Lehmann & Michels Gas Engine Indicator
notes: label: Description
content: Lehmann & Michels manufactured this gas engine indicator, serial number D. R. P. 416623. It is based on patent number 1532692, which was granted to Josef Geiger of Augsburg, Germany, on April 7, 1925.
label: Description
content: The L & M catalog describes this instrument as a Pi-Meter and instrument to measure the mean pressure. There is no record, pointer, or dial. Most likely, it contains a heavy fly wheel that is damped, an impulse of gas pressure each cycle does work on fly wheel, and the work determines the velocity which is transcribed into a mean pressure for each stroke of the piston.
label: Description
content: An engine indicator is an instrument for graphically recording the pressure versus piston displacement through an engine stroke cycle. Engineers use the resulting diagram to check the design and performance of the engine.
label: Description
content: A mechanical indicator consists of a piston, spring, stylus, and recording system. The gas pressure of the cylinder deflects the piston and pushes against the spring, creating a linear relationship between the gas pressure and the deflection of the piston against the spring. The deflection is recorded by the stylus on a rotating drum that is connected to the piston. Most indicators incorporate a mechanical linkage to amplify the movement of the piston to increase the scale of the record.
label: Description
content: When the ratio of the frequency of the pressure variation to the natural frequency of the system is small, then the dynamic deflection is equal to the static deflection. To design a system with a high natural frequency, the mass of the piston, spring, stylus, and mechanical linkage must be small, but the stiffness of the spring must be high. The indicator is subjected to high temperatures and pressures and rapid oscillations, imposing a limitation on the reduction in mass. Too stiff a spring will result in a small displacement of the indicator piston and a record too small to measure with accuracy. Multiplication of the displacement will introduce mechanical ad dynamic errors.
label: Description
content: The parameters of the problem for designing an accurate and trouble free recorder are such that there is no easy or simple solution. Studying the variety of indicators in the collection shows how different inventors made different compromises in their designs.
label: Location
content: Currently not on view
setName: label: See more items in
content: Work and Industry: Mechanical and Civil Engineering
label: See more items in
content: Engineering, Building, and Architecture
label: See more items in
content: Work
label: See more items in
content: Industry & Manufacturing
label: See more items in
content: Indicators
creditLine: label: Credit Line
content: Gift of the Estate of Kalman J. Dejuhasz, State College, Pennsylvania
dataSource: label: Data Source
content: National Museum of American History
identifier: label: ID Number
content: 1981.0217.19
label: accession number
content: 1981.0217
label: catalog number
content: 1981.0217.19
objectType: label: Object Name
content: indicator, gas engine
physicalDescription: label: Physical Description
content: metal (overall material)
label: Physical Description
content: wood (case material)
label: Measurements
content: overall - case: 3 1/2 in x 8 5/8 in x 9 3/4 in; 8.89 cm x 21.9075 cm x 24.765 cm