natgillin/translations-raw huggingface dataset:
the identification number of the notified body, where that body is involved in the production control phase, has been affixed in violation of Article 17 or has not been affixed;
identifikacijska številka priglašenega organa, ki je bil udeležen v fazi nadzora proizvodnje, je bila nameščena tako, da je kršen člen 17, ali ni bila nameščena;
the information referred to in Article 6(6) or 8(3) is absent, false or incomplete;
informacije iz člena 6(6) ali člena 8(3) manjkajo, so napačne ali nepopolne;
any other administrative requirement provided for in Article 6 or 8 is not fulfilled.
ni izpolnjena katera koli druga upravna zahteva iz člena 6 ali člena 8.
Where the non-compliance referred to in paragraph 1 persists, the Member State concerned shall take all appropriate measures to restrict or prohibit the instrument being made available on the market or ensure that it is recalled or withdrawn from the market.
Kadar se neskladnost iz odstavka 1 nadaljuje, zadevna država članica izvede vse ustrezne ukrepe za omejitev ali prepoved dostopnosti tehtnice na trgu ali pa zagotovi njen odpoklic ali umik s trga.
The Commission shall be assisted by the Committee on non-automatic weighing instruments.
Komisiji pomaga odbor za neavtomatske tehtnice.
Such rules may include criminal penalties for serious infringements
Taka pravila lahko v primeru resnih kršitev vključujejo kazenskopravne sankcije.
Member States shall not impede the making available on the market and/or the putting into service of instruments covered by Directive 2009/23/EC which are in conformity with that Directive and which were placed on the market before 20 April 2016.
Države članice ne ovirajo omogočanja dostopnosti na trgu in/ali dajanja v uporabo tehtnic, zajetih v Direktivi 2009/23/ES, ki so skladne z navedeno direktivo in so bile dane na trg pred 20. aprilom 2016.
Certificates issued under Directive 2009/23/EC shall be valid under this Directive.
Potrdila, izdana v skladu z Direktivo 2009/23/ES, so veljavna v skladu s to direktivo.
Member States shall adopt and publish, by 19 April 2016 the laws, regulations and administrative provisions necessary to comply with points (3) to (19) of Article 2, Articles 6 to 17, Articles 19 to 43 and Annex II, III and IV.
Države članice sprejmejo in objavijo zakone in druge predpise, potrebne za uskladitev s točkami (3) do (19) člena 2, členi 6 do 17, členi 19 do 43 ter prilogami II, III in IV do 19. aprila 2016.
Directive 2009/23/EC, as amended by the Regulation listed in Annex V, Part A, is repealed with effect from 20 April 2016, without prejudice to the obligations of the Member States relating to the time limits for transposition into national law and the dates of application of the Directives set out in Annex V, Part B.
Direktiva 2009/23/ES, kakor je bila spremenjena z uredbo iz dela A Priloge V, se razveljavi z učinkom od 20. aprila 2016, brez poseganja v obveznosti držav članic glede rokov za prenos v nacionalno pravo in datumov začetka uporabe direktive iz dela B Priloge V.
Article 1, points (1) and (2) of Article 2, Articles 3, 4, 5 and 18 and Annexes I, V and VI shall apply from 20 April 2016.
Člen 1, točki (1) in (2) člena 2, členi 3, 4, 5 in 18 ter priloge I, V in VI se uporabljajo od 20. aprila 2016.
OJ L 122, 16.5.2009, p.
UL L 122, 16.5.2009, str.
Directive 2009/23/EC is the codification of Council Directive 90/384/EEC of 20 June 1990 on the harmonization of the laws of the Member States relating to non-automatic weighing instruments (OJ L 189, 20.7.1990, p.
Direktiva 2009/23/ES je kodifikacija Direktive Sveta 90/384/EGS z dne 20. junija 1990 o usklajevanju zakonodaje držav članic v zvezi z neavtomatskimi tehtnicami (UL L 189, 20.7.1990, str.
The terminology used is that of the International Organisation of Legal Metrology.
Uporabljena je terminologija Mednarodne organizacije za zakonsko meroslovje.
