Turn on the Model 2461. Let it warm up for one hour. 2. Press QUICKSET and set Function to DIGITIZE. Source Voltage and Measure Current. 3. Press HOME. Select Voltage range (top display) to 100V. 4. Press MENU. 5. In the SENSE column, select Settings. 6. Select SENSE to 4 WIRE. 7. Press MENU. 8. In the SOURCE column , select Pulse. 9. Set Level to 10A. 10. Set Width to 1msec. 11. Set Off Time to 20msec. 12. Set Bias Limit to 5V. 13. Set Limit to 100V. 13. Set Count to 5. For five pulses. 15. Press Generate. 16. Press TRIGGER button. This will output the five 10A pulses. It should look like the attached picture.