The oxygen analyzer is maintenance-free but should be calibrated for systems for the first time after 15 months and then every 12 months.
The moisture sensor should be cleaned according to the system message (every 2000 hours) and re-passivated with 85% phosphoric acid. Then reset the hour counter to 'zero' (see description in the user manual). The water analyzer should also be calibrated regularly in production systems.