Where an instrument includes, or is connected to, more than one indicating or printing device used for the applications listed in points (a) to (f) of Article 1(2), those devices which repeat the results of the weighing operation and which cannot influence the correct functioning of the instrument shall not be subject to the essential requirements if the weighing results are printed or recorded correctly and indelibly by a part of the instrument which meets the essential requirements and the results are accessible to both parties concerned by the measurement.
Kadar tehtnica vključuje ali je priključena na več kot eno kazalno ali izpisovalno napravo, ki se uporablja za namene iz točk (a) do (f) člena 1(2), tiste naprave, ki ponavljajo rezultate tehtanja in ki ne morejo vplivati na pravilno delovanje tehtnice, niso predmet bistvenih zahtev, če so rezultati tehtanja izpisani ali zabeleženi pravilno in neizbrisno z delom tehtnice, ki izpolnjuje bistvene zahteve in so rezultati dosegljivi obema partnerjema v merjenju.
Metrological requirements
Meroslovne zahteve
The units of mass used shall be the legal units within the meaning of Council Directive 80/181/EEC of 20 December 1979 on the approximation of the laws of the Member States relating to units of measurement [1].Subject to compliance with this condition, the following units are permitted:
Uporabljene enote mase so zakonite enote v smislu Direktive Sveta 80/181/EGS z dne 20. decembra 1979 o približevanju zakonodaj držav članic, ki se nanašajo na merske enote [1].Če ustrezajo temu pogoju, so dovoljene naslednje enote:
SI units: kilogram, microgram, milligram, gram, tonne;
enote SI: kilogram, mikrogram, miligram, gram, tona;
imperial unit: troy ounce, if weighing precious metals;
enota imperialnega sistema: troy unča za tehtanje žlahtnih kovin;
For instruments that make use of the imperial unit of mass referred to above, the relevant essential requirements specified below shall be converted to that unit, using simple interpolation.
Za tehtnice, ki uporabljajo zgoraj omenjene enote mase imperialnega sistema, se ustrezne bistvene zahteve, ki so določene spodaj, pretvorijo v te enote s pomočjo linearne interpolacije.
I special
I – posebni
II high
II – visoki
III medium
III – srednji
IIII ordinary
IIII – navadni
Minimum capacity (Min)
Najmanjša zmogljivost (Min)
minimum value
Najmanjša vrednost
maximum value
Največja vrednost
0,001 g ≤ e ≤ 0,05 g
Pri tehtnicah razredov II in III se za ugotavljanje prenosne tarife najmanjša zmogljivost zmanjša na 5 e.
The minimum capacity is reduced to 5 e for instruments in classes II and III for determining a conveying tariff.
Vrednosti razdelka
Scale intervals
Dejanski razdelek (d) in preskusni razdelek (e) sta v obliki:
The actual scale interval (d) and the verification scale interval (e) shall be in the form:
1 × 10k, 2 × 10k ali 5 × 10k enot mase,
1 × 10k, 2 × 10k, or 5 × 10k mass units,
kjer je k celo število ali nič.
k being any integer or zero.
Za vse tehtnice, razen za tiste, ki imajo pomožne naprave za prikazovanje, je:
For all instruments other than those with auxiliary indicating devices:
Za vse tehtnice, ki imajo pomožne naprave za prikazovanje, je:
For instruments with auxiliary indicating devices the following conditions apply:
d < e ≤ 10 d.
d < e ≤ 10 d.
Navedeni pogoji se ne uporabljajo za tehtnice razreda I z d < 10–4 g, za katere je e = 10–3 g.
Those conditions do not apply for instruments of class I with d < 10–4 g, for which e = 10–3 g.
Razvrstitev
Classification
Tehtnice z enim tehtalnim območjem
Instruments with one weighing range
Tehtnice, opremljene s pomožno kazalno napravo, spadajo v razred I ali II.
Instruments equipped with an auxiliary indicating device shall belong to class I or class II.
Za te tehtnice spodnji meji najmanjše zmogljivosti za ta dva razreda dobimo iz tabele 1 tako, da v stolpcu 3 zamenjamo preskusni razdelek (e) z dejanskim razdelkom (d).
