To reduce the drift value, make sure all the openings on the titrator have been greased (e.g. tube inserts, electrodes, adapters, etc.) and that you use dry and fresh molecular sieves and silica gel in your drying tubes. We recommend exchanging your molecular sieves every 6 weeks. If you still have high values, try to gently shake the titration vessel to collect traces of water from the glass walls.