landuyt, J. P. Guigay, and M. Schlenker, “Holotomography: Quantitative phase tomography with micrometer resolution using hard synchrotron radiation X rays,”  (1999). 12. Y. Sung, W. Choi, C. Fang-Yen, K. Badizadegan, R. R. Dasari, and M. S. Feld, “Optical diffraction tomography for high resolution live cell imaging,” Opt. Express 17, 266–277 (2009). 13. Y. Cotte, F. Toy, P. Jourdain, N. Pavillon, D. Boss, P. Magistretti, P. Marquet, and C. Depeursinge, “Marker-free phase nanoscopy,”  (2013). 14. D. Brady, K. Choi, D. Marks, and R. Horisaki, “Compressive holography,” Opt. Express 17, 13040–13049 (2009). 15. Y. Sung and R. R. Dasari, “Deterministic regularization of three-dimensional optical diffraction tomography,”  (2011). 16. A. Bronnikov, “Theory of quantitative phase-contrast computed tomography,”  (2002). 17. L. Tian, J. C. Petruccelli, Q. Miao, H. Kudrolli, V. Nagarkar, and G. Barbastathis, “Compressive X-ray phase tomography based on the transport of intensity equation,” Opt. Lett. 38, 3418–3421 (2013). 18. E. Wolf, “Three-dimensional structure determination of semi-transparent objects from holographic data,” Opt. Commun. 1, 153–156 (1969). 19. G. Gbur, M. A. Anastasio, Y. Huang, and D. Shi, “Spherical-wave intensity diffraction tomography,”  (2005). 20. L. Tian and L. Waller, “3D intensity and phase imaging from light field measurements in an LED array microscope,” Optica 2, 104–111 (2015). 21. T. Kim, R. Zhou, M. Mir, S. D. Babacan, P. S. Carney, L. L. Goddard, and G. Popescu, “White-light diffraction tomography of unlabelled live cells,”  (2014). 22. D. Hamilton and C. Sheppard, “Differential phase contrast in scanning optical microscopy,” J. Microsc. 133, 27–39 (1984). 23. T. N. Ford, K. K. Chu, and J. Mertz, “Phase-gradient microscopy in thick tissue with oblique back-illumination,”  (2012). 24. S. B. Mehta and C. J. Sheppard, “Quantitative phase-gradient imaging at high resolution with asymmetric illumination-based differential phase contrast,” Opt. Lett. 34, 1924–1926 (2009). 25. L. Tian and L. Waller, “Quantitative differential phase contrast imaging in an LED array microscope,” Opt. Express 23, 11394–11403 (2015). 26. G. Zheng, C. Kolner, and C. Yang, “Microscopy refocusing and dark-field imaging by using a simple LED array,” Opt. Lett. 36, 3987–3989 (2011). 27. G. Zheng, R. Horstmeyer, and C. Yang, “Wide-field, high-resolution Fourier ptychographic microscopy,”  (2013). 28. Z. Liu, L. Tian, S. Liu, and L. Waller, “Real-time brightfield, darkfield, and phase contrast imaging in a lightemitting diode array microscope,” J. Biomed. Opt. 19, 106002–106002 (2014). 29. L. Tian, Z. Liu, L.-H. Yeh, M. Chen, J. Zhong, and L. Waller, “Computational illumination for high-speed in vitro Fourier ptychographic microscopy,” Optica 2, 904–911 (2015). 30. L. Tian, X. Li, K. Ramchandran, and L. Waller, “Multiplexed coded illumination for Fourier ptychography with an LED array microscope,” Biomed. Opt. Express 5, 2376–2389 (2014). 31. L. Tian, J. Wang, and L. Waller, “3D differential phase-contrast microscopy with computational illumination using an LED array,” Opt. Lett. 39, 1326–1329 (2014). 32. X. Ou, G. Zheng, and C. Yang, “Embedded pupil function recovery for Fourier ptychographic microscopy,” Opt. Express 22, 4960–4972 (2014). 33. M. Born and E. Wolf, Principles of Optics: Electromagnetic Theory of Propagation, Interference and Diffraction of Light ( 1999), 7th ed. 34. N. Streibl, “Three-dimensional imaging by a microscope,”  (1985). 35. Y. Sung and C. J. R. Sheppard, “Three-dimensional imaging by partially coherent light under non-paraxial condition,”  (2011). 36. T. H. Nguyen, C. Edwards, L. L. Goddard, and G. Popescu, “Quantitative phase imaging of weakly scattering objects using partially coherent illumination,” Opt. Express 24, 11683–11693 (2016). 37. R. A. Claus, P. P. Naulleau, A. R. Neureuther, and L. Waller, “Quantitative phase retrieval with arbitrary pupil and illumination,” Opt. Express 23, 26672–26682 (2015). 38. M. H. Jenkins and T. K. Gaylord, “Three-dimensional quantitative phase imaging via tomographic deconvolution phase microscopy,”  (2015). 39. J. Guigay, “Fourier transform analysis of Fresnel diffraction patterns and in-line holograms,” Optik 49, 121–125 (1977). 40. Y. I. Nesterests and T. E. Gureyev, “Partially coherent contrast-transfer-function approximation,”  (2016). 41. U. S. Kamilov, I. N. Papadopoulos, M. H. Shoreh, A. Goy, C. Vonesch, M. Unser, and D. Psaltis, “Learning approach to optical tomography,” Optica 2, 517–522 (2015). 42. T. L. Jensen, J. H. Joergensen, P. C. Hansen, and S. H. Jensen, “Implementation of an Optimal First-Order Method for Strongly Convex Total Variation Regularization,” BIT  (2012).  //www.imm.dtu.dk/ ̃pcha/TVReg/ 43. W. Choi, C. Fang-Yen, K. Badizadegan, S. Oh, N. Lue, R. R. Dasari, and M. S. Feld, “Tomographic phase microscopy,” Nat. Methods 4, 717–719 (2007). 44. S. Jones, M. King, and A. Ward, “Determining the unique refractive index properties of solid polystyrene aerosol using broadband Mie scattering from optically trapped beads,”  20735–20741 (2013). 45. Z. Jingshan, R. A. Claus, J. Dauwels, L. Tian, and L. Waller, “Transport of intensity phase imaging by intensity spectrum fitting of exponentially spaced defocus planes,” Opt. Express 22, 10661–10674 (2014).