For these instruments the minimum capacity lower limits for these two classes are obtained from Table 1 by replacement in column 3 of the verification scale interval (e) by the actual scale interval (d).
Če je d < 10–4 g, je lahko največja zmogljivost razreda I manj kot 50000 e.
If d < 10–4 g, the maximum capacity of class I may be less than 50000 e.
Tehtnice z več tehtalnimi območji
Instruments with multiple weighing ranges
Več tehtalnih območij je dovoljenih pod pogojem, da so na tehtnici jasno nakazani.
Multiple weighing ranges are permitted, provided they are clearly indicated on the instrument.
Vsako posamezno tehtalno območje je razvrščeno v skladu s točko 3.1.
Each individual weighing range is classified according to point 3.1.
Če tehtalna območja spadajo v različne razrede točnosti, tehtnica ustreza najstrožji od zahtev, ki veljajo za razrede točnosti, v katere spadajo tehtalna območja.
If the weighing ranges fall into different accuracy classes the instrument shall comply with the severest of the requirements that apply for the accuracy classes in which the weighing ranges fall.
Tehtnice z več vrednostmi razdelka
Multi-interval instruments
Tehtnice z enim tehtalnim območjem imajo lahko več delnih tehtalnih območij (tehtnice z več vrednostmi razdelkov).
Instruments with one weighing range may have several partial weighing ranges (multi-interval instruments).
Tehtnice z več vrednostmi razdelkov niso opremljene s pomožno kazalno napravo.
Multi-interval instruments shall not be equipped with an auxiliary indicating device.
Vsako delno tehtalno območje i tehtnic z več vrednostmi razdelkov je določeno:
Each partial weighing range i of multi-interval instruments is defined by:
s svojo vrednostjo preskusnega razdelka ei,
its verification scale interval ei
s tem da je e(i + 1) > ei
with e(i + 1) > ei
s svojo največjo zmogljivostjo Maxi,
its maximum capacity Maxi
s tem da je Maxr = Max
with Maxr = Max
s svojo najmanjšo zmogljivostjo Mini,
its minimum capacity Mini
s tem da je Mini = Max (i – 1)
with Mini = Max (i – 1)
in Min1 = Min
and Min1 = Min
kjer je:
where:
i številka delnega tehtalnega območja,
i partial weighing range number,
r skupno število delnih tehtalnih območij.
r the total number of partial weighing ranges.
Vse zmogljivosti so zmogljivosti neto bremena, ne glede na uporabljeno vrednost tare.
All capacities are capacities of net load, irrespective of the value of any tare used.
Delna tehtalna območja so razvrščena v skladu s tabelo 2.
The partial weighing ranges are classified according to Table 2.
Vsa delna tehtalna območja spadajo v isti razred točnosti, in to v razred točnosti tehtnice.
All partial weighing ranges shall fall into the same accuracy class, that class being the instrument’s accuracy class.
Tabela 2
Table 2
i številka delnega tehtalnega območja
i partial weighing range number
r skupno število delnih tehtalnih območij
r total number of partial weighing ranges
Najmanjša vrednost
Minimum value
Najmanjša vrednost [2]n
Minimum value [2]n
Največja vrednost
Maximum value
5 g ≤ ei
≤ ei
Točnost
Accuracy
Pri izvajanju postopkov iz člena 13 odklon v prikazu ne presega največjega dopustnega pogreška, ki je naveden v tabeli 3.
On implementation of the procedures laid down in Article 13, the error of indication shall not exceed the maximum permissible error of indication as shown in Table 3.
V primeru digitalnega prikaza se odklon v prikazu popravi za odklon, ki se zaokroži navzgor ali navzdol.
In the case of digital indication the error of indication shall be corrected for the rounding error.
Največji dopustni pogreški veljajo za neto in tara vrednost za vsa možna bremena, razen za prednastavljene vrednosti tare.
The maximum permissible errors apply to the net value and tare value for all possible loads, excluding preset tare values.
