Yes, but there are factors end users should consider because the reliable operation of the sensor depends upon a powerful light source and sensitive optics. Reflections from walls and bends in pipes can cause problems in accuracy, so the turbidity of the process stream, the diameter of the pipe and the material used will need to be taken into account. Larger diameter pipes and those made of less reflective materials are better; additionally, the higher the turbidity the more likely pipe insertion will be effective. At >200NTU, putting a TurbSense into a pipe shouldn’t be a problem.