The extension cables used for the connection must be shielded. We recommend using a 6-wire cable with reference/sense management, which can compensate for the voltage drop due to the distance between the devices. If possible, it should be inserted into the cable duct by itself and laid as far away as possible from the power cables.
When using 4-conductor connection cables, use a minimum cross-section of 1 square millimetre and preferably not more than 300 metres in length.
Share this FAQ