Tabela 3
Table 3
Največji dopustni pogrešek
Maximum permissible errors
Breme
Load
Največji dopustni pogrešek
Maximum permissible error
Razred I
Class I
Razred II
Class II
Razred III
Class III
Razred IIII
Class IIII
Največji dopustni pogreški v uporabi so dvakratni največji dopustni pogreški, določeni v točki 4.1.
The maximum permissible errors in service are twice the maximum permissible errors fixed in Section 4.1.
Rezultati tehtanja so ponovljivi in obnovljivi z drugimi prikazovalnimi napravami ter z drugimi načini uravnoteženja, ki jih je možno uporabiti.
Weighing results of an instrument shall be repeatable, and shall be reproducible by the other indicating devices used and in accordance with other methods of balancing used.
Rezultati tehtanja so dovolj neobčutljivi na spremembe položaja bremena na sprejemniku bremena.
The weighing results shall be sufficiently insensitive to changes in the position of the load on the load receptor.
Tehtnice se odzovejo na majhne spremembe v bremenu.
The instrument shall react to small variations in the load.
Vplivne veličine in čas
Influence quantities and time
Tehtnice razredov II, III in IIII, ki se uporabljajo v nagnjenem položaju, so dovolj neobčutljive na stopnjo nagnjenosti, do katere lahko pride pri običajni uporabi.
Instruments of classes II, III and IIII, liable to be used in a tilted position, shall be sufficiently insensitive to the degree of tilting that can occur in normal use.
Tehtnice izpolnjujejo meroslovne zahteve znotraj temperaturnega območja, ki ga določi proizvajalec.
The instruments shall meet the metrological requirements within the temperature range specified by the manufacturer.
Velikost tega območja je najmanj enaka:
The value of this range shall be at least equal to:
5 °C za tehtnice v razredu I;
5 °C for an instrument in class I;
15 °C za tehtnice v razredu II;
15 °C for an instrument in class II;
30 °C za tehtnice v razredu III ali IIII.
30 °C for an instrument in class III or IIII.
Če proizvajalec temperaturnega območja ni določil, velja temperaturno območje od – 10 °C do + 40 °C.
In the absence of a manufacturer’s specification, the temperature range of – 10 °C to + 40 °C applies.
Tehtnice, ki se napajajo iz električnega omrežja, izpolnjujejo meroslovne zahteve pod pogoji napajanja, ki so v mejah običajnih nihanj.
Instruments operated from a mains power supply shall meet the metrological requirements under conditions of power supply within the limits of normal fluctuation.
Tehtnice, ki se napajajo iz baterije ali akumulatorja, pokažejo, če pade napetost pod najmanjšo zahtevano vrednost in v takih okoliščinah še naprej pravilno delujejo ali pa se samodejno izklopijo.
Instruments operated from battery power shall indicate whenever the voltage drops below the minimum required value and shall under those circumstances either continue to function correctly or be automatically put out of service.
Elektronske tehtnice, razen iz razredov I in II, kadar je e manjši od 1 g, izpolnjujejo meroslovne zahteve pri pogojih visoke relativne vlažnosti na zgornji meji njihovega temperaturnega območja.
Electronic instruments, except those in class I and in class II if e is less than 1 g, shall meet the metrological requirements under conditions of high relative humidity at the upper limit of their temperature range.
Podaljšano trajanje obremenitve za tehtnice razredov točnosti II, III in IIII ima zanemarljiv vpliv na prikaz med obremenitvijo oziroma na ničlo takoj po razbremenitvi.
Loading an instrument in class II, III or IIII for a prolonged period of time shall have a negligible influence on the indication at load or on the zero indication immediately after removal of the load.
Pod drugimi vplivnimi pogoji tehtnica ali pravilno deluje ali pa se samodejno izključi.
Under other conditions the instruments shall either continue to function correctly or be automatically put out of service.
Zasnova in konstrukcija
Design and construction
Splošne zahteve
General requirements
Zasnova in konstrukcija tehtnic sta taki, da bodo tehtnice ob pravilni uporabi in pravilni namestitvi ter v okolju, za katerega so namenjene, ohranile svoje meroslovne lastnosti.
Design and construction of the instruments shall be such that the instruments will preserve their metrological qualities when properly used and installed and when used in an environment for which they are intended.
