The flute-shaped uniform velocity tube flowmeter is based on the pitot tube velocity measurement principle. It outputs a differential pressure signal. Used in conjunction with a differential pressure transmitter for measuring differential pressure. It can measure the flow of steam, liquid and gas. The structure is simple, the pressure loss is small, and the length of the upstream and downstream straight pipe sections is short. Easy to install, the range ratio can reach 10:1. It is resistant to high temperature and pressure, is not affected by abrasion, and has no leakage. The flute-shaped average velocity tube flowmeter also has its shortcomings. If the on-site installation conditions are demanding, the differential pressure generated is small, and some are only 20-30Pa. Must be equipped with high-precision differential pressure transmitter. Due to the structure, the flow rates of the pressure ports are different, and there is a certain pressure difference between the pressure ports. There is a flow of medium between the pressure holes. If there is a medium, it may be blocked and cause measurement errors. The fluid to be measured flows in the circular tube, and the pressure distribution caused by the circular tube is different when the fluid flows in due to the different separation points of the fluid. The flow coefficient is unstable and the flow is unstable. Extended reading: Use of high temperature melt pressure sensor