This unsafe is caused by Step Control Error Check setting on the Schedule global page. It will monitor the actual current/Voltage/Power output value and compare it with the schedule setting value, unsafe will happen when the discrepancy between the setting value and the actual value is larger than the setting limit. They are designed to detect open circuits or short circuits caused by loose IV cable connections, it also detects abnormal output when the system has problem.