Yes, the Model 2700 can support dozens of channels of digital input. While the 2700 mainframe has two digital inputs for triggering a measurement or scanning sequence, additional digital input lines can be created by using the Analog Input channels. Simply set the HI / LO limits on each channel to TTL levels, and configure the 2700 to look for the Pass / Fail flags while scanning.