During SRM, a valid calibration interval is defined, which covers a well-defined range (from the zero point to slightly above the highest results from the parallel measurement). AMS signals that are outside the valid calibration interval are valid and must be reported, but a new QAL2 is required if the AMS is outside the valid calibration interval for a longer period of time.