The RPM eliminates the need to re-cable and increases switching speed between Pulsed IV and CV measurements.  However, we do not specify the switching time for the RPMs in the data sheets. A simple test using a MDO3102 yeided the below approximate switch times: CVU to PMU measurements through RPM = 460ms. PMU to CVU measurements through RPM = 1700ms.