Prikazana mora biti vrednost mase.
The value of the mass must be indicated.
Če so elektronske tehtnice izpostavljene motnjam, ne prikažejo posledic pomembnejših napak, ali pa jih samodejno odkrijejo in prikažejo.
When exposed to disturbances, electronic instruments shall not display the effects of significant faults, or shall automatically detect and indicate them.
Pri samodejnem odkritju pomembnejše napake oddajo elektronske tehtnice svetlobni ali zvočni alarm, ki traja, dokler uporabnik ne izvede korektivnega ukrepa ali dokler napaka ne izgine.
Upon automatic detection of a significant fault, electronic instruments shall provide a visual or audible alarm that shall continue until the user takes corrective action or the fault disappears.
Zahteve iz točk 8.1 in 8.2 tehtnice izpolnjujejo v celotnem časovnem obdobju, predvidenem za uporabo takih tehtnic.
The requirements of points 8.1 and 8.2 shall be met on a lasting basis during a period of time that is normal in view of the intended use of such instruments.
Digitalne elektronske naprave vedno izvajajo ustrezen nadzor pravilnega delovanja merilnega procesa, naprave za prikazovanje ter shranjevanja in vseh prenosov podatkov.
Digital electronic devices shall always exercise adequate control of the correct operation of the measuring process, of the indicating device, and of all data storage and data transfer.
Ob samodejnem odkritju pomembnejše trajne napake oddajo elektronske tehtnice svetlobni ali zvočni signal, ki traja, dokler uporabnik ne izvede korektivnega ukrepa ali dokler napaka ne izgine.
Upon automatic detection of a significant durability error, electronic instruments shall provide a visual or audible alarm that shall continue until the user takes corrective action or the error disappears.
Če je na elektronsko tehtnico preko ustreznega vmesnika priključena zunanja oprema, ta ne vpliva negativno na meroslovne lastnosti tehtnice.
When external equipment is connected to an electronic instrument through an appropriate interface the metrological qualities of the instrument shall not be adversely influenced.
Tehtnice nimajo lastnosti, ki bi lahko omogočale nepošteno uporabo, možnosti nenamerne zlorabe pa so čim manjše.
The instruments shall have no characteristics likely to facilitate fraudulent use, whereas possibilities for unintentional misuse shall be minimal.
Elementi, ki jih uporabnik ne more razstaviti ali naravnati, so pred takimi dejanji zaščiteni.
Components that may not be dismantled or adjusted by the user shall be secured against such actions.
Tehtnice so zasnovane tako, da omogočajo enostavno izvajanje zakonskega nadzora, določenega s to direktivo.
Instruments shall be designed to permit ready execution of the statutory controls laid down by this Directive.
Prikaz rezultatov tehtanja in drugih vrednosti mase
Indication of weighing results and other weight values
Prikaz rezultatov tehtanja in drugih vrednosti mase je točen, nedvoumen in nezavajajoč, naprava za prikazovanje pa pod običajnimi pogoji uporabe omogoča enostavno branje prikaza.
The indication of the weighing results and other weight values shall be accurate, unambiguous and non-misleading and the indicating device shall permit easy reading of the indication under normal conditions of use.
Imena in simboli za enote iz odstavka 1 te priloge ustrezajo določbam Direktive 80/181/EGS z dodatkom simbola za metrični karat, ki je „ct“.
The names and symbols of the units referred to in point 1 of this Annex shall comply with the provisions of Directive 80/181/EEC with the addition of the symbol for the metric carat which shall be the symbol ‘ct’.
Prikaz nad največjo zmogljivostjo (Max), povečano za 9 e, ni mogoč.
Indication shall be impossible above the maximum capacity (Max), increased by 9 e.
Pomožna naprava za prikazovanje je dovoljena samo desno od decimalne oznake.
An auxiliary indicating device is permitted only to the right of the decimal mark.
Razširjena naprava za prikazovanje se sme uporabljati samo začasno, med njenim delovanjem pa je izpisovanje onemogočeno.
An extended indicating device may be used only temporarily, and printing shall be inhibited during its functioning.
