When measuring inverters always use high-speed amplifiers as the voltage is modulated with switching frequencies up to several hundred kHz, also ensure to use current sensors with high bandwidth capabilities as well as make sure they are capable to measure both DC and AC currents. Do not forget to set the current as of the frequency source and use either a star or delta connection. Never use an Aron or a V-connection for such measurements.