M100 Extended Frequency Transducer Selection Guide M100-FE1 Frequency range 0-10kHz Typical Applications The M100-FE1 measure the frequency of the input signal and provides a DC output that is directly proportional to the input frequency. The FE1 can accept frequency or pulse inputs over a wide range. The frequency range is 0-10kHz, with the minimum span being 100Hz and the maximum 10kHz. The M100-FE1 can be used in a wide variety of applications such as speed measurement using a proximity sensor, flow measurement etc. Isolation of 1.5kV is provided between the input and the output signal allowing the output to be fed to conventional analogue meters, digital meters, PLC and computer systems. Ordering Information Product code I/P Hz I/P Un Output Aux Freq. Options M100-FE1 600Hz 10V 4-20mA 230V 50Hz Options 1. Non standard inputs / outputs only as far as technically acceptable. 2. A.C. Auxiliary in range 57.7 to 450 volts 3. Calibration at temperature other than 23°C Technical Specification Input Input range Hz 0-100Hz minimum span Input range Hz 0-10kHz maximum span Working range ± 125% Hz Voltage range 0 - 600V Overload continuous 1.5 x Un Output Rated value mA 0-1/5/10/20 & 4-20mA Load resistance 12/2.4/1.2/0.6 kOhm Rated value volts 0-5 / 10 & 1-5 V Adjustments Zero ± 2% Span ± 10% Auxiliary A.C. Voltage 115 / 230 / 400 V (± 25% / 45-65 Hz / < 2 VA) D.C. Voltage 24 / 48 / 110 V ( ±20% / galvanically isolated / <3W) Weight and Case Approx. 0.4kg. 55mm case