Prikažejo se lahko tudi sekundarni prikazi, vendar pod pogojem, da jih ni mogoče zamenjati za primarne prikaze.
Secondary indications may be shown, provided that they cannot be mistaken for primary indications.
Izpisovanje rezultatov tehtanja in drugih vrednosti mase
Printing of weighing results and other weight values
Izpisani rezultati so pravilni, primerno prepoznavni in nedvoumni.
Printed results shall be correct, suitably identified and unambiguous.
Izpis je jasen, čitljiv, neizbrisen in trajen.
The printing shall be clear, legible, non-erasable and durable.
Niveliranje
Levelling
Če je potrebno, so tehtnice opremljene z nivelirno napravo in kazalnikom niveliranja, ki sta dovolj občutljiva, da omogočata pravo namestitev.
When appropriate, instruments shall be fitted with a levelling device and a level indicator, sufficiently sensitive to allow proper installation.
Ničliranje
Zeroing
Tehtnice so lahko opremljene z napravami za ničliranje.
Instruments may be equipped with zeroing devices.
Te naprave omogočajo točno ničliranje in ne povzročajo nepravilnih merilnih rezultatov.
The operation of these devices shall result in accurate zeroing and shall not cause incorrect measuring results.
Naprave za tariranje ali naprave za prednastavljivo tariranje
Tare devices and preset tare devices
Tehtnice imajo lahko eno ali več naprav za tariranje in eno napravo za prednastavljivo tariranje.
The instruments may have one or more tare devices and a preset tare device.
Naprava za tariranje omogoča točno ničliranje in zagotavlja pravilno neto tehtanje.
The operation of the tare devices shall result in accurate zeroing and shall ensure correct net weighing.
Naprava za prednastavljivo tariranje zagotavlja pravilno ugotavljanje izračunane neto vrednosti.
The operation of the preset tare device shall ensure correct determination of the calculated net value.
Tehtnice za neposredno prodajo javnosti z največjo zmogljivostjo do 100 kg: dodatne zahteve
Instruments for direct sales to the public, with a maximum capacity not greater than 100 kg: additional requirements
Tehtnice za neposredno prodajo javnosti prikazujejo vse bistvene informacije o tehtanju, v primeru tehtnic, ki kažejo znesek, pa kupcu jasno pokažejo izračunani znesek za proizvod, ki ga kupuje.
Instruments for direct sale to the public shall show all essential information about the weighing operation and, in the case of price-indicating instruments, shall clearly show the customer the price calculation of the product to be purchased.
Če je prikazan znesek za plačilo, je ta točen.
The price to pay, if indicated, shall be accurate.
Tehtnice, ki izračunavajo znesek, prikazujejo bistvene prikaze dovolj dolgo, da jih lahko kupec pravilno prebere.
Price-computing instruments shall display the essential indications long enough for the customer to read them properly.
Tehtnice, ki izračunavajo znesek, lahko poleg tehtanja posameznih artiklov in računanja cene izvajajo še druge naloge le, če so vsi prikazi, povezani z vsemi transakcijami, jasno in nedvoumno izpisani in primerno razporejeni na listku ali nalepki, namenjenima kupcu.
Price-computing instruments may perform functions other than per-article weighing and price computation only if all indications related to all transactions are printed clearly and unambiguously and are conveniently arranged on a ticket or label for the customer.
Tehtnice nimajo nobenih lastnosti, ki bi lahko neposredno ali posredno povzročile prikaze, ki ne bi bili preprosto in nedvoumno razumljivi.
Instruments shall bear no characteristics that can cause, directly or indirectly, indications the interpretation of which is not easy or straightforward.
Tehtnice varujejo kupce pred nepravilnimi prodajami, ki bi bile posledica njihovega nepravilnega delovanja.
Instruments shall safeguard customers against incorrect sales transactions due to their malfunctioning.
Pomožne naprave za prikazovanje in razširjene naprave za prikazovanje niso dovoljene.
Auxiliary indicating devices and extended indicating devices are not permitted.
Dodatne naprave so dovoljene le, če ne morejo povzročiti zlorabe.