Yes. You can put your WVR or WFM instrument into an SDI-to-composite emulation mode which presents the SDI waveform and vector as if they were composite signals and uses IRE units instead of mV. To enable the SDI->Composite mode on a WFM700 or WFM7100 (with the touch screen), bring up either a waveform or vectorscope screen and touch the Colorspace button at the bottom of the screen, then select "SDI->Composite". To enable the SDI->Composite mode in the WFM7120 or WVR7100/7120 series, select a waveform or vectorscope tile, then press and hold the button corresponding to that tile to bring up the context menu. In this menu, scroll down the Display Mode menu option and choose SDI->Composite.